# Message Decoder for Solo2, Alto and Deep Solo Floats; v241230
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11497    73  339 000000000000009  2961     0   5 2025/01/17, 00:27:40,  41.5603  -70.6398   3      0 13
SBD 11497    73  335 000000000000009  2963     0   3 2025/01/17, 00:28:22,  41.5683  -70.6851   3      0 8
SBD 11497    73  332 000000000000009  2964     0   2 2025/01/17, 00:28:33,  41.5683  -70.6851   4      0 5,14,0,1
SBD 11497    73  292 000000000000009  2962     0   4 2025/01/17, 00:27:57,  41.5683  -70.6851   3      0 8,15,13
SBD 11497    73  334 000000000000009  2966     0   0 2025/01/17, 00:34:23,  41.5334  -70.6398   6      0 0,4,14,14,9,3
SBD 11497    73  276 000000000000009  2960     0   6 2025/01/17, 00:27:21,  41.5683  -70.6851   3      0 6,14,14
SBD 11497    73  327 000000000000009  2965     0   1 2025/01/17, 00:29:56,  41.5603  -70.6398   8      0 10,11,7,2
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      22         2151           84        2235
# Sensor_ID bytes     % packets Description
          0    48  2.23       2         GPS
          1    29  1.35       1    Pressure
          2    36  1.67       1 Temperature
          3    29  1.35       1    Salinity
          4   153  7.11       1        Fall
          5   248 11.53       1        Rise
          6   169  7.86       1        Pump
          7    72  3.35       1 Drift_times
          8   466 21.66       2     Surface
          9    72  3.35       1  HR/Drift_P
         10   119  5.53       1  HR/Drift_T
         11    88  4.09       1  HR/Drift_S
         13   398 18.50       2  Parameters
         14   158  7.35       5 Engineering
         15    66  3.07       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   73 1 2025/01/14 19:40 +41.53427  -70.64686   30  5  0  0  0  1.9     0
G   73 2 2025/01/17 00:15 +41.53427  -70.64681   40  6  0  0  0  1.1     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0    73    9    1     3         2
#      Pres      Temp    Salinity
      1.120    17.8760    33.8220
      2.000    17.8750    33.8210
      3.000    17.8700    33.8210
      4.040    17.8210    33.8150
      5.000    17.7870    33.8140
      6.000    17.7440    33.8140
      7.000    17.6960    33.8140
      8.080    17.6080    33.8130
      9.280    17.3200    33.8110
# Drift Data
#      C Npts comp n_var
I0    73   96    1     4
#      Pres       Temp   Salinity     Time
      9.360    17.4710    33.7970        5
      9.360    17.4540    33.7960     1811
      9.360    17.4340    33.7960     3616
      9.360    17.4150    33.7960     5422
      9.360    17.3960    33.7950     7228
      9.360    17.3770    33.7960     9033
      9.360    17.3590    33.7950    10839
      9.360    17.3400    33.7950    12644
      9.360    17.3200    33.7950    14450
      9.360    17.2980    33.7950    16256
      9.360    17.2800    33.7950    18061
      9.360    17.2600    33.7940    19867
      9.360    17.2420    33.7930    21672
      9.360    17.2220    33.7930    23478
      9.360    17.2030    33.7930    25283
      9.360    17.1820    33.7930    27089
      9.360    17.1630    33.7930    28895
      9.360    17.1420    33.7930    30700
      9.360    17.1230    33.7920    32506
      9.360    17.1020    33.7920    34311
      9.360    17.0810    33.7920    36117
      9.360    17.5930    33.7970    37923
      9.360    17.5720    33.8070    39728
      9.360    17.5720    33.8050    41534
      9.360    17.5900    33.8050    43339
      9.400    17.6050    33.8060    45145
      9.360    17.6190    33.8050    46951
