# Message Decoder for Solo2, Alto and Deep Solo Floats; v231107
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11342   874  247 000000000881890  5608     0   6 2023/12/01, 00:43:55,  66.6975  -53.6195   4      0 6,14,14
SBD 11342   874  340 000000000881890  5609     0   5 2023/12/01, 00:44:08,  66.6780  -53.5466   4      0 13
SBD 11342   874  299 000000000881890  5610     0   4 2023/12/01, 00:44:20,  66.6780  -53.5466   4      0 8
SBD 11342   874  338 000000000881890  5611     0   3 2023/12/01, 00:44:46,  66.6780  -53.5466   3      0 2,3,15
SBD 11342   874  278 000000000881890  5612     0   2 2023/12/01, 00:45:02,  66.6780  -53.5466   3      0 5,14,0,1
SBD 11342   874  289 000000000881890  5613     0   1 2023/12/01, 00:45:36,  66.6780  -53.5466   3      0 10,11,7,13
SBD 11342   874  310 000000000881890  5614     0   0 2023/12/01, 00:45:47,  66.6780  -53.5466   3      0 0,4,14,14,9
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         2017           84        2101
# Sensor_ID bytes     % packets Description
          0    48  2.38       2         GPS
          1    90  4.46       1    Pressure
          2   146  7.24       1 Temperature
          3   114  5.65       1    Salinity
          4   153  7.59       1        Fall
          5   133  6.59       1        Rise
          6   147  7.29       1        Pump
          7    43  2.13       1 Drift_times
          8   287 14.23       1     Surface
          9    77  3.82       1  HR/Drift_P
         10    79  3.92       1  HR/Drift_T
         11    79  3.92       1  HR/Drift_S
         13   404 20.03       2  Parameters
         14   151  7.49       5 Engineering
         15    66  3.27       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  874 1 2023/11/27 17:57 +66.67429  -53.55566    0  4 18 29 38  2.0     0
G  874 2 2023/12/01 00:37 +66.66888  -53.57129   30  6 35 41 45  0.9     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   874  132    1     3         0
#      Pres      Temp    Salinity
      1.040     1.2960    32.3370
      2.000     1.3110    32.3420
      3.000     1.2780    32.3430
      4.000     1.2790    32.3390
      5.040     1.2740    32.3370
      6.000     1.2700    32.3360
      7.040     1.2770    32.3380
      8.040     1.2760    32.3360
      9.040     1.2750    32.3370
     10.000     1.2610    32.3340
     11.000     1.2620    32.3330
     12.040     1.2590    32.3340
     13.040     1.2650    32.3350
     14.040     1.2620    32.3350
     15.000     1.2630    32.3330
     16.000     1.2640    32.3360
     17.040     1.2590    32.3330
     18.000     1.2610    32.3330
     19.000     1.2590    32.3340
     20.000     1.2630    32.3340
     21.000     1.2670    32.3350
     22.040     1.2660    32.3350
     23.000     1.2680    32.3350
     23.960     1.2710    32.3360
     24.960     1.2720    32.3370
     25.960     1.2720    32.3380
     27.000     1.2700    32.3360
     28.000     1.2710    32.3360
     29.000     1.2730    32.3370
     30.040     1.2750    32.3380
     31.000     1.2760    32.3380
     31.960     1.2800    32.3380
     33.000     1.2830    32.3420
     34.040     1.2800    32.3410
     35.000     1.2850    32.3400
     36.000     1.2840    32.3410
     37.040     1.2830    32.3420
     38.000     1.2870    32.3450
     39.000     1.2950    32.3460
     40.000     1.2930    32.3470
     41.000     1.2950    32.3470
     42.040     1.2990    32.3450
     43.040     1.2990    32.3460
     44.040     1.2930    32.3510
     45.000     1.2900    32.3550
     46.000     1.2910    32.3570
     47.040     1.2940    32.3590
     48.000     1.2950    32.3590
     49.000     1.2950    32.3590
     50.040     1.2940    32.3590
     51.040     1.2950    32.3590
     52.040     1.2970    32.3610
     53.040     1.2970    32.3610
     54.000     1.2990    32.3610
     55.000     1.2900    32.3610
     56.000     1.2930    32.3610
     57.000     1.2940    32.3620
     58.040     1.3060    32.3640
     59.000     1.3200    32.3710
     59.960     1.3160    32.3760
     60.960     1.3090    32.3740
     62.040     1.3190    32.3750
     63.000     1.3150    32.3740
     63.960     1.3080    32.3760
     65.000     1.3080    32.3750
     66.000     1.3080    32.3760
