MOD 3.1 solo/Alto Message_Decoder for SBD messages, format version 0.5-2.1
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD  7164   818  329 000000000595640  1392     0   0 2020/06/21, 05:22:02, -31.9665  -30.6008   4      0 15,0,0,4,5
SBD  7164   818  144 000000000595640  1393     0   1 2020/06/21, 05:22:16, -31.9665  -30.6008   4      0 6,14
# Delayed Transmission - (2/2) Messages transmitted   0.1 days after surfacing on 2020/06/21 03:13:07 (refer to 1st Rise scan < 3dbar)
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0       7          449           24         473
# Sensor_ID bytes     % packets Description
          0    48 10.69       2         GPS
          4   208 46.33       1        Fall
          5    36  8.02       1        Rise
          6    34  7.57       1        Pump
         14    98 21.83       1 Engineering
         15    25  5.57       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  818 1 2020/06/20 09:29 -31.99592  -30.74665  100  4 33 36 40  3.6     0
G  818 2 2020/06/21 05:17 -31.98141  -30.56326  100  5 29 36 43  3.5     0
# Fall Time Series
#  Cycle npts     Start                Date
F00  818   50 645960557 2020/06/20 09:29:17
# time     Pres phase                Date
     0   -0.040    -9 2020/06/20 09:29:17
   100    0.160    -9 2020/06/20 09:30:57
   200    0.120    -9 2020/06/20 09:32:37
   300    0.160    -9 2020/06/20 09:34:17
   400    0.280    -9 2020/06/20 09:35:57
   500    0.160    -9 2020/06/20 09:37:37
   600    0.160    -9 2020/06/20 09:39:17
   700    0.200    -9 2020/06/20 09:40:57
   800    0.160    -9 2020/06/20 09:42:37
   900    0.360    -9 2020/06/20 09:44:17
  1000    0.160    -9 2020/06/20 09:45:57
  1100    0.320    -9 2020/06/20 09:47:37
  1200    0.200    -9 2020/06/20 09:49:17
  1300    0.360    -9 2020/06/20 09:50:57
  1400    0.160    -9 2020/06/20 09:52:37
  1500    0.160    -9 2020/06/20 09:54:17
  1600    0.240    -9 2020/06/20 09:55:57
  1700    0.160    -9 2020/06/20 09:57:37
  1800    0.160    -9 2020/06/20 09:59:17
  1900    0.160    -9 2020/06/20 10:00:57
  2000    0.160    -9 2020/06/20 10:02:37
  2100    0.160    -9 2020/06/20 10:04:17
  2200    0.160    -9 2020/06/20 10:05:57
  2300    0.320    -9 2020/06/20 10:07:37
  2400    0.160    -9 2020/06/20 10:09:17
  2500    0.440    -9 2020/06/20 10:10:57
  2600    0.160    -9 2020/06/20 10:12:37
  2700    0.160    -9 2020/06/20 10:14:17
  2800    0.320    -9 2020/06/20 10:15:57
  2900    0.200    -9 2020/06/20 10:17:37
  3000    0.160    -9 2020/06/20 10:19:17
  3100    0.160    -9 2020/06/20 10:20:57
  3200    0.280    -9 2020/06/20 10:22:37
  3300    0.160    -9 2020/06/20 10:24:17
  3400    0.200    -9 2020/06/20 10:25:57
  3500    0.200    -9 2020/06/20 10:27:37
  3600    0.160    -9 2020/06/20 10:29:17
  3700    0.160    -9 2020/06/20 10:30:57
  3800    0.200    -9 2020/06/20 10:32:37
  3900    0.320    -9 2020/06/20 10:34:17
  4000    0.160    -9 2020/06/20 10:35:57
  4100    0.160    -9 2020/06/20 10:37:37
  4200    0.200    -9 2020/06/20 10:39:17
  4300    0.240    -9 2020/06/20 10:40:57
  4400    0.320    -9 2020/06/20 10:42:37
  4500    0.160    -9 2020/06/20 10:44:17
  4600    0.160    -9 2020/06/20 10:45:57
  4700    0.240    -9 2020/06/20 10:47:37
  4800    0.160    -9 2020/06/20 10:49:17
 31415    0.200    -9 2020/06/20 18:12:52
# Rise Time Series
#  Cycle npts     Start                Date
R00  818    7 646024387 2020/06/21 03:13:07
# time     Pres phase                Date
     0    0.200    -9 2020/06/21 03:13:07
  1200    0.160    -9 2020/06/21 03:33:07
  2400    0.160    -9 2020/06/21 03:53:07
  3600    0.160    -9 2020/06/21 04:13:07
  4800    0.160    -9 2020/06/21 04:33:07
  6015    0.080    -9 2020/06/21 04:53:22
  6959    0.120    -9 2020/06/21 05:09:06
# Pump Time Series
#  Cycle  npts
B00  818   3
#  Pres P_time  Volts    mAmps  Vacuum        phase
#                               Before  After
    0.120   -16   14.19     245    6      6     -9
    0.080    41   13.16     230    6      6     -9
    0.120   401   13.90     231    6     75     -9
# Argo Mission Data
#  Cycle Data  Miss Depth       Time                       Ascent Seeks Max Time           Cycle
#        Form  No   Prof  Park  Rise to Park to Prof Drift Speed        Surface  Seek      Time
A2  818  1.3    3  1500  1000   600   400     100    540    11     1    4000     120       1.0097
# Engineering Data
# Normal Cycle
# Communications:
#    Cycle Tries Sent Reply_Status SBD_status  Time  Interrupt  Packets
EC02  818     2    2         0      1            61       0        2
# Pump:
#    Cycle Voltages          Vacuum                     Currents       Times                          Vent
#          CPU  Pump  End    @50db S_Before S_After     Pump  Max      To_Surf @_Surf To_Target Vent  Status
EP02  818  6.84 15.19 13.90  0.00   9.94    10.71       230.0  250.0    41     401    6528.6     0.0    0
# CTD:
#   Cycle  Surface Pres     Start of Profile        End of Profile          Scans  Status P offset Bad Scans
#          Before    After  Pres    Temp   Salinity Pres    Temp   Salinity
ES02  818   0.040    0.120    0.080  19.269   1.564   0.240  19.261   0.995      2   21     0.1462    0
# Cycle Info:
# Descent
#   Cycle P_Start P_end   Time   Speed    1st Seek        Except Ice Detect
#                                         Time  P_change  Flags  Status
EF02  818    0.120    0.200 65535     0.0    25.0     0.2  196      -9
# Drift and Rise
#   Cycle   Drift            Ascent   Drift Averages
#           P_Start P_end    P_end    1st half                2nd half
#                                     Pres    Temp   Salinity Pres    Temp   Salinity
ET02  818    0.200    0.160    0.120    0.120  19.398   0.925    0.120  19.350   1.223
