MOD 3.1 solo/Alto Message_Decoder for SBD messages, format version 0.5-2.1
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD  8805     0  339 000000000000009    10     0   0 2019/12/06, 17:06:43, -45.3984 -170.0335   4      0 0,1,2,3,5
SBD  8805     0  265 000000000000009    11     0   1 2019/12/06, 17:06:55, -45.3940 -169.9831   3      0 6,9,10,11,14
# Raw packet information
#  Packets Bytes
P0      10   580
# Sensor_ID bytes     % packets Description
          0    24  4.14       1         GPS
          1    67 11.55       1    Pressure
          2    99 17.07       1 Temperature
          3    99 17.07       1    Salinity
          5    38  6.55       1        Rise
          6    37  6.38       1        Pump
          9    36  6.21       1  HR/Drift_P
         10    36  6.21       1  HR/Drift_T
         11    62 10.69       1  HR/Drift_S
         14    82 14.14       1 Engineering
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G    0 0 2019/12/06 17:04 -45.39749 -169.99434   20  8 29 36 45  0.8     0
# Profile Data
#      C Npts comp n_var
D0     0   60    1     3
#      Pres      Temp      Salinity
      0.320    14.3850    32.2430
      1.040    14.3850    34.7110
      1.960    14.3850    34.6910
      2.920    14.3840    34.7260
      3.920    14.3830    34.7100
      4.920    14.3820    34.7140
      6.080    14.3760    34.7070
      7.080    14.3810    34.6860
      8.000    14.2870    34.7360
      8.960    14.0140    34.7450
      9.920    13.8860    34.7320
     10.720    13.8260    34.7270
     12.080    13.7740    34.7240
     14.040    13.3410    34.7760
     16.000    13.1650    34.7570
     17.960    13.1140    34.7590
     19.960    13.0850    34.7610
     22.040    13.0730    34.7510
     24.120    13.0560    34.7560
     26.000    13.0250    34.7560
     27.960    13.0120    34.7600
     30.080    12.9810    34.7600
     32.040    12.9420    34.7630
     34.040    12.8340    34.7680
     36.040    12.7630    34.7650
     38.040    12.6840    34.7610
     40.040    12.5680    34.7690
     42.000    12.4530    34.7680
     43.960    12.4200    34.7600
     46.040    12.3630    34.7630
     48.080    12.2820    34.7640
     50.000    12.1520    34.7670
     51.920    12.0150    34.7990
     53.880    11.7720    34.8610
     55.880    11.5640    34.8150
     57.840    11.5150    34.8230
     59.880    11.4490    34.8280
     61.880    11.4370    34.8350
     63.920    11.3870    34.8380
     65.960    11.3880    34.8470
     68.000    11.3400    34.8480
     70.080    11.1930    34.8280
     72.120    11.1600    34.8140
     74.040    11.1810    34.8180
     75.960    11.1690    34.8300
     78.000    11.0640    34.8110
     80.040    11.0230    34.8090
     82.040    10.9920    34.8010
     84.040    10.9790    34.8030
     86.080    10.9820    34.8120
     88.000    10.9720    34.8120
     89.880    10.9670    34.8110
     91.920    10.9520    34.8120
     93.960    10.9330    34.8110
     96.000    10.9060    34.8060
     98.000    10.8840    34.8060
    100.000    10.8630    34.8030
    101.920    10.8530    34.8040
    104.000    10.8380    34.8030
    105.400    10.8330    34.8030
# High Resolution Data
#     C   Npts comp
H0     0   23    1
#      Pres      Temp      Salinity
      0.240    14.3840    29.7570
      0.400    14.3850    34.7300
      0.640    14.3860    34.7300
      0.920    14.3850    34.7300
      1.040    14.3850    34.7300
      1.320    14.3840    34.7130
      1.400    14.3850    34.6550
      1.640    14.3860    34.7290
      1.800    14.3860    34.7280
      2.120    14.3850    34.7160
      2.320    14.3830    34.5920
      2.520    14.3830    34.7290
      2.640    14.3840    34.7270
      2.880    14.3850    34.7210
      3.120    14.3850    34.7270
      3.400    14.3830    34.7280
      3.600    14.3800    34.7280
      3.840    14.3820    34.7250
      3.960    14.3850    34.7250
      4.320    14.3850    34.6620
      4.520    14.3850    34.7260
      4.720    14.3800    34.6930
      4.960    14.3790    34.7240
# Rise Time Series
#  Cycle  npts  Start
#               seconds        Date
R00    0   6    628966090 2019/12/06 16:48:10
# time    Pres   phase Date
     0   91.360   10    2019/12/06 16:48:10
    61  104.400    7    2019/12/06 16:49:11
   535    4.480   13    2019/12/06 16:57:05
   655    0.120   13    2019/12/06 16:59:05
   677    0.120    8    2019/12/06 16:59:27
   925    0.200   14    2019/12/06 17:03:35
# Pump Time Series
#  Cycle  npts
B00    0   3
#  Pres P_time  Volts    mAmps  Vacuum        phase
#                               Before  After
   91.360    22   15.52     380    3      3      2
  104.400    74   15.49     387    3      3      7
    0.120   240   15.47     231    3     11      8
# Argo Mission Data
#  Cycle Data  Miss Depth       Time                       Ascent Seeks Max Time           Cycle
#        Form  No   Prof  Park  Rise to Park to Prof Drift Speed        Surface  Seek      Time
A2    0  2.0    1   150   150   180   360       0      0    11     1    2160       0       0.2908
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Engineering Data
# Verification Cycle
# Communications:
#    Cycle Packets Time Interrupt
EC00     0       2    0         0
# Pump:
#   Cycle Voltages         Vacuum                 Currents      Times                          Vent
#         CPU  Pump  End   @50db S_Before S_After Pump   Max    To_Surf @_Surf To_Target Vent  Status
EP00    0 7.24 15.53 15.49 9.91  9.99     -9.90   387.0  411.0  96      240    -9.9      0.0   0     
# CTD:
#   Cycle Surface Pres    Start of Profile         End of Profile
#         Before  After   Pres    Temp    Salinity Pres    Temp    Salinity Scans Status
ES00    0 0.000   0.000   93.880  10.951  34.825   0.240   14.384  29.756   431   17    
# BIT Test:
#   Cycle Pcase Pump                        CPU     Except Ascent   Abort
#         Vac   Current Vac   Voltage Time  Voltage Flags  End Pres Code
E100    0 9.91  223.0   3.00  15.53   379   7.22    0      0.1200   0   
