# Message Decoder for Solo2, Alto and Deep Solo Floats; v230712
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11342   464  257 000000000881890  3119     0   5 2022/12/12, 23:12:38,  64.8851  -56.7317 176      2 13,6,14,14
SBD 11342   464  340 000000000881890  3120     0   4 2022/12/12, 23:12:50,  64.8579  -56.7780 183      2 13
SBD 11342   464  299 000000000881890  3121     0   3 2022/12/12, 23:13:06,  64.8757  -56.8487 181      2 8
SBD 11342   464  278 000000000881890  3122     0   2 2022/12/12, 23:13:27,  65.0176  -53.1301   6      0 2,3,0
SBD 11342   464  324 000000000881890  3123     0   1 2022/12/12, 23:13:37,  65.0176  -53.1301   6      0 11,7,5,14,1,15
SBD 11342   464  306 000000000881890  3124     0   0 2022/12/12, 23:13:48,  65.0371  -53.1981   6      0 0,4,14,14,9,10
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1732           72        1804
# Sensor_ID bytes     % packets Description
          0    48  2.77       2         GPS
          1    84  4.85       1    Pressure
          2   142  8.20       1 Temperature
          3   100  5.77       1    Salinity
          4   138  7.97       1        Fall
          5    73  4.21       1        Rise
          6    81  4.68       1        Pump
          7    35  2.02       1 Drift_times
          8   287 16.57       1     Surface
          9    44  2.54       1  HR/Drift_P
         10    44  2.54       1  HR/Drift_T
         11    35  2.02       1  HR/Drift_S
         13   404 23.33       2  Parameters
         14   151  8.72       5 Engineering
         15    66  3.81       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  464 1 2022/12/12 17:49 +64.99004  -53.10825    0  4 28 32 38  3.9     0
G  464 2 2022/12/12 23:06 +65.00569  -53.12703   60  4 27 32 40  7.1     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   464  120    1     3         0
#      Pres      Temp    Salinity
      1.080     2.1160    33.2650
      1.960     2.1150    33.2660
      3.040     2.1080    33.2660
      4.000     2.1240    33.2660
      5.000     2.1600    33.2750
      5.960     2.1740    33.2760
      7.040     2.1950    33.2820
      8.000     2.2320    33.2810
      9.000     2.2410    33.2900
     10.000     2.3570    33.3010
     11.000     2.3630    33.3080
     12.000     2.3630    33.3090
     13.080     2.3680    33.3110
     14.000     2.3790    33.3130
     15.000     2.3840    33.3170
     16.000     2.3920    33.3240
     17.000     2.3890    33.3330
     18.000     2.3850    33.3380
     18.960     2.4540    33.3680
     19.960     2.4710    33.3850
     21.040     2.5250    33.4080
     22.000     2.5340    33.4140
     22.960     2.5680    33.4260
     24.000     2.6150    33.4460
     25.040     2.6250    33.4480
     26.000     2.6210    33.4480
     27.000     2.6310    33.4500
     28.000     2.6570    33.4550
     29.000     2.6670    33.4590
     30.040     2.6860    33.4670
     31.040     2.6830    33.4720
     32.000     2.6780    33.4760
     32.960     2.6910    33.4800
     34.000     2.6950    33.4830
     35.040     2.7040    33.4850
     36.040     2.6980    33.4920
     37.000     2.6960    33.4930
     37.960     2.7140    33.5010
     38.960     2.7150    33.5060
     39.960     2.7200    33.5080
     41.040     2.7260    33.5110
     42.000     2.7370    33.5160
     43.000     2.7410    33.5180
     44.000     2.7460    33.5210
     45.040     2.7600    33.5240
     46.000     2.7700    33.5310
     47.000     2.7750    33.5340
     48.000     2.7840    33.5370
     49.000     2.7910    33.5400
     50.040     2.8030    33.5450
     51.000     2.8060    33.5480
     52.000     2.8220    33.5540
     53.000     2.8310    33.5570
     54.000     2.8430    33.5620
     54.960     2.8470    33.5630
     56.000     2.8580    33.5650
     57.000     2.8660    33.5690
     58.000     2.8760    33.5710
     58.960     2.8930    33.5740
     59.960     2.9030    33.5760
     60.960     2.9060    33.5780
     61.960     2.9200    33.5800
     62.960     2.9390    33.5840
     63.960     2.9510    33.5880
     64.960     2.9590    33.5900
     66.000     2.9690    33.5930
     67.000     2.9740    33.5940
     68.000     2.9780    33.5960
     69.000     2.9850    33.5960
     70.000     2.9890    33.5980
     71.000     2.9890    33.5990
     72.040     2.9920    33.5990
     73.040     2.9930    33.6000
     74.000     2.9960    33.6000
     75.000     3.0000    33.6010
     76.040     3.0100    33.6030
     77.040     3.0120    33.6040
     78.000     3.0130    33.6050
     79.040     3.0170    33.6060
     80.000     3.0190    33.6070
     80.960     3.0190    33.6070
     82.000     3.0160    33.6080
