# Message Decoder for Solo2, Alto and Deep Solo Floats; v240529
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11342   982  290 000000000881890  6253     0   5 2024/08/01, 10:06:38,  66.7586  -53.1846   3      0 13,6,14,14
SBD 11342   982  340 000000000881890  6254     0   4 2024/08/01, 10:06:51,  66.7388  -53.1119   3      0 13
SBD 11342   982  288 000000000881890  6255     0   3 2024/08/01, 10:07:05,  66.7586  -53.1846   4      0 3,0,8
SBD 11342   982  316 000000000881890  6256     0   2 2024/08/01, 10:07:20,  66.7586  -53.1846   4      0 2,15
SBD 11342   982  293 000000000881890  6257     0   1 2024/08/01, 10:07:38,  66.7586  -53.1846   4      0 7,5,14,1
SBD 11342   982  325 000000000881890  6258     0   0 2024/08/01, 10:07:53,  66.7586  -53.1846   5      0 0,4,14,14,9,10,11
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1780           72        1852
# Sensor_ID bytes     % packets Description
          0    48  2.70       2         GPS
          1   123  6.91       1    Pressure
          2   238 13.37       1 Temperature
          3   190 10.67       1    Salinity
          4    98  5.51       1        Fall
          5   103  5.79       1        Rise
          6   114  6.40       1        Pump
          7    36  2.02       1 Drift_times
          8    62  3.48       1     Surface
          9    57  3.20       1  HR/Drift_P
         10    46  2.58       1  HR/Drift_T
         11    44  2.47       1  HR/Drift_S
         13   404 22.70       2  Parameters
         14   151  8.48       5 Engineering
         15    66  3.71       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  982 1 2024/07/31 20:18 +66.74243  -53.10507    0  8 36 41 46  1.1     0
G  982 2 2024/08/01 10:06 +66.74395  -53.10589  140  5 33 39 44  1.2     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   982  188    1     3         0
#      Pres      Temp    Salinity
      1.000     6.6330    32.6420
      1.960     6.6320    32.6420
      3.000     6.6320    32.6420
      4.000     6.6270    32.6410
      5.040     6.6140    32.6400
      6.040     6.6100    32.6400
      7.040     6.6020    32.6390
      8.000     6.5820    32.6360
      9.040     6.5450    32.6320
     10.040     6.4450    32.6400
     10.960     6.1730    32.6550
     12.000     5.8750    32.6730
     13.000     5.7170    32.7020
     14.000     5.5270    32.7070
     15.000     5.4920    32.7000
     15.960     5.2270    32.7440
     17.000     5.0770    32.7290
     18.000     5.0040    32.7290
     19.000     4.9450    32.7280
     20.040     4.9030    32.7240
     21.040     4.8760    32.7230
     22.040     4.8520    32.7260
     23.040     4.6990    32.7290
     24.000     4.6310    32.7360
     25.000     4.4870    32.7230
     26.000     4.4690    32.7190
     27.000     4.3610    32.7320
     28.000     4.3040    32.7240
     29.000     4.3080    32.7180
     30.040     4.3320    32.7430
     31.000     4.0580    32.7630
     31.960     3.7480    32.7130
     33.000     3.7180    32.7210
     34.040     3.6860    32.7190
     35.000     3.6780    32.7160
     36.000     3.7580    32.7230
     37.000     3.7060    32.7340
     38.000     3.5090    32.7350
     38.960     3.4050    32.7210
     39.960     3.3730    32.7260
     40.960     3.3550    32.7310
     42.000     3.3590    32.7310
     43.000     3.3450    32.7400
     43.960     3.3120    32.7480
     45.000     3.2810    32.6730
     46.040     3.2730    32.7600
     47.000     3.2640    32.7670
     48.000     3.1520    32.7890
     49.000     3.2120    32.8070
     50.000     3.1680    32.8500
     51.000     3.1220    32.8520
     52.000     3.0500    32.8750
     53.000     3.0020    32.8900
     54.000     2.9300    32.9010
     55.000     2.7820    32.8960
     56.000     2.7680    32.9100
     57.000     2.7300    32.9360
     58.040     2.6210    32.9640
     59.000     2.5770    32.9710
     59.960     2.5720    32.9820
