# Message Decoder for Solo2, Alto and Deep Solo Floats; v260317
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11732   103  237 000000000000009  1515     0   4 2026/04/13, 00:27:27,  41.5683  -70.6851  10      0 13,6,14,14
SBD 11732   103  339 000000000000009  1516     0   3 2026/04/13, 00:27:38,  41.5762  -70.7303  12      0 13
SBD 11732   103  275 000000000000009  1517     0   2 2026/04/13, 00:27:54,  41.5840  -70.7757  14      0 8,15
SBD 11732   103  335 000000000000009  1518     0   1 2026/04/13, 00:52:54,  41.5334  -70.6398   2      0 8
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0       8         1138           48        1186
# Sensor_ID bytes     % packets Description
          6    37  3.25       1        Pump
          8   520 45.69       2     Surface
         13   415 36.47       2  Parameters
         14   100  8.79       2 Engineering
         15    66  5.80       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
# Surface Pressure Time Series
#    C npts n_var               Start
S  103    56     6 2026/04/13 00:14:07
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
     0      0.280     20.592      0.277      14 2026/04/13 00:14:07
    11      0.300     20.593      0.295      14 2026/04/13 00:14:18
    22      0.280     20.591      0.281      14 2026/04/13 00:14:29
    33      0.290     20.588      0.294      14 2026/04/13 00:14:40
    44      0.290     20.590      0.293      14 2026/04/13 00:14:51
    55      0.280     20.587      0.284      14 2026/04/13 00:15:02
    65      0.290     20.585      0.288      14 2026/04/13 00:15:12
    76      0.280     20.582      0.280      14 2026/04/13 00:15:23
    88      0.280     20.579      0.275      14 2026/04/13 00:15:35
    98      0.280     20.579      0.284      14 2026/04/13 00:15:45
   109      0.270     20.582      0.274      14 2026/04/13 00:15:56
   120      0.290     20.585      0.287      14 2026/04/13 00:16:07
   131      0.280     20.589      0.284      14 2026/04/13 00:16:18
   141      0.270     20.592      0.269      14 2026/04/13 00:16:28
   152      0.280     20.590      0.280      14 2026/04/13 00:16:39
   163      0.280     20.588      0.285      14 2026/04/13 00:16:50
   174      0.280     20.583      0.284      14 2026/04/13 00:17:01
   184      0.280     20.580      0.281      14 2026/04/13 00:17:11
   195      0.280     20.583      0.281      14 2026/04/13 00:17:22
   206      0.270     20.588      0.270      14 2026/04/13 00:17:33
   217      0.280     20.589      0.282      14 2026/04/13 00:17:44
   227      0.280     20.587      0.282      14 2026/04/13 00:17:54
   238      0.280     20.584      0.285      14 2026/04/13 00:18:05
   249      0.280     20.583      0.275      14 2026/04/13 00:18:16
   260      0.280     20.584      0.283      14 2026/04/13 00:18:27
   270      0.280     20.581      0.280      14 2026/04/13 00:18:37
   281      0.280     20.578      0.279      14 2026/04/13 00:18:48
   292      0.280     20.581      0.284      14 2026/04/13 00:18:59
   303      0.280     20.576      0.278      14 2026/04/13 00:19:10
   313      0.290     20.577      0.293      14 2026/04/13 00:19:20
   324      0.290     20.580      0.288      14 2026/04/13 00:19:31
   335      0.280     20.580      0.282      14 2026/04/13 00:19:42
   345      0.280     20.578      0.285      14 2026/04/13 00:19:52
   356      0.270     20.577      0.271      14 2026/04/13 00:20:03
   367      0.280     20.570      0.277      14 2026/04/13 00:20:14
     0      0.280     20.564      0.280      14 2026/04/13 00:20:25
    11      0.290     20.566      0.291      14 2026/04/13 00:20:36
    21      0.280     20.571      0.277      14 2026/04/13 00:20:46
    32      0.280     20.576      0.284      14 2026/04/13 00:20:57
    43      0.280     20.572      0.282      14 2026/04/13 00:21:08
    54      0.280     20.568      0.276      14 2026/04/13 00:21:19
    64      0.280     20.566      0.282      14 2026/04/13 00:21:29
    75      0.290     20.564      0.288      14 2026/04/13 00:21:40
    86      0.280     20.558      0.278      14 2026/04/13 00:21:51
    97      0.280     20.553      0.284      14 2026/04/13 00:22:02
   107      0.280     20.546      0.280      14 2026/04/13 00:22:12
   118      0.280     20.538      0.280      14 2026/04/13 00:22:23
   129      0.280     20.532      0.283      14 2026/04/13 00:22:34
   140      0.280     20.531      0.283      14 2026/04/13 00:22:45
   151      0.270     20.536      0.268      14 2026/04/13 00:22:56
   162      0.280     20.535      0.278      14 2026/04/13 00:23:07
   173      0.290     20.533      0.285      14 2026/04/13 00:23:18
   183      0.290     20.529      0.288      14 2026/04/13 00:23:28
   194      0.280     20.531      0.281      14 2026/04/13 00:23:39
   205      0.290     20.533      0.292      14 2026/04/13 00:23:50
