# Message Decoder for Solo2, Alto and Deep Solo Floats; v230712
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11342   377  236 000000000881890  2543     0   6 2022/11/22, 13:35:35,  64.4229  -52.7149   6      0 6,14,14
SBD 11342   377  340 000000000881890  2544     0   5 2022/11/22, 13:35:53,  64.4425  -52.7811   5      0 13
SBD 11342   377  299 000000000881890  2545     0   4 2022/11/22, 13:36:11,  64.4229  -52.7149   5      0 8
SBD 11342   377  276 000000000881890  2546     0   3 2022/11/22, 13:36:22,  64.4229  -52.7149   5      0 3,0,15
SBD 11342   377  270 000000000881890  2547     0   2 2022/11/22, 13:36:35,  64.4229  -52.7149   5      0 2,13
SBD 11342   377  295 000000000881890  2548     0   1 2022/11/22, 13:36:46,  64.4229  -52.7149   5      0 11,7,5,14,1
SBD 11342   377  311 000000000881890  2549     0   0 2022/11/22, 13:36:56,  64.4229  -52.7149   5      0 0,4,14,14,9,10
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1943           84        2027
# Sensor_ID bytes     % packets Description
          0    48  2.47       2         GPS
          1   102  5.25       1    Pressure
          2   182  9.37       1 Temperature
          3   174  8.96       1    Salinity
          4   133  6.85       1        Fall
          5    83  4.27       1        Rise
          6   136  7.00       1        Pump
          7    35  1.80       1 Drift_times
          8   287 14.77       1     Surface
          9    54  2.78       1  HR/Drift_P
         10    44  2.26       1  HR/Drift_T
         11    44  2.26       1  HR/Drift_S
         13   404 20.79       2  Parameters
         14   151  7.77       5 Engineering
         15    66  3.40       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  377 1 2022/11/22 08:08 +64.37975  -52.72889    0  5 39 41 44  1.3     0
G  377 2 2022/11/22 13:30 +64.40899  -52.68293  120  5 30 37 44  1.0     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   377  143    1     3         0
#      Pres      Temp    Salinity
      1.040     1.4040    31.9610
      2.000     1.4250    31.9590
      3.000     1.4410    31.9610
      4.000     1.4420    31.9610
      5.000     1.4420    31.9610
      6.040     1.4390    31.9610
      7.040     1.4400    31.9610
      8.000     1.4410    31.9610
      9.000     1.4460    31.9620
     10.040     1.4470    31.9630
     11.000     1.4460    31.9630
     12.000     1.4470    31.9640
     13.040     1.4500    31.9670
     13.960     1.4590    31.9750
     14.960     1.5140    32.0070
     16.040     1.7620    32.1750
     17.000     2.0200    32.3490
     18.000     2.1930    32.4640
     19.000     2.2940    32.5430
     20.000     2.4440    32.6600
     21.000     2.5190    32.7270
     22.000     2.7050    32.8430
     23.000     2.9590    33.0320
     24.040     3.0600    33.1070
     25.040     3.1790    33.1730
     26.040     3.4020    33.3100
     27.040     3.6700    33.4370
     28.040     3.9260    33.6190
     29.040     4.0050    33.6700
     30.040     4.1610    33.7770
     31.040     4.2010    33.8080
     32.000     4.2260    33.8260
     33.000     4.2330    33.8310
     34.040     4.2610    33.8460
     35.000     4.3020    33.8710
     36.000     4.3150    33.8800
     37.000     4.3320    33.8910
     38.040     4.3490    33.9020
     39.040     4.3650    33.9100
     40.040     4.3950    33.9300
     41.000     4.4070    33.9380
     42.000     4.4230    33.9480
     43.000     4.4420    33.9590
     44.000     4.4810    33.9840
     44.960     4.5020    34.0010
     46.000     4.5180    34.0100
     47.000     4.5400    34.0250
     48.000     4.5560    34.0360
     49.040     4.5580    34.0380
     50.040     4.5690    34.0440
     51.040     4.5800    34.0520
     52.040     4.6010    34.0640
     53.000     4.6120    34.0720
     54.000     4.6240    34.0780
     55.040     4.6500    34.0960
     56.000     4.6670    34.1080
     56.960     4.6720    34.1120
     58.000     4.6870    34.1190
     59.000     4.7010    34.1280
     60.000     4.7030    34.1290
     60.960     4.7080    34.1300
     62.000     4.7120    34.1320
     63.000     4.7170    34.1350
     64.000     4.7250    34.1370
