MOD 3.1 solo/Alto Message_Decoder for SBD messages, format version 0.5-2.1
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD  7164   776  329 000000000595640  1326     0   0 2020/05/04, 18:22:25, -30.5044  -31.2141   3      0 15,0,0,4,5
SBD  7164   776  144 000000000595640  1327     0   1 2020/05/04, 18:22:34, -30.5355  -31.2141   3      0 6,14
# Delayed Transmission - (2/2) Messages transmitted   0.1 days after surfacing on 2020/05/04 16:13:32 (refer to 1st Rise scan < 3dbar)
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0       7          449           24         473
# Sensor_ID bytes     % packets Description
          0    48 10.69       2         GPS
          4   208 46.33       1        Fall
          5    36  8.02       1        Rise
          6    34  7.57       1        Pump
         14    98 21.83       1 Engineering
         15    25  5.57       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  776 1 2020/05/03 22:29 -30.60416  -31.16520   90  5 37 40 43  1.1     0
G  776 2 2020/05/04 18:18 -30.54631  -31.24235  100  4 26 32 37  1.3     0
# Fall Time Series
#  Cycle  npts  Start
#               seconds        Date
F00  776  50    641860182 2020/05/03 22:29:42
# time    Pres   phase Date
     0   -0.040   -9    2020/05/03 22:29:42
   100    0.200   -9    2020/05/03 22:31:22
   200    0.160   -9    2020/05/03 22:33:02
   300    0.160   -9    2020/05/03 22:34:42
   400    0.120   -9    2020/05/03 22:36:22
   500    0.280   -9    2020/05/03 22:38:02
   600    0.160   -9    2020/05/03 22:39:42
   700    0.160   -9    2020/05/03 22:41:22
   800    0.120   -9    2020/05/03 22:43:02
   900    0.120   -9    2020/05/03 22:44:42
  1000    0.160   -9    2020/05/03 22:46:22
  1100    0.160   -9    2020/05/03 22:48:02
  1200    0.120   -9    2020/05/03 22:49:42
  1300    0.160   -9    2020/05/03 22:51:22
  1400    0.160   -9    2020/05/03 22:53:02
  1500    0.160   -9    2020/05/03 22:54:42
  1600    0.160   -9    2020/05/03 22:56:22
  1700    0.160   -9    2020/05/03 22:58:02
  1800    0.160   -9    2020/05/03 22:59:42
  1900    0.160   -9    2020/05/03 23:01:22
  2000    0.200   -9    2020/05/03 23:03:02
  2100    0.160   -9    2020/05/03 23:04:42
  2200    0.120   -9    2020/05/03 23:06:22
  2300    0.160   -9    2020/05/03 23:08:02
  2400    0.160   -9    2020/05/03 23:09:42
  2500    0.160   -9    2020/05/03 23:11:22
  2600    0.120   -9    2020/05/03 23:13:02
  2700    0.200   -9    2020/05/03 23:14:42
  2800    0.160   -9    2020/05/03 23:16:22
  2900    0.120   -9    2020/05/03 23:18:02
  3000    0.120   -9    2020/05/03 23:19:42
  3100    0.160   -9    2020/05/03 23:21:22
  3200    0.160   -9    2020/05/03 23:23:02
  3300    0.160   -9    2020/05/03 23:24:42
  3400    0.160   -9    2020/05/03 23:26:22
  3500    0.160   -9    2020/05/03 23:28:02
  3600    0.160   -9    2020/05/03 23:29:42
  3700    0.120   -9    2020/05/03 23:31:22
  3800    0.160   -9    2020/05/03 23:33:02
  3900    0.160   -9    2020/05/03 23:34:42
  4000    0.120   -9    2020/05/03 23:36:22
  4100    0.160   -9    2020/05/03 23:38:02
  4200    0.200   -9    2020/05/03 23:39:42
  4300    0.160   -9    2020/05/03 23:41:22
  4400    0.160   -9    2020/05/03 23:43:02
  4500    0.120   -9    2020/05/03 23:44:42
  4600    0.160   -9    2020/05/03 23:46:22
  4700    0.160   -9    2020/05/03 23:48:02
  4800    0.120   -9    2020/05/03 23:49:42
 31415    0.280   -9    2020/05/04 07:13:17
# Rise Time Series
#  Cycle  npts  Start
#               seconds        Date
R00  776   7    641924012 2020/05/04 16:13:32
# time    Pres   phase Date
     0    0.160   -9    2020/05/04 16:13:32
  1200    0.200   -9    2020/05/04 16:33:32
  2400    0.240   -9    2020/05/04 16:53:32
  3600    0.160   -9    2020/05/04 17:13:32
  4800    0.200   -9    2020/05/04 17:33:32
  6015    0.080   -9    2020/05/04 17:53:47
  6959    0.160   -9    2020/05/04 18:09:31
# Pump Time Series
#  Cycle  npts
B00  776   3
#  Pres P_time  Volts    mAmps  Vacuum        phase
#                               Before  After
    0.120   -16   14.23     227    6      6     -9
    0.080    41   12.93     218    6      7     -9
    0.240   401   13.93     221    6     73     -9
# Argo Mission Data
#  Cycle Data  Miss Depth       Time                       Ascent Seeks Max Time           Cycle
#        Form  No   Prof  Park  Rise to Park to Prof Drift Speed        Surface  Seek      Time
A2  776  1.3    3  1500  1000   600   400     100    540    11     1    4000     120       1.0097
# Engineering Data
# Normal Cycle
# Communications:
#    Cycle Tries Sent Reply_Status SBD_status  Time  Interrupt  Packets
EC02  776     2    2         0      1            55       0        2
# Pump:
#    Cycle Voltages          Vacuum                     Currents       Times                          Vent
#          CPU  Pump  End    @50db S_Before S_After     Pump  Max      To_Surf @_Surf To_Target Vent  Status
EP02  776  6.88 15.19 13.93  0.00   9.84    10.61       218.0  230.0    41     401    6528.6     0.0    0
# CTD:
#   Cycle  Surface Pres     Start of Profile        End of Profile          Scans  Status P offset Bad Scans
#          Before    After  Pres    Temp   Salinity Pres    Temp   Salinity
ES02  776   0.120    0.120    0.200  19.105   1.486   0.120  18.870   1.197      2   21     0.2437    0
# Cycle Info:
# Descent
#   Cycle P_Start P_end   Time   Speed    1st Seek        Except Ice Detect
#                                         Time  P_change  Flags  Status
EF02  776    0.120    0.120 65535     0.0    25.0     0.0  196      -9
# Drift and Rise
#   Cycle   Drift            Ascent   Drift Averages
#           P_Start P_end    P_end    1st half                2nd half
#                                     Pres    Temp   Salinity Pres    Temp   Salinity
ET02  776    0.280    0.120    0.240    0.160  19.234   1.552    0.120  19.626   1.616
