# Message Decoder for Solo2, Alto and Deep Solo Floats; v230712
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11342   508   88 000000000881890  3393     0   6 2022/12/22, 22:52:01,  65.1778  -52.8739   3      0 14
SBD 11342   508  340 000000000881890  3394     0   5 2022/12/22, 22:52:12,  65.1778  -52.8739   3      0 13
SBD 11342   508  269 000000000881890  3395     0   4 2022/12/22, 22:52:23,  65.1778  -52.8739   4      0 15,13,6,14
SBD 11342   508  299 000000000881890  3396     0   3 2022/12/22, 22:52:35,  65.1778  -52.8739   4      0 8
SBD 11342   508  338 000000000881890  3397     0   2 2022/12/22, 22:52:46,  65.1974  -52.9421   4      0 2,3
SBD 11342   508  322 000000000881890  3398     0   1 2022/12/22, 22:53:44,  65.1367  -52.9254   5      0 11,7,5,14,1,0
SBD 11342   508  315 000000000881890  3399     0   0 2022/12/22, 22:53:57,  65.1974  -52.9421   8      0 0,4,14,14,9,10
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1887           84        1971
# Sensor_ID bytes     % packets Description
          0    48  2.54       2         GPS
          1   100  5.30       1    Pressure
          2   179  9.49       1 Temperature
          3   147  7.79       1    Salinity
          4   138  7.31       1        Fall
          5    88  4.66       1        Rise
          6   103  5.46       1        Pump
          7    35  1.85       1 Drift_times
          8   287 15.21       1     Surface
          9    54  2.86       1  HR/Drift_P
         10    43  2.28       1  HR/Drift_T
         11    44  2.33       1  HR/Drift_S
         13   404 21.41       2  Parameters
         14   151  8.00       5 Engineering
         15    66  3.50       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  508 1 2022/12/22 17:24 +65.11347  -52.88621   40  5 31 38 48  3.0     0
G  508 2 2022/12/22 22:46 +65.13412  -52.90013   90  7 19 35 48  1.5     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   508  149    1     3         0
#      Pres      Temp    Salinity
      1.080     0.9210    32.7690
      2.000     0.9280    32.7680
      3.040     0.9490    32.7710
      4.040     0.9690    32.7720
      5.000     0.9860    32.7780
      6.000     1.0080    32.7870
      7.000     1.0380    32.8030
      8.000     1.1080    32.8220
      9.000     1.1650    32.8720
     10.040     1.1370    32.8830
     11.000     1.1410    32.8870
     12.000     1.1330    32.8860
     13.040     1.0950    32.8940
     14.000     1.0610    32.9010
     15.040     1.0370    32.9020
     16.040     0.9930    32.9130
     17.040     0.9830    32.9190
     18.000     0.9810    32.9210
     19.000     0.9790    32.9220
     20.000     0.9790    32.9240
     21.000     0.9850    32.9280
     21.960     1.0070    32.9300
     22.960     1.1050    32.9380
     24.000     1.1320    32.9590
     25.000     1.1320    32.9600
     25.960     1.1310    32.9610
     27.000     1.1370    32.9600
     28.040     1.1660    32.9660
     29.000     1.1830    32.9660
     29.960     1.2290    32.9760
     31.000     1.2620    32.9840
     32.040     1.2810    32.9900
     33.000     1.2800    33.0050
     33.960     1.2940    33.0170
     34.960     1.5190    33.0390
     35.960     1.8350    33.0590
     36.960     2.1160    33.1120
     38.000     2.1360    33.1270
     39.000     2.1650    33.1430
     40.000     2.1870    33.1500
     41.040     2.2170    33.1660
     42.040     2.2380    33.1760
     43.000     2.2530    33.1820
     44.000     2.2730    33.1880
     45.000     2.3020    33.1980
     46.000     2.3150    33.2050
     47.000     2.3320    33.2090
     48.000     2.3420    33.2140
     49.000     2.3520    33.2170
     50.040     2.3640    33.2210
     51.040     2.3700    33.2240
     52.000     2.3760    33.2270
     53.000     2.3850    33.2310
     54.040     2.4050    33.2380