     83.000     3.0120    33.6100
     84.040     3.0120    33.6120
     85.040     3.0120    33.6130
     86.000     3.0080    33.6130
     87.000     3.0010    33.6130
     88.000     3.0080    33.6150
     89.040     3.0110    33.6160
     90.040     3.0110    33.6170
     91.000     3.0140    33.6170
     92.000     3.0190    33.6180
     93.000     3.0320    33.6210
     94.040     3.0340    33.6230
     95.000     3.0380    33.6250
     96.040     3.0430    33.6270
     97.040     3.0420    33.6270
     98.000     3.0430    33.6280
     98.960     3.0470    33.6290
    100.000     3.0550    33.6300
    101.000     3.0780    33.6350
    102.000     3.0840    33.6360
    103.000     3.0880    33.6380
    104.000     3.0900    33.6390
    105.000     3.0930    33.6390
    106.040     3.1010    33.6420
    107.000     3.1030    33.6440
    108.000     3.1090    33.6460
    109.040     3.1100    33.6480
    110.000     3.1100    33.6490
    111.000     3.1130    33.6500
    112.000     3.1170    33.6510
    113.040     3.1300    33.6540
    114.040     3.1470    33.6590
    115.040     3.1560    33.6640
    116.040     3.1590    33.6670
    117.000     3.1610    33.6690
    118.000     3.1620    33.6690
    119.000     3.1630    33.6690
    119.880  -999.0000  -999.0000
# Drift Data
#      C Npts comp n_var
I0   464   21    1     4
#      Pres       Temp   Salinity     Time
    121.520     3.1620    33.6670        6
    119.840     3.1810    33.6740      611
    121.840     3.1840    33.6740     1216
    123.000     3.1790    33.6780     1822
    122.120     3.1820    33.6770     2427
    121.960     3.1780    33.6780     3033
    123.000     3.1810    33.6770     3638
    119.120     3.1830    33.6770     4244
    117.520     3.1840    33.6770     4849
    117.560     3.1770    33.6790     5455
    117.520     3.1780    33.6800     6060
    120.840     3.1800    33.6810     6666
    122.880     3.1850    33.6780     7271
    122.160     3.1760    33.6740     7877
    121.600     3.1690    33.6720     8482
    122.160     3.1620    33.6700     9087
    123.880     3.1570    33.6680     9693
    122.480     3.1470    33.6650    10298
    122.360     3.1620    33.6680    10904
    123.840     3.1630    33.6680    11509
    122.160     3.1620    33.6680    12115
# Surface Pressure Time Series
#    C npts n_var               Start
S  464    31     6 2022/12/12 23:05:30
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    70      0.530      2.039   -999.000      14 2022/12/12 23:06:40
    83      0.560      1.952   -999.000      14 2022/12/12 23:06:53
    94      0.550      1.895   -999.000      14 2022/12/12 23:07:04
   105      0.580      1.946   -999.000      14 2022/12/12 23:07:15
   116      0.550      1.911   -999.000      14 2022/12/12 23:07:26
   127      0.620      1.897   -999.000      14 2022/12/12 23:07:37
   138      0.550      1.877   -999.000      14 2022/12/12 23:07:48
   149      0.550      1.961   -999.000      14 2022/12/12 23:07:59
   160      0.510      1.807   -999.000      14 2022/12/12 23:08:10
   172      0.550      1.706   -999.000      14 2022/12/12 23:08:22
   183      0.490      1.343   -999.000      14 2022/12/12 23:08:33
   194      0.500      1.081   -999.000      14 2022/12/12 23:08:44
   205      0.560      0.563   -999.000      14 2022/12/12 23:08:55
   216      0.550      1.696   -999.000      14 2022/12/12 23:09:06
   227      0.540      1.765   -999.000      14 2022/12/12 23:09:17
   238      0.580      1.862   -999.000      14 2022/12/12 23:09:28
   249      0.580      1.564   -999.000      14 2022/12/12 23:09:39
   260      0.510      1.815   -999.000      14 2022/12/12 23:09:50
   271      0.490      1.846   -999.000      14 2022/12/12 23:10:01
   282      0.520      1.699   -999.000      14 2022/12/12 23:10:12
   294      0.600      1.864   -999.000      14 2022/12/12 23:10:24
   305      0.470      1.939   -999.000      14 2022/12/12 23:10:35
   316      0.550      1.875   -999.000      14 2022/12/12 23:10:46
   327      0.440      1.210   -999.000      14 2022/12/12 23:10:57
   338      0.540      1.428   -999.000      14 2022/12/12 23:11:08
   349      0.570      1.378   -999.000      14 2022/12/12 23:11:19
   360      0.570      1.765   -999.000      14 2022/12/12 23:11:30
   371      0.550      1.839   -999.000      14 2022/12/12 23:11:41
   382      0.540      1.283   -999.000      14 2022/12/12 23:11:52
   393      0.510      1.847   -999.000      14 2022/12/12 23:12:03
   404      0.520      1.665   -999.000      14 2022/12/12 23:12:14
# Fall Time Series
#  Cycle npts     Start                Date
F00  464   26 724182580 2022/12/12 17:49:40