     65.000     4.7290    34.1420
     66.040     4.7340    34.1450
     67.000     4.7540    34.1560
     68.000     4.7780    34.1670
     69.000     4.8100    34.1890
     70.000     4.8110    34.1900
     71.040     4.8060    34.1920
     72.000     4.8030    34.1920
     72.960     4.8040    34.1930
     74.000     4.8040    34.1930
     75.000     4.8130    34.1950
     76.000     4.8230    34.1990
     77.000     4.8310    34.2020
     77.960     4.8330    34.2030
     79.000     4.8340    34.2040
     80.040     4.8370    34.2050
     81.040     4.8530    34.2080
     82.040     4.8870    34.2180
     83.000     4.9270    34.2260
     84.000     4.9480    34.2360
     85.000     4.9690    34.2450
     86.000     5.0040    34.2550
     86.960     5.0440    34.2780
     88.000     5.0480    34.2820
     89.000     5.0480    34.2830
     90.000     5.0500    34.2850
     91.040     5.0630    34.2940
     92.040     5.0690    34.3060
     93.040     5.0670    34.3120
     94.040     5.0660    34.3150
     95.040     5.0670    34.3190
     96.040     5.0720    34.3260
     97.000     5.0860    34.3400
     98.000     5.0900    34.3460
     99.040     5.0910    34.3470
    100.040     5.1010    34.3530
    101.000     5.1050    34.3580
    102.000     5.1060    34.3590
    103.040     5.1070    34.3610
    104.040     5.1070    34.3610
    105.040     5.1070    34.3610
    106.000     5.1070    34.3610
    107.000     5.1070    34.3620
    108.000     5.1080    34.3630
    109.000     5.1080    34.3620
    110.000     5.1080    34.3630
    111.000     5.1090    34.3630
    112.040     5.1100    34.3650
    113.040     5.1130    34.3700
    114.000     5.1150    34.3720
    115.000     5.1240    34.3790
    116.000     5.1300    34.3880
    117.000     5.1430    34.3950
    118.040     5.1500    34.4010
    119.040     5.1530    34.4070
    120.000     5.1550    34.4090
    121.000     5.1560    34.4110
    122.000     5.1660    34.4190
    123.000     5.1740    34.4240
    124.000     5.1960    34.4370
    125.000     5.2160    34.4510
    126.000     5.2290    34.4580
    127.000     5.2420    34.4630
    128.000     5.2500    34.4670
    129.040     5.2510    34.4690
    130.040     5.2500    34.4670
    131.040     5.2540    34.4700
    132.000     5.2590    34.4730
    133.000     5.2750    34.4810
    134.000     5.2850    34.4870
    135.040     5.2960    34.4940
    136.000     5.3050    34.5010
    137.000     5.3130    34.5030
    138.000     5.3160    34.5070
    139.000     5.3170    34.5080
    140.000     5.3150    34.5070
    141.040     5.3170    34.5080
    142.080     5.3310    34.5170
    142.800     5.3540    34.5350
# Drift Data
#      C Npts comp n_var
I0   377   21    1     4
#      Pres       Temp   Salinity     Time
    127.000     5.3650    34.5390        6
    127.880     5.3880    34.5480      611
    125.560     5.3950    34.5540     1217
    133.080     5.3960    34.5540     1822
    140.080     5.3720    34.5400     2427
    135.960     5.3550    34.5300     3033
    136.520     5.3520    34.5280     3638
    138.600     5.3540    34.5290     4244
    136.960     5.3490    34.5240     4849
    142.560     5.3360    34.5150     5455
    133.440     5.3280    34.5100     6060
    129.720     5.3590    34.5290     6666
    133.440     5.3660    34.5380     7271
    131.440     5.3640    34.5350     7877
    131.120     5.3650    34.5360     8482
    136.160     5.3480    34.5270     9087
    137.560     5.3180    34.5080     9693
    139.240     5.3440    34.5230    10298
    152.240     5.2620    34.4810    10904
    152.680     5.3040    34.5000    11509
    135.800     5.3130    34.4990    12115
# Surface Pressure Time Series
#    C npts n_var               Start
S  377    31     6 2022/11/22 13:28:23
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    72      0.260      1.314   -999.000      14 2022/11/22 13:29:35
    85      0.170      1.327   -999.000      14 2022/11/22 13:29:48
    96      0.290      1.366   -999.000      14 2022/11/22 13:29:59
   107      0.210      1.297   -999.000      14 2022/11/22 13:30:10