      9.360    17.6370    33.8060    48756
      9.400    17.6530    33.8060    50562
      9.400    17.6720    33.8060    52367
      9.360    17.6870    33.8060    54173
      9.400    17.7020    33.8080    55978
      9.400    17.7200    33.8070    57784
      9.400    17.7190    33.8080    59590
      9.400    17.7030    33.8050    61395
      9.400    17.6750    33.8070    63201
      9.400    17.6530    33.8060    65006
      9.400    17.6300    33.8070    66812
      9.400    17.6090    33.8060    68618
      9.400    17.5890    33.8060    70423
      9.400    17.5690    33.8050    72229
      9.400    17.5470    33.8060    74034
      9.400    17.5280    33.8050    75840
      9.400    17.5060    33.8050    77646
      9.400    17.4850    33.8050    79451
      9.400    17.4660    33.8050    81257
      9.400    17.4460    33.8050    83062
      9.400    17.4270    33.8030    84868
      9.400    17.4070    33.8040    86674
      9.400    17.3880    33.8030    88479
      9.400    17.3670    33.8030    90285
      9.440    17.3470    33.8040    92090
      9.440    17.3290    33.8020    93896
      9.440    17.3090    33.8020    95702
      9.440    17.2890    33.8020    97507
      9.400    17.2700    33.8020    99313
      9.440    17.2500    33.8020   101118
      9.440    17.2310    33.8010   102924
      9.440    17.2100    33.8010   104730
      9.400    17.1880    33.8020   106535
      9.440    17.1690    33.8010   108341
      9.400    17.1470    33.8010   110146
      9.400    17.1280    33.8010   111952
      9.400    17.1070    33.8010   113757
      9.440    17.0850    33.8000   115563
      9.440    17.0640    33.8000   117369
      9.400    17.0440    33.7990   119174
      9.440    17.0210    33.8000   120980
      9.440    17.0000    33.7990   122785
      9.400    17.6410    33.8100   124591
      9.440    17.5950    33.8140   126397
      9.440    17.5970    33.8130   128202
      9.400    17.6130    33.8130   130008
      9.440    17.6330    33.8140   131813
      9.440    17.6550    33.8120   133619
      9.440    17.6730    33.8140   135425
      9.440    17.6890    33.8140   137230
      9.400    17.7050    33.8140   139036
      9.440    17.7210    33.8150   140841
      9.440    17.7430    33.8160   142647
      9.400    17.7600    33.8140   144452
      9.440    17.7520    33.8160   146258
      9.400    17.7320    33.8150   148064
      9.440    17.7050    33.8150   149869
      9.400    17.6820    33.8140   151675
      9.400    17.6610    33.8130   153480
      9.400    17.6410    33.8130   155286
      9.400    17.6210    33.8120   157092
      9.400    17.5980    33.8130   158897
      9.400    17.5790    33.8130   160703
      9.360    17.5590    33.8120   162508
      9.360    17.5370    33.8130   164314
      9.360    17.5180    33.8110   166119
      9.360    17.4980    33.8110   167925
      9.360    17.4770    33.8110   169731
      9.360    17.4560    33.8110   171536
# Surface Pressure Time Series
#    C npts n_var               Start
S   73    50     6 2025/01/17 00:22:18
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
     0      0.180     17.860   -999.000      14 2025/01/17 00:15:48
    13      0.170     17.709   -999.000      14 2025/01/17 00:16:01
    24      0.160     17.614   -999.000      14 2025/01/17 00:16:12
    35      0.180     17.520   -999.000      14 2025/01/17 00:16:23
    46      0.160     17.375   -999.000      14 2025/01/17 00:16:34
    57      0.160     17.212   -999.000      14 2025/01/17 00:16:45
    69      0.160     17.054   -999.000      14 2025/01/17 00:16:57
