# Message Decoder for Solo2, Alto and Deep Solo Floats; v260317
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11732   104  249 000000000000009  1519     0   4 2026/04/13, 02:58:13,  41.5413  -70.6851   3      0 13,6,14,14
SBD 11732   104  339 000000000000009  1520     0   3 2026/04/13, 02:58:29,  41.5603  -70.6398   4      0 13
SBD 11732   104  275 000000000000009  1521     0   2 2026/04/13, 02:58:40,  41.5603  -70.6398   4      0 8,15
SBD 11732   104  335 000000000000009  1522     0   1 2026/04/13, 03:07:02,  41.5603  -70.6398   3      0 8
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0       8         1150           48        1198
# Sensor_ID bytes     % packets Description
          6    37  3.22       1        Pump
          8   520 45.22       2     Surface
         13   415 36.09       2  Parameters
         14   112  9.74       2 Engineering
         15    66  5.74       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
# Surface Pressure Time Series
#    C npts n_var               Start
S  104    56     6 2026/04/13 02:41:14
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
     0      0.280     20.056      0.277      14 2026/04/13 02:41:14
    11      0.270     20.061      0.270      14 2026/04/13 02:41:25
    22      0.280     20.070      0.275      14 2026/04/13 02:41:36
    33      0.280     20.067      0.278      14 2026/04/13 02:41:47
    44      0.260     20.072      0.265      14 2026/04/13 02:41:58
    55      0.280     20.074      0.276      14 2026/04/13 02:42:09
    65      0.270     20.074      0.274      14 2026/04/13 02:42:19
    76      0.280     20.071      0.282      14 2026/04/13 02:42:30
    87      0.270     20.067      0.273      14 2026/04/13 02:42:41
    98      0.270     20.069      0.270      14 2026/04/13 02:42:52
   109      0.270     20.074      0.270      14 2026/04/13 02:43:03
   120      0.270     20.074      0.269      14 2026/04/13 02:43:14
   130      0.280     20.073      0.277      14 2026/04/13 02:43:24
   141      0.270     20.081      0.269      14 2026/04/13 02:43:35
   152      0.270     20.075      0.272      14 2026/04/13 02:43:46
   163      0.280     20.078      0.278      14 2026/04/13 02:43:57
   173      0.280     20.080      0.280      14 2026/04/13 02:44:07
   184      0.270     20.085      0.269      14 2026/04/13 02:44:18
   195      0.270     20.085      0.274      14 2026/04/13 02:44:29
   206      0.280     20.072      0.279      14 2026/04/13 02:44:40
   216      0.280     20.072      0.276      14 2026/04/13 02:44:50
   227      0.280     20.076      0.283      14 2026/04/13 02:45:01
   238      0.270     20.076      0.272      14 2026/04/13 02:45:12
   249      0.270     20.074      0.270      14 2026/04/13 02:45:23
   259      0.280     20.068      0.279      14 2026/04/13 02:45:33
   270      0.270     20.067      0.267      14 2026/04/13 02:45:44
   281      0.270     20.065      0.273      14 2026/04/13 02:45:55
   292      0.270     20.061      0.274      14 2026/04/13 02:46:06
   302      0.270     20.061      0.273      14 2026/04/13 02:46:16
   313      0.280     20.070      0.280      14 2026/04/13 02:46:27
   324      0.280     20.070      0.281      14 2026/04/13 02:46:38
   335      0.270     20.065      0.271      14 2026/04/13 02:46:49
   345      0.280     20.066      0.284      14 2026/04/13 02:46:59
   356      0.280     20.072      0.277      14 2026/04/13 02:47:10
   367      0.270     20.075      0.273      14 2026/04/13 02:47:21
     0      0.270     20.078      0.271      14 2026/04/13 02:47:32
    11      0.280     20.081      0.283      14 2026/04/13 02:47:43
    21      0.280     20.079      0.281      14 2026/04/13 02:47:53
    32      0.270     20.080      0.269      14 2026/04/13 02:48:04
    43      0.280     20.079      0.278      14 2026/04/13 02:48:15
    53      0.270     20.079      0.270      14 2026/04/13 02:48:25
    64      0.280     20.084      0.280      14 2026/04/13 02:48:36
    76      0.270     20.080      0.274      14 2026/04/13 02:48:48
    87      0.280     20.074      0.278      14 2026/04/13 02:48:59
    97      0.270     20.074      0.274      14 2026/04/13 02:49:09
   108      0.280     20.082      0.277      14 2026/04/13 02:49:20
   119      0.270     20.081      0.272      14 2026/04/13 02:49:31
   130      0.280     20.074      0.281      14 2026/04/13 02:49:42
   140      0.280     20.071      0.275      14 2026/04/13 02:49:52
   151      0.270     20.062      0.269      14 2026/04/13 02:50:03
   162      0.280     20.068      0.278      14 2026/04/13 02:50:14
   173      0.270     20.067      0.267      14 2026/04/13 02:50:25
   183      0.270     20.069      0.271      14 2026/04/13 02:50:35
   194      0.270     20.077      0.272      14 2026/04/13 02:50:46
   205      0.270     20.076      0.272      14 2026/04/13 02:50:57
   216      0.280     20.075      0.281      14 2026/04/13 02:51:08
