# Message Decoder for Solo2, Alto and Deep Solo Floats; v240529
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11342   931  225 000000000881890  5938     0   6 2024/07/02, 17:20:05,  66.7652  -53.0574   4      0 6,14,14
SBD 11342   931  340 000000000881890  5939     0   5 2024/07/02, 17:20:23,  66.7388  -53.1119   4      0 13
SBD 11342   931  290 000000000881890  5940     0   4 2024/07/02, 17:20:34,  66.7586  -53.1846  15      0 3,8
SBD 11342   931  298 000000000881890  5941     0   3 2024/07/02, 17:20:46,  66.7454  -52.9849  14      0 2
SBD 11342   931  314 000000000881890  5942     0   2 2024/07/02, 17:20:59,  66.7247  -53.1119  12      0 1,15,13
SBD 11342   931  270 000000000881890  5943     0   1 2024/07/02, 17:21:13,  66.7388  -53.1119   4      0 11,7,5,14,0
SBD 11342   931  319 000000000881890  5944     0   0 2024/07/02, 17:21:26,  66.7717  -52.9301  10      0 0,4,14,14,9,10
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1972           84        2056
# Sensor_ID bytes     % packets Description
          0    48  2.43       2         GPS
          1   160  8.11       1    Pressure
          2   286 14.50       1 Temperature
          3   216 10.95       1    Salinity
          4   143  7.25       1        Fall
          5   143  7.25       1        Rise
          6   125  6.34       1        Pump
          7    36  1.83       1 Drift_times
          8    62  3.14       1     Surface
          9    52  2.64       1  HR/Drift_P
         10    44  2.23       1  HR/Drift_T
         11    36  1.83       1  HR/Drift_S
         13   404 20.49       2  Parameters
         14   151  7.66       5 Engineering
         15    66  3.35       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  931 1 2024/07/02 03:18 +66.74425  -53.06095    0  4 40 43 47  4.6     0
G  931 2 2024/07/02 17:18 +66.74490  -53.05883   60  6 35 39 43  2.0     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   931  271    1     3         0
#      Pres      Temp    Salinity
      1.040     6.8350    32.7800
      2.040     6.8310    32.7800
      3.040     6.8210    32.7790
      4.000     6.8150    32.7770
      5.000     6.8040    32.7750
      6.000     6.7880    32.7670
      7.000     6.7800    32.7590
      7.960     6.3660    32.8190
      8.960     5.5080    32.9310
     10.040     5.1120    32.9680
     11.000     4.7790    32.9710
     12.000     4.4790    33.0100
     13.040     4.1380    33.0140
     14.000     4.0140    33.0090
     15.000     3.9470    33.0100
     16.000     3.7920    33.0180
     17.000     3.3080    33.0760
     17.960     3.1430    33.0470
     18.960     3.0780    33.0520
     20.040     2.7170    33.0800
     21.000     2.6090    33.0710
     21.960     2.3450    33.0980
     23.000     2.2590    33.0870
     23.960     2.0880    33.1150
     24.960     1.9430    33.1140
     25.960     1.8990    33.1100
     27.000     1.8870    33.1100
     28.040     1.8790    33.1100
     29.000     1.8240    33.1150
     30.040     1.7750    33.1130
     31.000     1.7580    33.1110
     32.000     1.7500    33.1120
     33.000     1.7200    33.1110
     34.000     1.6770    33.1120
     35.000     1.6330    33.1160
     36.000     1.5780    33.1180
     37.000     1.5030    33.1070
     38.000     1.4960    33.1060
     39.000     1.5400    33.1010
     40.000     1.4010    33.1250
     41.000     1.3750    33.1120
     41.960     1.3740    33.1120
     43.000     1.3640    33.1120
     44.000     1.3590    33.1120
     45.000     1.3540    33.1120
     46.000     1.3360    33.1160
     46.960     1.3230    33.1160
     48.000     1.3070    33.1200
     49.000     1.2890    33.1220
     50.000     1.2080    33.1440
     51.000     1.1600    33.1370
     52.040     1.1620    33.1370
     53.040     1.1630    33.1360
     54.040     1.1690    33.1340
     55.000     1.1700    33.1320