    80      0.160     16.886   -999.000      14 2025/01/17 00:17:08
    91      0.150     16.744   -999.000      14 2025/01/17 00:17:19
   102      0.140     16.648   -999.000      14 2025/01/17 00:17:30
   113      0.150     16.550   -999.000      14 2025/01/17 00:17:41
   124      0.150     16.419   -999.000      14 2025/01/17 00:17:52
   135      0.140     16.300   -999.000      14 2025/01/17 00:18:03
   146      0.150     16.152   -999.000      14 2025/01/17 00:18:14
   157      0.140     16.056   -999.000      14 2025/01/17 00:18:25
   168      0.120     16.018   -999.000      14 2025/01/17 00:18:36
   179      0.140     15.893   -999.000      14 2025/01/17 00:18:47
   190      0.130     15.855   -999.000      14 2025/01/17 00:18:58
   202      0.140     15.847   -999.000      14 2025/01/17 00:19:10
   213      0.140     15.810   -999.000      14 2025/01/17 00:19:21
   224      0.130     15.735   -999.000      14 2025/01/17 00:19:32
   235      0.130     15.629   -999.000      14 2025/01/17 00:19:43
   246      0.130     15.531   -999.000      14 2025/01/17 00:19:54
   257      0.130     15.494   -999.000      14 2025/01/17 00:20:05
   268      0.140     15.424   -999.000      14 2025/01/17 00:20:16
   279      0.130     15.373   -999.000      14 2025/01/17 00:20:27
   290      0.130     15.336   -999.000      14 2025/01/17 00:20:38
   301      0.150     15.308   -999.000      14 2025/01/17 00:20:49
   312      0.140     15.250   -999.000      14 2025/01/17 00:21:00
   324      0.140     15.169   -999.000      14 2025/01/17 00:21:12
   335      0.140     15.052   -999.000      14 2025/01/17 00:21:23
   346      0.150     14.981   -999.000      14 2025/01/17 00:21:34
   357      0.130     14.977   -999.000      14 2025/01/17 00:21:45
   368      0.120     14.953   -999.000      14 2025/01/17 00:21:56
   379      0.140     14.937   -999.000      14 2025/01/17 00:22:07
     0      0.140     14.934   -999.000      14 2025/01/17 00:22:18
    11      0.140     14.953   -999.000      14 2025/01/17 00:22:29
    22      0.140     14.958   -999.000      14 2025/01/17 00:22:40
    33      0.120     14.942   -999.000      14 2025/01/17 00:22:51
    44      0.130     14.943   -999.000      14 2025/01/17 00:23:02
    56      0.150     14.936   -999.000      14 2025/01/17 00:23:14
    67      0.130     14.918   -999.000      14 2025/01/17 00:23:25
    78      0.140     14.897   -999.000      14 2025/01/17 00:23:36
    89      0.140     14.880   -999.000      14 2025/01/17 00:23:47
   100      0.140     14.871   -999.000      14 2025/01/17 00:23:58
   111      0.130     14.856   -999.000      14 2025/01/17 00:24:09
   122      0.140     14.846   -999.000      14 2025/01/17 00:24:20
   133      0.140     14.803   -999.000      14 2025/01/17 00:24:31
   144      0.140     14.755   -999.000      14 2025/01/17 00:24:42
   155      0.160     14.726   -999.000      14 2025/01/17 00:24:53
# Fall Time Series
#  Cycle npts     Start                Date
F00   73   29 790198858 2025/01/14 19:40:58
# time     Pres phase                Date
     0    0.120     1 2025/01/14 19:40:58
    74    1.120    10 2025/01/14 19:42:12
    86    2.440    10 2025/01/14 19:42:24
    95    3.400    10 2025/01/14 19:42:33
   100    4.120     2 2025/01/14 19:42:38
   103    4.400    10 2025/01/14 19:42:41
   111    5.480    10 2025/01/14 19:42:49
   115    6.000    10 2025/01/14 19:42:53
   126    7.320    10 2025/01/14 19:43:04
   133    8.160    10 2025/01/14 19:43:11
   143    9.200    10 2025/01/14 19:43:21
