# Message Decoder for Solo2, Alto and Deep Solo Floats; v260727
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 12161    -1  115 000000000000009    18     0   0 2026/08/13, 21:39:26,  47.7005 -122.2258   3      0 15,14
SBD 12161    -1  328 000000000000009    22     0   4 2026/08/13, 21:40:03,  47.7005 -122.2258   3      0 13
SBD 12161    -1  328 000000000000009    23     0   5 2026/08/13, 21:40:14,  47.6863 -122.2709   3      0 13
SBD 12161    -1  261 000000000000009    24     0   6 2026/08/13, 21:42:52,  47.7005 -122.2258   8      0 13,0
# Delayed Transmission - (4/4) Messages transmitted 9721.9 days after surfacing on 2000/01/01 00:00:00 (refer to Surface Pressure Start Time)
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0       6          984           48        1032
# Sensor_ID bytes     % packets Description
          0    24  2.44       1         GPS
         13   857 87.09       3  Parameters
         14    62  6.30       1 Engineering
         15    41  4.17       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   -1 5 2026/08/13 21:38 +47.68678 -122.25946   80 13 28 37 45  1.1     0
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    -1  2.2    1  1000   600   300   300     3   240      0     4    2700  120  0.9500
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           10      10      1000       5      1000       1
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           10      10      1000       5      1000       1
# Engineering Data
# Successful Bit Test
# Communications:
#    Cycle Sent Interrupt
EC05   -1  7     6
# Pump:
#    Cycle Voltages          Current  Pcase Vacuum Pump
#          CPU  No Load  End (Avg)    Start End    Time
EP05   -1  7.07 15.42  15.13  347.0   10.23 11.54   837
# CTD:
#   Cycle  P offset
ES05   -1  -1.860
# Hydraulic Test:
#   Cycle  Pump Vacuum         Valve Tries
#          Before   After      Open   Close
EB05   -1   0.0     1.25         0      0
#   Cycle  RH CTD_SN SBE String
ER05   -1  58   6257 0.1,22.5734,0.000
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   -1        65    2026/08/13 21:42:52
Ventop =    2|
SkSLsc =  240|
AsSLsc =  360|
Ndives = 9999|
BinMod =    2|
UNBINd =    2|
UBZmax =   20|
   UBn =   60|
DrfDat =    1|
  dZ_0 =   10|
  dZ_1 =   10|
  dZ_2 =   10|
  dZ_3 =   10|
  dZ_4 =   10|
 Z_C2D = 2000|
 Z_D2C = 1000|
Z_DP_1 = 3000|
Z_DP_2 = 4000|
Z_DP_3 = 5000|
Z_DP_4 = 5200|
PTS_T0 =    2|
PTS_T1 =    2|
PTS_T2 =    2|
PTS_T3 =    2|
PTS_T4 =    2|
   AZ1 = 1000|
   AZ2 = 2000|
   AZ3 = 3000|
 Adz_1 =   10|
 Adz_2 =   20|
 Adz_3 =   40|
   AT1 =    3|
   AT2 =    3|
   AT3 =    3|
 dT_CP =   60|
 dT_DP =  900|
T_WAIT =    0|
FP_CHK =   60|
dT_PMP =   20|
Z_dPMP =   50|
Z1_RIS = 2000|
Z2_RIS = 1000|
W0_RIS =    5|
W1_RIS =    8|
W2_RIS =   12|
MAX_XL =   32|
CCzero =  732|
 CC_km =   22|
CCC_CF =   20|
 ccCOR = -700|
 WD_EN =    1|
WDhour =  360|
Ice_Pd =   40|
Ice_Ps =   20|
Ice_Mn =    0|
Ice_Tc = -165|
Ice_Sc =   20|
 IceIn =    5|
IceOut =    3|
PrkMin =  200|
 RstPr =    1|
ptRise =    1|
ptFall =    2|
tmTurn =  240|
 tmChk =   60|