     56.000     1.0950    33.1490
     57.000     1.0250    33.1500
     57.960     1.0090    33.1490
     58.960     0.9910    33.1490
     59.960     0.9600    33.1530
     60.960     0.9000    33.1570
     62.000     0.8260    33.1570
     63.000     0.8090    33.1560
     64.000     0.8020    33.1550
     65.040     0.7970    33.1540
     66.040     0.7910    33.1540
     67.000     0.7710    33.1520
     68.000     0.7360    33.1540
     68.960     0.7120    33.1520
     70.000     0.6710    33.1580
     71.040     0.5370    33.1740
     72.000     0.4840    33.1650
     73.000     0.4750    33.1640
     74.000     0.4520    33.1650
     74.960     0.4250    33.1630
     76.000     0.3890    33.1640
     77.000     0.3100    33.1660
     78.000     0.2220    33.1670
     78.960     0.2120    33.1620
     79.960     0.2140    33.1620
     81.000     0.2100    33.1620
     82.040     0.1900    33.1620
     83.000     0.1840    33.1590
     84.000     0.1660    33.1610
     85.000     0.0710    33.1620
     86.000    -0.0190    33.1560
     87.000    -0.0570    33.1560
     88.000    -0.1720    33.1630
     88.960    -0.2000    33.1540
     89.960    -0.1970    33.1550
     91.000    -0.1980    33.1560
     92.040    -0.1940    33.1530
     93.000    -0.1420    33.1550
     94.000    -0.0680    33.1560
     95.000    -0.0490    33.1650
     96.000    -0.0710    33.1680
     97.000    -0.0970    33.1660
     97.960    -0.1250    33.1650
     99.000    -0.1400    33.1650
    100.000    -0.1640    33.1660
    101.000    -0.2080    33.1640
    102.040    -0.2180    33.1640
    103.040    -0.2510    33.1630
    104.000    -0.2560    33.1610
    104.960    -0.3150    33.1690
    106.000    -0.3530    33.1630
    107.040    -0.3560    33.1630
    108.000    -0.3680    33.1640
    109.000    -0.3660    33.1640
    110.000    -0.3720    33.1650
    110.960    -0.3750    33.1650
    112.000    -0.3780    33.1640
    113.000    -0.3750    33.1650
    114.000    -0.3790    33.1640
    115.000    -0.3790    33.1650
    115.960    -0.3780    33.1650
    116.960    -0.3770    33.1630
    118.000    -0.3850    33.1650
    119.000    -0.3900    33.1630
    120.000    -0.4100    33.1660
    121.000    -0.4280    33.1630
    122.000    -0.4440    33.1640
    123.000    -0.4500    33.1630
    124.000    -0.4900    33.1660
    125.000    -0.5020    33.1650
    126.000    -0.5060    33.1650
    127.000    -0.5100    33.1640
    128.000    -0.5290    33.1650
    129.040    -0.5290    33.1640
    130.000    -0.5390    33.1650
    130.960    -0.5490    33.1640
    132.000    -0.5550    33.1650
    133.040    -0.5620    33.1660
    134.000    -0.5630    33.1650
    135.000    -0.5620    33.1660
    136.040    -0.5590    33.1660
    137.040    -0.5570    33.1660
    138.040    -0.5620    33.1670
    139.040    -0.5770    33.1680
    139.960    -0.5970    33.1690
    141.000    -0.5660    33.1660
    142.000    -0.5520    33.1690
    143.000    -0.5510    33.1700
    144.040    -0.5520    33.1700
    145.000    -0.5520    33.1700
    145.960    -0.5550    33.1700
    147.000    -0.5620    33.1700
    148.000    -0.5640    33.1690
    148.960    -0.5730    33.1710
    149.960    -0.5800    33.1700
    150.960    -0.5820    33.1700
    152.000    -0.5830    33.1700
    153.000    -0.5860    33.1700
    154.000    -0.5890    33.1700
    155.040    -0.5860    33.1700
    156.000    -0.5850    33.1700
    156.960    -0.5840    33.1700
    158.000    -0.5890    33.1710
    159.000    -0.6090    33.1710
    160.000    -0.6140    33.1700
    161.040    -0.6160    33.1700
    162.040    -0.6150    33.1700
    163.000    -0.6290    33.1690
    163.960    -0.6340    33.1680
    164.960    -0.6500    33.1680
    166.000    -0.6650    33.1690
    167.000    -0.6710    33.1680
    168.040    -0.6730    33.1680
    169.000    -0.6740    33.1680
