# Message Decoder for Solo2, Alto and Deep Solo Floats; v230908
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11342   849  247 000000000881890  5445     0   5 2023/09/09, 21:48:57,  66.6710  -53.6732  21      0 6,14,14
SBD 11342   849  340 000000000881890  5446     0   4 2023/09/09, 21:49:08,  66.6975  -53.6195   4      0 13
SBD 11342   849  299 000000000881890  5447     0   3 2023/09/09, 21:49:48,  66.5446  -56.9346 160      2 8
SBD 11342   849  332 000000000881890  5448     0   2 2023/09/09, 21:50:05,  66.5446  -56.9346 161      2 2,3,15,13
SBD 11342   849  296 000000000881890  5449     0   1 2023/09/09, 21:50:16,  66.5446  -56.9346 160      2 11,7,5,14,0,1
SBD 11342   849  321 000000000881890  5450     0   0 2023/09/09, 21:50:31,  66.5170  -56.9837 162      2 0,4,14,14,9,10
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1763           72        1835
# Sensor_ID bytes     % packets Description
          0    48  2.72       2         GPS
          1    53  3.01       1    Pressure
          2    89  5.05       1 Temperature
          3    89  5.05       1    Salinity
          4    93  5.28       1        Fall
          5    68  3.86       1        Rise
          6   147  8.34       1        Pump
          7    43  2.44       1 Drift_times
          8   287 16.28       1     Surface
          9    69  3.91       1  HR/Drift_P
         10    79  4.48       1  HR/Drift_T
         11    77  4.37       1  HR/Drift_S
         13   404 22.92       2  Parameters
         14   151  8.56       5 Engineering
         15    66  3.74       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  849 1 2023/09/06 15:15 +66.67271  -53.57534    0  5 30 39 47  2.2     0
G  849 2 2023/09/09 21:42 +66.67330  -53.57601   50  7 28 37 42  0.8     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   849   58    1     3         0
#      Pres      Temp    Salinity
      1.040     5.7400    31.1250
      2.040     5.7410    31.1250
      3.000     5.7400    31.1250
      3.960     5.7380    31.1230
      5.000     5.7440    31.1230
      6.000     5.7480    31.1230
      7.000     5.7370    31.1240
      7.960     5.7200    31.1210
      9.000     5.6980    31.1240
     10.040     5.6640    31.1160
     11.040     5.6450    31.1200
     12.040     5.6390    31.1190
     13.000     5.6460    31.1180
     14.000     5.6640    31.1210
     15.000     5.6480    31.1200
     16.000     5.6450    31.1200
     17.040     5.6400    31.1200
     18.000     5.6370    31.1190
     19.000     5.6320    31.1180
     20.000     5.6220    31.1190
     20.960     5.6210    31.1170
     21.960     5.6180    31.1180
     22.960     5.6130    31.1150
     24.000     5.6030    31.1190
     25.040     5.5560    31.1550
     26.040     5.5320    31.1740
     27.000     5.5310    31.1750
     28.000     5.5310    31.1760
     29.000     5.5040    31.1990
     30.000     5.4300    31.2620
     31.040     5.4170    31.2750
     32.040     5.4000    31.2930
     33.000     5.3970    31.2970
     34.000     5.3960    31.3000
     35.000     5.3890    31.3090
     35.960     5.3790    31.3190
     37.000     5.3770    31.3190
     38.040     5.3080    31.3610
     39.040     5.2670    31.3840
     40.000     5.2550    31.3870
     41.000     5.0920    31.4870
     42.040     4.6690    31.7480
     43.040     4.5260    31.8370
     44.000     4.5180    31.8410
     45.040     4.5100    31.8430
     46.000     4.4310    31.8850
     47.000     4.2600    31.9820
     48.040     4.0360    32.1120
     49.000     3.9280    32.1720
     50.000     3.8130    32.2250
     51.000     3.7430    32.2540
     52.000     3.6700    32.2870
     53.000     3.6330    32.2980
     54.000     3.4920    32.3540
     55.040     3.4230    32.3820
     56.040     3.3810    32.4070
     57.040     3.3440    32.4230
     57.880     3.3340    32.4270
# Drift Data
#      C Npts comp n_var
I0   849   38    1     4
#      Pres       Temp   Salinity     Time
     50.040     4.3690    31.8710        6
     49.600     4.5310    31.7760     7212
     51.000     4.5680    31.7520    14417
     51.040     4.9290    31.5510    21623
     56.280     4.7380    31.6600    28828
     56.880     4.9720    31.5280    36034
     56.800     4.8840    31.5750    43240
     56.240     4.6260    31.7170    50445
     55.600     4.6720    31.6920    57651
     55.600     4.6840    31.6870    64857
     56.240     4.4210    31.8260    72062