     60.960     2.5630    32.9840
     62.000     2.5370    32.9890
     63.040     2.4930    33.0030
     64.040     2.4400    33.0100
     65.000     2.4350    33.0110
     65.960     2.4230    33.0170
     67.000     2.3970    33.0300
     68.000     2.3830    33.0310
     69.040     2.3750    33.0290
     70.000     2.3180    33.0450
     71.000     2.2330    33.0530
     72.000     2.1060    33.0700
     72.960     2.0610    33.0630
     73.960     2.0180    33.0720
     75.000     1.9140    33.0800
     76.040     1.8100    33.0840
     77.000     1.7870    33.0830
     78.000     1.7680    33.0820
     79.040     1.7620    33.0800
     79.960     1.7160    33.0890
     80.960     1.6600    33.0850
     81.960     1.6470    33.0800
     83.000     1.5990    33.0860
     84.000     1.5050    33.0880
     85.040     1.4560    33.0910
     86.040     1.3780    33.0890
     87.040     1.2530    33.1110
     88.000     1.0410    33.1160
     89.040     0.9910    33.1130
     90.040     0.9420    33.1150
     91.040     0.8970    33.1160
     92.000     0.8660    33.1190
     93.000     0.8240    33.1210
     94.000     0.7590    33.1180
     94.960     0.7500    33.1200
     96.000     0.7220    33.1220
     97.000     0.6640    33.1250
     98.000     0.6330    33.1230
     99.040     0.6220    33.1220
    100.040     0.6090    33.1220
    101.000     0.5580    33.1330
    102.000     0.4870    33.1280
    102.960     0.4740    33.1270
    103.960     0.4410    33.1340
    105.000     0.3800    33.1330
    106.040     0.3580    33.1300
    107.000     0.3550    33.1330
    108.000     0.3010    33.1440
    109.040     0.2610    33.1330
    110.040     0.2640    33.1340
    111.040     0.2200    33.1450
    112.000     0.1650    33.1380
    113.000     0.1410    33.1430
    114.040     0.1280    33.1400
    115.040     0.1170    33.1410
    116.040     0.1100    33.1410
    117.040     0.1020    33.1410
    118.000     0.0700    33.1450
    119.040     0.0210    33.1480
    120.080    -0.0040    33.1480
    121.040    -0.0110    33.1470
    122.000    -0.0120    33.1480
    123.000    -0.0170    33.1490
    124.000    -0.0200    33.1480
    125.040    -0.0210    33.1480
    126.000    -0.0220    33.1480
    127.000    -0.0190    33.1480
    128.040    -0.0190    33.1480
    129.040    -0.0180    33.1470
    130.000    -0.0200    33.1470
    131.000    -0.0220    33.1470
    132.040    -0.0230    33.1460
    133.040    -0.0360    33.1500
    134.000    -0.1000    33.1500
    134.960    -0.1430    33.1590
    136.000    -0.1630    33.1530
    137.040    -0.1650    33.1540
    138.040    -0.1730    33.1550
    139.040    -0.1740    33.1530
    140.000    -0.1820    33.1550
    140.960    -0.1930    33.1550
    142.000    -0.1970    33.1550
    143.000    -0.1980    33.1560
    143.960    -0.2000    33.1550
    145.000    -0.2070    33.1560
    146.000    -0.2110    33.1560
    147.040    -0.2120    33.1560
    148.000    -0.2120    33.1560
    148.960    -0.2130    33.1560
    150.000    -0.2130    33.1560
    151.000    -0.2130    33.1560
    151.960    -0.2120    33.1560
    152.960    -0.2150    33.1560
    153.960    -0.2140    33.1570
    155.040    -0.2110    33.1550
    156.040    -0.2050    33.1560
    157.040    -0.2110    33.1560
    158.040    -0.2090    33.1560
    159.040    -0.2090    33.1570
    160.040    -0.2100    33.1560
    161.000    -0.2170    33.1570
    162.000    -0.2100    33.1550
    163.000    -0.2100    33.1560
    164.040    -0.2100    33.1560
    165.000    -0.2110    33.1560
    166.040    -0.2120    33.1560
    167.040    -0.2100    33.1560
    168.000    -0.2110    33.1560
    169.000    -0.2110    33.1560
    169.960    -0.2110    33.1560
    171.000    -0.2120    33.1560
    172.000    -0.2120    33.1560
    173.000    -0.2120    33.1560
    174.000    -0.2110    33.1560
    175.000    -0.2110    33.1560