    170.000    -0.6740    33.1680
    171.000    -0.6670    33.1680
    172.040    -0.6650    33.1690
    173.000    -0.6640    33.1680
    174.000    -0.6600    33.1680
    175.000    -0.6600    33.1690
    175.960    -0.6590    33.1690
    177.000    -0.6660    33.1700
    178.000    -0.6700    33.1690
    179.000    -0.6680    33.1680
    180.040    -0.6760    33.1690
    181.040    -0.6820    33.1690
    182.000    -0.6820    33.1690
    183.000    -0.6880    33.1690
    184.040    -0.6880    33.1690
    185.000    -0.6850    33.1690
    185.960    -0.6860    33.1700
    187.040    -0.6870    33.1700
    188.000    -0.6870    33.1700
    189.000    -0.6840    33.1700
    190.040    -0.6830    33.1700
    191.000    -0.6850    33.1710
    192.000    -0.6870    33.1710
    193.000    -0.6830    33.1710
    194.000    -0.6810    33.1720
    195.000    -0.6810    33.1720
    195.960    -0.6790    33.1720
    197.000    -0.6790    33.1720
    198.080    -0.6770    33.1720
    199.000    -0.6760    33.1720
    200.000    -0.6750    33.1730
    201.000    -0.6740    33.1730
    201.960    -0.6740    33.1730
    203.000    -0.6730    33.1730
    204.000    -0.6720    33.1740
    205.000    -0.6730    33.1740
    206.040    -0.6710    33.1740
    207.040    -0.6710    33.1740
    208.000    -0.6730    33.1750
    208.960    -0.6710    33.1750
    210.000    -0.6720    33.1750
    211.040    -0.6710    33.1750
    212.000    -0.6700    33.1750
    212.960    -0.6690    33.1750
    213.960    -0.6690    33.1750
    214.960    -0.6700    33.1760
    216.000    -0.6680    33.1760
    217.000    -0.6690    33.1760
    218.000    -0.6690    33.1760
    219.000    -0.6660    33.1760
    220.000    -0.6640    33.1770
    221.000    -0.6630    33.1770
    221.960    -0.6630    33.1780
    223.000    -0.6630    33.1770
    224.040    -0.6610    33.1780
    225.040    -0.6600    33.1780
    226.000    -0.6590    33.1790
    226.960    -0.6610    33.1780
    228.000    -0.6570    33.1780
    229.040    -0.6520    33.1790
    230.040    -0.6470    33.1800
    231.040    -0.6460    33.1800
    232.040    -0.6450    33.1800
    233.040    -0.6450    33.1800
    234.040    -0.6460    33.1800
    235.000    -0.6470    33.1800
    236.000    -0.6470    33.1800
    237.000    -0.6480    33.1800
    238.000    -0.6480    33.1800
    239.000    -0.6480    33.1800
    240.000    -0.6470    33.1800
    241.000    -0.6440    33.1810
    242.000    -0.6400    33.1810
    243.000    -0.6390    33.1810
    243.960    -0.6370    33.1820
    244.960    -0.6350    33.1820
    246.000    -0.6340    33.1820
    247.000    -0.6330    33.1820
    248.040    -0.6330    33.1820
    249.000    -0.6320    33.1820
    249.960    -0.6310    33.1830
    250.960    -0.6310    33.1830
    252.000    -0.6300    33.1830
    253.000    -0.6270    33.1830
    253.960    -0.6240    33.1840
    255.000    -0.6210    33.1840
    256.000    -0.6180    33.1850
    257.000    -0.6140    33.1860
    257.960    -0.6140    33.1850
    259.000    -0.5990    33.1860
    260.000    -0.6020    33.1870
    261.000    -0.6090    33.1870
    262.000    -0.5980    33.1870
    263.000    -0.5950    33.1880
    264.000    -0.5940    33.1880
    265.000    -0.5930    33.1880
    266.000    -0.5920    33.1880
    267.000    -0.5910    33.1880
    267.960    -0.5880    33.1880
    269.000    -0.5860    33.1890
    270.040    -0.5820    33.1890
    270.840    -0.5800    33.1890
# Drift Data
#      C Npts comp n_var
I0   931   23    1     4
#      Pres       Temp   Salinity     Time
    240.800    -0.6330    33.1810        6
    271.360    -0.5730    33.1890     1812
    271.560    -0.5730    33.1900     3617
    271.720    -0.5730    33.1900     5423
    271.840    -0.5760    33.1900     7228
    271.960    -0.5750    33.1890     9034
    272.000    -0.5760    33.1900    10840
