# Message Decoder for Solo2, Alto and Deep Solo Floats; v241107
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11682    -4  272 000000000000009    20     0   2 2024/11/12, 13:41:54,  41.5334  -70.6398   3      0 14
SBD 11682    -4  100 000000000000009    21     0   1 2024/11/12, 13:42:01,  41.5334  -70.6398   3      0 13
SBD 11682    -4  339 000000000000009    22     0   0 2024/11/12, 13:42:40,  41.5334  -70.6398   3      0 13
# Delayed Transmission - (3/3) No surface time available, transmission was most likely delayed
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0       3          675           36         711
# Sensor_ID bytes     % packets Description
         13   415 61.48       2  Parameters
         14   260 38.52       1 Engineering
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    -4  2.0    1   150   150   180   360     0     0     11     1    2160    0  0.2908
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS           25      10      1000       5      1000       1 
# Engineering Data
#       Alto BIT Test
#    cycle software   BIT         Total
#          version    result      time
E010   -4  0x0a070101 0x00000000  0
 # Skyview           BIT_Start             BIT_End
         0 2024/05/23 15:41:38 2024/05/23 15:58:22
# Successful Bit Test
# CTD Status temp    salinity
 0x73746f70  27.639   0.000
# Pressure offsets 
# before   now     after [dBar]
    0.139    0.000    0.141
# Pcase vacuum [inHg]
   5.680   6.210
# Battery Voltages [V]
  15.430  15.270
# Air Pump test
# Air Valve Positions [V]
   1.175   0.066   0.757
# Air Valve Calibrations [V]
   1.118   0.122   0.620
# duration count max P max currents      max voltage
     360     1    7.74    0.09    0.09   15.40
# Oil Pump test
# Oil Valve Positions [V]
   1.889   0.848
# Oil Valve Calibrations [V]
   1.835   0.899
# duration max currents [mAmp]      max voltage [V]
     600      0.20    0.16        15.31
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   -4         0    2024/11/12 13:42:40