    176.000    -0.2120    33.1560
    177.000    -0.2120    33.1570
    178.040    -0.2110    33.1550
    179.000    -0.2110    33.1560
    180.000    -0.2100    33.1560
    181.000    -0.2110    33.1570
    182.000    -0.2120    33.1560
    183.000    -0.2120    33.1560
    184.000    -0.2110    33.1560
    185.000    -0.2120    33.1560
    186.000    -0.2140    33.1560
    187.040    -0.2150    33.1560
    187.760    -0.2120    33.1550
# Drift Data
#      C Npts comp n_var
I0   982   23    1     4
#      Pres       Temp   Salinity     Time
    155.440    -0.1950    33.1580        6
    171.560    -0.2410    33.1610     1811
    185.000    -0.3080    33.1610     3617
    184.880    -0.3050    33.1590     5423
    188.960    -0.3070    33.1570     7228
    188.600    -0.3020    33.1570     9034
    189.800    -0.2970    33.1560    10840
    188.760    -0.2830    33.1570    12645
    186.720    -0.2830    33.1570    14451
    189.200    -0.2940    33.1570    16256
    194.000    -0.2830    33.1570    18062
    196.680    -0.2780    33.1560    19868
    194.400    -0.2330    33.1530    21673
    191.400    -0.1970    33.1580    23479
    185.760    -0.1790    33.1550    25285
    185.840    -0.1940    33.1590    27090
    184.440    -0.2400    33.1570    28896
    190.000    -0.2850    33.1570    30702
    188.440    -0.2780    33.1540    32507
    188.360    -0.2820    33.1560    34313
    187.360    -0.2800    33.1570    36118
    188.080    -0.2780    33.1570    37924
    191.920    -0.2820    33.1590    39730
# Surface Pressure Time Series
#    C npts n_var               Start
S  982     6     6 2024/08/01 10:03:40
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    74      0.150      6.641   -999.000      14 2024/08/01 10:04:54
    87      0.190      6.639   -999.000      14 2024/08/01 10:05:07
    98      0.210      6.628   -999.000      14 2024/08/01 10:05:18
   109      0.210      6.635   -999.000      14 2024/08/01 10:05:29
   120      0.210      6.632   -999.000      14 2024/08/01 10:05:40
   132      0.200      6.634   -999.000      14 2024/08/01 10:05:52
# Fall Time Series
#  Cycle npts     Start                Date
F00  982   18 775772295 2024/07/31 20:18:15
# time     Pres phase                Date
     0    0.200     1 2024/07/31 20:18:15
   132   10.200    10 2024/07/31 20:20:27
   197   20.280    10 2024/07/31 20:21:32
   258   30.240    10 2024/07/31 20:22:33
   323   40.440    10 2024/07/31 20:23:38
   363   46.840     2 2024/07/31 20:24:18
   385   50.400    10 2024/07/31 20:24:40
   447   60.040    10 2024/07/31 20:25:42
   521   70.000    10 2024/07/31 20:26:56
   609   80.080    10 2024/07/31 20:28:24
   745   90.080    10 2024/07/31 20:30:40
  1260  100.000    10 2024/07/31 20:39:15
  1646  110.040    10 2024/07/31 20:45:41
  2077  120.040    10 2024/07/31 20:52:52
  2654  130.040    10 2024/07/31 21:02:29
  3290  140.040    10 2024/07/31 21:13:05
  4010  150.000    10 2024/07/31 21:25:05
  4354  155.400    10 2024/07/31 21:30:49
# Rise Time Series
#  Cycle npts     Start                Date
R00  982   19 775819858 2024/08/01 09:30:58
# time     Pres phase                Date
     0  187.800     7 2024/08/01 09:30:58
   196  177.560    13 2024/08/01 09:34:14
   319  166.680    13 2024/08/01 09:36:17
   421  156.840    13 2024/08/01 09:37:59
   524  146.880    13 2024/08/01 09:39:42
   626  136.560    13 2024/08/01 09:41:24
   729  125.680    13 2024/08/01 09:43:07
   831  114.720    13 2024/08/01 09:44:49
   934  104.120    13 2024/08/01 09:46:32
  1036   93.360    13 2024/08/01 09:48:14
  1139   82.840    13 2024/08/01 09:49:57
  1241   71.920    13 2024/08/01 09:51:39
  1344   61.520    13 2024/08/01 09:53:22
  1446   50.680    13 2024/08/01 09:55:04
  1549   40.160    13 2024/08/01 09:56:47
  1651   28.840    13 2024/08/01 09:58:29
  1754   18.320    13 2024/08/01 10:00:12
  1856    8.080    13 2024/08/01 10:01:54