    271.920    -0.5770    33.1900    12645
    271.840    -0.5780    33.1890    14451
    271.720    -0.5780    33.1890    16257
    271.560    -0.5740    33.1900    18062
    271.320    -0.5820    33.1890    19868
    271.040    -0.5860    33.1890    21674
    270.760    -0.5870    33.1880    23479
    270.480    -0.5760    33.1900    25285
    270.240    -0.5780    33.1890    27091
    270.040    -0.5760    33.1890    28896
    269.920    -0.5790    33.1900    30702
    269.840    -0.5760    33.1890    32507
    269.880    -0.5760    33.1900    34313
    269.960    -0.5760    33.1900    36119
    270.120    -0.5770    33.1890    37924
    270.360    -0.5780    33.1890    39730
# Surface Pressure Time Series
#    C npts n_var               Start
S  931     6     6 2024/07/02 17:17:27
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    80      0.110      6.831   -999.000      14 2024/07/02 17:18:47
    93      0.110      6.834   -999.000      14 2024/07/02 17:19:00
   104      0.060      6.821   -999.000      14 2024/07/02 17:19:11
   115      0.080      6.802   -999.000      14 2024/07/02 17:19:22
   126      0.070      6.827   -999.000      14 2024/07/02 17:19:33
   137      0.050      6.796   -999.000      14 2024/07/02 17:19:44
# Fall Time Series
#  Cycle npts     Start                Date
F00  931   27 773205526 2024/07/02 03:18:46
# time     Pres phase                Date
     0    0.040     1 2024/07/02 03:18:46
   132   10.040    10 2024/07/02 03:20:58
   197   20.160    10 2024/07/02 03:22:03
   261   30.520    10 2024/07/02 03:23:07
   323   40.120    10 2024/07/02 03:24:09
   366   46.920     2 2024/07/02 03:24:52
   388   50.240    10 2024/07/02 03:25:14
   456   60.480    10 2024/07/02 03:26:22
   528   70.360    10 2024/07/02 03:27:34
   612   80.040    10 2024/07/02 03:28:58
   751   90.120    10 2024/07/02 03:31:17
   974  100.080    10 2024/07/02 03:35:00
  1166  110.120    10 2024/07/02 03:38:12
  1344  120.040    10 2024/07/02 03:41:10
  1526  130.160    10 2024/07/02 03:44:12
  1707  140.000    10 2024/07/02 03:47:13
  1912  150.120    10 2024/07/02 03:50:38
  2142  160.120    10 2024/07/02 03:54:28
  2379  170.040    10 2024/07/02 03:58:25
  2625  180.040    10 2024/07/02 04:02:31
  2875  190.080    10 2024/07/02 04:06:41
  3115  200.000    10 2024/07/02 04:10:41
  3374  210.040    10 2024/07/02 04:15:00
  3647  220.040    10 2024/07/02 04:19:33
  3948  230.080    10 2024/07/02 04:24:34
  4308  240.000    10 2024/07/02 04:30:34
  4331  240.640    10 2024/07/02 04:30:57
# Rise Time Series
#  Cycle npts     Start                Date
R00  931   27 773253066 2024/07/02 16:31:06
# time     Pres phase                Date
     0  270.960     7 2024/07/02 16:31:06
   237  259.800    13 2024/07/02 16:35:03
   360  249.600    13 2024/07/02 16:37:06
   463  239.960    13 2024/07/02 16:38:49
   565  229.680    13 2024/07/02 16:40:31
   668  219.000    13 2024/07/02 16:42:14
   770  208.400    13 2024/07/02 16:43:56
   873  198.000    13 2024/07/02 16:45:39
   996  185.840    13 2024/07/02 16:47:42
  1098  175.200    13 2024/07/02 16:49:24
  1201  164.600    13 2024/07/02 16:51:07
  1303  154.240    13 2024/07/02 16:52:49
  1406  144.080    13 2024/07/02 16:54:32
  1508  133.360    13 2024/07/02 16:56:14
  1611  122.400    13 2024/07/02 16:57:57
  1713  111.720    13 2024/07/02 16:59:39
  1816  101.160    13 2024/07/02 17:01:22
  1918   90.880    13 2024/07/02 17:03:04
  2021   80.400    13 2024/07/02 17:04:47
  2123   69.400    13 2024/07/02 17:06:29
  2226   57.960    13 2024/07/02 17:08:12
  2328   46.800    13 2024/07/02 17:09:54
  2431   35.680    13 2024/07/02 17:11:37
  2533   24.400    13 2024/07/02 17:13:19
  2636   13.920    13 2024/07/02 17:15:02
  2759    2.840    13 2024/07/02 17:17:05
  2780    0.720     8 2024/07/02 17:17:26
# Pump Time Series
#   Cycle npts
B02   931   11
