# Message Decoder for Solo2, Alto and Deep Solo Floats; v230712
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11342   797   88 000000000881890  5130     0   4 2023/06/17, 23:47:46,  66.6107  -53.6539   3      0 14
SBD 11342   797  340 000000000881890  5131     0   3 2023/06/17, 23:47:57,  66.6107  -53.6539   3      0 13
SBD 11342   797  320 000000000881890  5132     0   2 2023/06/17, 23:48:33,  66.6302  -53.7267   4      0 8,15,13,6,14
SBD 11342   797  287 000000000881890  5133     0   1 2023/06/17, 23:48:56,  66.6302  -53.7267   5      0 7,5,14,1,2,3,0
SBD 11342   797  307 000000000881890  5134     0   0 2023/06/17, 23:49:42,  66.6037  -53.7801   7      0 0,4,14,14,9,10,11
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1282           60        1342
# Sensor_ID bytes     % packets Description
          0    48  3.74       2         GPS
          1    42  3.28       1    Pressure
          2    58  4.52       1 Temperature
          3    58  4.52       1    Salinity
          4    58  4.52       1        Fall
          5    33  2.57       1        Rise
          6    92  7.18       1        Pump
          7    41  3.20       1 Drift_times
          8    62  4.84       1     Surface
          9    41  3.20       1  HR/Drift_P
         10    72  5.62       1  HR/Drift_T
         11    56  4.37       1  HR/Drift_S
         13   404 31.51       2  Parameters
         14   151 11.78       5 Engineering
         15    66  5.15       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  797 1 2023/06/17 05:20 +66.61969  -53.65754   10  5 35 39 44  2.2     0
G  797 2 2023/06/17 23:46 +66.61996  -53.65827   60  4 41 43 46  2.6     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   797   35    1     3         0
#      Pres      Temp    Salinity
      1.040     2.5430    33.2450
      2.000     2.5430    33.2450
      3.040     2.5440    33.2440
      4.040     2.5420    33.2450
      5.040     2.4940    33.2550
      6.040     2.4880    33.2560
      7.000     2.4820    33.2560
      8.000     2.4810    33.2570
      8.960     2.4430    33.2650
      9.960     2.4150    33.2720
     10.960     2.3900    33.2760
     12.000     2.3840    33.2770
     13.000     2.3790    33.2780
     14.000     2.3770    33.2780
     15.000     2.3820    33.2770
     16.000     2.3950    33.2750
     17.000     2.3990    33.2750
     18.000     2.3860    33.2760
     19.000     2.3610    33.2780
     20.080     2.3360    33.2810
     21.000     2.3200    33.2840
     22.000     2.3040    33.2900
     23.000     2.2920    33.2930
     23.960     2.2860    33.2970
     25.000     2.2700    33.3030
     26.040     2.2540    33.3060
     27.000     2.2480    33.3060
     27.960     2.2460    33.3070
     29.040     2.2420    33.3070
     30.040     2.2280    33.3110
     31.000     2.1880    33.3210
     32.000     2.1430    33.3310
     33.040     2.1400    33.3310
     34.040     2.1320    33.3320
     34.880     2.1070    33.3370
# Drift Data
#      C Npts comp n_var
I0   797   33    1     4
#      Pres       Temp   Salinity     Time
     32.520     2.3300    33.2900        5
     32.800     2.3300    33.2900     1811
     33.120     2.3350    33.2900     3617
     33.360     2.3390    33.2900     5422
     33.600     2.3490    33.2900     7228
     33.800     2.3720    33.2890     9033
     33.920     2.3760    33.2900    10839
     33.960     2.2740    33.2990    12645
     33.960     2.3070    33.2980    14450
     33.840     2.3350    33.2940    16256
     33.720     2.3020    33.3000    18062
     33.480     2.3560    33.2890    19867
     33.240     2.3770    33.2870    21673
     32.920     2.3900    33.2850    23479
     32.600     2.3820    33.2870    25284
     32.320     2.3780    33.2870    27090
     32.080     2.4030    33.2840    28895
     31.880     2.4090    33.2850    30701
     31.760     2.4310    33.2860    32507
     31.720     2.4280    33.2880    34312
     31.800     2.4260    33.2880    36118
     31.960     2.4270    33.2880    37924
     32.160     2.4170    33.2890    39729
     32.440     2.3800    33.2940    41535
     32.800     2.3830    33.2900    43341
     33.200     2.3130    33.2970    45146
     33.600     2.2800    33.3030    46952
     33.880     2.1570    33.3300    48757
     34.240     2.0410    33.3590    50563
     34.560     2.0740    33.3490    52369
     34.720     2.0040    33.3670    54174
     34.880     2.1310    33.3360    55980
     34.920     2.1050    33.3360    57786
# Surface Pressure Time Series
#    C npts n_var               Start
S  797     6     6 2023/06/17 23:45:27
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    65      0.400      2.461   -999.000      14 2023/06/17 23:46:32
    77      0.390      2.488   -999.000      14 2023/06/17 23:46:44
    88      0.390      2.410   -999.000      14 2023/06/17 23:46:55
