# Message Decoder for Solo2, Alto and Deep Solo Floats; v231212
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD  8572   244  335 000000000624060  4390     0   0 2024/04/07, 19:14:26, -34.0401 -164.6917   2      0 15,0,9,10,11,0,4,11,14
SBD  8572   244  324 000000000624060  4391     0   1 2024/04/07, 19:14:39, -34.0348 -164.6499   2      0 6
SBD  8572   244  331 000000000624060  4392     0   2 2024/04/07, 19:14:51, -34.0401 -164.6917   2      0 5,6,9,10
# Delayed Transmission - (3/3) No surface time available, transmission was most likely delayed
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      14          954           36         990
# Sensor_ID bytes     % packets Description
          0    48  5.03       2         GPS
          4    33  3.46       1        Fall
          5   173 18.13       1        Rise
          6   404 42.35       2        Pump
          9    52  5.45       2  HR/Drift_P
         10    52  5.45       2  HR/Drift_T
         11    53  5.56       2  HR/Drift_S
         14   102 10.69       1 Engineering
         15    37  3.88       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  244 1 9999/99/99 99:99 +99.99000 +999.99000  600  0 250  0  0  0.0     0
G  244 2 9999/99/99 99:99 +99.99000 +999.99000  900  0 250  0  0  0.0     0
# High Resolution Data
#     C   Npts comp
H0   244    5     1
#      Pres      Temp    Salinity
   2240.400     2.0820    34.6774
   2240.400     2.0890    34.6806
   2240.400     2.0910    34.6806
   2240.400     2.0930    34.6790
   2240.400     2.0930    34.6792
# Drift Data
#      C Npts comp n_var
I0   244    1    1     3
#      Pres       Temp   Salinity
   2240.400     2.0690    34.6824
# Fall Time Series
#  Cycle npts     Start                Date
F00  244    5 764943904 2024/03/28 12:25:04
# time     Pres phase                Date
     0 2240.400     1 2024/03/28 12:25:04
   100 2240.400    10 2024/03/28 12:26:44
   155 2240.400     3 2024/03/28 12:27:39
  7373 2240.400     3 2024/03/28 14:27:57
 14600 2240.400     4 2024/03/28 16:28:24
# Rise Time Series
#  Cycle npts     Start                Date
R00  244   33 764958573 2024/03/28 16:29:33
# time     Pres phase                Date
     0 2240.400     7 2024/03/28 16:29:33
  1269 2240.400    13 2024/03/28 16:50:42
  2469 2240.400    13 2024/03/28 17:10:42
  3669 2240.400    13 2024/03/28 17:30:42
  4869 2240.400    13 2024/03/28 17:50:42
  6069 2240.400    13 2024/03/28 18:10:42
  7269 2240.400    13 2024/03/28 18:30:42
  8469 2240.400    13 2024/03/28 18:50:42
  9669 2240.400    13 2024/03/28 19:10:42
 10869 2240.400    13 2024/03/28 19:30:42
 12069 2240.400    13 2024/03/28 19:50:42
 13269 2240.400    13 2024/03/28 20:10:42
 14469 2240.400    13 2024/03/28 20:30:42
 15669 2240.400    13 2024/03/28 20:50:42
 16869 2240.400    13 2024/03/28 21:10:42
 18069 2240.400    13 2024/03/28 21:30:42
 19269 2240.400    13 2024/03/28 21:50:42
 20469 2240.400    13 2024/03/28 22:10:42
 21669 2240.400    13 2024/03/28 22:30:42
 22869 2240.400    13 2024/03/28 22:50:42
 24069 2240.400    13 2024/03/28 23:10:42
 25269 2240.400    13 2024/03/28 23:30:42
 26469 2240.400    13 2024/03/28 23:50:42
 27669 2240.400    13 2024/03/29 00:10:42
 28869 2240.400    13 2024/03/29 00:30:42
 30069 2240.400    13 2024/03/29 00:50:42
 31269 2240.400    13 2024/03/29 01:10:42
 32469 2240.400    13 2024/03/29 01:30:42
 33669 2240.400    13 2024/03/29 01:50:42
 34869 2240.400    13 2024/03/29 02:10:42
 36069 2240.400    13 2024/03/29 02:30:42
 36212 2240.400     8 2024/03/29 02:33:05