   443    9.360    10 2025/01/14 19:48:21
   743    9.360    10 2025/01/14 19:53:21
  1043    9.360    10 2025/01/14 19:58:21
  1343    9.360    10 2025/01/14 20:03:21
  1643    9.360    10 2025/01/14 20:08:21
  1924    9.360     3 2025/01/14 20:13:02
  1945    9.360    10 2025/01/14 20:13:23
  2245    9.360    10 2025/01/14 20:18:23
  2545    9.360    10 2025/01/14 20:23:23
  2845    9.320    10 2025/01/14 20:28:23
  2861    9.360     3 2025/01/14 20:28:39
  3147    9.360    10 2025/01/14 20:33:25
  3447    9.320    10 2025/01/14 20:38:25
  3748    9.360    10 2025/01/14 20:43:26
  4050    9.360    10 2025/01/14 20:48:28
  4350    9.320    10 2025/01/14 20:53:28
  4650    9.360    10 2025/01/14 20:58:28
  4699    9.360    10 2025/01/14 20:59:17
# Rise Time Series
#  Cycle npts     Start                Date
R00   73   48 790376364 2025/01/16 20:59:24
# time     Pres phase                Date
     0    9.360     6 2025/01/16 20:59:24
   324    9.360    12 2025/01/16 21:04:48
   636    9.360    12 2025/01/16 21:10:00
   948    9.360    12 2025/01/16 21:15:12
  1260    9.360    12 2025/01/16 21:20:24
  1572    9.360    12 2025/01/16 21:25:36
  1884    9.360    12 2025/01/16 21:30:48
  2196    9.360    12 2025/01/16 21:36:00
  2508    9.360    12 2025/01/16 21:41:12
  2820    9.360    12 2025/01/16 21:46:24
  3132    9.360    12 2025/01/16 21:51:36
  3445    9.360    12 2025/01/16 21:56:49
  3757    9.360    12 2025/01/16 22:02:01
  4069    9.360    12 2025/01/16 22:07:13
  4381    9.360    12 2025/01/16 22:12:25
  4693    9.360    12 2025/01/16 22:17:37
  5005    9.360    12 2025/01/16 22:22:49
  5317    9.360    12 2025/01/16 22:28:01
  5629    9.360    12 2025/01/16 22:33:13
  5941    9.360    12 2025/01/16 22:38:25
  6253    9.360    12 2025/01/16 22:43:37
  6566    9.360    12 2025/01/16 22:48:50
  6878    9.360    12 2025/01/16 22:54:02
  7190    9.360    12 2025/01/16 22:59:14
  7502    9.360    12 2025/01/16 23:04:26
  7814    9.360    12 2025/01/16 23:09:38
  8126    9.360    12 2025/01/16 23:14:50
  8438    9.360    12 2025/01/16 23:20:02
  8750    9.360    12 2025/01/16 23:25:14
  9062    9.360    12 2025/01/16 23:30:26
  9374    9.360    12 2025/01/16 23:35:38
  9687    9.360    12 2025/01/16 23:40:51
  9999    9.360    12 2025/01/16 23:46:03
 10311    9.360    12 2025/01/16 23:51:15
 10623    9.360    12 2025/01/16 23:56:27
 10838    9.360     7 2025/01/17 00:00:02
 11154    9.320    13 2025/01/17 00:05:18
 11459    9.160    13 2025/01/17 00:10:23
 11574    8.960    13 2025/01/17 00:12:18
 11627    7.760    13 2025/01/17 00:13:11
 11648    6.680    13 2025/01/17 00:13:32
 11658    5.920    13 2025/01/17 00:13:42
 11679    4.320    13 2025/01/17 00:14:03
 11690    3.480    13 2025/01/17 00:14:14
 11700    2.560    13 2025/01/17 00:14:24
 11711    1.760    13 2025/01/17 00:14:35
 11721    0.840    13 2025/01/17 00:14:45
 11722    0.840     8 2025/01/17 00:14:46
# Pump Time Series
#   Cycle npts
B02    73   15
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
     4.120   22   15.08     175    577    -9     2
     9.360   22   15.18     214    581    -9     7
     9.360   11   15.18     213    582    -9    13
     9.360   11   15.16     197    583    -9    13
     9.320   11   15.17     214    584    -9    13
     9.320   11   15.17     216    584    -9    13
     9.360   11   15.15     183    585    -9    13
     9.360   11   15.16     205    586    -9    13
     9.360   11   15.16     196    587    -9    13
     9.360   11   15.14     176    588    -9    13
     9.280   11   15.15     182    589    -9    13
