# Message Decoder for Solo2, Alto and Deep Solo Floats; v231212
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD  6065   230   64 000000000000009  2315    21   0 2024/01/10, 16:44:22, -34.0785 -157.0633   6      0 0,14
SBD  6065   230   12 000000000000009  2316     0   1 2024/01/10, 16:44:50, -34.0864 -157.1032   6      0 
SBD  6065   230  328 000000000000009  2317     0   2 2024/01/10, 16:44:59, -34.0864 -157.1032   7      0 13
SBD  6065   230  328 000000000000009  2318     0   3 2024/01/10, 16:46:41, -34.1270 -157.0464   4      0 13
SBD  6065   230  328 000000000000009  2319     0   4 2024/01/10, 16:47:42, -34.0785 -157.0633   4      0 13
SBD  6065   230  328 000000000000009  2320     0   5 2024/01/10, 16:48:09, -34.1270 -157.0464   3      0 13
SBD  6065   230  120 000000000000009  2321     0   6 2024/01/10, 16:48:24, -34.1270 -157.0464   3      0 13
SBD  6065   230   18 000000000000009  2322     0   7 2024/01/10, 16:48:51, -34.1569 -157.0464   8      0 13
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0       8         1430           96        1526
# Sensor_ID bytes     % packets Description
          0    24  1.68       1         GPS
         13  1378 96.36       6  Parameters
         14    28  1.96       1 Engineering
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  230 3 2024/01/10 16:43 -34.12051 -157.05647   90  7 37 40 45  1.7     0
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   230  1.1    9  5700  5000  1500  1500     3 12000     12     2    2700  300 10.3748
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           10      10      1000       5      1000       1
# Engineering Data
# Abort Cycle
# Abort code 0 Description: no error
# Communications:
#   Cycle Tries Sent Reply_Status Reply_Error SBD_status Tx_time(s) last_Interrupt msg_version
EC03  230     0    1           -9          -9          0          1           1089           6
# Pump:
#   Cycle Voltages(v) Vacuum(inHg) Abort
#         CPU   Pump  @Tx   @abort  Code
EP03  230 00.00 07.29 11.00  00.00     0
# Upload command:
C P;
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  230       105    2024/01/10 16:48:51
 PROup =    3|
  BLOK =    1|
   PB1 =   10|
   PB2 = 1005|
   AV1 =    2|
   AV2 =    4|
PchSec =  600|
MaxHrs = 4320|
dBarGo =   50|
ABcymn =   60|
  Cyc0 =  999|
  Cyc1 =  999|
  Cyc2 =  999|
Zpark0 =  600|
Zpark1 = 3500|
Zpark2 = 5000|
 Zpro0 = 1000|
 Zpro1 = 4000|
 Zpro2 = 5700|
Tlast0 =  460|
Tlast1 =  165|
Tlast2 =   95|
ccPro0 = 3240|
ccPro1 = 1895|
ccPro2 = 1108|
ccPrk0 = 3500|
ccPrk1 = 2225|
ccPrk2 = 1380|
 Rise0 =  300|
 Rise1 = 1200|
 Rise2 = 1500|
 Fall0 =  300|
 Fall1 = 1200|
 Fall2 = 1500|
SAMmn0 =   15|
SAMmn1 =   60|
SAMmn2 =  240|
 Nsam0 =    4|
 Nsam1 =    8|
 Nsam2 =   50|
PmpPrk =   80|
PmpSlo =   80|
 MxHiP = 1200|
 MnSfP =  100|
 MxSfP =  600|
GPSsec =  600|
IRIsec =  900|
mxSeek =   60|
 Nseek =    2|
STLmin =  300|
  dDdZ = 3600|
 dDdV2 =  700|
Ventsc =    2|
Ventop =    2|
SkSLsc =  240|
AsSLsc =  360|
Ndives = 9999|
BinMod =    2|
UNBINd =    2|
UBZmax =   20|
   UBn =   60|
DrfDat =    1|
  dZ_0 =   10|
  dZ_1 =   10|
  dZ_2 =   10|
  dZ_3 =   10|
  dZ_4 =   10|
 Z_C2D = 2000|
 Z_D2C =  750|
Z_DP_1 = 3000|
Z_DP_2 = 4000|
Z_DP_3 = 5000|
Z_DP_4 = 5200|
PTS_T0 =    2|
PTS_T1 =    2|
PTS_T2 =    2|
PTS_T3 =    2|
PTS_T4 =    2|
 dT_CP =   60|
 dT_DP =  900|
T_WAIT =    0|
FP_CHK =   60|
dT_PMP =   20|
Z_dPMP =   50|
Z1_RIS = 2000|
Z2_RIS = 1000|
W0_RIS =    5|
W1_RIS =    8|
W2_RIS =   12|
MAX_XL =   32|
CCzero =  732|
 CC_km =   22|
CCC_CF =   70|
 ccCOR = -284|
 WD_EN =    1|
WDhour =  360|
Ice_Pd =   40|
Ice_Ps =   20|
Ice_Mn =    0|
Ice_Tc = -165|
Ice_Sc =   20|
 IceIn =    5|
IceOut =    3|
PrkMin =  200|
Dump_Param = 1 |

