MOD 3.1 solo/Alto Message_Decoder for SBD messages, format version 0.5-2.1
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD  8184    -1  296 000000000475370    14     2   0 2013/12/09, 12:31:45, -50.9764  163.7808   3      0 0,2,3,1,5,6,14
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0       7          284           12         296
# Sensor_ID bytes     % packets Description
          0    24  8.45       1         GPS
          1    42 14.79       1    Pressure
          2    42 14.79       1 Temperature
          3    42 14.79       1    Salinity
          5    24  8.45       1        Rise
          6    34 11.97       1        Pump
         14    76 26.76       1 Engineering
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   -1 0 2013/12/09 12:30 -50.99113 +163.80550  210  3 33 41 46  2.9     0
# Profile Data
#      C Npts comp n_var
D0    -1   34    0     3
#      Pres      Temp    Salinity
      1.120    10.1140    34.6400
      2.000    10.1130    34.6450
      3.000    10.1150    34.6510
      4.040    10.1150    34.6480
      5.000    10.1140    34.6550
      6.000    10.1160    34.6530
      7.000    10.1160    34.6560
      8.040    10.1160    34.6570
      9.040    10.1150    34.6490
     10.000    10.1160    34.6530
     12.040    10.1080    34.6540
     14.000    10.1080    34.6540
     16.040    10.1050    34.6560
     18.000    10.1070    34.6550
     20.000    10.1090    34.6560
     21.960    10.1080    34.6550
     23.960    10.1060    34.6560
     26.000    10.0930    34.6560
     28.040    10.0890    34.6570
     30.000    10.0860    34.6580
     31.960    10.0850    34.6580
     33.960    10.0870    34.6570
     35.960    10.0740    34.6560
     37.960    10.0720    34.6580
     40.000    10.0580    34.6580
     42.000    10.0520    34.6580
     44.000    10.0490    34.6590
     46.040    10.0490    34.6590
     48.040    10.0480    34.6590
     50.040    10.0480    34.6590
     52.000    10.0480    34.6590
     54.000    10.0490    34.6590
     56.000    10.0490    34.6590
     58.280    10.0490    34.6600
# Rise Time Series
#  Cycle npts     Start                Date
R00   -1    4 439906333 2013/12/09 12:12:13
# time     Pres phase                Date
     0   53.560    -9 2013/12/09 12:12:13
   363   25.720    -9 2013/12/09 12:18:16
   488   13.000    -9 2013/12/09 12:20:21
   608    0.400    -9 2013/12/09 12:22:21
# Pump Time Series
#  Cycle  npts
B00   -1   3
#  Pres P_time  Volts    mAmps  Vacuum        phase
#                               Before  After
   53.560    70   14.82     433  159      7     -9
   13.000    25   14.70     374    6      7     -9
    0.240   250   14.34     329    6     40     -9
# Argo Mission Data
#  Cycle Data  Miss Depth       Time                       Ascent Seeks Max Time           Cycle
#        Form  No   Prof  Park  Rise to Park to Prof Drift Speed        Surface  Seek      Time
A2   -1  1.5    1   150   150   180   360       0      0    11     1    2160       0       0.2908
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Engineering Data
# Verification Cycle
# Communications:
#    Cycle Packets Time Interrupt
EC00    -1       0   40         0
# Pump:
#   Cycle Voltages         Vacuum                 Currents      Times                          Vent
#         CPU  Pump  End   @50db S_Before S_After Pump   Max    To_Surf @_Surf To_Target Vent  Status
EP00   -1 7.61 15.00 14.82 10.82 10.66    -9.90   433.0  464.0  95      250    -9.9      0.0   0     
# CTD:
#   Cycle Surface Pres    Start of Profile         End of Profile
#         Before  After   Pres    Temp    Salinity Pres    Temp    Salinity Scans Status
ES00   -1 0.000   0.000   51.840  10.049  34.664   0.840   10.114  34.658   562   17    
# BIT Test:
#   Cycle Pcase Pump                        CPU     Except Ascent   Abort
#         Vac   Current Vac   Voltage Time  Voltage Flags  End Pres Code
E100   -1 11.32 314.0   6.00  14.48   213   7.65    0      0.2400   0   