# Pump Time Series
#   Cycle npts
B02   104    3
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
     0.280   20   15.04     159    551    -9     2
     0.280    1   15.11     203    551    -9     7
     0.000   60   15.07     188    684    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   104  2.4    4  2000    10   600    31   180  2880      5     2       2   15  2.5841
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  104 2026/04/13 02:40:59
# number of packets previous packets
     14             119
# CTD Status pres offset
 0x64697361   0.000
# Pcase vacuums
   5.510   9.610
# Air Pump max current time  max pressure
      62       205     8.236
# Air Valve Positions
   1.140   0.000
# Air Valve Calibrations
   1.110   0.098   0.604
# Oil Valve Positions
   1.741   0.903
# Oil Valve Calibrations
   1.720   0.969
# Encountered Error Codes
#    Cycle Errors
EE20   104      4
#               Date   Code Description
 2026/04/13 01:24:29 0x0311 ValveTimeout
 2026/04/13 01:24:29 0x0613 DescendOilValveClose
 2026/04/13 02:40:55 0x0615 ParkDepthMin
 2026/04/13 02:40:56 0x0615 ParkDepthMin
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  104  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.240   1806   5.500
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  104       199    2026/04/13 02:58:29
Parameter Table Param_Table-V18.txt |
     IFV = 10.12.0.1	(IFV) |
     VER =       18	(version) |
     STY =        0	(Sensor 0=RBR,1=SBE) |
     MSN =      104	(cycle number) |
     MPS =        2	(phase) |
     MPC =       92	(count) |
  MNM[0] =        6	(count) |
  MNM[1] =        6	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       10	(dBar) |
  DTD[1] =       10	(dBar) |
  DTD[2] =       10	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        2	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        1	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =       20	(dBar) |
  CPN[2] =       10	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =     1000	(ms) |
  CPP[2] =     1000	(ms) |
  CPP[3] =     1000	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =        2	(dBar) |
     MTT =    30000	(ms) |
     MTS =    10800	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        0	(sec) |
    MAT2 =        0	(sec) |
     MAP =        0	(sec) |
     ACA =        0	(gain) |
     ACG =        0	(scale) |
     ACR =        0	(sampling rate) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    56641	(vac 0 inHg cal) |
     PV5 =    32815	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =    -1.65	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        1	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        1	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      103	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =        3	(dBar) |
  DSD[1] =        3	(dBar) |
  DSD[2] =        3	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     1800	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      300	(sec) |
     DDC =        0	(sec) |
     DSI =     2000	(ms) |
     DDD =        1	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =       20	(sec) |
  DFT[1] =       20	(sec) |
  DFT[2] =       20	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =     1800	(sec) |
     DCA =        2	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        2	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      900	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =        3	(dBar) |
     DDS =       60	(sec) |
     PNS =      103	(cycle) |
  PTO[0] =    10800	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   172800	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =    65552	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        1	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =     1800	(sec) |
  PFT[2] =    10800	(sec) |
  PFD[0] =      500	(dBar) |
  PFD[1] =       20	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        5	(sec) |
  PFV[2] =        5	(sec) |
     PDT =     1800	(sec) |
     PSM =      256	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    10800	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      103	(cycle number) |
     ARR =        5	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       20	(sec) |
     APN =       10	(sec) |
     APF =       20	(sec) |
     ASI =    10000	(ms) |
     ASD =        1	(dBar) |
     ADD =        1	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        2	(dBar) |
     SFN =      103	(cycle number) |
     SDC =    65526	(dBar) |
     SGT =       15	(min) |
     SET =       30	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      600	(sec) |
     SSP =       60	(sec) |
  PPC[0] =   0.0000	(1) |
  PPC[1] =   0.0000	(1) |
  PPC[2] =   0.0000	(sec) |
  PPC[3] =   0.0000	(1) |
     PPE =        0	(1) |

