# Message Decoder for Solo2, Alto and Deep Solo Floats; v230712
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11342   624  340 000000000881890  4143     0   5 2023/01/18, 11:14:47,  65.4510  -53.3890   3      0 13
SBD 11342   624  323 000000000881890  4144     0   4 2023/01/18, 11:15:11,  65.4705  -53.4583   3      0 15,13,6,14,14
SBD 11342   624  299 000000000881890  4145     0   3 2023/01/18, 11:15:22,  65.4510  -53.3890   3      0 8
SBD 11342   624  334 000000000881890  4146     0   2 2023/01/18, 11:15:32,  65.4705  -53.4583   3      0 2,3
SBD 11342   624  311 000000000881890  4147     0   1 2023/01/18, 11:15:43,  65.4705  -53.4583   3      0 11,7,5,14,0,1
SBD 11342   624  313 000000000881890  4148     0   0 2023/01/18, 11:15:56,  65.4705  -53.4583   3      0 0,4,14,14,9,10
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1848           72        1920
# Sensor_ID bytes     % packets Description
          0    48  2.60       2         GPS
          1    95  5.14       1    Pressure
          2   164  8.87       1 Temperature
          3   158  8.55       1    Salinity
          4   138  7.47       1        Fall
          5    83  4.49       1        Rise
          6    81  4.38       1        Pump
          7    35  1.89       1 Drift_times
          8   287 15.53       1     Surface
          9    52  2.81       1  HR/Drift_P
         10    43  2.33       1  HR/Drift_T
         11    43  2.33       1  HR/Drift_S
         13   404 21.86       2  Parameters
         14   151  8.17       5 Engineering
         15    66  3.57       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  624 1 2023/01/18 05:47 +65.45343  -53.40615  120  9 24 36 44  0.9     0
G  624 2 2023/01/18 11:08 +65.46518  -53.41644   30  4 29 37 46  4.2     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   624  141    1     3         0
#      Pres      Temp    Salinity
      1.080    -0.5250    33.0200
      2.040    -0.5030    33.0240
      3.000    -0.4890    33.0210
      4.000    -0.4840    33.0250
      5.000    -0.4610    33.0220
      6.000    -0.3730    33.0260
      7.000    -0.3510    33.0320
      8.000    -0.2660    33.0330
      9.040    -0.1550    33.0400
     10.040    -0.1170    33.0500
     11.000    -0.1110    33.0510
     11.960    -0.0700    33.0500
     13.000    -0.0530    33.0560
     14.000    -0.0460    33.0560
     15.040    -0.0390    33.0590
     16.040    -0.0540    33.0590
     17.040    -0.0510    33.0600
     18.040    -0.0340    33.0610
     19.040    -0.0090    33.0250
     20.040     0.0180    33.0680
     21.040     0.0240    33.0700
     22.040     0.0290    33.0710
     23.040     0.0320    33.0730
     24.040     0.0680    33.0760
     25.000     0.1290    33.0850
     26.000     0.1780    33.0960
     27.000     0.1890    33.1000
     28.000     0.2090    33.1030
     29.000     0.2500    33.1070
     30.000     0.2830    33.1150
     31.000     0.3080    33.1180
     32.000     0.3330    33.1220
     33.000     0.3490    33.1250
     34.040     0.3710    33.1290
     35.000     0.4280    33.1340
     36.000     0.4820    33.1490
     37.000     0.5070    33.1580
     37.960     0.5450    33.1670
     38.960     0.5920    33.1770
     40.000     0.6150    33.1840
     41.000     0.6300    33.1860
     41.960     0.6400    33.1930
     43.000     0.6480    33.1960
     44.000     0.6800    33.2070
     45.040     0.6770    33.2100
     46.040     0.6840    33.2130
     47.000     0.6880    33.2170
     48.000     0.7000    33.2200
     49.000     0.6970    33.2230
     49.960     0.6970    33.2230
     51.000     0.6990    33.2240
     52.000     0.7000    33.2240
     52.960     0.7090    33.2250
     54.000     0.7260    33.2280
     55.000     0.7520    33.2300
     56.000     0.7350    33.2320
     57.000     0.7760    33.2340
     58.040     0.8070    33.2400
     59.040     0.8230    33.2420
     60.000     0.8380    33.2440
     60.960     0.8990    33.2450
     62.000     0.9680    33.2640
     63.000     0.9630    33.2680
     63.960     0.9680    33.2640
     65.000     0.9720    33.2660
     66.040     0.9770    33.2660
     67.000     0.9760    33.2670
     68.000     0.9860    33.2670
     69.040     1.0070    33.2710
     70.040     1.0450    33.2790
     71.000     1.0570    33.2810
     72.000     1.0630    33.2840
     73.000     1.0740    33.2900
     74.040     1.0910    33.2980
     75.000     1.1530    33.3100
     76.000     1.2640    33.3210
     77.040     1.4260    33.3550
     78.000     1.4950    33.3740
     79.000     1.5190    33.3910
     80.040     1.5670    33.4000
     81.000     1.6280    33.4190
     82.000     1.7090    33.4360
     83.000     1.7860    33.4480
     84.040     1.8270    33.4500
     85.040     1.8330    33.4510
     86.000     1.8340    33.4510
     87.000     1.8350    33.4530
     88.000     1.8400    33.4550
     88.960     1.8460    33.4580