     57.080     4.1570    31.9710    79268
     57.400     3.7660    32.1880    86474
     57.440     3.7200    32.1920    93679
     56.760     4.1400    31.9790   100885
     56.280     4.4780    31.8010   108090
     56.320     4.8670    31.5990   115296
     56.800     4.6750    31.6960   122502
     57.240     4.6880    31.6920   129707
     57.200     4.1220    31.9820   136913
     56.800     4.2120    31.9360   144119
     56.520     4.2150    31.9350   151324
     56.640     4.0990    31.9940   158530
     57.280     3.8390    32.1230   165736
     57.880     3.7340    32.1820   172941
     58.040     3.6160    32.2500   180147
     57.680     3.5570    32.2810   187352
     57.080     3.6590    32.2330   194558
     56.680     4.3370    31.9050   201764
     56.680     4.6760    31.7280   208969
     56.960     4.5210    31.8140   216175
     57.240     4.1300    32.0020   223381
     57.160     3.6190    32.2670   230586
     56.800     3.6680    32.2440   237792
     56.600     3.5780    32.2950   244998
     56.880     3.6030    32.2840   252203
     57.560     3.4790    32.3470   259409
     58.120     3.3980    32.3860   266614
# Surface Pressure Time Series
#    C npts n_var               Start
S  849    31     6 2023/09/09 21:42:00
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    64      0.350      5.689   -999.000      14 2023/09/09 21:43:04
    76      0.400      5.645   -999.000      14 2023/09/09 21:43:16
    88      0.340      5.663   -999.000      14 2023/09/09 21:43:28
    99      0.340      5.636   -999.000      14 2023/09/09 21:43:39
   110      0.370      5.630   -999.000      14 2023/09/09 21:43:50
   121      0.350      5.616   -999.000      14 2023/09/09 21:44:01
   132      0.350      5.516   -999.000      14 2023/09/09 21:44:12
   143      0.400      5.616   -999.000      14 2023/09/09 21:44:23
   154      0.370      5.586   -999.000      14 2023/09/09 21:44:34
   165      0.360      5.604   -999.000      14 2023/09/09 21:44:45
   176      0.350      5.614   -999.000      14 2023/09/09 21:44:56
   187      0.360      5.420   -999.000      14 2023/09/09 21:45:07
   198      0.340      5.258   -999.000      14 2023/09/09 21:45:18
   209      0.360      5.110   -999.000      14 2023/09/09 21:45:29
   221      0.350      5.308   -999.000      14 2023/09/09 21:45:41
   232      0.350      5.504   -999.000      14 2023/09/09 21:45:52
   243      0.300      5.490   -999.000      14 2023/09/09 21:46:03
   254      0.340      5.490   -999.000      14 2023/09/09 21:46:14
   265      0.320      5.322   -999.000      14 2023/09/09 21:46:25
   276      0.330      5.227   -999.000      14 2023/09/09 21:46:36
   287      0.320      5.051   -999.000      14 2023/09/09 21:46:47
   298      0.310      4.928   -999.000      14 2023/09/09 21:46:58
   309      0.380      4.878   -999.000      14 2023/09/09 21:47:09
   320      0.290      5.394   -999.000      14 2023/09/09 21:47:20
   331      0.370      5.382   -999.000      14 2023/09/09 21:47:31
   343      0.340      5.581   -999.000      14 2023/09/09 21:47:43
   354      0.320      5.404   -999.000      14 2023/09/09 21:47:54
   365      0.340      5.208   -999.000      14 2023/09/09 21:48:05
   376      0.360      5.111   -999.000      14 2023/09/09 21:48:16
   387      0.330      5.095   -999.000      14 2023/09/09 21:48:27
   398      0.340      5.004   -999.000      14 2023/09/09 21:48:38
# Fall Time Series
#  Cycle npts     Start                Date
F00  849   17 747328528 2023/09/06 15:15:28
# time     Pres phase                Date
     0    0.160     1 2023/09/06 15:15:28
   125    5.000    10 2023/09/06 15:17:33
   154   10.080    10 2023/09/06 15:18:02
   184   15.240    10 2023/09/06 15:18:32
   213   20.200    10 2023/09/06 15:19:01
   242   25.000    10 2023/09/06 15:19:30
   271   30.200    10 2023/09/06 15:19:59
   300   35.440    10 2023/09/06 15:20:28
   330   40.480    10 2023/09/06 15:20:58
   359   45.480    10 2023/09/06 15:21:27
   367   46.800     2 2023/09/06 15:21:35
   388   50.120    10 2023/09/06 15:21:56
  1289   50.160    10 2023/09/06 15:36:57
  2191   50.120    10 2023/09/06 15:51:59
  3093   50.120    10 2023/09/06 16:07:01
  3995   50.080    10 2023/09/06 16:22:03
  4313   50.040    10 2023/09/06 16:27:21
# Rise Time Series
#  Cycle npts     Start                Date
R00  849   12 747610127 2023/09/09 21:28:47
