# Message Decoder for Solo2, Alto and Deep Solo Floats; v230712
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11342   798  257 000000000881890  5135     0   3 2023/06/18, 18:15:25,  66.5516  -51.1291 144      2 13,6,14,14
SBD 11342   798  340 000000000881890  5136     0   2 2023/06/18, 18:15:37,  66.5309  -51.0590 143      2 13
SBD 11342   798  338 000000000881890  5137     0   1 2023/06/18, 18:16:24,  66.5358  -50.9324 144      2 5,14,0,1,2,3,8,15
SBD 11342   798  340 000000000881890  5138     0   0 2023/06/18, 18:16:37,  66.5613  -50.8755 143      2 0,4,14,14,9,10,11,7
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1227           48        1275
# Sensor_ID bytes     % packets Description
          0    48  3.91       2         GPS
          1    38  3.10       1    Pressure
          2    51  4.16       1 Temperature
          3    38  3.10       1    Salinity
          4    58  4.73       1        Fall
          5    28  2.28       1        Rise
          6    81  6.60       1        Pump
          7    41  3.34       1 Drift_times
          8    62  5.05       1     Surface
          9    41  3.34       1  HR/Drift_P
         10    64  5.22       1  HR/Drift_T
         11    56  4.56       1  HR/Drift_S
         13   404 32.93       2  Parameters
         14   151 12.31       5 Engineering
         15    66  5.38       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  798 1 2023/06/17 23:49 +66.62004  -53.65822    0  4 31 38 45  2.6     0
G  798 2 2023/06/18 18:13 +66.62054  -53.65815   20  9 25 38 45  0.8     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   798   28    1     3         0
#      Pres      Temp    Salinity
      1.080     2.6810    33.2700
      2.040     2.6390    33.2650
      3.000     2.6330    33.2630
      4.040     2.5940    33.2650
      5.040     2.5710    33.2650
      6.000     2.4980    33.2670
      7.040     2.4810    33.2660
      8.040     2.4640    33.2660
      9.040     2.4620    33.2670
     10.000     2.4620    33.2670
     11.000     2.4740    33.2700
     12.040     2.4700    33.2710
     13.040     2.4680    33.2710
     14.040     2.4680    33.2700
     15.040     2.4670    33.2700
     16.040     2.4640    33.2720
     17.040     2.4610    33.2740
     18.000     2.4560    33.2750
     19.000     2.4460    33.2790
     20.000     2.4370    33.2810
     21.000     2.4130    33.2830
     22.000     2.4090    33.2840
     22.960     2.4110    33.2830
     24.000     2.4090    33.2830
     25.040     2.4020    33.2840
     26.040     2.3790    33.2880
     27.040     2.3740    33.2900
     28.040     2.3670    33.2910
# Drift Data
#      C Npts comp n_var
I0   798   33    1     4
#      Pres       Temp   Salinity     Time
     29.960     2.3060    33.2960        5
     29.640     2.3780    33.2890     1811
     29.280     2.4450    33.2770     3616
     28.880     2.3330    33.3000     5422
     28.560     2.3810    33.2910     7228
     28.240     2.4130    33.2890     9033
     28.040     2.4080    33.2870    10839
     27.920     2.4000    33.2870    12645
     27.880     2.3710    33.2880    14450
     27.880     2.3580    33.2880    16256
     27.960     2.3570    33.2880    18062
     28.120     2.3620    33.2860    19867
     28.360     2.3480    33.2900    21673
     28.640     2.3150    33.2970    23478
     28.960     2.3100    33.2980    25284
     29.280     2.3220    33.2960    27090
     29.520     2.3310    33.2940    28895
     29.720     2.3400    33.2930    30701
     29.880     2.3390    33.2930    32507
     29.920     2.2990    33.3020    34312
     29.960     2.3090    33.3010    36118
     29.880     2.3160    33.3000    37924
     29.760     2.3150    33.3000    39729
     29.560     2.3080    33.3030    41535
     29.280     2.3360    33.2950    43340
     29.000     2.3540    33.2920    45146
     28.680     2.3770    33.2870    46952
     28.360     2.3930    33.2860    48757
     28.120     2.3870    33.2860    50563
     27.920     2.3930    33.2860    52369
     27.800     2.4010    33.2830    54174
     27.720     2.4170    33.2840    55980
     27.800     2.4230    33.2840    57786
# Surface Pressure Time Series
#    C npts n_var               Start
S  798     6     6 2023/06/18 18:13:05
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    64      0.410      2.938   -999.000      14 2023/06/18 18:14:09
    76      0.420      2.893   -999.000      14 2023/06/18 18:14:21
    88      0.400      2.862   -999.000      14 2023/06/18 18:14:33
    99      0.390      2.885   -999.000      14 2023/06/18 18:14:44
   110      0.390      2.926   -999.000      14 2023/06/18 18:14:55