     90.000     1.8560    33.4600
     91.000     1.8590    33.4620
     92.000     1.8740    33.4690
     92.960     1.9150    33.4730
     94.000     1.9400    33.4840
     95.040     1.9820    33.4840
     96.040     1.9890    33.4900
     97.000     2.0090    33.4910
     97.960     2.0180    33.4950
     99.000     2.0250    33.4940
    100.000     2.0270    33.4950
    101.000     2.0270    33.4960
    102.000     2.0310    33.4960
    103.000     2.0320    33.4960
    104.000     2.0330    33.4970
    105.000     2.0350    33.4990
    106.000     2.0420    33.5000
    107.040     2.0420    33.5000
    108.000     2.0530    33.5030
    108.960     2.0620    33.5040
    110.000     2.0640    33.5040
    111.000     2.0670    33.5060
    112.000     2.0750    33.5060
    113.040     2.0810    33.5150
    114.000     2.1260    33.5160
    115.000     2.1320    33.5180
    116.000     2.1390    33.5200
    117.000     2.1470    33.5240
    118.000     2.1630    33.5250
    119.000     2.1720    33.5280
    120.000     2.1950    33.5360
    121.000     2.2250    33.5400
    121.960     2.2360    33.5420
    122.960     2.2510    33.5510
    123.960     2.2740    33.5570
    124.960     2.2880    33.5650
    126.000     2.3340    33.5760
    127.000     2.3670    33.5790
    128.040     2.3750    33.5820
    129.040     2.3840    33.5830
    130.000     2.3890    33.5850
    131.040     2.3940    33.5880
    132.040     2.4020    33.5890
    133.040     2.4050    33.5910
    134.000     2.4160    33.5940
    135.000     2.4240    33.5950
    136.000     2.4260    33.5970
    136.960     2.4330    33.5980
    137.960     2.4330    33.5970
    139.000     2.4340    33.5990
    140.000     2.4390    33.6000
    141.000     2.4430  -999.0000
# Drift Data
#      C Npts comp n_var
I0   624   21    1     4
#      Pres       Temp   Salinity     Time
    138.640     2.3750    33.5810        6
    126.520     2.3730    33.5790      611
    121.520     2.4810    33.6140     1217
    126.400     2.4740    33.6090     1822
    128.040     2.4720    33.6090     2428
    130.680     2.4500    33.6030     3033
    132.520     2.4510    33.6040     3638
    138.320     2.4440    33.6010     4244
    134.080     2.4440    33.6010     4849
    134.160     2.4350    33.5990     5455
    135.640     2.4420    33.5990     6060
    136.920     2.4420    33.6000     6666
    137.280     2.4410    33.6000     7271
    137.800     2.4420    33.6000     7877
    135.360     2.4230    33.5970     8482
    133.960     2.4200    33.5930     9088
    133.360     2.4330    33.5960     9693
    136.080     2.4380    33.5990    10299
    137.240     2.4400    33.5990    10904
    139.680     2.4410    33.6000    11509
    139.040     2.4380    33.5990    12115
# Surface Pressure Time Series
#    C npts n_var               Start
S  624    31     6 2023/01/18 11:07:32
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    71      0.440     -0.616   -999.000      14 2023/01/18 11:08:43
    84      0.350     -0.750   -999.000      14 2023/01/18 11:08:56
    95      0.360     -1.391   -999.000      14 2023/01/18 11:09:07
   106      0.350     -0.807   -999.000      14 2023/01/18 11:09:18
   117      0.330     -0.816   -999.000      14 2023/01/18 11:09:29
   129      0.330     -0.837   -999.000      14 2023/01/18 11:09:41
   140      0.350     -0.880   -999.000      14 2023/01/18 11:09:52
   151      0.380     -1.446   -999.000      14 2023/01/18 11:10:03
   162      0.360     -0.958   -999.000      14 2023/01/18 11:10:14
   173      0.350     -0.934   -999.000      14 2023/01/18 11:10:25
   184      0.310     -1.732   -999.000      14 2023/01/18 11:10:36
   195      0.290     -1.931   -999.000      14 2023/01/18 11:10:47
   206      0.340     -1.611   -999.000      14 2023/01/18 11:10:58
   217      0.330     -1.919   -999.000      14 2023/01/18 11:11:09
   228      0.340   -999.000   -999.000      14 2023/01/18 11:11:20
   239      0.290   -999.000   -999.000      14 2023/01/18 11:11:31
   251      0.310   -999.000   -999.000      14 2023/01/18 11:11:43
   262      0.320   -999.000   -999.000      14 2023/01/18 11:11:54
   273      0.400   -999.000   -999.000      14 2023/01/18 11:12:05
   284      0.310     -1.541   -999.000      14 2023/01/18 11:12:16
   295      0.340     -1.857   -999.000      14 2023/01/18 11:12:27
   306      0.360   -999.000   -999.000      14 2023/01/18 11:12:38
   317      0.360   -999.000   -999.000      14 2023/01/18 11:12:49
   328      0.350     -1.659   -999.000      14 2023/01/18 11:13:00
   339      0.390     -1.219   -999.000      14 2023/01/18 11:13:11
   350      0.310     -1.837   -999.000      14 2023/01/18 11:13:22
   361      0.330   -999.000   -999.000      14 2023/01/18 11:13:33
   373      0.300   -999.000   -999.000      14 2023/01/18 11:13:45
   384      0.380   -999.000   -999.000      14 2023/01/18 11:13:56
