# Message Decoder for Solo2, Alto and Deep Solo Floats; v260304
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11732     2  237 000000000000009    42     0   4 2025/12/19, 21:56:26,  41.5524  -70.5945   5      0 13,6,14,14
SBD 11732     2  339 000000000000009    43     0   3 2025/12/19, 21:56:36,  41.5603  -70.6398   6      0 13
SBD 11732     2  275 000000000000009    44     0   2 2025/12/19, 21:57:10,  41.5683  -70.6851   8      0 8,15
SBD 11732     2  335 000000000000009    45     0   1 2025/12/19, 21:57:27,  41.5334  -70.6398   9      0 8
SBD 11732     2  279 000000000000009    46     0   0 2025/12/19, 21:59:31,  41.5108  -70.6557   6      0 0,4,14,14,5,14,0
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      15         1405           60        1465
# Sensor_ID bytes     % packets Description
          0    48  3.42       2         GPS
          4   138  9.82       1        Fall
          5    18  1.28       1        Rise
          6    37  2.63       1        Pump
          8   520 37.01       2     Surface
         13   415 29.54       2  Parameters
         14   163 11.60       5 Engineering
         15    66  4.70       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G    2 1 2025/12/19 19:59 +41.53425  -70.64687    0  8  0  0  0  0.9     0
G    2 2 2025/12/19 21:49 +41.53432  -70.64681  190 10  0  0  0  0.9     0
# Surface Pressure Time Series
#    C npts n_var               Start
S    2    56     6 2025/12/19 21:46:24
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
     0      0.620     16.173      0.620      14 2025/12/19 21:46:24
    11      0.630     16.174      0.634      14 2025/12/19 21:46:35
    21      0.630     16.173      0.632      14 2025/12/19 21:46:45
    33      0.620     16.173      0.617      14 2025/12/19 21:46:57
    44      0.620     16.173      0.624      14 2025/12/19 21:47:08
    55      0.630     16.173      0.627      14 2025/12/19 21:47:19
    65      0.620     16.173      0.616      14 2025/12/19 21:47:29
    76      0.620     16.173      0.618      14 2025/12/19 21:47:40
    87      0.620     16.173      0.619      14 2025/12/19 21:47:51
    98      0.620     16.173      0.616      14 2025/12/19 21:48:02
   108      0.630     16.173      0.633      14 2025/12/19 21:48:12
   119      0.630     16.173      0.631      14 2025/12/19 21:48:23
   130      0.620     16.173      0.622      14 2025/12/19 21:48:34
   141      0.570     16.175      0.569      14 2025/12/19 21:48:45
   151      0.240     16.220      0.240      14 2025/12/19 21:48:55
   162     -0.060     16.309     -0.061      14 2025/12/19 21:49:06
   173     -0.060     16.175     -0.055      14 2025/12/19 21:49:17
   184     -0.060     16.057     -0.064      14 2025/12/19 21:49:28
   194     -0.060     15.955     -0.063      14 2025/12/19 21:49:38
   205     -0.060     15.833     -0.057      14 2025/12/19 21:49:49
   216     -0.090     15.782     -0.090      14 2025/12/19 21:50:00
   227     -0.090     15.702     -0.085      14 2025/12/19 21:50:11
   237     -0.080     15.615     -0.082      14 2025/12/19 21:50:21
   248     -0.090     15.572     -0.088      14 2025/12/19 21:50:32
   259     -0.080     15.512     -0.084      14 2025/12/19 21:50:43
   269     -0.070     15.526     -0.071      14 2025/12/19 21:50:53
   280     -0.080     15.514     -0.083      14 2025/12/19 21:51:04
   291     -0.080     15.352     -0.080      14 2025/12/19 21:51:15
   302     -0.080     15.351     -0.078      14 2025/12/19 21:51:26
   312     -0.080     15.250     -0.085      14 2025/12/19 21:51:36
   323     -0.080     15.153     -0.079      14 2025/12/19 21:51:47
   334     -0.090     15.115     -0.090      14 2025/12/19 21:51:58
   345     -0.090     15.103     -0.091      14 2025/12/19 21:52:09
   355     -0.090     15.091     -0.086      14 2025/12/19 21:52:19
   366     -0.090     15.035     -0.086      14 2025/12/19 21:52:30
     0     -0.080     14.863     -0.079      14 2025/12/19 21:52:41
    11     -0.090     14.919     -0.089      14 2025/12/19 21:52:52
