# Message Decoder for Solo2, Alto and Deep Solo Floats; v260317
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11732    66  237 000000000000009  1337     0   4 2026/04/09, 08:48:18,  41.5603  -70.6398   7      0 13,6,14,14
SBD 11732    66  339 000000000000009  1338     0   3 2026/04/09, 08:57:24,  41.5750  -70.5786  16      0 13
SBD 11732    66  275 000000000000009  1339     0   2 2026/04/09, 08:58:04,  41.5750  -70.5786  23      0 8,15
SBD 11732    66  335 000000000000009  1340     0   1 2026/04/09, 08:58:33,  41.5881  -70.3655  32      0 8
SBD 11732    66  279 000000000000009  1341     0   0 2026/04/09, 08:58:50,  41.5881  -70.3655  37      0 0,4,14,14,5,14,0
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      15         1405           60        1465
# Sensor_ID bytes     % packets Description
          0    48  3.42       2         GPS
          4   138  9.82       1        Fall
          5    18  1.28       1        Rise
          6    37  2.63       1        Pump
          8   520 37.01       2     Surface
         13   415 29.54       2  Parameters
         14   163 11.60       5 Engineering
         15    66  4.70       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   66 1 2026/04/09 06:44 +41.53446  -70.64751   30  5  0  0  0  1.7     0
G   66 2 2026/04/09 08:31 +41.53462  -70.64731   30  3  0  0  0  3.3     0
# Surface Pressure Time Series
#    C npts n_var               Start
S   66    56     6 2026/04/09 08:31:28
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
     0      0.340     20.204      0.341      14 2026/04/09 08:31:28
    11      0.340     20.203      0.341      14 2026/04/09 08:31:39
    22      0.360     20.203      0.356      14 2026/04/09 08:31:50
    32      0.340     20.201      0.343      14 2026/04/09 08:32:00
    43      0.340     20.198      0.341      14 2026/04/09 08:32:11
    54      0.350     20.195      0.351      14 2026/04/09 08:32:22
    64      0.340     20.194      0.341      14 2026/04/09 08:32:32
    75      0.360     20.195      0.359      14 2026/04/09 08:32:43
    86      0.350     20.196      0.353      14 2026/04/09 08:32:54
    98      0.340     20.197      0.339      14 2026/04/09 08:33:06
   108      0.340     20.198      0.335      14 2026/04/09 08:33:16
   119      0.350     20.196      0.347      14 2026/04/09 08:33:27
   130      0.350     20.195      0.351      14 2026/04/09 08:33:38
   140      0.350     20.195      0.353      14 2026/04/09 08:33:48
   151      0.350     20.194      0.354      14 2026/04/09 08:33:59
   162      0.350     20.193      0.349      14 2026/04/09 08:34:10
   173      0.350     20.193      0.351      14 2026/04/09 08:34:21
   183      0.350     20.193      0.353      14 2026/04/09 08:34:31
   194      0.350     20.193      0.351      14 2026/04/09 08:34:42
   205      0.340     20.192      0.343      14 2026/04/09 08:34:53
   216      0.360     20.193      0.359      14 2026/04/09 08:35:04
   226      0.350     20.193      0.347      14 2026/04/09 08:35:14
   237      0.360     20.193      0.356      14 2026/04/09 08:35:25
   248      0.350     20.193      0.347      14 2026/04/09 08:35:36
   259      0.350     20.193      0.349      14 2026/04/09 08:35:47
   269      0.350     20.194      0.350      14 2026/04/09 08:35:57
   280      0.350     20.194      0.354      14 2026/04/09 08:36:08
   291      0.350     20.194      0.352      14 2026/04/09 08:36:19
   302      0.340     20.193      0.345      14 2026/04/09 08:36:30
   312      0.340     20.192      0.344      14 2026/04/09 08:36:40
   323      0.350     20.192      0.349      14 2026/04/09 08:36:51
   334      0.350     20.192      0.351      14 2026/04/09 08:37:02
   345      0.350     20.192      0.350      14 2026/04/09 08:37:13
   355      0.350     20.192      0.352      14 2026/04/09 08:37:23
   366      0.350     20.192      0.347      14 2026/04/09 08:37:34
     0      0.360     20.191      0.358      14 2026/04/09 08:37:45
    11      0.340     20.191      0.340      14 2026/04/09 08:37:56