     9.200   11   15.15     200    589    -9    13
     9.080   11   15.15     181    589    -9    13
     8.080   11   15.14     183    590    -9    13
     0.160  239   15.11     195    604    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    73  2.4    4  2000    10   600    31   180  2880      5     2       2   15  2.5841
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17   73 2025/01/14 19:40:56
# Pcase vacuums
   6.030   5.780
# Oil valve positions
   1.895   0.715   0.539
# Seek attempts and pump time
      2      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18   73 2025/01/14 20:59:18 9999/99/99 99:99:99
# Pcase vacuums
   5.780   5.790
# Average 
#   P      T      S
    9.400  17.452  33.804
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18   73 2025/01/17 00:00:02 2025/01/16 20:59:30
# Pcase vacuums
   5.800   5.870
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20   73 2025/01/17 00:14:43
# number of packets previous packets
      7              48
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   5.870   6.020
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.137   0.139   0.638
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.822   0.777
# Encountered Error Codes
#    Cycle Errors
EE20    73      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00   73  2      0
# MC    Pres      Time  Pcase Vacuum
   1     0.160     96   5.750
  12     9.320      6   5.790
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   73       195    2025/01/17 00:27:57
Parameter Table Param_Table-V18.txt |
     IFV = 10.8.0.4	(IFV) |
     VER =       18	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =       73	(cycle number) |
     MPS =        2	(phase) |
     MPC =       61	(count) |
  MNM[0] =        6	(count) |
  MNM[1] =        6	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       10	(dBar) |
  DTD[1] =       10	(dBar) |
  DTD[2] =       10	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        7	(dBar) |
     CRS =     1000	(HRD samples) |
     MTD =        2	(dBar) |
     MTT =    30000	(ms) |
     MTS =    21600	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        0	(sec) |
    MAT2 =        0	(sec) |
     MAP =        0	(sec) |
     ACA =        0	(gain) |
     ACG =        0	(scale) |
     ACR =        0	(sampling rate) |
     PNB =    43668	(V) |
     PNC =    50958	(inHg) |
     PXC =        0	(inHg) |
     PBL =    25960	(12V cal) |
     PBH =     1790	(14V cal) |
     PP0 =     2089	(oil pump 0 mA cal) |
     PP1 =    61614	(oil pump 1000 mA cal) |
     PVL =    32000	(V12 12V cal) |
     PVH =        0	(V12 14V cal) |
     PV0 =    16600	(vac 0 inHg cal) |
     PV5 =        0	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =     1280	(V12_sig 1000mA cal) |
     ICM =        5	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        1	(ice month mask) |
  ICA[0] =        1	(summer attempts) |
  ICA[1] =       70	(winter attempts) |
  ICB[0] =       70	(summer cycles) |
  ICB[1] =      180	(winter cycles) |
  IMT[0] =      180	(summer message threshold) |
  IMT[1] =       20	(winter message threshold) |
  ILT[0] =       40	(summer time threshold) |
  ILT[1] =    65371	(winter time threshold) |
     ICN =       30	(shallow limit for test dBar) |
     ICX =      514	(deep limit for test dBar) |
     ICT =     2.56	(deg C) |
     ICS =     5120	(sec) |
  ISS[0] =      180	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        0	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     2.56	(PSU) |
     ISO =    25600	(sec) |