    99      0.370      2.442   -999.000      14 2023/06/17 23:47:06
   111      0.370      2.416   -999.000      14 2023/06/17 23:47:18
   122      0.440      2.316   -999.000      14 2023/06/17 23:47:29
# Fall Time Series
#  Cycle npts     Start                Date
F00  797   10 740294461 2023/06/17 05:21:01
# time     Pres phase                Date
     0    0.440     1 2023/06/17 05:21:01
   164   10.160    10 2023/06/17 05:23:45
   229   20.040    10 2023/06/17 05:24:50
   301   30.480    10 2023/06/17 05:26:02
   607   32.000     2 2023/06/17 05:31:08
  1202   32.080    10 2023/06/17 05:41:03
  2104   32.200    10 2023/06/17 05:56:05
  3006   32.280    10 2023/06/17 06:11:07
  3907   32.400    10 2023/06/17 06:26:08
  4530   32.520    10 2023/06/17 06:36:31
# Rise Time Series
#  Cycle npts     Start                Date
R00  797    5 740360201 2023/06/17 23:36:41
# time     Pres phase                Date
     0   34.920     7 2023/06/17 23:36:41
   278   23.400    13 2023/06/17 23:41:19
   380   12.680    13 2023/06/17 23:43:01
   483    2.600    13 2023/06/17 23:44:44
   524    0.440     8 2023/06/17 23:45:25
# Pump Time Series
#   Cycle npts
B02   797    8
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
     0.400  180   15.24     187    683    -9    14
    32.000  318   15.19     204    655    -9     2
    34.920  134   15.28     220    664    -9     7
    33.520   20   15.27     234    666    -9    13
    31.000   20   15.28     241    668    -9    13
    27.640   20   15.26     242    669    -9    13
    10.760   20   15.26     218    671    -9    13
     6.840   20   15.26     203    672    -9    13
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   797  2.4   22    90   120   600    80    10  1020      9     1    4020   15  0.8394
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  797 2023/06/17 05:20:58
# Pcase vacuums
   6.830   6.530
# Oil valve positions
   1.822   0.801   0.631
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  797 2023/06/17 06:36:32 9999/99/99 99:99:99
# Pcase vacuums
   6.520   6.540
# Average 
#   P      T      S
   33.200   2.313  33.301
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  797 2023/06/17 23:36:41 9999/99/99 99:99:99
# Pcase vacuums
   6.540   6.720
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  797 2023/06/17 23:45:26
# number of packets previous packets
      4               4
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.720   6.830
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.176   0.158   0.667
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.740   0.876
# Encountered Error Codes
#    Cycle Errors
EE20   797      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  797  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.400    602   6.280
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  797       197    2023/06/17 23:48:33
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      797	(cycle number) |
     MPS =        0	(phase) |
     MPC =      797	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      120	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    42104	(vac 0 inHg cal) |
     PV5 =    18698	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =     4095	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        0	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        0	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       99	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      999	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       10	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        1	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =      600	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      796	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       46	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       10	(dBar) |
     DDT =      900	(sec) |
  DFT[0] =      315	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        0	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65531	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      900	(sec) |
     DUC =       67	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      796	(cycle) |
  PTO[0] =    61200	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  1800000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =     1800	(sec) |
     PSM =        1	(0=dis 1=en) |
     PSD =      100	(dBar) |
     PST =     7200	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      796	(cycle number) |
     ARR =        9	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =       20	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =       10	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      790	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =       60	(sec) |
     SSP =      180	(sec) |