     67.000     1.3030    32.3750
     68.080     1.3100    32.3740
     69.000     1.3010    32.3720
     70.000     1.2920    32.3730
     71.080     1.2870    32.3720
     72.000     1.2870    32.3700
     73.040     1.2900    32.3730
     74.000     1.2940    32.3750
     75.000     1.3000    32.3750
     76.000     1.3060    32.3780
     77.040     1.2940    32.3690
     78.000     1.2920    32.3690
     79.040     1.2960    32.3690
     80.000     1.3020    32.3700
     81.000     1.3020    32.3710
     82.040     1.3010    32.3710
     83.040     1.2980    32.3720
     84.040     1.3060    32.3820
     85.040     1.3380    32.4030
     86.040     1.3880    32.4250
     87.080     1.3940    32.4290
     88.000     1.3960    32.4350
     88.960     1.4010    32.4340
     90.040     1.4160    32.4450
     91.000     1.4230    32.4470
     92.000     1.4400    32.4540
     93.000     1.4910    32.4750
     94.000     1.5050    32.4880
     95.000     1.5160    32.4970
     96.000     1.5190    32.4970
     96.960     1.5230    32.5000
     98.040     1.5240    32.5000
     99.040     1.5340    32.5060
    100.040     1.5380    32.5080
    100.960     1.5420    32.5100
    101.960     1.5420    32.5110
    102.960     1.5420    32.5110
    103.960     1.5450    32.5130
    105.000     1.5490    32.5150
    106.040     1.5490    32.5140
    107.040     1.5520    32.5170
    108.000     1.5700    32.5240
    109.000     1.5770    32.5320
    109.960     1.5820    32.5340
    111.000     1.5820    32.5340
    111.960     1.5880    32.5380
    113.040     1.5890    32.5390
    114.000     1.5900    32.5390
    114.960     1.5960    32.5420
    116.040     1.6020    32.5470
    117.000     1.6160    32.5550
    118.000     1.6520    32.5730
    119.080     1.6630    32.5830
    119.960     1.6640    32.5840
    121.000     1.6630    32.5830
    122.040     1.6620    32.5830
    123.080     1.6630    32.5830
    124.000     1.6600    32.5820
    125.040     1.6590    32.5800
    126.040     1.6590    32.5810
    126.960     1.6580    32.5800
    127.960     1.6570    32.5800
    129.000     1.6610    32.5820
    130.000     1.6610    32.5820
    131.080     1.6590    32.5810
    132.200     1.6580    32.5800
# Drift Data
#      C Npts comp n_var
I0   874   38    1     4
#      Pres       Temp   Salinity     Time
    117.560     1.5710    32.4530        5
    118.960     1.5100    32.4080     7211
    131.680     1.4350    32.3490    14417
    131.760     1.6690    32.5170    21622
    131.160     1.6420    32.4860    28828
    131.080     1.6120    32.4630    36034
    132.200     1.5150    32.4090    43239
     83.600     1.2520    32.3130    50445
     84.560     1.1710    32.3040    57650
     83.800     1.7230    32.5870    64856
    132.000     1.6240    32.5220    72062
    131.600     1.5800    32.4920    79267
    128.600     1.4790    32.4310    86473
    123.800     1.3510    32.3550    93679
    124.640     1.2620    32.3290   100884
    134.080     1.5690    32.5160   108090
    125.840     1.5590    32.4870   115296
    126.000     1.6060    32.5080   122501
    131.520     1.4340    32.3980   129707
    132.360     1.2840    32.3370   136912
    132.960     1.2360    32.3440   144118
    131.760     1.6390    32.5690   151324
    134.120     1.6040    32.5360   158529
    132.880     1.6190    32.5590   165735
    132.760     1.5200    32.4850   172941
    132.920     1.3940    32.4030   180146
    134.640     1.3520    32.3780   187352
    133.440     1.7240    32.6190   194558
    132.720     1.7640    32.6490   201763
    132.160     1.6220    32.5530   208969
    132.040     1.6480    32.5730   216174
    133.600     1.6520    32.5710   223380
    134.760     1.4570    32.4400   230586
    134.600     1.4740    32.4740   237791
    131.600     1.7420    32.6450   244997
    131.880     1.6640    32.5920   252203
    131.440     1.6370    32.5730   259408
    132.000     1.7070    32.6150   266614
# Surface Pressure Time Series
#    C npts n_var               Start
S  874    31     6 2023/12/01 00:36:50
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    69      0.370      1.418   -999.000      14 2023/12/01 00:37:59