    22      0.350     20.190      0.345      14 2026/04/09 08:38:07
    32      0.350     20.191      0.348      14 2026/04/09 08:38:17
    43      0.360     20.190      0.359      14 2026/04/09 08:38:28
    54      0.340     20.189      0.342      14 2026/04/09 08:38:39
    65      0.340     20.187      0.344      14 2026/04/09 08:38:50
    75      0.350     20.187      0.349      14 2026/04/09 08:39:00
    86      0.360     20.189      0.355      14 2026/04/09 08:39:11
    97      0.350     20.190      0.346      14 2026/04/09 08:39:22
   108      0.340     20.189      0.343      14 2026/04/09 08:39:33
   118      0.350     20.189      0.348      14 2026/04/09 08:39:43
   129      0.350     20.188      0.352      14 2026/04/09 08:39:54
   140      0.350     20.188      0.346      14 2026/04/09 08:40:05
   151      0.350     20.188      0.348      14 2026/04/09 08:40:16
   161      0.340     20.188      0.340      14 2026/04/09 08:40:26
   172      0.350     20.188      0.345      14 2026/04/09 08:40:37
   183      0.350     20.188      0.349      14 2026/04/09 08:40:48
   194      0.340     20.188      0.344      14 2026/04/09 08:40:59
   204      0.340     20.188      0.345      14 2026/04/09 08:41:09
   215      0.350     20.188      0.355      14 2026/04/09 08:41:20
# Fall Time Series
#  Cycle npts     Start                Date
F00   66   26 829032276 2026/04/09 06:44:36
# time     Pres phase                Date
     0    0.320     1 2026/04/09 06:44:36
   302    0.320    10 2026/04/09 06:49:38
   604    0.320    10 2026/04/09 06:54:40
   906    0.360    10 2026/04/09 06:59:42
  1206    0.360    10 2026/04/09 07:04:42
  1508    0.360    10 2026/04/09 07:09:44
  1803    0.320     2 2026/04/09 07:14:39
  1809    0.320    10 2026/04/09 07:14:45
  2109    0.360    10 2026/04/09 07:19:45
  2409    0.360    10 2026/04/09 07:24:45
  2709    0.360    10 2026/04/09 07:29:45
  3009    0.360    10 2026/04/09 07:34:45
  3309    0.360    10 2026/04/09 07:39:45
  3609    0.360    10 2026/04/09 07:44:45
  3626    0.360     3 2026/04/09 07:45:02
  3909    0.360    10 2026/04/09 07:49:45
  4209    0.360    10 2026/04/09 07:54:45
  4509    0.360    10 2026/04/09 07:59:45
  4559    0.360     3 2026/04/09 08:00:35
  4809    0.360    10 2026/04/09 08:04:45
  5109    0.360    10 2026/04/09 08:09:45
  5409    0.360    10 2026/04/09 08:14:45
  5709    0.360    10 2026/04/09 08:19:45
  6009    0.360    10 2026/04/09 08:24:45
  6309    0.360    10 2026/04/09 08:29:45
  6392    0.360    10 2026/04/09 08:31:08
# Rise Time Series
#  Cycle npts     Start                Date
R00   66    2 829038671 2026/04/09 08:31:11
# time     Pres phase                Date
     0    0.360     7 2026/04/09 08:31:11
     2    0.360     8 2026/04/09 08:31:13
# Pump Time Series
#   Cycle npts
B02    66    3
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
     0.320   22   15.08     178    557    -9     2
     0.360    1   15.13     220    557    -9     7
     0.000   60   15.08     216    689    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    66  2.4    4  2000    10   600    31   180  2880      5     2       2   15  2.5841
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17   66 2026/04/09 06:44:35
# Pcase vacuums
   7.680   5.570
# Oil valve positions
   1.722   0.912   0.473
# Seek attempts and pump time
      2      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18   66 2026/04/09 08:31:08 9999/99/99 99:99:99
# Pcase vacuums
   5.570   5.570
# Average 
#   P      T      S
    0.000   0.000   0.000
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18   66 2026/04/09 08:31:11 9999/99/99 99:99:99
# Pcase vacuums
   5.570   5.570
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20   66 2026/04/09 08:31:13
# number of packets previous packets
      5             147
# CTD Status pres offset
 0x64697361   0.000
# Pcase vacuums
   5.570   9.520
# Air Pump max current time  max pressure
      66       190     8.114
# Air Valve Positions
   1.142   0.000
# Air Valve Calibrations
   1.110   0.098   0.604
