# Message Decoder for Solo2, Alto and Deep Solo Floats; v230712
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11342   511   88 000000000881890  3412     0   6 2022/12/23, 15:17:04,  65.1906  -53.4889 681      2 14
SBD 11342   511  340 000000000881890  3413     0   5 2022/12/23, 15:17:16,  65.2943  -55.7389 178      2 13
SBD 11342   511  311 000000000881890  3414     0   4 2022/12/23, 15:17:29,  65.1838  -53.6084 681      2 8,14
SBD 11342   511  338 000000000881890  3415     0   3 2022/12/23, 15:17:44,  65.1768  -53.7278 197      2 3,0,15,13
SBD 11342   511  289 000000000881890  3416     0   2 2022/12/23, 15:17:58,  65.1961  -53.7967 180      2 2,6
SBD 11342   511  289 000000000881890  3417     0   1 2022/12/23, 15:18:17,  65.1650  -53.1130  38      0 11,7,5,14,1
SBD 11342   511  316 000000000881890  3418     0   0 2022/12/23, 15:18:50,  65.1714  -52.9935  29      0 0,4,14,14,9,10
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1887           84        1971
# Sensor_ID bytes     % packets Description
          0    48  2.54       2         GPS
          1    96  5.09       1    Pressure
          2   174  9.22       1 Temperature
          3   160  8.48       1    Salinity
          4   138  7.31       1        Fall
          5    83  4.40       1        Rise
          6   103  5.46       1        Pump
          7    35  1.85       1 Drift_times
          8   287 15.21       1     Surface
          9    54  2.86       1  HR/Drift_P
         10    44  2.33       1  HR/Drift_T
         11    44  2.33       1  HR/Drift_S
         13   404 21.41       2  Parameters
         14   151  8.00       5 Engineering
         15    66  3.50       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  511 1 2022/12/23 09:49 +65.15336  -52.93221    0  4 33 39 44  1.5     0
G  511 2 2022/12/23 15:11 +65.14677  -52.92686   80  5 35 39 46  1.7     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   511  144    1     3         0
#      Pres      Temp    Salinity
      1.040     0.2170    32.7270
      2.000     0.2530    32.7450
      3.000     0.2610    32.7420
      4.040     0.3260    32.7610
      5.040     0.3550    32.7640
      6.040     0.4130    32.7790
      7.040     0.5030    32.8010
      8.040     0.5350    32.8090
      9.040     0.7040    32.8400
     10.000     0.7720    32.8480
     11.000     0.8640    32.8660
     12.000     0.9140    32.8810
     13.000     0.9460    32.8880
     14.000     0.9860    32.8940
     15.000     1.0070    32.8980
     15.960     1.0110    32.9000
     16.960     1.0240    32.9040
     17.960     1.0500    32.9040
     18.960     1.0450    32.9100
     20.000     1.0360    32.9090
     21.040     1.0320    32.9100
     22.000     1.0290    32.9100
     23.000     1.0390    32.9130
     24.000     1.0430    32.9150
     25.000     1.0860    32.9170
     25.960     1.1430    32.9320
     27.000     1.1440    32.9320
     28.000     1.1720    32.9380
     29.040     1.1060    32.9290
     30.000     1.0990    32.9230
     31.000     1.1670    32.9360
     32.000     1.2060    32.9460
     33.000     1.2420    32.9590
     34.000     1.3000    32.9650
     35.000     1.3930    32.9850
     36.040     1.4410    32.9890
     37.000     1.5280    33.0080
     38.000     1.5800    33.0240
     39.000     1.6060    33.0380
     39.960     1.6650    33.0520
     41.040     1.6780    33.0590
     42.000     1.6820    33.0600
     43.000     1.6850    33.0620
     44.040     1.6910    33.0630
     45.040     1.6980    33.0630
     46.040     1.7430    33.0710
     47.040     1.7830    33.0790
     48.000     1.8090    33.0860
     48.960     1.8330    33.0910
     50.000     1.8600    33.0970
     51.000     1.8870    33.1030
     52.000     1.9130    33.1100
