# Message Decoder for Solo2, Alto and Deep Solo Floats; v260609
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11698     3  289 000000000726540    62     0   7 2026/06/13, 21:06:38,  35.5648  -70.1413   4      0 6,14,14
SBD 11698     3  339 000000000726540    63     0   6 2026/06/13, 21:06:51,  35.5774  -70.0848   4      0 13
SBD 11698     3  335 000000000726540    64     0   5 2026/06/13, 21:07:01,  35.5648  -70.1413   3      0 8
SBD 11698     3  335 000000000726540    65     0   4 2026/06/13, 21:07:14,  35.5648  -70.1413   3      0 8
SBD 11698     3  335 000000000726540    66     0   3 2026/06/13, 21:07:27,  35.5577  -70.0998   3      0 8
SBD 11698     3  320 000000000726540    67     0   2 2026/06/13, 21:07:47,  35.5774  -70.0848   3      0 5,14,0,8,15
SBD 11698     3  335 000000000726540    68     0   1 2026/06/13, 21:08:02,  35.5648  -70.1413   2      0 4
SBD 11698     3  338 000000000726540    69     0   0 2026/06/13, 21:08:16,  35.5648  -70.1413   2      0 0,4,14,14,9,10,11,7,13
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      22         2530           96        2626
# Sensor_ID bytes     % packets Description
          0    48  1.90       2         GPS
          4   341 13.48       2        Fall
          5   173  6.84       1        Rise
          6   169  6.68       1        Pump
          7    33  1.30       1 Drift_times
          8   995 39.33       4     Surface
          9    48  1.90       1  HR/Drift_P
         10    48  1.90       1  HR/Drift_T
         11    40  1.58       1  HR/Drift_S
         13   398 15.73       2  Parameters
         14   171  6.76       5 Engineering
         15    66  2.61       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G    3 1 2026/06/11 12:01 +35.87613  -69.53952  180  4  0  0  0  1.7     0
G    3 2 2026/06/13 21:01 +35.56975  -70.11494   80  3  0  0  0  2.4     0
# Drift Data
#      C Npts comp n_var
I0     3   17    1     4
#      Pres       Temp   Salinity     Time
    971.480     8.4940    35.1750        6
    972.800     8.4000    35.1700     7212
    975.720     8.4930    35.1750    14417
    970.520     8.4070    35.1700    21623
    971.760     8.3840    35.1680    28828
    972.360     8.4690    35.1730    36034
    970.800     8.3690    35.1690    43240
    967.840     8.2340    35.1510    50445
    970.000     8.2960    35.1640    57651
    972.800     8.2910    35.1640    64856
    975.880     8.5360    35.1780    72062
    972.560     8.3570    35.1660    79268
    973.040     8.2670    35.1620    86473
    972.400     8.3610    35.1690    93679
    974.600     8.3300    35.1660   100884
    970.880     8.3810    35.1690   108090
    969.160     8.1560    35.1590   115296
# Surface Pressure Time Series
#    C npts n_var               Start
S    3   106     6 2026/06/13 21:01:30
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
     0      0.080     22.953      0.000      14 2026/06/13 21:01:30
     4      0.140     22.987      0.000      14 2026/06/13 21:01:34
     7      0.040     22.935      0.000      14 2026/06/13 21:01:37
     9      0.220     22.966      0.000      14 2026/06/13 21:01:39
    12      0.150     22.945      0.000      14 2026/06/13 21:01:42
    15      0.140     22.908      0.000      14 2026/06/13 21:01:45
    18      0.160     22.955      0.000      14 2026/06/13 21:01:48
    20      0.160     22.980      0.000      14 2026/06/13 21:01:50
    23      0.130     22.998      0.000      14 2026/06/13 21:01:53
    26      0.050     22.987      0.000      14 2026/06/13 21:01:56
    29      0.130     23.011      0.000      14 2026/06/13 21:01:59
    32      0.070     23.053      0.000      14 2026/06/13 21:02:02
    35      0.150     23.082      0.000      14 2026/06/13 21:02:05
    38      0.080     23.039      0.000      14 2026/06/13 21:02:08
    41      0.070     23.051      0.000      14 2026/06/13 21:02:11
    43      0.190     23.075      0.000      14 2026/06/13 21:02:13
    46      0.040     23.099      0.000      14 2026/06/13 21:02:16