# time     Pres phase                Date
     0    0.520     1 2022/12/12 17:49:40
   168   10.400    10 2022/12/12 17:52:28
   239   20.400    10 2022/12/12 17:53:39
   307   30.120    10 2022/12/12 17:54:47
   375   40.320    10 2022/12/12 17:55:55
   422   47.160     2 2022/12/12 17:56:42
   444   50.440    10 2022/12/12 17:57:04
   515   60.080    10 2022/12/12 17:58:15
   599   70.160    10 2022/12/12 17:59:39
   716   80.000    10 2022/12/12 18:01:36
  1017   88.240    10 2022/12/12 18:06:37
  1317   93.680    10 2022/12/12 18:11:37
  1617   97.680    10 2022/12/12 18:16:37
  1917  102.200    10 2022/12/12 18:21:37
  2217  107.920    10 2022/12/12 18:26:37
  2517  115.080    10 2022/12/12 18:31:37
  2817  121.400    10 2022/12/12 18:36:37
  3117  123.880    10 2022/12/12 18:41:37
  3417  123.160    10 2022/12/12 18:46:37
  3717  122.880    10 2022/12/12 18:51:37
  4017  123.520    10 2022/12/12 18:56:37
  4317  123.840    10 2022/12/12 19:01:37
  4336  123.800     3 2022/12/12 19:01:56
  4617  123.120    10 2022/12/12 19:06:37
  4917  121.800    10 2022/12/12 19:11:37
  4973  121.560    10 2022/12/12 19:12:33
# Rise Time Series
#  Cycle npts     Start                Date
R00  464   13 724200163 2022/12/12 22:42:43
# time     Pres phase                Date
     0  120.120     7 2022/12/12 22:42:43
   175  109.840    13 2022/12/12 22:45:38
   298   98.200    13 2022/12/12 22:47:41
   401   88.720    13 2022/12/12 22:49:24
   524   77.360    13 2022/12/12 22:51:27
   626   67.720    13 2022/12/12 22:53:09
   749   56.560    13 2022/12/12 22:55:12
   872   45.480    13 2022/12/12 22:57:15
   975   35.880    13 2022/12/12 22:58:58
  1077   25.840    13 2022/12/12 23:00:40
  1200   14.160    13 2022/12/12 23:02:43
  1303    3.920    13 2022/12/12 23:04:26
  1365    0.560     8 2022/12/12 23:05:28
# Pump Time Series
#   Cycle npts
B02   464    7
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    47.160  311   15.18     203    651    -9     2
   120.120   93   15.28     242    652    -9     7
   113.520   21   15.28     233    655    -9    13
    51.240   20   15.27     220    658    -9    13
    35.880   20   15.27     212    660    -9    13
    16.080   20   15.26     206    663    -9    13
     0.480  180   15.23     179    680    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   464  2.4   15    90   160   600    80    10   210      8     1    4020    5  0.2714
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  464 2022/12/12 17:49:37
# Pcase vacuums
   6.780   6.470
# Oil valve positions
   1.815   0.809   0.631
# Seek attempts and pump time
      1     -3
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  464 2022/12/12 19:12:34 9999/99/99 99:99:99
# Pcase vacuums
   6.460   6.470
# Average 
#   P      T      S
  121.400   3.173  33.674
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  464 2022/12/12 22:42:43 9999/99/99 99:99:99
# Pcase vacuums
   6.460   6.650
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  464 2022/12/12 23:05:28
# number of packets previous packets
      6               6
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.640   6.810
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.176   0.158   0.667
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.740   0.876
# Encountered Error Codes
#    Cycle Errors
EE20   464      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  464  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.520    417   6.260
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  464       197    2022/12/12 23:12:50
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      464	(cycle number) |
     MPS =        0	(phase) |
     MPC =      464	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      160	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    42104	(vac 0 inHg cal) |
     PV5 =    18698	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      463	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       46	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       10	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      309	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      463	(cycle) |
  PTO[0] =    12600	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        1	(0=dis 1=en) |
     PSD =      100	(dBar) |
     PST =     7200	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      463	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =       10	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      463	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      340	(sec) |
     SSP =      180	(sec) |