     53.000     1.8820    33.1420
     54.040     1.8990    33.1690
     55.040     1.9410    33.1810
     56.000     2.0500    33.2080
     57.000     2.0850    33.2190
     58.040     2.1320    33.2310
     59.000     2.1540    33.2370
     59.960     2.1700    33.2420
     61.000     2.1710    33.2450
     62.000     2.1450    33.2560
     62.960     2.1050    33.2740
     64.000     2.0940    33.2770
     65.000     2.0830    33.2820
     66.000     2.0800    33.2850
     67.000     2.0930    33.2950
     68.000     2.1050    33.2980
     69.000     2.1230    33.3070
     70.000     2.1580    33.3100
     71.040     2.1950    33.3160
     72.000     2.2300    33.3250
     72.960     2.2310    33.3260
     74.000     2.2350    33.3280
     75.040     2.2550    33.3310
     76.000     2.2820    33.3350
     76.960     2.3030    33.3380
     78.000     2.3140    33.3430
     79.000     2.3330    33.3460
     80.000     2.3890    33.3540
     81.040     2.3860    33.3580
     82.040     2.3810    33.3560
     83.040     2.4500    33.3640
     84.000     2.5360    33.3890
     85.000     2.5410    33.3960
     86.040     2.5560    33.3980
     87.000     2.6010    33.4100
     88.000     2.6480    33.4200
     89.000     2.6940    33.4350
     90.040     2.6980    33.4400
     91.000     2.7120    33.4460
     91.960     2.7260    33.4480
     92.960     2.7370    33.4550
     94.000     2.7400    33.4570
     95.000     2.7430    33.4590
     96.000     2.7460    33.4610
     96.960     2.7550    33.4660
     98.040     2.7730    33.4850
     99.040     2.7770    33.4930
    100.000     2.7760    33.5060
    101.040     2.7860    33.5140
    102.040     2.8110    33.5230
    103.040     2.8170    33.5280
    104.040     2.8260    33.5340
    105.040     2.8490    33.5420
    106.000     2.8730    33.5530
    107.000     2.8820    33.5590
    108.000     2.8830    33.5590
    109.000     2.8840    33.5590
    110.040     2.8850    33.5600
    111.040     2.8850    33.5600
    112.000     2.8840    33.5610
    113.040     2.8830    33.5600
    114.000     2.8840    33.5620
    115.040     2.9160    33.5740
    116.040     2.9430    33.5850
    116.960     2.9660    33.5960
    118.000     2.9700    33.6000
    119.040     2.9700    33.6020
    120.000     2.9720    33.6040
    121.000     2.9740    33.6040
    122.000     2.9810    33.6070
    123.000     2.9900    33.6100
    124.000     2.9970    33.6130
    125.000     3.0120    33.6210
    126.000     3.0250    33.6270
    127.000     3.0330    33.6290
    128.000     3.0380    33.6310
    129.000     3.0410    33.6340
    130.000     3.0420    33.6360
    130.960     3.0400    33.6350
    132.000     3.0420    33.6340
    133.000     3.0460    33.6360
    134.000     3.0450    33.6360
    135.000     3.0480    33.6370
    136.000     3.0500    33.6380
    137.000     3.0510    33.6380
    138.000     3.0480    33.6380
    139.040     3.0500    33.6390
    140.000     3.0590    33.6420
    141.000     3.0600    33.6430
    142.000     3.0660    33.6460
    143.040     3.0720    33.6490
    143.840  -999.0000  -999.0000
# Drift Data
#      C Npts comp n_var
I0   511   21    1     4
#      Pres       Temp   Salinity     Time
    138.640     3.1160    33.6710        5
    142.240     3.1130    33.6700      611
    143.680     3.0900    33.6600     1216
    141.960     3.0690    33.6460     1822
    147.800     3.0530    33.6430     2427
    151.720     3.0430    33.6390     3033
    154.040     3.0370    33.6340     3638
    150.760     3.0380    33.6350     4244
    149.240     3.0340    33.6310     4849
    152.760     3.0300    33.6300     5455
    154.480     3.0330    33.6300     6060
    155.600     3.0260    33.6270     6666
    149.960     3.0260    33.6280     7271
