# Message Decoder for Solo2, Alto and Deep Solo Floats; v220120
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD  6023     0  335 000000000753410    30     5   0 2016/10/23, 16:37:54, -38.0010  114.9756   3      0 0,5,6
SBD  6023     0   88 000000000753410    31     0   1 2016/10/23, 16:38:18, -38.0010  114.9756   4      0 14
SBD  6023     0   55 000000000753410    32     0   2 2016/10/23, 16:38:31, -38.0010  114.9756   4      0 13
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0       5          442           36         478
# Sensor_ID bytes     % packets Description
          0    24  5.43       1         GPS
          5   113 25.57       1        Rise
          6   186 42.08       1        Pump
         13    43  9.73       1  Parameters
         14    76 17.19       1 Engineering
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G    0 0 2016/10/23 16:37 -37.98613 +114.99206  100  6 35 41 47  1.0     0
# Rise Time Series
#  Cycle npts     Start                Date
R00    0   21 530550426 2016/10/23 15:07:06
# time     Pres phase                Date
     0  132.800    10 2016/10/23 15:07:06
    60  150.600     6 2016/10/23 15:08:06
   500  238.100     7 2016/10/23 15:15:26
   880  275.600    13 2016/10/23 15:21:46
  1180  302.600    13 2016/10/23 15:26:46
  1480  322.100    13 2016/10/23 15:31:46
  1780  330.700    13 2016/10/23 15:36:46
  2080  319.300    13 2016/10/23 15:41:46
  2380  299.700    13 2016/10/23 15:46:46
  2680  276.400    13 2016/10/23 15:51:46
  2980  250.400    13 2016/10/23 15:56:46
  3280  220.700    13 2016/10/23 16:01:46
  3580  187.100    13 2016/10/23 16:06:46
  3880  150.400    13 2016/10/23 16:11:46
  4180  113.000    13 2016/10/23 16:16:46
  4480   75.600    13 2016/10/23 16:21:46
  4780   36.800    13 2016/10/23 16:26:46
  4960   13.900    13 2016/10/23 16:29:46
  5080    0.700    13 2016/10/23 16:31:46
  5140    0.800     8 2016/10/23 16:32:46
  5180    0.800    14 2016/10/23 16:33:26
# Pump Time Series
#   Cycle npts
B01     0   14
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
   150.600  420   14.30     451      2    35    13
   238.100   80   14.22     469      3    37     7
   302.600   40   14.72     521      2    37    13
   322.100   40   14.65     532      2    35    13
   330.700   40   14.63     541      2    36    13
   319.300   40   14.63     537      2    39    13
   299.700   40   14.62     541      2    40    13
   276.400   40   14.63     525      3    41    13
   250.400   40   14.64     507      2    45    13
   220.700   40   14.65     494      3    51    13
   187.100   40   14.66     467      5    79    13
     0.700   40   14.96     366     22   243    13
     0.800    1   14.81     376    243   243    14
     0.800   99   14.59     346    243   243    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2     0  0.5    1   150   150   180   360     0     0     11     1    2160    0  0.2908
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           10      10      1000       5      1000       1
# Engineering Data
# Verification Cycle
# Communications:
#    Cycle Packets Time Interrupt
EC00     0       2    0         0
# Pump:
#   Cycle Voltages         Vacuum                 Currents      Times                          Vent
#         CPU  Pump  End   @50db S_Before S_After Pump   Max    To_Surf @_Surf To_Target Vent  Status
EP00    0 7.68 14.98 14.22 10.49 11.75    -9.90   469.0  0.0    480     100    -9.9      -99.0 0     
# CTD:
#   Cycle Surface Pres    Start of Profile         End of Profile
#         Before  After   Pres    Temp    Salinity Pres    Temp    Salinity Scans Status
ES00    0 0.500   0.600   127.900 11.609  35.042   -10.000 0.000   0.000    0     0     
# BIT Test:
#   Cycle Pcase Pump                        CPU     Except Ascent   Abort
#         Vac   Current Vac   Voltage Time  Voltage Flags  End Pres Code
E100    0 10.50 344.0   5.00  7.68    815   0.00    3207   0.8000   0   
# Upload command:
C L 750 40 50 6600;Z0 9999 960 1000 2700;
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M    0         9    2016/10/23 16:38:31
SrfDrf = 750 |
SrfLon = 40 |
SrfLat = 50 |
SrfMxN = 6600 |
SrfInt = 39 |
  Cyc0 = 9999 |
 Ztar0 = 960 |
 Zpro0 = 1000 |
 Tlast = 2700 |

