# Message Decoder for Solo2, Alto and Deep Solo Floats; v260317
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11732    90  237 000000000000009  1453     0   4 2026/04/11, 18:58:05,  41.5180  -68.7594 169      2 13,6,14,14
SBD 11732    90  339 000000000000009  1454     0   3 2026/04/11, 19:05:34,  41.5213  -71.1557 105      2 13
SBD 11732    90  275 000000000000009  1455     0   2 2026/04/11, 19:06:23,  41.5361  -71.0948  84      2 8,15
SBD 11732    90  335 000000000000009  1456     0   1 2026/04/11, 19:06:36,  41.5361  -71.0948  79      2 8
SBD 11732    90  279 000000000000009  1457     0   0 2026/04/11, 19:16:56,  41.5334  -70.6398   4      0 0,4,14,14,5,14,0
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      15         1405           60        1465
# Sensor_ID bytes     % packets Description
          0    48  3.42       2         GPS
          4   138  9.82       1        Fall
          5    18  1.28       1        Rise
          6    37  2.63       1        Pump
          8   520 37.01       2     Surface
         13   415 29.54       2  Parameters
         14   163 11.60       5 Engineering
         15    66  4.70       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   90 1 2026/04/11 17:01 +41.53446  -70.64747   30  6  0  0  0  2.5     0
G   90 2 2026/04/11 18:48 +41.53436  -70.64753   30  4  0  0  0  2.6     0
# Surface Pressure Time Series
#    C npts n_var               Start
S   90    56     6 2026/04/11 18:47:56
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
     0      0.190     22.770      0.185      14 2026/04/11 18:47:56
    11      0.190     22.772      0.191      14 2026/04/11 18:48:07
    22      0.190     22.770      0.189      14 2026/04/11 18:48:18
    34      0.190     22.772      0.193      14 2026/04/11 18:48:30
    44      0.190     22.774      0.187      14 2026/04/11 18:48:40
    55      0.180     22.774      0.184      14 2026/04/11 18:48:51
    66      0.190     22.777      0.195      14 2026/04/11 18:49:02
    77      0.200     22.779      0.197      14 2026/04/11 18:49:13
    87      0.190     22.780      0.191      14 2026/04/11 18:49:23
    98      0.190     22.779      0.188      14 2026/04/11 18:49:34
   109      0.190     22.781      0.189      14 2026/04/11 18:49:45
   120      0.190     22.783      0.187      14 2026/04/11 18:49:56
   131      0.180     22.785      0.178      14 2026/04/11 18:50:07
   142      0.190     22.788      0.190      14 2026/04/11 18:50:18
   152      0.180     22.791      0.184      14 2026/04/11 18:50:28
   164      0.190     22.791      0.186      14 2026/04/11 18:50:40
   175      0.190     22.792      0.187      14 2026/04/11 18:50:51
   185      0.200     22.791      0.195      14 2026/04/11 18:51:01
   196      0.190     22.792      0.191      14 2026/04/11 18:51:12
   207      0.190     22.794      0.190      14 2026/04/11 18:51:23
   217      0.170     22.794      0.175      14 2026/04/11 18:51:33
   228      0.190     22.794      0.188      14 2026/04/11 18:51:44
   240      0.190     22.795      0.191      14 2026/04/11 18:51:56
   250      0.180     22.793      0.184      14 2026/04/11 18:52:06
   261      0.180     22.794      0.183      14 2026/04/11 18:52:17
   272      0.190     22.796      0.191      14 2026/04/11 18:52:28
   283      0.190     22.797      0.189      14 2026/04/11 18:52:39
   293      0.190     22.800      0.188      14 2026/04/11 18:52:49
   304      0.190     22.799      0.191      14 2026/04/11 18:53:00
   315      0.190     22.797      0.186      14 2026/04/11 18:53:11
   326      0.200     22.799      0.198      14 2026/04/11 18:53:22
   336      0.190     22.801      0.194      14 2026/04/11 18:53:32
   347      0.190     22.804      0.193      14 2026/04/11 18:53:43
   358      0.190     22.802      0.195      14 2026/04/11 18:53:54
   369      0.190     22.800      0.187      14 2026/04/11 18:54:05
     0      0.190     22.798      0.193      14 2026/04/11 18:54:15
    11      0.180     22.798      0.182      14 2026/04/11 18:54:26