# Oil Valve Positions
   1.740   0.898
# Oil Valve Calibrations
   1.720   0.969
# Encountered Error Codes
#    Cycle Errors
EE20    66      2
#               Date   Code Description
 2026/04/09 08:31:09 0x0615 ParkDepthMin
 2026/04/09 08:31:10 0x0615 ParkDepthMin
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00   66  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.320   1802   5.560
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   66       199    2026/04/09 08:57:24
Parameter Table Param_Table-V18.txt |
     IFV = 10.12.0.1	(IFV) |
     VER =       18	(version) |
     STY =        0	(Sensor 0=RBR,1=SBE) |
     MSN =       66	(cycle number) |
     MPS =        2	(phase) |
     MPC =       54	(count) |
  MNM[0] =        6	(count) |
  MNM[1] =        6	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       10	(dBar) |
  DTD[1] =       10	(dBar) |
  DTD[2] =       10	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        2	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        1	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =       20	(dBar) |
  CPN[2] =       10	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =     1000	(ms) |
  CPP[2] =     1000	(ms) |
  CPP[3] =     1000	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =        2	(dBar) |
     MTT =    30000	(ms) |
     MTS =    10800	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        0	(sec) |
    MAT2 =        0	(sec) |
     MAP =        0	(sec) |
     ACA =        0	(gain) |
     ACG =        0	(scale) |
     ACR =        0	(sampling rate) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    56641	(vac 0 inHg cal) |
     PV5 =    32815	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =    -1.65	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        1	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        1	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =       65	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =        3	(dBar) |
  DSD[1] =        3	(dBar) |
  DSD[2] =        3	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     1800	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      300	(sec) |
     DDC =        0	(sec) |
     DSI =     2000	(ms) |
     DDD =        1	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =       20	(sec) |
  DFT[1] =       20	(sec) |
  DFT[2] =       20	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =     1800	(sec) |
     DCA =        2	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        2	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      900	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =        3	(dBar) |
     DDS =       60	(sec) |
     PNS =       65	(cycle) |
  PTO[0] =    10800	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   172800	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =    65552	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        1	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =     1800	(sec) |
  PFT[2] =    10800	(sec) |
  PFD[0] =      500	(dBar) |
  PFD[1] =       20	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        5	(sec) |
  PFV[2] =        5	(sec) |
     PDT =     1800	(sec) |
     PSM =      256	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    10800	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =       65	(cycle number) |
     ARR =        5	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       20	(sec) |
     APN =       10	(sec) |
     APF =       20	(sec) |
     ASI =    10000	(ms) |
     ASD =        1	(dBar) |
     ADD =        1	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        2	(dBar) |
     SFN =       65	(cycle number) |
     SDC =    65526	(dBar) |
     SGT =       15	(min) |
     SET =       30	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      600	(sec) |
     SSP =       60	(sec) |
  PPC[0] =   0.0000	(1) |
  PPC[1] =   0.0000	(1) |
  PPC[2] =   0.0000	(sec) |
  PPC[3] =   0.0000	(1) |
     PPE =        0	(1) |