    49      0.110     23.119      0.000      14 2026/06/13 21:02:19
    52      0.090     23.144      0.000      14 2026/06/13 21:02:22
    54      0.120     23.167      0.000      14 2026/06/13 21:02:24
    57      0.090     23.188      0.000      14 2026/06/13 21:02:27
    60      0.060     23.207      0.000      14 2026/06/13 21:02:30
    63      0.240     23.203      0.000      14 2026/06/13 21:02:33
    65      0.030     23.136      0.000      14 2026/06/13 21:02:35
    68      0.110     23.151      0.000      14 2026/06/13 21:02:38
    71      0.040     23.171      0.000      14 2026/06/13 21:02:41
    74      0.150     23.128      0.000      14 2026/06/13 21:02:44
    76      0.080     23.073      0.000      14 2026/06/13 21:02:46
    79      0.040     23.031      0.000      14 2026/06/13 21:02:49
    82      0.150     23.034      0.000      14 2026/06/13 21:02:52
    85      0.110     23.048      0.000      14 2026/06/13 21:02:55
    87      0.260     23.042      0.000      14 2026/06/13 21:02:57
    90      0.120     22.875      0.000      14 2026/06/13 21:03:00
    93      0.090     22.920      0.000      14 2026/06/13 21:03:03
     0      0.050     22.973      0.000      14 2026/06/13 21:03:06
     3      0.120     23.022      0.000      14 2026/06/13 21:03:09
     6      0.100     23.070      0.000      14 2026/06/13 21:03:12
     9      0.100     23.104      0.000      14 2026/06/13 21:03:15
    12      0.220     23.122      0.000      14 2026/06/13 21:03:18
    15      0.040     23.019      0.000      14 2026/06/13 21:03:21
    17      0.040     23.030      0.000      14 2026/06/13 21:03:23
    20      0.040     23.066      0.000      14 2026/06/13 21:03:26
    23      0.290     23.065      0.000      14 2026/06/13 21:03:29
    26      0.040     22.919      0.000      14 2026/06/13 21:03:32
    28      0.120     22.959      0.000      14 2026/06/13 21:03:34
    31      0.030     22.999      0.000      14 2026/06/13 21:03:37
    34      0.300     23.021      0.000      14 2026/06/13 21:03:40
    37      0.050     22.918      0.000      14 2026/06/13 21:03:43
    39      0.110     22.965      0.000      14 2026/06/13 21:03:45
    42      0.040     23.003      0.000      14 2026/06/13 21:03:48
    45      0.060     23.050      0.000      14 2026/06/13 21:03:51
    48      0.040     23.089      0.000      14 2026/06/13 21:03:54
    50      0.110     23.122      0.000      14 2026/06/13 21:03:56
    53      0.060     23.157      0.000      14 2026/06/13 21:03:59
    56      0.280     23.157      0.000      14 2026/06/13 21:04:02
    59      0.070     23.128      0.000      14 2026/06/13 21:04:05
    61      0.050     23.138      0.000      14 2026/06/13 21:04:07
    64      0.090     23.156      0.000      14 2026/06/13 21:04:10
    67      0.140     23.179      0.000      14 2026/06/13 21:04:13
    70      0.070     23.203      0.000      14 2026/06/13 21:04:16
    72      0.040     23.226      0.000      14 2026/06/13 21:04:18
    75      0.120     23.223      0.000      14 2026/06/13 21:04:21
    78      0.080     23.237      0.000      14 2026/06/13 21:04:24
    81      0.080     23.250      0.000      14 2026/06/13 21:04:27
    84      0.160     23.273      0.000      14 2026/06/13 21:04:30
    87      0.090     23.300      0.000      14 2026/06/13 21:04:33
    90      0.120     23.316      0.000      14 2026/06/13 21:04:36
    92      0.050     23.342      0.000      14 2026/06/13 21:04:38
    95      0.140     23.363      0.000      14 2026/06/13 21:04:41
     0      0.040     23.388      0.000      14 2026/06/13 21:04:44
     3      0.060     23.409      0.000      14 2026/06/13 21:04:47
     6      0.150     23.310      0.000      14 2026/06/13 21:04:50
     8      0.040     23.271      0.000      14 2026/06/13 21:04:52
    11      0.060     23.276      0.000      14 2026/06/13 21:04:55
    14      0.050     23.295      0.000      14 2026/06/13 21:04:58
    17      0.230     23.300      0.000      14 2026/06/13 21:05:01