    145.080     3.0570    33.6400     7876
    145.040     3.0380    33.6350     8482
    147.160     3.1000    33.6680     9087
    150.160     3.0630    33.6480     9693
    155.240     3.0590    33.6430    10298
    148.680     3.0410    33.6360    10904
    147.040     3.0450    33.6370    11509
    145.920     3.0520    33.6370    12115
# Surface Pressure Time Series
#    C npts n_var               Start
S  511    31     6 2022/12/23 15:09:56
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    72      0.550      0.124   -999.000      14 2022/12/23 15:11:08
    85      0.530      0.143   -999.000      14 2022/12/23 15:11:21
    95      0.500      0.066   -999.000      14 2022/12/23 15:11:31
   106      0.440     -0.009   -999.000      14 2022/12/23 15:11:42
   118      0.590      0.047   -999.000      14 2022/12/23 15:11:54
   129      0.450      0.065   -999.000      14 2022/12/23 15:12:05
   140      0.490      0.024   -999.000      14 2022/12/23 15:12:16
   151      0.510      0.041   -999.000      14 2022/12/23 15:12:27
   162      0.480     -0.062   -999.000      14 2022/12/23 15:12:38
   173      0.430     -0.148   -999.000      14 2022/12/23 15:12:49
   184      0.450     -0.068   -999.000      14 2022/12/23 15:13:00
   195      0.510     -0.190   -999.000      14 2022/12/23 15:13:11
   206      0.480     -0.067   -999.000      14 2022/12/23 15:13:22
   217      0.490     -0.387   -999.000      14 2022/12/23 15:13:33
   228      0.430     -0.813   -999.000      14 2022/12/23 15:13:44
   240      0.460     -0.178   -999.000      14 2022/12/23 15:13:56
   251      0.490     -0.029   -999.000      14 2022/12/23 15:14:07
   262      0.470      0.027   -999.000      14 2022/12/23 15:14:18
   273      0.440      0.121   -999.000      14 2022/12/23 15:14:29
   284      0.500      0.049   -999.000      14 2022/12/23 15:14:40
   295      0.480      0.021   -999.000      14 2022/12/23 15:14:51
   306      0.490      0.046   -999.000      14 2022/12/23 15:15:02
   317      0.450     -0.052   -999.000      14 2022/12/23 15:15:13
   328      0.470     -0.011   -999.000      14 2022/12/23 15:15:24
   339      0.500     -0.196   -999.000      14 2022/12/23 15:15:35
   350      0.460     -0.031   -999.000      14 2022/12/23 15:15:46
   362      0.480     -0.047   -999.000      14 2022/12/23 15:15:58
   373      0.490     -0.039   -999.000      14 2022/12/23 15:16:09
   384      0.450      0.033   -999.000      14 2022/12/23 15:16:20
   395      0.460     -0.346   -999.000      14 2022/12/23 15:16:31
   406      0.440     -0.425   -999.000      14 2022/12/23 15:16:42
# Fall Time Series
#  Cycle npts     Start                Date
F00  511   26 725104192 2022/12/23 09:49:52
# time     Pres phase                Date
     0    0.520     1 2022/12/23 09:49:52
   174   10.400    10 2022/12/23 09:52:46
   239   20.080    10 2022/12/23 09:53:51
   304   30.160    10 2022/12/23 09:54:56
   369   40.160    10 2022/12/23 09:56:01
   415   47.080     2 2022/12/23 09:56:47
   437   50.040    10 2022/12/23 09:57:09
   515   60.320    10 2022/12/23 09:58:27
   602   70.040    10 2022/12/23 09:59:54
   744   80.080    10 2022/12/23 10:02:16
  1011   90.080    10 2022/12/23 10:06:43
  1306  100.040    10 2022/12/23 10:11:38
  1606  109.200    10 2022/12/23 10:16:38
  1906  115.960    10 2022/12/23 10:21:38
  2206  122.960    10 2022/12/23 10:26:38
  2506  127.240    10 2022/12/23 10:31:38
  2806  128.560    10 2022/12/23 10:36:38
  3107  131.600    10 2022/12/23 10:41:39
  3407  134.920    10 2022/12/23 10:46:39
  3707  136.640    10 2022/12/23 10:51:39
  4007  138.000    10 2022/12/23 10:56:39
  4307  137.840    10 2022/12/23 11:01:39
  4323  137.840     3 2022/12/23 11:01:55
  4607  137.720    10 2022/12/23 11:06:39