# time     Pres phase                Date
     0   57.920     7 2023/09/09 21:28:47
   237   52.080    13 2023/09/09 21:32:44
   319   46.560    13 2023/09/09 21:34:06
   401   40.400    13 2023/09/09 21:35:28
   463   35.200    13 2023/09/09 21:36:30
   524   29.240    13 2023/09/09 21:37:31
   565   24.880    13 2023/09/09 21:38:12
   606   19.920    13 2023/09/09 21:38:53
   647   14.800    13 2023/09/09 21:39:34
   708    7.680    13 2023/09/09 21:40:35
   749    2.800    13 2023/09/09 21:41:16
   791    0.320     8 2023/09/09 21:41:58
# Pump Time Series
#   Cycle npts
B02   849   13
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    46.800  340   15.16     205    640    -9     2
    57.920  135   15.24     265    657    -9     7
    56.840   21   15.24     224    659    -9    13
    54.760   21   15.24     230    661    -9    13
    52.080   21   15.23     224    662    -9    13
    49.440   21   15.23     238    663    -9    13
    46.560   21   15.22     242    665    -9    13
    43.800   21   15.21     241    665    -9    13
    40.400   21   15.22     224    665    -9    13
    37.160   21   15.21     215    666    -9    13
    33.240   21   15.20     217    666    -9    13
    29.240   21   15.20     230    667    -9    13
     0.160  180   15.16     220    679    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   849  2.4   26    90   120   600    80    10  4620      9     1    4020   15  3.3394
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  849 2023/09/06 15:15:25
# Pcase vacuums
   6.740   6.390
# Oil valve positions
   1.822   0.798   0.631
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  849 2023/09/06 16:27:21 9999/99/99 99:99:99
# Pcase vacuums
   6.390   6.470
# Average 
#   P      T      S
   56.200   4.209  31.946
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  849 2023/09/09 21:28:47 9999/99/99 99:99:99
# Pcase vacuums
   6.470   6.650
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  849 2023/09/09 21:41:56
# number of packets previous packets
      6               6
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.650   6.770
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.176   0.158   0.667
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.740   0.876
# Encountered Error Codes
#    Cycle Errors
EE20   849      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  849  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.160    362   6.080
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  849       197    2023/09/09 21:50:05
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      849	(cycle number) |
     MPS =        0	(phase) |
     MPC =      849	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      120	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    42104	(vac 0 inHg cal) |
     PV5 =    18698	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =     4095	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        0	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        0	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       99	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      999	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       10	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        1	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =      600	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      848	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       46	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      900	(sec) |
  DFT[0] =      340	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        0	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65531	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      900	(sec) |
     DUC =       67	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      848	(cycle) |
  PTO[0] =   277277	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  7200000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =     7200	(sec) |
     PSM =        1	(0=dis 1=en) |
     PSD =      100	(dBar) |
     PST =     7200	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      848	(cycle number) |
     ARR =        9	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =       20	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      842	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      340	(sec) |
     SSP =      180	(sec) |