    22     -0.080     14.934     -0.076      14 2025/12/19 21:53:03
    32     -0.090     14.920     -0.095      14 2025/12/19 21:53:13
    43     -0.080     14.862     -0.082      14 2025/12/19 21:53:24
    54     -0.080     14.841     -0.082      14 2025/12/19 21:53:35
    65     -0.080     14.857     -0.082      14 2025/12/19 21:53:46
    75     -0.090     14.924     -0.094      14 2025/12/19 21:53:56
    86     -0.080     14.551     -0.084      14 2025/12/19 21:54:07
    97     -0.100     14.667     -0.097      14 2025/12/19 21:54:18
   108     -0.080     14.587     -0.082      14 2025/12/19 21:54:29
   119     -0.090     14.639     -0.092      14 2025/12/19 21:54:40
   129     -0.090     14.629     -0.086      14 2025/12/19 21:54:50
   140     -0.080     14.770     -0.080      14 2025/12/19 21:55:01
   151     -0.080     14.821     -0.080      14 2025/12/19 21:55:12
   162     -0.080     14.937     -0.082      14 2025/12/19 21:55:23
   172     -0.080     14.946     -0.079      14 2025/12/19 21:55:33
   183     -0.100     14.974     -0.096      14 2025/12/19 21:55:44
   194     -0.090     15.057     -0.088      14 2025/12/19 21:55:55
   204     -0.080     15.059     -0.079      14 2025/12/19 21:56:05
   215     -0.080     15.085     -0.084      14 2025/12/19 21:56:16
# Fall Time Series
#  Cycle npts     Start                Date
F00    2   26 819489573 2025/12/19 19:59:33
# time     Pres phase                Date
     0   -0.120     1 2025/12/19 19:59:33
   302    0.600    10 2025/12/19 20:04:35
   604    0.600    10 2025/12/19 20:09:37
   906    0.600    10 2025/12/19 20:14:39
  1206    0.600    10 2025/12/19 20:19:39
  1508    0.600    10 2025/12/19 20:24:41
  1803    0.600     2 2025/12/19 20:29:36
  1809    0.600    10 2025/12/19 20:29:42
  2109    0.600    10 2025/12/19 20:34:42
  2409    0.640    10 2025/12/19 20:39:42
  2709    0.600    10 2025/12/19 20:44:42
  3009    0.600    10 2025/12/19 20:49:42
  3309    0.600    10 2025/12/19 20:54:42
  3609    0.600    10 2025/12/19 20:59:42
  3626    0.600     3 2025/12/19 20:59:59
  3909    0.600    10 2025/12/19 21:04:42
  4209    0.640    10 2025/12/19 21:09:42
  4509    0.640    10 2025/12/19 21:14:42
  4559    0.640     3 2025/12/19 21:15:32
  4809    0.640    10 2025/12/19 21:19:42
  5109    0.600    10 2025/12/19 21:24:42
  5409    0.640    10 2025/12/19 21:29:42
  5709    0.640    10 2025/12/19 21:34:42
  6009    0.640    10 2025/12/19 21:39:42
  6309    0.640    10 2025/12/19 21:44:42
  6391    0.600    10 2025/12/19 21:46:04
# Rise Time Series
#  Cycle npts     Start                Date
R00    2    2 819495967 2025/12/19 21:46:07
# time     Pres phase                Date
     0    0.640     7 2025/12/19 21:46:07
     2    0.640     8 2025/12/19 21:46:09
# Pump Time Series
#   Cycle npts
B02     2    3
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
     0.600   22   15.13     190    585    -9     2
     0.640    1   15.19     211    584    -9     7
     0.000   60   15.14     199    695    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2     2  2.4    2   500    10   600    31     0   180      5     2       2   15  0.2831
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17    2 2025/12/19 19:59:32
# Pcase vacuums
   9.430   5.840
# Oil valve positions
   1.749   0.965   0.493
# Seek attempts and pump time
      2      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18    2 2025/12/19 21:46:04 9999/99/99 99:99:99
# Pcase vacuums
   5.840   5.840
# Average 
#   P      T      S
    0.000   0.000   0.000
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18    2 2025/12/19 21:46:07 9999/99/99 99:99:99
# Pcase vacuums
   5.840   5.840
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20    2 2025/12/19 21:46:09
# number of packets previous packets
      5              12
# CTD Status pres offset
 0x64697361   0.000
# Pcase vacuums
   5.840   9.170
# Air Pump max current time  max pressure
      74       155     8.466
# Air Valve Positions
   1.154   0.000
# Air Valve Calibrations
   1.110   0.098   0.604