    19      0.170     23.269      0.000      14 2026/06/13 21:05:03
    22      0.040     23.283      0.000      14 2026/06/13 21:05:06
    25      0.130     23.296      0.000      14 2026/06/13 21:05:09
    28      0.050     23.235      0.000      14 2026/06/13 21:05:12
    30      0.050     23.235      0.000      14 2026/06/13 21:05:14
    33      0.150     23.236      0.000      14 2026/06/13 21:05:17
    36      0.040     23.131      0.000      14 2026/06/13 21:05:20
    39      0.140     23.114      0.000      14 2026/06/13 21:05:23
    41      0.050     23.149      0.000      14 2026/06/13 21:05:25
    44      0.140     23.181      0.000      14 2026/06/13 21:05:28
    47      0.060     23.151      0.000      14 2026/06/13 21:05:31
    50      0.050     23.154      0.000      14 2026/06/13 21:05:34
    52      0.230     23.071      0.000      14 2026/06/13 21:05:36
    55      0.130     23.090      0.000      14 2026/06/13 21:05:39
    58      0.050     23.127      0.000      14 2026/06/13 21:05:42
    61      0.160     23.170      0.000      14 2026/06/13 21:05:45
    63      0.040     23.207      0.000      14 2026/06/13 21:05:47
    66      0.120     23.245      0.000      14 2026/06/13 21:05:50
    69      0.110     23.283      0.000      14 2026/06/13 21:05:53
    72      0.120     23.326      0.000      14 2026/06/13 21:05:56
    75      0.050     23.349      0.000      14 2026/06/13 21:05:59
    77      0.040     23.375      0.000      14 2026/06/13 21:06:01
    80      0.220     23.267      0.000      14 2026/06/13 21:06:04
    83      0.050     23.258      0.000      14 2026/06/13 21:06:07
    86      0.050     23.305      0.000      14 2026/06/13 21:06:10
    88      0.070     23.356      0.000      14 2026/06/13 21:06:12
    91      0.220     23.381      0.000      14 2026/06/13 21:06:15
    94      0.070     23.399      0.000      14 2026/06/13 21:06:18
     0      0.060     23.433      0.000      14 2026/06/13 21:06:21
     3      0.050     23.470      0.000      14 2026/06/13 21:06:24
# Fall Time Series
#  Cycle npts     Start                Date
F00    3   65 834494487 2026/06/11 12:01:27
# time     Pres phase                Date
     0    0.160     1 2026/06/11 12:01:27
   379   30.720    10 2026/06/11 12:07:46
   495   51.160     2 2026/06/11 12:09:42
   548   60.600    10 2026/06/11 12:10:35
   772   90.200    10 2026/06/11 12:14:19
  1037  120.160    10 2026/06/11 12:18:44
  1304  150.040    10 2026/06/11 12:23:11
  1579  180.160    10 2026/06/11 12:27:46
  1860  210.120    10 2026/06/11 12:32:27
  2147  240.080    10 2026/06/11 12:37:14
  2449  270.200    10 2026/06/11 12:42:16
  2745  300.120    10 2026/06/11 12:47:12
  3031  330.040    10 2026/06/11 12:51:58
  3324  360.160    10 2026/06/11 12:56:51
  3628  390.160    10 2026/06/11 13:01:55
  3945  420.000    10 2026/06/11 13:07:12
  4307  450.160    10 2026/06/11 13:13:14
  4701  480.160    10 2026/06/11 13:19:48
  5079  510.040    10 2026/06/11 13:26:06
  5438  540.200    10 2026/06/11 13:32:05
  5783  570.000    10 2026/06/11 13:37:50
  6146  600.040    10 2026/06/11 13:43:53
  6571  630.080    10 2026/06/11 13:50:58
  7005  660.160    10 2026/06/11 13:58:12
  7418  690.120    10 2026/06/11 14:05:05
  7909  720.080    10 2026/06/11 14:13:16
  8498  750.040    10 2026/06/11 14:23:05
  9131  780.080    10 2026/06/11 14:33:38
  9831  810.080    10 2026/06/11 14:45:18
 10683  840.000    10 2026/06/11 14:59:30
 11622  870.040    10 2026/06/11 15:15:09
 13333  900.040    10 2026/06/11 15:43:40
 15133  910.440    10 2026/06/11 16:13:40
 16933  915.880    10 2026/06/11 16:43:40
 18733  916.400    10 2026/06/11 17:13:40
 20533  918.240    10 2026/06/11 17:43:40
 22282  915.560     3 2026/06/11 18:12:49
 23232  933.000    10 2026/06/11 18:28:39
 25032  952.080    10 2026/06/11 18:58:39
 26832  952.200    10 2026/06/11 19:28:39
 28632  950.560    10 2026/06/11 19:58:39
 30432  951.280    10 2026/06/11 20:28:39
 32233  956.680    10 2026/06/11 20:58:40
 34032  955.240     3 2026/06/11 21:28:39