  1960    0.240     8 2024/08/01 10:03:38
# Pump Time Series
#   Cycle npts
B02   982   10
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    46.840  386   15.10     220    661    -9     2
   187.800   73   15.24     289    674    -9     7
   179.080   21   15.25     302    676    -9    13
   168.560   21   15.24     300    678    -9    13
   156.840   21   15.23     279    679    -9    13
   138.680   21   15.22     302    681    -9    13
    61.520   21   15.22     248    674    -9    13
    44.520   21   15.21     246    672    -9    13
     9.960   21   15.21     219    664    -9    13
     0.200   60   15.19     198    663    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   982  2.4   29    90   120   600    81    10   720      9     1    4020   15  0.6317
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  982 2024/07/31 20:18:12
# Pcase vacuums
   6.610   6.680
# Oil valve positions
   1.822   0.795   0.631
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  982 2024/07/31 21:30:49 9999/99/99 99:99:99
# Pcase vacuums
   6.670   6.700
# Average 
#   P      T      S
  186.760  -0.265  33.157
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  982 2024/08/01 09:30:58 9999/99/99 99:99:99
# Pcase vacuums
   6.690   6.600
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  982 2024/08/01 10:03:38
# number of packets previous packets
      6               6
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.610   6.640
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.176   0.158   0.667
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.740   0.876
# Encountered Error Codes
#    Cycle Errors
EE20   982      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  982  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.200    359   6.200
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  982       197    2024/08/01 10:06:51
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      982	(cycle number) |
     MPS =        0	(phase) |
     MPC =      982	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      120	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    42104	(vac 0 inHg cal) |
     PV5 =    18698	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =     4095	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        0	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        0	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       99	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      999	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       10	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        1	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =      600	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      981	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       46	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       10	(dBar) |
     DDT =     1800	(sec) |
  DFT[0] =      385	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        0	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65531	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      900	(sec) |
     DUC =       67	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      981	(cycle) |
  PTO[0] =    43200	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  1800000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =     1800	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      100	(dBar) |
     PST =     7200	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      981	(cycle number) |
     ARR =        9	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       60	(sec) |
     APN =       20	(sec) |
     APF =      100	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =       10	(dBar) |
     ADT =     1800	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      975	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =       60	(sec) |
     SSP =       60	(sec) |

