# Message Decoder for Solo2, Alto and Deep Solo Floats; v220120
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD  6034    -1  328 000000000528590    12     0   0 2017/10/01, 21:07:46, -41.2929  174.7839   3      0 13
SBD  6034    -1  328 000000000528590    13     0   1 2017/10/01, 21:07:56, -41.2951  174.8315   3      0 13
SBD  6034    -1  328 000000000528590    14     0   2 2017/10/01, 21:08:09, -41.2951  174.8315   3      0 13
SBD  6034    -1  302 000000000528590    15     0   3 2017/10/01, 21:10:38, -41.2951  174.8315   3      0 13
SBD  6034    -1  131 000000000528590    16     0   4 2017/10/01, 21:10:48, -41.2951  174.8315   3      0 15,0,14
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0       7         1357           60        1417
# Sensor_ID bytes     % packets Description
          0    24  1.77       1         GPS
         13  1238 91.23       4  Parameters
         14    58  4.27       1 Engineering
         15    37  2.73       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   -1 5 2017/10/01 21:07 -41.30297 +174.80565   40 11 33 38 46  0.8     0
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    -1  0.7    1  1000   960   300   300     0    60      0     2    2700  120  0.6562
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           10      10      1000       5      1000       1
# Engineering Data
# Successful Bit Test
# Communications:
#    Cycle Sent Interrupt
EC05   -1  6     6
# Pump:
#    Cycle Voltages          Current  Pcase Vacuum Pump
#          CPU  No Load  End (Avg)    Start End    Time
EP05   -1  7.48 15.65  15.36  399.0   10.40 11.44   811
# CTD:
#   Cycle  P offset
ES05   -1  -0.589
# Hydraulic Test:
#   Cycle  Pump Vacuum         Valve Tries
#          Before   After      Open   Close
EB05   -1   0.0     1.03         0      0
#   Cycle  SBE String
ER05   -1 0.4,14.1799,0.007
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   -1        94    2017/10/01 21:10:38
  BLOK =    1|
   PB1 =   10|
   PB2 = 1005|
   AV1 =    2|
   AV2 =    4|
PchSec =  600|
MaxHrs = 4320|
dBarGo =   50|
ABcymn =   60|
  Cyc0 = 9999|
  Cyc1 = 9999|
  Cyc2 = 9999|
Zpark0 =  960|
Zpark1 = 3960|
Zpark2 = 4960|
 Zpro0 = 1000|
 Zpro1 = 4000|
 Zpro2 = 5000|
Tlast0 =  460|
Tlast1 =  165|
Tlast2 =   95|
ccPro0 = 3150|
ccPro1 = 1750|
ccPro2 = 1360|
ccPrk0 = 3170|
ccPrk1 = 1765|
ccPrk2 = 1375|
 Rise0 =  300|
 Rise1 = 1200|
 Rise2 = 1500|
 Fall0 =  300|
 Fall1 = 1200|
 Fall2 = 1500|
SAMmn0 =   15|
SAMmn1 =   30|
SAMmn2 =   60|
 Nsam0 =    4|
 Nsam1 =    8|
 Nsam2 =   12|
PmpPrk =   80|
PmpSlo =   80|
 MxHiP = 1200|
 MnSfP =  100|
 MxSfP =  600|
GPSsec =  600|
IRIsec =  900|
mxSeek =   60|
 Nseek =    2|
STLmin =  120|
  dDdZ = 3600|
 dDdV2 =  700|
Ventsc =    2|
Ventop =    2|
SkSLsc =  240|
AsSLsc =  300|
Ndives = 9999|
BinMod =    2|
UNBINd =    2|
UBZmax =    0|
   UBn =   60|
DrfDat =    1|
  dZ_0 =    5|
  dZ_1 =   10|
  dZ_2 =   20|
  dZ_3 =   50|
  dZ_4 =   10|
 Z_C2D =  500|
Z_DP_1 = 1000|
Z_DP_2 = 2000|
Z_DP_3 = 3000|
Z_DP_4 = 3800|
PTS_T0 =   20|
PTS_T1 =   10|
PTS_T2 =    5|
PTS_T3 =    2|
PTS_T4 =    2|
 dT_CP =   60|
 dT_DP =  900|
T_WAIT =    0|
FP_CHK =   60|
dT_PMP =   20|
Z_dPMP =   50|
Z1_RIS = 2000|
Z2_RIS = 1000|
W0_RIS =    5|
W1_RIS =    8|
W2_RIS =   12|
MAX_XL =   32|
CCzero =  732|
 CC_km =   22|
CCC_CF =   50|
 ccCOR =    0|
 WD_EN =    1|
WDhour =  360|