 34035  955.280    10 2026/06/11 21:28:42
 35836  968.640    10 2026/06/11 21:58:43
 37637  971.760    10 2026/06/11 22:28:44
 39438  969.640    10 2026/06/11 22:58:45
 41239  968.600    10 2026/06/11 23:28:46
 43039  974.200    10 2026/06/11 23:58:46
 44840  971.080    10 2026/06/12 00:28:47
 45781  972.640     3 2026/06/12 00:44:28
 46642  972.440    10 2026/06/12 00:58:49
 48442  970.120    10 2026/06/12 01:28:49
 50244  971.920    10 2026/06/12 01:58:51
 52045  969.920    10 2026/06/12 02:28:52
 53845  970.560    10 2026/06/12 02:58:52
 55647  970.000    10 2026/06/12 03:28:54
 57447  972.280    10 2026/06/12 03:58:54
 59248  976.160    10 2026/06/12 04:28:55
 61048  970.000    10 2026/06/12 04:58:55
 62848  975.360    10 2026/06/12 05:28:55
 64648  974.640    10 2026/06/12 05:58:55
 66449  974.640    10 2026/06/12 06:28:56
 67418  971.480    10 2026/06/12 06:45:05
# Rise Time Series
#  Cycle npts     Start                Date
R00    3   33 834691514 2026/06/13 18:45:14
# time     Pres phase                Date
     0  969.600     7 2026/06/13 18:45:14
   360  938.840    13 2026/06/13 18:51:14
   626  906.520    13 2026/06/13 18:55:40
   872  875.760    13 2026/06/13 18:59:46
  1139  844.760    13 2026/06/13 19:04:13
  1405  814.680    13 2026/06/13 19:08:39
  1672  783.840    13 2026/06/13 19:13:06
  1918  753.680    13 2026/06/13 19:17:12
  2164  721.800    13 2026/06/13 19:21:18
  2389  691.480    13 2026/06/13 19:25:03
  2635  659.760    13 2026/06/13 19:29:09
  2902  629.120    13 2026/06/13 19:33:36
  3168  598.120    13 2026/06/13 19:38:02
  3414  568.920    13 2026/06/13 19:42:08
  3681  536.520    13 2026/06/13 19:46:35
  3927  505.360    13 2026/06/13 19:50:41
  4152  476.000    13 2026/06/13 19:54:26
  4398  443.440    13 2026/06/13 19:58:32
  4644  411.440    13 2026/06/13 20:02:38
  4890  380.440    13 2026/06/13 20:06:44
  5157  349.000    13 2026/06/13 20:11:11
  5423  316.560    13 2026/06/13 20:15:37
  5690  285.760    13 2026/06/13 20:20:04
  5936  254.680    13 2026/06/13 20:24:10
  6182  223.880    13 2026/06/13 20:28:16
  6428  193.640    13 2026/06/13 20:32:22
  6694  162.520    13 2026/06/13 20:36:48
  6961  132.120    13 2026/06/13 20:41:15
  7207  100.600    13 2026/06/13 20:45:21
  7453   69.040    13 2026/06/13 20:49:27
  7719   39.600    13 2026/06/13 20:53:53
  7986    9.000    13 2026/06/13 20:58:20
  8089    0.200     8 2026/06/13 21:00:03
# Pump Time Series
#   Cycle npts
B02     3   15
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    51.160  183   15.03     194    535    -9     2
   969.600   83   15.06     560    589    -9     7
   956.440   21   15.07     555    592    -9    13
   943.440   21   15.03     567    592    -9    13
   849.320   21   15.05     488    586    -9    13
   807.960   21   15.01     484    583    -9    13
   629.120   21   15.02     431    569    -9    13
   586.800   21   15.00     414    569    -9    13
   349.000   21   15.02     324    564    -9    13
   285.760   21   15.01     282    564    -9    13
   142.080   21   15.03     221    563    -9    13
    57.520   21   15.03     213    560    -9    13
    37.480   21   15.04     194    559    -9    13
     6.800   21   15.04     182    557    -9    13
     0.120   60   15.00     197    699    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2     3  2.4    3  1500  1000   600   422   180  2160     10     3       3  180  2.4667
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17    3 2026/06/11 12:01:25
# Pcase vacuums
   8.990   5.820
# Oil valve positions
   1.743   0.876   0.561
# Seek attempts and pump time
      3     -7
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18    3 2026/06/12 06:45:05 9999/99/99 99:99:99
# Pcase vacuums
   5.810   5.830
# Average 
#   P      T      S
  972.040   8.366  35.167
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18    3 2026/06/13 18:45:14 9999/99/99 99:99:99
# Pcase vacuums
   5.830   5.570
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20    3 2026/06/13 21:00:02