 37233 2240.400    14 2024/03/29 02:50:06
# Pump Time Series
#   Cycle npts
B01   244   36
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
  2240.400   17   14.79     390      2     2    10
  2240.400    2   14.85     405      2     2     3
  2240.400    2   15.01     413      2     2     3
  2240.400   44   14.93     401      2     2     7
  2240.400   17   14.94     404      2     2    13
  2240.400   17   14.92     404      2     2    13
  2240.400   17   14.90     408      2     2    13
  2240.400   17   14.88     405      2     2    13
  2240.400   17   14.87     404      2     2    13
  2240.400   17   14.86     408      2     2    13
  2240.400   17   14.85     405      2     2    13
  2240.400   17   14.84     405      2     2    13
  2240.400   17   14.83     402      2     2    13
  2240.400   17   14.83     405      2     2    13
  2240.400   17   14.82     404      2     2    13
  2240.400   17   14.81     403      2     2    13
  2240.400   17   14.81     402      2     2    13
  2240.400   17   14.81     407      2     2    13
  2240.400   17   14.81     405      2     2    13
  2240.400   17   14.80     404      2     2    13
  2240.400   17   14.80     407      2     2    13
  2240.400   17   14.79     404      2     2    13
  2240.400   17   14.80     410      2     2    13
  2240.400   17   14.79     409      2     2    13
  2240.400   17   14.79     403      2     2    13
  2240.400   17   14.79     405      2     2    13
  2240.400   17   14.79     405      2     2    13
  2240.400   17   14.79     403      2     2    13
  2240.400   17   14.80     406      2     2    13
  2240.400   17   14.82     407      2     2    13
  2240.400   17   14.84     407      2     2    13
  2240.400   17   14.85     413      2     2    13
  2240.400   17   14.86     413      2     2    13
  2240.400   17   14.87     409      2     2    13
  2240.400  180   14.79     392      2     2     8
  2240.400   20   14.78     397      2     2    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   244  2.4    6  2000  1000   600   400   330 13200     11     2    2160  120 10.0757
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      5000     -25
# Engineering Data
# Normal Cycle
# Communications:
#    Cycle Tries Sent Reply_Status SBD_status  Time  Interrupt  Packets
EC02  244    19    0         0      8           1213       0        3
# Pump:
#    Cycle Voltages          Vacuum                     Currents       Times                          Vent
#          CPU  Pump  End    @50db S_Before S_After     Pump  Max      To_Surf @_Surf To_Target Vent  Status
EP02  244  7.10 15.05 14.78 10.43  10.43    10.43       401.0  415.0   554     200     44.6    10.0    7
# CTD:
#   Cycle  Surface Pres     Start of Profile        End of Profile          Scans  Status P offset Bad Scans
#          Before    After  Pres    Temp   Salinity Pres    Temp   Salinity
ES02  244 2240.400 2240.400 2240.400   2.069  34.682 2240.400   2.082  34.677   30008   17     0.0600    0
# Cycle Info:
# Descent
#   Cycle P_Start P_end   Time   Speed    1st Seek        Except Ice Detect
#                                         Time  P_change  Flags  Status
EF02  244 2240.400 2240.400  100     0.0     0.2     0.0 13594       0
# Drift and Rise
#   Cycle   Drift            Ascent   Drift Averages
#           P_Start P_end    P_end    1st half                2nd half
#                                     Pres    Temp   Salinity Pres    Temp   Salinity
ET02  244 2240.400   -9.000 2240.400 2240.400   2.069  34.682 -999.000 -999.000 -999.000
