# Message Decoder for Solo2, Alto and Deep Solo Floats; v260609
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD  2115    -1  328 000000000000009     6     0   0 2026/02/24, 12:20:42,  41.9511  -90.7729 241      0 13
SBD  2115    -1  328 000000000000009     7     0   1 2026/02/24, 12:21:55,  41.9894  -87.9718   4      0 13
SBD  2115    -1  328 000000000000009     8     0   2 2026/02/24, 12:22:00,  42.0135  -87.9701   4      0 13
SBD  2115    -1  228 000000000000009     9     0   3 2026/02/24, 12:22:09,  41.9894  -87.9718   4      0 13,0,14
SBD  2115    -1   72 000000000000009    10     0   0 2026/06/08, 19:54:27,  41.5413  -70.6851   4      0 14
SBD  2115    -1  328 000000000000009    11     0   0 2026/06/11, 16:36:09,  41.5187  -70.7009   4      0 13
SBD  2115    -1  328 000000000000009    12     0   2 2026/06/11, 16:38:52,  41.5187  -70.7009   2      0 13
SBD  2115    -1  328 000000000000009    13     0   1 2026/06/11, 16:42:48,  41.5334  -70.6398   3      0 13
# Delayed Transmission - (4/8) Messages transmitted 9551.5 days after surfacing on 2000/01/01 00:00:00 (refer to Surface Pressure Start Time)
# Delayed Transmission - (1/8) Messages transmitted 9655.8 days after surfacing on 2000/01/01 00:00:00 (refer to Surface Pressure Start Time)
# Delayed Transmission - (3/8) Messages transmitted 9658.7 days after surfacing on 2000/01/01 00:00:00 (refer to Surface Pressure Start Time)
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      10         2172           96        2268
# Sensor_ID bytes     % packets Description
          0    24  1.10       1         GPS
         13  2030 93.46       7  Parameters
         14   118  5.43       2 Engineering
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   -1 5 2026/02/24 12:20 +41.98994  -87.98548   40 11 40 43 49  0.9     0
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    -1  2.0    1   150   150   180   360     0     0     11     1    2160    0  0.2908
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Engineering Data
# Failed Bit Test
# Communications:
#    Cycle Sent Interrupt BIT Status (hex)
EC06   -1  1     6      0008
# Pump:
#    Cycle Voltages          Current  Pcase Vacuum Pump
#          CPU  No Load  End (Avg)    Start End    Time
EP06   -1  0.00 15.48  15.22  197.0   10.21 10.20   393
# CTD:
#   Cycle  P offset
ES06   -1  -0.050
# BIT Test:
#   Cycle  Pump Vacuum         Valve Tries
#   Cycle  Before   After      Open   Close
EB06   -1   0.1     2.55         0      0
#   Cycle  SBE StringER06   -1   -0.26, 16.5761,  0.0085
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   -1        82    2026/06/11 16:42:48
 PROup =    1|
  BLOK =    1|
   PB1 =   10|
   PB2 = 1005|
   AV1 =    2|
   AV2 =    2|
PchSec =  600|
MaxHrs = 9999|
dBarGo =   50|
ABcymn =   60|
  Cyc0 =    1|
  Cyc1 =    3|
  Cyc2 = 9999|
 Ztar0 = 1000|
 Ztar1 = 1000|
 Ztar2 = 1000|
 Zpro0 = 1100|
 Zpro1 = 1500|
 Zpro2 = 2000|
Tlast0 =  308|
Tlast1 =  308|
Tlast2 =  308|
 Rise0 =  180|
 Rise1 =  600|
 Rise2 =  600|
 Fall0 =  360|
 Fall1 =  400|
 Fall2 =  400|
SAMmn0 =    0|
SAMmn1 =  120|
SAMmn2 =  120|
 Nsam0 =    0|
 Nsam1 =   10|
 Nsam2 =  115|
Pwait0 =    0|
Pwait1 =  100|
Pwait2 =  400|
PmpBtm =   40|
Pmpslo =   20|
MinRis =   11|
 MxHiP =  600|
VlvDly =    0|
 MxSfP =  400|
GPSsec = 1200|
IRIsec = 1200|
 dTadZ = 2440|
 dTsdZ = 1560|
FuseMd =    0|
dTfldZ = 1900|
mxSeek =   60|
 Nseek =    1|
STLmin =  120|
 MnSfP =  200|
Ventsc =   10|
Ventop =    2|
SkSLsc = 1200|
AsSLsc =  900|
Ndives = 9999|
CTDofZ =    1|
SrfDft =    0|
SrfLon =    0|
SrfLat =    0|
SrfMxN =    0|
SrfInt =   20|
BinMod =    2|
UNBINd =    1|
UBZmax =   50|
   UBn = 1020|
DrfDat =    1|
Ice_Pd =   40|
Ice_Ps =   20|
Ice_Mn =    0|
Ice_Tc = -165|
Ice_Sc =   30|
 IceIn =   -1|
IceOut =   -1|
   XP0 =    0|
   XP1 =    0|
   XP2 =    0|
 XPdly =    0|
 XP0dZ =   -1|
FstSrf =   -1|