  4907  138.240    10 2022/12/23 11:11:39
  4960  138.600    10 2022/12/23 11:12:32
# Rise Time Series
#  Cycle npts     Start                Date
R00  511   15 725121762 2022/12/23 14:42:42
# time     Pres phase                Date
     0  144.080     7 2022/12/23 14:42:42
   175  134.280    13 2022/12/23 14:45:37
   298  123.520    13 2022/12/23 14:47:40
   421  112.600    13 2022/12/23 14:49:43
   524  102.880    13 2022/12/23 14:51:26
   626   91.760    13 2022/12/23 14:53:08
   729   81.560    13 2022/12/23 14:54:51
   831   71.560    13 2022/12/23 14:56:33
   954   60.280    13 2022/12/23 14:58:36
  1057   50.920    13 2022/12/23 15:00:19
  1180   39.320    13 2022/12/23 15:02:22
  1303   28.320    13 2022/12/23 15:04:25
  1426   17.040    13 2022/12/23 15:06:28
  1528    7.160    13 2022/12/23 15:08:10
  1631    0.560     8 2022/12/23 15:09:53
# Pump Time Series
#   Cycle npts
B02   511    9
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    47.080  305   15.16     222    654    -9     2
   144.080   93   15.28     239    653    -9     7
   137.520   21   15.27     230    655    -9    13
   129.000   20   15.27     233    657    -9    13
   110.960   20   15.26     231    659    -9    13
    50.920   20   15.25     213    664    -9    13
    35.640   20   15.25     219    668    -9    13
    28.320   20   15.24     214    671    -9    13
     0.480  180   15.21     178    691    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   511  2.4   15    90   160   600    80    10   210      8     1    4020    5  0.2714
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  511 2022/12/23 09:49:49
# Pcase vacuums
   6.860   6.470
# Oil valve positions
   1.816   0.807   0.631
# Seek attempts and pump time
      1     -2
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  511 2022/12/23 11:12:33 9999/99/99 99:99:99
# Pcase vacuums
   6.470   6.470
# Average 
#   P      T      S
  148.440   3.055  33.642
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  511 2022/12/23 14:42:42 9999/99/99 99:99:99
# Pcase vacuums
   6.460   6.750
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  511 2022/12/23 15:09:54
# number of packets previous packets
      6               8
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.750   6.920
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.176   0.158   0.667
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.740   0.876
# Encountered Error Codes
#    Cycle Errors
EE20   511      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  511  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.440    411   6.320
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  511       197    2022/12/23 15:17:44
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      511	(cycle number) |
     MPS =        0	(phase) |
     MPC =      511	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      160	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    42104	(vac 0 inHg cal) |
     PV5 =    18698	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      510	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       46	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       10	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      303	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      510	(cycle) |
  PTO[0] =    12600	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        1	(0=dis 1=en) |
     PSD =      100	(dBar) |
     PST =     7200	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      510	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =       10	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      510	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      340	(sec) |
     SSP =      180	(sec) |