    82      0.290      1.356   -999.000      14 2023/12/01 00:38:12
    93      0.340      1.327   -999.000      14 2023/12/01 00:38:23
   104      0.330      1.330   -999.000      14 2023/12/01 00:38:34
   115      0.310      1.401   -999.000      14 2023/12/01 00:38:45
   127      0.300      1.433   -999.000      14 2023/12/01 00:38:57
   138      0.300      1.506   -999.000      14 2023/12/01 00:39:08
   149      0.320      1.566   -999.000      14 2023/12/01 00:39:19
   160      0.280      1.616   -999.000      14 2023/12/01 00:39:30
   171      0.290      1.393   -999.000      14 2023/12/01 00:39:41
   182      0.240      1.383   -999.000      14 2023/12/01 00:39:52
   193      0.310      1.432   -999.000      14 2023/12/01 00:40:03
   204      0.310      1.462   -999.000      14 2023/12/01 00:40:14
   215      0.320      1.469   -999.000      14 2023/12/01 00:40:25
   226      0.340      1.560   -999.000      14 2023/12/01 00:40:36
   237      0.270      1.689   -999.000      14 2023/12/01 00:40:47
   248      0.300      1.763   -999.000      14 2023/12/01 00:40:58
   260      0.320      1.796   -999.000      14 2023/12/01 00:41:10
   271      0.270      1.748   -999.000      14 2023/12/01 00:41:21
   282      0.260      1.727   -999.000      14 2023/12/01 00:41:32
   293      0.330      1.541   -999.000      14 2023/12/01 00:41:43
   304      0.320      1.439   -999.000      14 2023/12/01 00:41:54
   315      0.250      1.504   -999.000      14 2023/12/01 00:42:05
   326      0.300      1.520   -999.000      14 2023/12/01 00:42:16
   337      0.320      1.522   -999.000      14 2023/12/01 00:42:27
   348      0.290      1.545   -999.000      14 2023/12/01 00:42:38
   359      0.290      1.593   -999.000      14 2023/12/01 00:42:49
   370      0.290      1.638   -999.000      14 2023/12/01 00:43:00
   382      0.300      1.692   -999.000      14 2023/12/01 00:43:12
   393      0.290      1.706   -999.000      14 2023/12/01 00:43:23
   404      0.290      1.688   -999.000      14 2023/12/01 00:43:34
# Fall Time Series
#  Cycle npts     Start                Date
F00  874   29 754423083 2023/11/27 17:58:03
# time     Pres phase                Date
     0    0.280     1 2023/11/27 17:58:03
   180    5.400    10 2023/11/27 18:01:03
   203   10.160    10 2023/11/27 18:01:26
   232   15.400    10 2023/11/27 18:01:55
   271   20.360    10 2023/11/27 18:02:34
   304   25.040    10 2023/11/27 18:03:07
   336   30.240    10 2023/11/27 18:03:39
   365   35.480    10 2023/11/27 18:04:08
   391   40.640    10 2023/11/27 18:04:34
   414   45.040    10 2023/11/27 18:04:57
   425   46.760     2 2023/11/27 18:05:08
   450   50.120    10 2023/11/27 18:05:33
   486   55.560    10 2023/11/27 18:06:09
   518   60.360    10 2023/11/27 18:06:41
   557   65.200    10 2023/11/27 18:07:20
   599   70.280    10 2023/11/27 18:08:02
   638   75.360    10 2023/11/27 18:08:41
   674   80.040    10 2023/11/27 18:09:17
   719   85.240    10 2023/11/27 18:10:02
   774   90.320    10 2023/11/27 18:10:57
   819   95.280    10 2023/11/27 18:11:42
  1056  100.000    10 2023/11/27 18:15:39
  1169  105.040    10 2023/11/27 18:17:32
  1237  110.120    10 2023/11/27 18:18:40
  1325  115.160    10 2023/11/27 18:20:08
  2227  117.200    10 2023/11/27 18:35:10
  3129  117.360    10 2023/11/27 18:50:12
  4031  117.480    10 2023/11/27 19:05:14
  4371  117.560    10 2023/11/27 19:10:54
# Rise Time Series
#  Cycle npts     Start                Date
R00  874   25 754704741 2023/12/01 00:12:21
# time     Pres phase                Date
     0  132.320     7 2023/12/01 00:12:21
   278  127.080    13 2023/12/01 00:16:59
   339  121.680    13 2023/12/01 00:18:00
   401  115.680    13 2023/12/01 00:19:02
   442  110.680    13 2023/12/01 00:19:43
   483  105.960    13 2023/12/01 00:20:24
   544   99.000    13 2023/12/01 00:21:25
   606   91.680    13 2023/12/01 00:22:27
   647   86.960    13 2023/12/01 00:23:08
   708   79.760    13 2023/12/01 00:24:09
   749   74.680    13 2023/12/01 00:24:50
   790   69.400    13 2023/12/01 00:25:31
   831   64.000    13 2023/12/01 00:26:12
   872   59.320    13 2023/12/01 00:26:53
   913   54.600    13 2023/12/01 00:27:34
   975   47.920    13 2023/12/01 00:28:36
