# Message Decoder for Solo2, Alto and Deep Solo Floats; v241204
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11696    -3  276 000000000000009    12     0   2 2024/12/04, 20:38:30,  42.0814  -90.4839 216      0 14
SBD 11696    -3   83 000000000000009    13     0   1 2024/12/04, 20:38:40,  41.9911  -88.0683   6      0 13
SBD 11696    -3  339 000000000000009    14     0   0 2024/12/04, 20:38:50,  42.0816  -90.4355 212      0 13
# Delayed Transmission - (3/3) No surface time available, transmission was most likely delayed
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0       3          662           36         698
# Sensor_ID bytes     % packets Description
         13   398 60.12       2  Parameters
         14   264 39.88       1 Engineering
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    -3  2.0    1   150   150   180   360     0     0     11     1    2160    0  0.2908
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS           25      10      1000       5      1000       1 
# Engineering Data
#       Alto BIT Test
#    cycle software   BIT         Total
#          version    result      time
E011   -3  0x0a080004 0x00000001  0
 # Skyview           BIT_Start             BIT_End
         0 1970/01/01 00:00:00 2024/12/03 20:40:03
# BIT error 0x01: Air pump failure
# Air pump sensor pressure low
# CTD Status temp    salinity
 0x3c2a64c3   0.010  21.054
# Pressure offsets 
# before   now     after [dBar]
 979611822018554411089920.000   -0.220    0.480
# Pcase vacuum [inHg]
 809500672.000   5.390
# Battery Voltages [V]
   5.870  15.400
# Air Pump test
# Air Valve Positions [V]
  15.250   1.068   0.108
# Air Valve Calibrations [V]
   1.122   0.055   0.699
# duration count max P max currents      max voltage
   16150   146 -142283693359104.00    0.98    0.09    0.05
# Oil Pump test
# Oil Valve Positions [V]
  15.320   1.725
# Oil Valve Calibrations [V]
   1.770   0.820
# duration max currents [mAmp]      max voltage [V]
   16223      0.00    0.18         0.10
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   -3         0    2024/12/04 20:38:50