   121      0.400      2.996   -999.000      14 2023/06/18 18:15:06
# Fall Time Series
#  Cycle npts     Start                Date
F00  798   10 740361001 2023/06/17 23:50:01
# time     Pres phase                Date
     0    0.400     1 2023/06/17 23:50:01
   171   10.240    10 2023/06/17 23:52:52
   245   20.360    10 2023/06/17 23:54:06
   313   30.280    10 2023/06/17 23:55:14
   607   30.640     2 2023/06/18 00:00:08
  1215   30.560    10 2023/06/18 00:10:16
  2117   30.400    10 2023/06/18 00:25:18
  3018   30.280    10 2023/06/18 00:40:19
  3920   30.120    10 2023/06/18 00:55:21
  4530   29.960    10 2023/06/18 01:05:31
# Rise Time Series
#  Cycle npts     Start                Date
R00  798    4 740426741 2023/06/18 18:05:41
# time     Pres phase                Date
     0   28.080     7 2023/06/18 18:05:41
   237   18.440    13 2023/06/18 18:09:38
   339    8.480    13 2023/06/18 18:11:20
   442    0.440     8 2023/06/18 18:13:03
# Pump Time Series
#   Cycle npts
B02   798    7
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
     0.440  179   15.22     224    687    -9    14
    30.640  318   15.17     198    657    -9     2
    28.080  134   15.30     232    663    -9     7
    24.920   20   15.28     225    664    -9    13
    21.880   20   15.29     218    666    -9    13
    18.440   20   15.28     230    668    -9    13
    14.800   20   15.27     221    670    -9    13
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   798  2.4   22    90   120   600    80    10  1020      9     1    4020   15  0.8394
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  798 2023/06/17 23:49:58
# Pcase vacuums
   6.870   6.530
# Oil valve positions
   1.821   0.801   0.631
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  798 2023/06/18 01:05:32 9999/99/99 99:99:99
# Pcase vacuums
   6.530   6.530
# Average 
#   P      T      S
   28.800   2.359  33.291
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  798 2023/06/18 18:05:41 9999/99/99 99:99:99
# Pcase vacuums
   6.520   6.700
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  798 2023/06/18 18:13:04
# number of packets previous packets
      4               7
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.700   6.800
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.176   0.158   0.667
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.740   0.876
# Encountered Error Codes
#    Cycle Errors
EE20   798      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  798  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.400    602   6.300
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  798       197    2023/06/18 18:15:37
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      798	(cycle number) |
     MPS =        0	(phase) |
     MPC =      798	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      120	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    42104	(vac 0 inHg cal) |
     PV5 =    18698	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =     4095	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        0	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        0	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       99	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      999	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       10	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        1	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =      600	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      797	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       46	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       10	(dBar) |
     DDT =      900	(sec) |
  DFT[0] =      315	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        0	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65531	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      900	(sec) |
     DUC =       67	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      797	(cycle) |
  PTO[0] =    61200	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  1800000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =     1800	(sec) |
     PSM =        1	(0=dis 1=en) |
     PSD =      100	(dBar) |
     PST =     7200	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      797	(cycle number) |
     ARR =        9	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =       20	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =       10	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      791	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =       60	(sec) |
     SSP =      180	(sec) |