     55.000     2.4830    33.2760
     55.960     2.5050    33.2900
     56.960     2.5210    33.3060
     58.000     2.5330    33.3160
     59.040     2.5490    33.3210
     60.040     2.5650    33.3290
     61.040     2.5680    33.3320
     61.960     2.5820    33.3350
     62.960     2.5930    33.3420
     64.000     2.5600    33.3470
     65.000     2.5180    33.3530
     66.040     2.5260    33.3650
     67.040     2.5120    33.3730
     68.000     2.4930    33.3740
     69.000     2.4930    33.3750
     70.000     2.5060    33.3870
     71.000     2.5210    33.4010
     72.040     2.5220    33.4060
     73.040     2.5270    33.4070
     74.000     2.5300    33.4110
     75.000     2.5310    33.4140
     76.040     2.5480    33.4210
     77.040     2.6060    33.4390
     78.000     2.6310    33.4530
     79.000     2.6640    33.4640
     80.040     2.7500    33.4930
     81.040     2.7730    33.5030
     82.000     2.7820    33.5070
     83.040     2.7890    33.5100
     84.040     2.7950    33.5120
     85.040     2.8360    33.5260
     86.000     2.8540    33.5370
     87.000     2.8770    33.5440
     88.000     2.8820    33.5480
     88.960     2.8830    33.5480
     90.000     2.8820    33.5490
     91.040     2.8820    33.5490
     92.000     2.8820    33.5500
     93.000     2.8850    33.5520
     94.040     2.8870    33.5520
     95.000     2.8910    33.5540
     96.000     2.8990    33.5570
     97.000     2.9050    33.5610
     98.000     2.9070    33.5620
     99.000     2.9160    33.5650
    100.040     2.9220    33.5690
    101.040     2.9260    33.5690
    102.040     2.9290    33.5760
    103.000     2.9280    33.5760
    104.000     2.9240    33.5790
    105.000     2.9240    33.5810
    106.000     2.9240    33.5800
    106.960     2.9250    33.5800
    108.000     2.9260    33.5820
    109.040     2.9260    33.5830
    110.000     2.9250    33.5830
    111.000     2.9260    33.5830
    112.000     2.9370    33.5860
    113.000     2.9430    33.5890
    114.040     2.9430    33.5890
    115.040     2.9430    33.5900
    115.960     2.9430    33.5900
    116.960     2.9450    33.5900
    117.960     2.9480    33.5910
    119.000     2.9620    33.5980
    120.000     2.9840    33.6060
    121.000     2.9890    33.6080
    122.000     2.9990    33.6130
    123.000     3.0090    33.6180
    124.000     3.0140    33.6210
    125.000     3.0200    33.6240
    126.040     3.0390    33.6320
    127.000     3.0420    33.6350
    128.000     3.0420    33.6350
    129.000     3.0440    33.6360
    130.000     3.0470    33.6380
    131.000     3.0520    33.6410
    132.040     3.0550    33.6420
    133.040     3.0600    33.6440
    134.000     3.0650    33.6470
    135.000     3.0740    33.6500
    136.000     3.0800    33.6540
    137.000     3.0830    33.6550
    137.960     3.0950    33.6590
    139.000     3.1120    33.6660
    140.000     3.1230    33.6730
    141.000     3.1310    33.6760
    142.000     3.1390    33.6800
    143.000     3.1420    33.6800
    144.000     3.1480    33.6830
    145.040     3.1520    33.6850
    146.040     3.1550    33.6860
    147.040     3.1570    33.6880
    148.160     3.1580    33.6880
    148.520     3.1600    33.6890
# Drift Data
#      C Npts comp n_var
I0   508   21    1     4
#      Pres       Temp   Salinity     Time
    152.440     3.1300    33.6720        6
    153.120     3.1310    33.6750      612
    154.320     3.1270    33.6750     1217
    154.200     3.1290    33.6740     1822
    152.600     3.1290    33.6750     2428
    153.920     3.1280    33.6740     3033
    153.200     3.1270    33.6750     3639
    157.720     3.1180    33.6690     4244
    149.960     3.1140    33.6700     4850
    146.280     3.1240    33.6710     5455
    150.360     3.1310    33.6740     6061