  1036   41.560    13 2023/12/01 00:29:37
  1098   35.640    13 2023/12/01 00:30:39
  1159   30.480    13 2023/12/01 00:31:40
  1221   24.280    13 2023/12/01 00:32:42
  1262   19.680    13 2023/12/01 00:33:23
  1323   13.480    13 2023/12/01 00:34:24
  1385    7.280    13 2023/12/01 00:35:26
  1426    2.720    13 2023/12/01 00:36:07
  1467    0.360     8 2023/12/01 00:36:48
# Pump Time Series
#   Cycle npts
B02   874   13
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    46.760  341   15.08     252    661    -9     2
   132.320  134   15.31     278    666    -9     7
   132.280   20   15.31     290    668    -9    13
   131.360   20   15.30     288    670    -9    13
   129.840   20   15.29     269    672    -9    13
   127.080   20   15.28     271    673    -9    13
   123.560   20   15.28     279    675    -9    13
   117.800   21   15.27     265    677    -9    13
    35.640   20   15.27     234    679    -9    13
    32.120   20   15.27     230    681    -9    13
    28.640   20   15.26     236    682    -9    13
     7.280   20   15.25     239    685    -9    13
     0.200  180   15.22     208    700    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   874  2.4   26    90   120   600    80    10  4620      9     1    4020   15  3.3394
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  874 2023/11/27 17:58:00
# Pcase vacuums
   7.110   6.570
# Oil valve positions
   1.822   0.800   0.631
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  874 2023/11/27 19:10:55 9999/99/99 99:99:99
# Pcase vacuums
   6.570   6.560
# Average 
#   P      T      S
  127.080   1.534  32.480
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  874 2023/12/01 00:12:21 9999/99/99 99:99:99
# Pcase vacuums
   6.560   6.850
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  874 2023/12/01 00:36:47
# number of packets previous packets
      7               7
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.850   7.000
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.176   0.158   0.667
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.740   0.876
# Encountered Error Codes
#    Cycle Errors
EE20   874      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  874  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.240    421   6.330
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  874       197    2023/12/01 00:45:36
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      874	(cycle number) |
     MPS =        0	(phase) |
     MPC =      874	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      120	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    42104	(vac 0 inHg cal) |
     PV5 =    18698	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =     4095	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        0	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        0	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       99	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      999	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       10	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        1	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =      600	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      873	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       46	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      900	(sec) |
  DFT[0] =      340	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        0	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65531	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      900	(sec) |
     DUC =       67	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      873	(cycle) |
  PTO[0] =   277277	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  7200000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =     7200	(sec) |
     PSM =        1	(0=dis 1=en) |
     PSD =      100	(dBar) |
     PST =     7200	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      873	(cycle number) |
     ARR =        9	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =       20	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      867	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      340	(sec) |
     SSP =      180	(sec) |