# number of packets previous packets
      8              10
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   5.570   8.630
# Air Pump max current time  max pressure
      79       130     7.868
# Air Valve Positions
   1.211   0.000
# Air Valve Calibrations
   1.155   0.134   0.645
# Oil Valve Positions
   1.756   0.866
# Oil Valve Calibrations
   1.700   0.911
# Encountered Error Codes
#    Cycle Errors
EE20     3      1
#               Date   Code Description
 2026/06/13 21:01:28 0x0216 SurfaceCtdProfile
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00    3  3      0
# MC    Pres      Time  Pcase Vacuum
   1     0.120    489   5.170
   3   915.520      5   5.720
   3   955.320      3   5.780
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M    3       194    2026/06/13 21:08:16
Parameter Table Param_Table-V18.txt |
     IFV = 10.12.0.4	(IFV) |
     VER =       18	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =        3	(cycle number) |
     MPS =        0	(phase) |
     MPC =        3	(count) |
  MNM[0] =        3	(count) |
  MNM[1] =        2	(count) |
  MNM[2] =     1000	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      500	(dBar) |
  DTD[1] =     1000	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2310	(dBar) |
  CPB[2] =     1010	(dBar) |
  CPB[3] =      310	(dBar) |
  CPN[1] =       20	(dBar) |
  CPN[2] =       20	(dBar) |
  CPN[3] =       10	(dBar) |
  CPP[1] =     1000	(ms) |
  CPP[2] =     1000	(ms) |
  CPP[3] =     1000	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   600000	(ms) |
     MTS = 15552000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        0	(sec) |
    MAT2 =        0	(sec) |
     MAP =        0	(sec) |
     ACA =        0	(gain) |
     ACG =        0	(scale) |
     ACR =        0	(sampling rate) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    57657	(vac 0 inHg cal) |
     PV5 =    33813	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =    -1.65	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        1	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        1	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =        2	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =      200	(dBar) |
  DSD[1] =       50	(dBar) |
  DSD[2] =       50	(dBar) |
  DST[0] =     3600	(sec) |
  DST[1] =     1800	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      360	(sec) |
     DDC =        0	(sec) |
     DSI =     2000	(ms) |
     DDD =       30	(dBar) |
     DDT =     1800	(sec) |
  DFT[0] =      257	(sec) |
  DFT[1] =      180	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     7200	(sec) |
  DCT[1] =    21600	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        9	(sec) |
     DCX =     1020	(sec) |
     DCP =        2	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =    10800	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =      100	(dBar) |
     DDS =     3600	(sec) |
     PNS =        2	(cycle) |
  PTO[0] =    21600	(sec) |
  PTO[1] =   129600	(sec) |
  PTO[2] =   734400	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  7200000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =    10800	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =     1000	(dBar) |
  PFD[1] =     1500	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =      420	(sec) |
  PFV[2] =      600	(sec) |
     PDT =     7200	(sec) |
     PSM =      256	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =        2	(cycle number) |
     ARR =       10	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =      100	(sec) |
     ASI =    20000	(ms) |
     ASD =        2	(dBar) |
     ADD =       30	(dBar) |
     ADT =     1800	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =        2	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       15	(min) |
     SET =       30	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =     1000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =       60	(sec) |