   216      0.280     20.537      0.276      14 2026/04/13 00:24:01
# Pump Time Series
#   Cycle npts
B02   103    3
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
     0.280   22   15.04     172    549    -9     2
     0.280    1   15.12     185    550    -9     7
     0.000   60   15.06     202    681    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   103  2.4    4  2000    10   600    31   180  2880      5     2       2   15  2.5841
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  103 2026/04/13 00:13:52
# number of packets previous packets
     13             118
# CTD Status pres offset
 0x64697361   0.000
# Pcase vacuums
   5.500   9.450
# Air Pump max current time  max pressure
      65       190     8.037
# Air Valve Positions
   1.140   0.000
# Air Valve Calibrations
   1.110   0.098   0.604
# Oil Valve Positions
   1.742   0.901
# Oil Valve Calibrations
   1.720   0.969
# Encountered Error Codes
#    Cycle Errors
EE20   103      2
#               Date   Code Description
 2026/04/13 00:13:48 0x0615 ParkDepthMin
 2026/04/13 00:13:49 0x0615 ParkDepthMin
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  103  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.280   1802   5.480
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  103       199    2026/04/13 00:27:38
Parameter Table Param_Table-V18.txt |
     IFV = 10.12.0.1	(IFV) |
     VER =       18	(version) |
     STY =        0	(Sensor 0=RBR,1=SBE) |
     MSN =      103	(cycle number) |
     MPS =        2	(phase) |
     MPC =       91	(count) |
  MNM[0] =        6	(count) |
  MNM[1] =        6	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       10	(dBar) |
  DTD[1] =       10	(dBar) |
  DTD[2] =       10	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        2	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        1	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =       20	(dBar) |
  CPN[2] =       10	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =     1000	(ms) |
  CPP[2] =     1000	(ms) |
  CPP[3] =     1000	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =        2	(dBar) |
     MTT =    30000	(ms) |
     MTS =    10800	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        0	(sec) |
    MAT2 =        0	(sec) |
     MAP =        0	(sec) |
     ACA =        0	(gain) |
     ACG =        0	(scale) |
     ACR =        0	(sampling rate) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    56641	(vac 0 inHg cal) |
     PV5 =    32815	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =    -1.65	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        1	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        1	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      102	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =        3	(dBar) |
  DSD[1] =        3	(dBar) |
  DSD[2] =        3	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     1800	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      300	(sec) |
     DDC =        0	(sec) |
     DSI =     2000	(ms) |
     DDD =        1	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =       20	(sec) |
  DFT[1] =       20	(sec) |
  DFT[2] =       20	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =     1800	(sec) |
     DCA =        2	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        2	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      900	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =        3	(dBar) |
     DDS =       60	(sec) |
     PNS =      102	(cycle) |
  PTO[0] =    10800	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   172800	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =    65552	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        1	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =     1800	(sec) |
  PFT[2] =    10800	(sec) |
  PFD[0] =      500	(dBar) |
  PFD[1] =       20	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        5	(sec) |
  PFV[2] =        5	(sec) |
     PDT =     1800	(sec) |
     PSM =      256	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    10800	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      102	(cycle number) |
     ARR =        5	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       20	(sec) |
     APN =       10	(sec) |
     APF =       20	(sec) |
     ASI =    10000	(ms) |
     ASD =        1	(dBar) |
     ADD =        1	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        2	(dBar) |
     SFN =      102	(cycle number) |
     SDC =    65526	(dBar) |
     SGT =       15	(min) |
     SET =       30	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      600	(sec) |
     SSP =       60	(sec) |
  PPC[0] =   0.0000	(1) |
  PPC[1] =   0.0000	(1) |
  PPC[2] =   0.0000	(sec) |
  PPC[3] =   0.0000	(1) |
     PPE =        0	(1) |