    147.280     3.1390    33.6790     6666
    146.600     3.1450    33.6820     7272
    137.640     3.1490    33.6810     7877
    141.520     3.1770    33.6940     8483
    151.160     3.1550    33.6860     9088
    144.640     3.1560    33.6850     9693
    144.960     3.1580    33.6900    10299
    149.320     3.1610    33.6870    10904
    147.640     3.1620    33.6900    11510
    149.840     3.1610    33.6890    12115
# Surface Pressure Time Series
#    C npts n_var               Start
S  508    31     6 2022/12/22 22:44:58
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    73      0.480      0.850   -999.000      14 2022/12/22 22:46:11
    85      0.520      0.800   -999.000      14 2022/12/22 22:46:23
    96      0.430      0.728   -999.000      14 2022/12/22 22:46:34
   107      0.470      0.767   -999.000      14 2022/12/22 22:46:45
   119      0.390      0.768   -999.000      14 2022/12/22 22:46:57
   130      0.440      0.739   -999.000      14 2022/12/22 22:47:08
   141      0.480      0.581   -999.000      14 2022/12/22 22:47:19
   152      0.420      0.722   -999.000      14 2022/12/22 22:47:30
   163      0.410      0.576   -999.000      14 2022/12/22 22:47:41
   174      0.480      0.267   -999.000      14 2022/12/22 22:47:52
   185      0.460      0.398   -999.000      14 2022/12/22 22:48:03
   196      0.490      0.295   -999.000      14 2022/12/22 22:48:14
   207      0.410      0.598   -999.000      14 2022/12/22 22:48:25
   218      0.490      0.647   -999.000      14 2022/12/22 22:48:36
   229      0.470      0.699   -999.000      14 2022/12/22 22:48:47
   240      0.420      0.407   -999.000      14 2022/12/22 22:48:58
   252      0.430     -0.326   -999.000      14 2022/12/22 22:49:10
   263      0.430      0.394   -999.000      14 2022/12/22 22:49:21
   274      0.410      0.286   -999.000      14 2022/12/22 22:49:32
   285      0.400      0.713   -999.000      14 2022/12/22 22:49:43
   296      0.380      0.603   -999.000      14 2022/12/22 22:49:54
   307      0.460      0.147   -999.000      14 2022/12/22 22:50:05
   318      0.460      0.504   -999.000      14 2022/12/22 22:50:16
   329      0.430      0.650   -999.000      14 2022/12/22 22:50:27
   340      0.430      0.519   -999.000      14 2022/12/22 22:50:38
   351      0.440      0.307   -999.000      14 2022/12/22 22:50:49
   362      0.450     -0.208   -999.000      14 2022/12/22 22:51:00
   374      0.450     -0.840   -999.000      14 2022/12/22 22:51:12
   385      0.480     -1.449   -999.000      14 2022/12/22 22:51:23
   396      0.420     -0.629   -999.000      14 2022/12/22 22:51:34
   407      0.410      0.413   -999.000      14 2022/12/22 22:51:45
# Fall Time Series
#  Cycle npts     Start                Date
F00  508   26 725045048 2022/12/22 17:24:08
# time     Pres phase                Date
     0    0.480     1 2022/12/22 17:24:08
   177   10.400    10 2022/12/22 17:27:05
   242   20.080    10 2022/12/22 17:28:10
   307   30.160    10 2022/12/22 17:29:15
   375   40.280    10 2022/12/22 17:30:23
   422   47.040     2 2022/12/22 17:31:10
   444   50.160    10 2022/12/22 17:31:32
   521   60.320    10 2022/12/22 17:32:49
   616   70.280    10 2022/12/22 17:34:24
   767   80.040    10 2022/12/22 17:36:55
  1022   90.000    10 2022/12/22 17:41:10
  1268  100.040    10 2022/12/22 17:45:16
  1568  108.920    10 2022/12/22 17:50:16
  1868  116.320    10 2022/12/22 17:55:16
  2168  123.840    10 2022/12/22 18:00:16
  2468  132.640    10 2022/12/22 18:05:16
  2769  140.840    10 2022/12/22 18:10:17
  3069  143.960    10 2022/12/22 18:15:17
  3369  144.240    10 2022/12/22 18:20:17
  3669  146.680    10 2022/12/22 18:25:17
  3969  152.040    10 2022/12/22 18:30:17
  4269  154.600    10 2022/12/22 18:35:17
  4330  154.640     3 2022/12/22 18:36:18
  4569  154.000    10 2022/12/22 18:40:17