# Oil Valve Positions
   1.761   0.949
# Oil Valve Calibrations
   1.720   0.969
# Encountered Error Codes
#    Cycle Errors
EE20     2      2
#               Date   Code Description
 2025/12/19 21:46:05 0x0615 ParkDepthMin
 2025/12/19 21:46:06 0x0615 ParkDepthMin
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00    2  1      0
# MC    Pres      Time  Pcase Vacuum
   1    -0.120   1801   5.830
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M    2       199    2025/12/19 21:56:36
Parameter Table Param_Table-V18.txt |
     IFV = 10.12.0.1	(IFV) |
     VER =       18	(version) |
     STY =        0	(Sensor 0=RBR,1=SBE) |
     MSN =        2	(cycle number) |
     MPS =        0	(phase) |
     MPC =        2	(count) |
  MNM[0] =        6	(count) |
  MNM[1] =        6	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       10	(dBar) |
  DTD[1] =       10	(dBar) |
  DTD[2] =       10	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        2	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        1	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =       20	(dBar) |
  CPN[2] =       10	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =     1000	(ms) |
  CPP[2] =     1000	(ms) |
  CPP[3] =     1000	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =        2	(dBar) |
     MTT =    30000	(ms) |
     MTS =    10800	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        0	(sec) |
    MAT2 =        0	(sec) |
     MAP =        0	(sec) |
     ACA =        0	(gain) |
     ACG =        0	(scale) |
     ACR =        0	(sampling rate) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    56641	(vac 0 inHg cal) |
     PV5 =    32815	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =    -1.65	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        1	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        1	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =        1	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =        3	(dBar) |
  DSD[1] =        3	(dBar) |
  DSD[2] =        3	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     1800	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      300	(sec) |
     DDC =        0	(sec) |
     DSI =     2000	(ms) |
     DDD =        1	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =       20	(sec) |
  DFT[1] =       20	(sec) |
  DFT[2] =       20	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =     1800	(sec) |
     DCA =        2	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        2	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      900	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =        3	(dBar) |
     DDS =       60	(sec) |
     PNS =        1	(cycle) |
  PTO[0] =    10800	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   172800	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =    65552	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        1	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =     1800	(sec) |
  PFT[2] =    10800	(sec) |
  PFD[0] =      500	(dBar) |
  PFD[1] =       20	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        5	(sec) |
  PFV[2] =        5	(sec) |
     PDT =     1800	(sec) |
     PSM =      256	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    10800	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =        1	(cycle number) |
     ARR =        5	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       20	(sec) |
     APN =       10	(sec) |
     APF =       20	(sec) |
     ASI =    10000	(ms) |
     ASD =        1	(dBar) |
     ADD =        1	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        2	(dBar) |
     SFN =        1	(cycle number) |
     SDC =    65526	(dBar) |
     SGT =       15	(min) |
     SET =       30	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      600	(sec) |
     SSP =       60	(sec) |
  PPC[0] =   0.0000	(1) |
  PPC[1] =   0.0000	(1) |
  PPC[2] =   0.0000	(sec) |
  PPC[3] =   0.0000	(1) |
     PPE =        0	(1) |