     ISP =      100	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =       30	(summer descent depth dBar) |
  ICD[1] =        0	(winter descent depth dBar) |
  ICR[0] =  1966080	(summer minutes) |
  ICR[1] = 16779780	(winter minutes) |
  ICG[0] =      176	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =       30	(attempts) |
     ISD =     1200	(dBar) |
     IAT =       45	(secs) |
     ISV =       45	(secs) |
     ISW =        0	(secs) |
  ICO[0] =        0	(summer secs) |
  ICO[1] =        0	(winter secs) |
     ICF =       72	(summer mask) |
     ICW =  1310723	(winter ice mask) |
     DNS =        3	(cycle number) |
     DRA =        3	(target descent rate dBar/sec) |
  DSD[0] =     1800	(dBar) |
  DSD[1] =     1800	(dBar) |
  DSD[2] =     1800	(dBar) |
  DST[0] =      300	(sec) |
  DST[1] =        0	(sec) |
  DST[2] =        0	(sec) |
     DDF =     2000	(sec) |
     DDC =        1	(sec) |
     DSI = 19660820	(ms) |
     DDD =       20	(dBar) |
     DDT =       20	(sec) |
  DFT[0] =      257	(sec) |
  DFT[1] =     1800	(sec) |
  DFT[2] =     1800	(sec) |
     DCM =        7	(0=dis 1=en) |
     DAL =        8	(0=cal 1=seek) |
  DCT[0] =      512	(sec) |
  DCT[1] =      768	(sec) |
  DCT[2] =        3	(sec) |
     DCA =      252	(max attempts) |
     DRC =        2	(cal rate calculation) |
     DRT =        2	(sec) |
     DCX =      200	(sec) |
     DCP =        0	(sec) |
     DCF =      900	(sec) |
     DCD =       50	(dBar) |
     DCR =       10	(dBar/sec) |
     DSW =       10	(sec) |
     DUC =        5	(under damping factor) |
     DPT =       30	(sec) |
     DUT =        5	(sec) |
     DPD =      900	(sec) |
     DVT =       19	(sec) |
     DVD =       16	(dBar) |
     DVW =        3	(sec) |
     PFR =       60	(cc/min) |
     PFC =       72	(.001 cc/dBar) |
     DDP =        0	(dBar) |
     DDS =    10800	(sec) |
     PNS =        0	(cycle) |
  PTO[0] =        2	(sec) |
  PTO[1] = -1560281088	(sec) |
  PTO[2] =        1	(sec) |
  PTH[0] =  1048576	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =      256	(24 LSB) |
     PTD = 460800000	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI = 16777479	(ms) |
  PFM[0] =        8	(0=dis 1=en) |
  PFM[1] =       42	(0=dis 1=en) |
  PFM[2] =       48	(0=dis 1=en) |
  PFT[0] =      500	(sec) |
  PFT[1] =       20	(sec) |
  PFT[2] =     2000	(sec) |
  PFD[0] =        1	(dBar) |
  PFD[1] =        5	(dBar) |
  PFD[2] =        5	(dBar) |
  PFV[0] =     1800	(sec) |
  PFV[1] =      256	(sec) |
  PFV[2] =      200	(sec) |
     PDT =    10800	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =     7682	(dBar) |
     PST =    22535	(sec) |
     PSV =       72	(sec) |
     PSW =        5	(sec) |
     PSS =        9	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =    24576	(cycle number) |
     ARR =     5120	(target rise rate cm/sec) |
     APM =     2560	(sec) |
     APS =    12800	(sec) |
     APN =       39	(sec) |
     APF =     4096	(sec) |
     ASI = 16777473	(ms) |
     ASD =    11268	(dBar) |
     ADD =    19459	(dBar) |
     ADT =    59392	(sec) |
     ARC =      512	(?) |
     AHR =    18432	(cm/sec) |
     AHD =        0	(dBar) |
     SFN =     3840	(cycle number) |
     SDC =     3840	(dBar) |
     SGT =      300	(min) |
     SET =        0	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =       39	(sec) |
     SSA =     4096	(0=dis 1=en) |
     SSI = 167779660	(ms) |
     SDS =     6500	(sec) |
     SAM =        4	(0=dis 1=en) |
     SAX =    14338	(.001 psi) |
     SAN =    22528	(.001 psi) |
     SAO =    61440	(sec) |