   395      0.370     -1.833   -999.000      14 2023/01/18 11:14:07
   406      0.310     -1.245   -999.000      14 2023/01/18 11:14:18
# Fall Time Series
#  Cycle npts     Start                Date
F00  624   26 727336060 2023/01/18 05:47:40
# time     Pres phase                Date
     0    0.400     1 2023/01/18 05:47:40
   190   10.200    10 2023/01/18 05:50:50
   258   20.200    10 2023/01/18 05:51:58
   323   30.120    10 2023/01/18 05:53:03
   391   40.440    10 2023/01/18 05:54:11
   435   47.000     2 2023/01/18 05:54:55
   457   50.280    10 2023/01/18 05:55:17
   528   60.080    10 2023/01/18 05:56:28
   606   70.160    10 2023/01/18 05:57:46
   706   80.120    10 2023/01/18 05:59:26
   902   90.040    10 2023/01/18 06:02:42
  1202   97.120    10 2023/01/18 06:07:42
  1502  104.200    10 2023/01/18 06:12:42
  1802  112.800    10 2023/01/18 06:17:42
  2102  119.360    10 2023/01/18 06:22:42
  2402  124.080    10 2023/01/18 06:27:42
  2702  123.880    10 2023/01/18 06:32:42
  3002  125.200    10 2023/01/18 06:37:42
  3302  131.000    10 2023/01/18 06:42:42
  3602  132.760    10 2023/01/18 06:47:42
  3902  132.440    10 2023/01/18 06:52:42
  4202  133.880    10 2023/01/18 06:57:42
  4352  135.520     3 2023/01/18 07:00:12
  4502  137.400    10 2023/01/18 07:02:42
  4802  139.000    10 2023/01/18 07:07:42
  4990  138.680    10 2023/01/18 07:10:50
# Rise Time Series
#  Cycle npts     Start                Date
R00  624   15 727353659 2023/01/18 10:40:59
# time     Pres phase                Date
     0  140.960     7 2023/01/18 10:40:59
   196  130.800    13 2023/01/18 10:44:15
   319  119.840    13 2023/01/18 10:46:18
   442  109.160    13 2023/01/18 10:48:21
   544   99.960    13 2023/01/18 10:50:03
   667   88.600    13 2023/01/18 10:52:06
   790   77.120    13 2023/01/18 10:54:09
   893   67.800    13 2023/01/18 10:55:52
   995   57.880    13 2023/01/18 10:57:34
  1098   47.800    13 2023/01/18 10:59:17
  1200   37.400    13 2023/01/18 11:00:59
  1303   26.640    13 2023/01/18 11:02:42
  1405   15.760    13 2023/01/18 11:04:24
  1508    5.360    13 2023/01/18 11:06:07
  1591    0.440     8 2023/01/18 11:07:30
# Pump Time Series
#   Cycle npts
B02   624    7
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    47.000  315   15.16     217    662    -9     2
   140.960   93   15.26     261    656    -9     7
   135.920   21   15.25     255    659    -9    13
   127.200   20   15.25     240    661    -9    13
   112.640   20   15.24     222    663    -9    13
   105.640   20   15.23     235    665    -9    13
     0.360  180   15.21     182    692    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   624  2.4   15    90   160   600    80    10   210      8     1    4020    5  0.2714
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  624 2023/01/18 05:47:37
# Pcase vacuums
   6.950   6.510
# Oil valve positions
   1.819   0.806   0.631
# Seek attempts and pump time
      1     -2
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  624 2023/01/18 07:10:50 9999/99/99 99:99:99
# Pcase vacuums
   6.510   6.510
# Average 
#   P      T      S
  133.960   2.438  33.599
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  624 2023/01/18 10:40:59 9999/99/99 99:99:99
# Pcase vacuums
   6.500   6.760
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  624 2023/01/18 11:07:30
# number of packets previous packets
      6               7
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.760   6.930
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.176   0.158   0.667
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.740   0.876
# Encountered Error Codes
#    Cycle Errors
EE20   624      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  624  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.400    430   6.390
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  624       197    2023/01/18 11:15:11
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      624	(cycle number) |
     MPS =        0	(phase) |
     MPC =      624	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      160	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    42104	(vac 0 inHg cal) |
     PV5 =    18698	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      623	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       46	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       10	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      312	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      623	(cycle) |
  PTO[0] =    12600	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        1	(0=dis 1=en) |
     PSD =      100	(dBar) |
     PST =     7200	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      623	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =       10	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      623	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      340	(sec) |
     SSP =      180	(sec) |