    22      0.190     22.798      0.193      14 2026/04/11 18:54:37
    33      0.190     22.800      0.188      14 2026/04/11 18:54:48
    43      0.190     22.804      0.195      14 2026/04/11 18:54:58
    54      0.190     22.808      0.187      14 2026/04/11 18:55:09
    65      0.180     22.810      0.182      14 2026/04/11 18:55:20
    76      0.190     22.814      0.186      14 2026/04/11 18:55:31
    86      0.180     22.816      0.179      14 2026/04/11 18:55:41
    97      0.190     22.816      0.186      14 2026/04/11 18:55:52
   108      0.180     22.817      0.185      14 2026/04/11 18:56:03
   119      0.200     22.819      0.197      14 2026/04/11 18:56:14
   129      0.180     22.819      0.182      14 2026/04/11 18:56:24
   140      0.190     22.818      0.185      14 2026/04/11 18:56:35
   151      0.190     22.819      0.190      14 2026/04/11 18:56:46
   162      0.180     22.820      0.181      14 2026/04/11 18:56:57
   172      0.190     22.820      0.189      14 2026/04/11 18:57:07
   183      0.180     22.820      0.184      14 2026/04/11 18:57:18
   194      0.190     22.821      0.191      14 2026/04/11 18:57:29
   205      0.180     22.819      0.178      14 2026/04/11 18:57:40
   215      0.180     22.818      0.179      14 2026/04/11 18:57:50
# Fall Time Series
#  Cycle npts     Start                Date
F00   90   26 829242065 2026/04/11 17:01:05
# time     Pres phase                Date
     0    0.160     1 2026/04/11 17:01:05
   302    0.200    10 2026/04/11 17:06:07
   604    0.160    10 2026/04/11 17:11:09
   904    0.200    10 2026/04/11 17:16:09
  1206    0.200    10 2026/04/11 17:21:11
  1508    0.200    10 2026/04/11 17:26:13
  1803    0.200     2 2026/04/11 17:31:08
  1809    0.200    10 2026/04/11 17:31:14
  2109    0.160    10 2026/04/11 17:36:14
  2409    0.200    10 2026/04/11 17:41:14
  2709    0.160    10 2026/04/11 17:46:14
  3009    0.200    10 2026/04/11 17:51:14
  3309    0.200    10 2026/04/11 17:56:14
  3609    0.200    10 2026/04/11 18:01:14
  3626    0.200     3 2026/04/11 18:01:31
  3909    0.200    10 2026/04/11 18:06:14
  4209    0.200    10 2026/04/11 18:11:14
  4509    0.200    10 2026/04/11 18:16:14
  4559    0.200     3 2026/04/11 18:17:04
  4809    0.200    10 2026/04/11 18:21:14
  5109    0.200    10 2026/04/11 18:26:14
  5409    0.200    10 2026/04/11 18:31:14
  5709    0.200    10 2026/04/11 18:36:14
  6009    0.200    10 2026/04/11 18:41:14
  6309    0.200    10 2026/04/11 18:46:14
  6391    0.200    10 2026/04/11 18:47:36
# Rise Time Series
#  Cycle npts     Start                Date
R00   90    2 829248459 2026/04/11 18:47:39
# time     Pres phase                Date
     0    0.200     7 2026/04/11 18:47:39
     2    0.200     8 2026/04/11 18:47:41
# Pump Time Series
#   Cycle npts
B02    90    3
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
     0.200   22   15.07     168    553    -9     2
     0.200    1   15.11     205    549    -9     7
     0.000   61   15.07     204    684    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    90  2.4    4  2000    10   600    31   180  2880      5     2       2   15  2.5841
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17   90 2026/04/11 17:01:04
# Pcase vacuums
   7.630   5.490
# Oil valve positions
   1.725   0.914   0.469
# Seek attempts and pump time
      2      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18   90 2026/04/11 18:47:37 9999/99/99 99:99:99
# Pcase vacuums
   5.490   5.490
# Average 
#   P      T      S
    0.000   0.000   0.000
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18   90 2026/04/11 18:47:39 9999/99/99 99:99:99
# Pcase vacuums
   5.490   5.490
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20   90 2026/04/11 18:47:41
# number of packets previous packets
     10             114
# CTD Status pres offset
 0x64697361   0.000
# Pcase vacuums
   5.500   9.680
# Air Pump max current time  max pressure
      61       210     8.123
# Air Valve Positions
   1.139   0.000
# Air Valve Calibrations
   1.110   0.098   0.604