  4869  152.520    10 2022/12/22 18:45:17
  4967  152.440    10 2022/12/22 18:46:55
# Rise Time Series
#  Cycle npts     Start                Date
R00  508   16 725062624 2022/12/22 22:17:04
# time     Pres phase                Date
     0  148.600     7 2022/12/22 22:17:04
   196  137.320    13 2022/12/22 22:20:20
   299  127.920    13 2022/12/22 22:22:03
   422  116.200    13 2022/12/22 22:24:06
   524  106.680    13 2022/12/22 22:25:48
   647   95.440    13 2022/12/22 22:27:51
   750   85.760    13 2022/12/22 22:29:34
   873   75.400    13 2022/12/22 22:31:37
   996   64.480    13 2022/12/22 22:33:40
  1119   53.200    13 2022/12/22 22:35:43
  1221   43.440    13 2022/12/22 22:37:25
  1324   33.680    13 2022/12/22 22:39:08
  1447   22.120    13 2022/12/22 22:41:11
  1549   12.160    13 2022/12/22 22:42:53
  1652    2.760    13 2022/12/22 22:44:36
  1673    0.840     8 2022/12/22 22:44:57
# Pump Time Series
#   Cycle npts
B02   508    9
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    47.040  305   15.16     229    655    -9     2
   148.600   94   15.26     250    652    -9     7
   142.560   20   15.26     245    654    -9    13
   133.720   21   15.26     266    656    -9    13
    80.560   21   15.26     232    659    -9    13
    66.320   21   15.25     240    662    -9    13
    51.480   21   15.25     225    664    -9    13
     4.640   21   15.24     210    672    -9    13
     0.400  180   15.22     180    688    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   508  2.4   15    90   160   600    80    10   210      8     1    4020    5  0.2714
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  508 2022/12/22 17:24:05
# Pcase vacuums
   6.860   6.470
# Oil valve positions
   1.816   0.808   0.631
# Seek attempts and pump time
      1      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  508 2022/12/22 18:46:55 9999/99/99 99:99:99
# Pcase vacuums
   6.460   6.460
# Average 
#   P      T      S
  149.440   3.140  33.679
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  508 2022/12/22 22:17:04 9999/99/99 99:99:99
# Pcase vacuums
   6.460   6.720
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  508 2022/12/22 22:44:57
# number of packets previous packets
      6               7
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.720   6.880
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.176   0.158   0.667
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.740   0.876
# Encountered Error Codes
#    Cycle Errors
EE20   508      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  508  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.480    417   6.310
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  508       197    2022/12/22 22:52:23
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      508	(cycle number) |
     MPS =        0	(phase) |
     MPC =      508	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      160	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    42104	(vac 0 inHg cal) |
     PV5 =    18698	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      507	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       46	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       10	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      303	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      507	(cycle) |
  PTO[0] =    12600	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        1	(0=dis 1=en) |
     PSD =      100	(dBar) |
     PST =     7200	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      507	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =       10	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      507	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      340	(sec) |
     SSP =      180	(sec) |