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    46.920  361   15.10     226    666    -9     2
   270.960   73   15.26     340    674    -9     7
   264.320   20   15.25     338    676    -9    13
   254.920   20   15.24     321    678    -9    13
   243.880   20   15.24     327    679    -9    13
   192.120   21   15.23     316    681    -9    13
   146.120   21   15.22     314    683    -9    13
    86.880   21   15.21     291    683    -9    13
    16.000   21   15.20     248    674    -9    13
     4.680   21   15.19     249    667    -9    13
     0.080   60   15.18     215    668    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   931  2.4   29    90   120   600    81    10   720      9     1    4020   15  0.6317
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  931 2024/07/02 03:18:43
# Pcase vacuums
   6.620   6.690
# Oil valve positions
   1.822   0.795   0.631
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  931 2024/07/02 04:30:57 9999/99/99 99:99:99
# Pcase vacuums
   6.680   6.700
# Average 
#   P      T      S
  269.680  -0.580  33.189
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  931 2024/07/02 16:31:06 9999/99/99 99:99:99
# Pcase vacuums
   6.690   6.650
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  931 2024/07/02 17:17:26
# number of packets previous packets
      7               8
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.650   6.660
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.176   0.158   0.667
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.740   0.876
# Encountered Error Codes
#    Cycle Errors
EE20   931      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  931  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.040    362   6.290
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  931       197    2024/07/02 17:20:59
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      931	(cycle number) |
     MPS =        0	(phase) |
     MPC =      931	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      120	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    42104	(vac 0 inHg cal) |
     PV5 =    18698	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =     4095	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        0	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        0	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       99	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      999	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       10	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        1	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =      600	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      930	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       46	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       10	(dBar) |
     DDT =     1800	(sec) |
  DFT[0] =      360	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        0	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65531	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      900	(sec) |
     DUC =       67	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      930	(cycle) |
  PTO[0] =    43200	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  1800000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =     1800	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      100	(dBar) |
     PST =     7200	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      930	(cycle number) |
     ARR =        9	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       60	(sec) |
     APN =       20	(sec) |
     APF =      100	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =       10	(dBar) |
     ADT =     1800	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      924	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =       60	(sec) |
     SSP =       60	(sec) |