# Oil Valve Positions
   1.742   0.902
# Oil Valve Calibrations
   1.720   0.969
# Encountered Error Codes
#    Cycle Errors
EE20    90      2
#               Date   Code Description
 2026/04/11 18:47:37 0x0615 ParkDepthMin
 2026/04/11 18:47:39 0x0615 ParkDepthMin
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00   90  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.160   1801   5.510
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   90       199    2026/04/11 19:05:34
Parameter Table Param_Table-V18.txt |
     IFV = 10.12.0.1	(IFV) |
     VER =       18	(version) |
     STY =        0	(Sensor 0=RBR,1=SBE) |
     MSN =       90	(cycle number) |
     MPS =        2	(phase) |
     MPC =       78	(count) |
  MNM[0] =        6	(count) |
  MNM[1] =        6	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       10	(dBar) |
  DTD[1] =       10	(dBar) |
  DTD[2] =       10	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        2	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        1	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =       20	(dBar) |
  CPN[2] =       10	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =     1000	(ms) |
  CPP[2] =     1000	(ms) |
  CPP[3] =     1000	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =        2	(dBar) |
     MTT =    30000	(ms) |
     MTS =    10800	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        0	(sec) |
    MAT2 =        0	(sec) |
     MAP =        0	(sec) |
     ACA =        0	(gain) |
     ACG =        0	(scale) |
     ACR =        0	(sampling rate) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    56641	(vac 0 inHg cal) |
     PV5 =    32815	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =    -1.65	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        1	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        1	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =       89	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =        3	(dBar) |
  DSD[1] =        3	(dBar) |
  DSD[2] =        3	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     1800	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      300	(sec) |
     DDC =        0	(sec) |
     DSI =     2000	(ms) |
     DDD =        1	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =       20	(sec) |
  DFT[1] =       20	(sec) |
  DFT[2] =       20	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =     1800	(sec) |
     DCA =        2	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        2	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      900	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =        3	(dBar) |
     DDS =       60	(sec) |
     PNS =       89	(cycle) |
  PTO[0] =    10800	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   172800	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =    65552	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        1	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =     1800	(sec) |
  PFT[2] =    10800	(sec) |
  PFD[0] =      500	(dBar) |
  PFD[1] =       20	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        5	(sec) |
  PFV[2] =        5	(sec) |
     PDT =     1800	(sec) |
     PSM =      256	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    10800	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =       89	(cycle number) |
     ARR =        5	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       20	(sec) |
     APN =       10	(sec) |
     APF =       20	(sec) |
     ASI =    10000	(ms) |
     ASD =        1	(dBar) |
     ADD =        1	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        2	(dBar) |
     SFN =       89	(cycle number) |
     SDC =    65526	(dBar) |
     SGT =       15	(min) |
     SET =       30	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      600	(sec) |
     SSP =       60	(sec) |
  PPC[0] =   0.0000	(1) |
  PPC[1] =   0.0000	(1) |
  PPC[2] =   0.0000	(sec) |
  PPC[3] =   0.0000	(1) |
     PPE =        0	(1) |