   118      0.250      1.350   -999.000      14 2022/11/22 13:30:21
   129      0.260      1.334   -999.000      14 2022/11/22 13:30:32
   140      0.200      1.305   -999.000      14 2022/11/22 13:30:43
   151      0.260      1.342   -999.000      14 2022/11/22 13:30:54
   162      0.220      1.319   -999.000      14 2022/11/22 13:31:05
   173      0.260      1.269   -999.000      14 2022/11/22 13:31:16
   184      0.190      1.227   -999.000      14 2022/11/22 13:31:27
   195      0.280      1.310   -999.000      14 2022/11/22 13:31:38
   207      0.170      1.270   -999.000      14 2022/11/22 13:31:50
   218      0.200      1.293   -999.000      14 2022/11/22 13:32:01
   229      0.260      1.277   -999.000      14 2022/11/22 13:32:12
   240      0.250      1.251   -999.000      14 2022/11/22 13:32:23
   251      0.180      1.318   -999.000      14 2022/11/22 13:32:34
   262      0.240      1.308   -999.000      14 2022/11/22 13:32:45
   273      0.310      1.363   -999.000      14 2022/11/22 13:32:56
   284      0.180      1.329   -999.000      14 2022/11/22 13:33:07
   295      0.190      1.344   -999.000      14 2022/11/22 13:33:18
   306      0.210      1.280   -999.000      14 2022/11/22 13:33:29
   317      0.100      1.307   -999.000      14 2022/11/22 13:33:40
   329      0.250      1.246   -999.000      14 2022/11/22 13:33:52
   340      0.210      1.328   -999.000      14 2022/11/22 13:34:03
   351      0.270      1.244   -999.000      14 2022/11/22 13:34:14
   362      0.180      1.233   -999.000      14 2022/11/22 13:34:25
   373      0.300      1.270   -999.000      14 2022/11/22 13:34:36
   384      0.240      1.249   -999.000      14 2022/11/22 13:34:47
   395      0.210      1.228   -999.000      14 2022/11/22 13:34:58
   406      0.200      1.240   -999.000      14 2022/11/22 13:35:09
# Fall Time Series
#  Cycle npts     Start                Date
F00  377   25 722419692 2022/11/22 08:08:12
# time     Pres phase                Date
     0    0.440     1 2022/11/22 08:08:12
   171   10.200    10 2022/11/22 08:11:03
   239   20.160    10 2022/11/22 08:12:11
   317   30.040    10 2022/11/22 08:13:29
   395   40.240    10 2022/11/22 08:14:47
   444   46.560     2 2022/11/22 08:15:36
   473   50.120    10 2022/11/22 08:16:05
   564   60.120    10 2022/11/22 08:17:36
   677   70.040    10 2022/11/22 08:19:29
   897   80.040    10 2022/11/22 08:23:09
  1136   90.080    10 2022/11/22 08:27:08
  1392  100.040    10 2022/11/22 08:31:24
  1678  110.040    10 2022/11/22 08:36:10
  1978  117.840    10 2022/11/22 08:41:10
  2278  120.720    10 2022/11/22 08:46:10
  2578  123.000    10 2022/11/22 08:51:10
  2878  125.600    10 2022/11/22 08:56:10
  3178  125.440    10 2022/11/22 09:01:10
  3478  122.840    10 2022/11/22 09:06:10
  3778  122.000    10 2022/11/22 09:11:10
  4078  124.240    10 2022/11/22 09:16:10
  4310  125.360     3 2022/11/22 09:20:02
  4378  125.360    10 2022/11/22 09:21:10
  4678  125.840    10 2022/11/22 09:26:10
  4948  127.000    10 2022/11/22 09:30:40
# Rise Time Series
#  Cycle npts     Start                Date
R00  377   15 722437249 2022/11/22 13:00:49
# time     Pres phase                Date
     0  142.880     7 2022/11/22 13:00:49
   237  132.440    13 2022/11/22 13:04:46
   360  121.360    13 2022/11/22 13:06:49
   483  110.160    13 2022/11/22 13:08:52
   585   99.640    13 2022/11/22 13:10:34
   688   89.280    13 2022/11/22 13:12:17
   790   78.600    13 2022/11/22 13:13:59
   893   67.640    13 2022/11/22 13:15:42
   995   56.680    13 2022/11/22 13:17:24
  1118   45.120    13 2022/11/22 13:19:27
  1241   34.360    13 2022/11/22 13:21:30
  1344   23.560    13 2022/11/22 13:23:13
  1467   13.560    13 2022/11/22 13:25:16
  1590    3.400    13 2022/11/22 13:27:19
  1652    0.280     8 2022/11/22 13:28:21
# Pump Time Series
#   Cycle npts
B02   377   12
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    46.560  263   15.20     203    638    -9     2
   142.880   93   15.31     254    636    -9     7
   139.200   21   15.30     249    638    -9    13
   135.520   20   15.30     241    639    -9    13
   114.000   20   15.30     242    642    -9    13
    47.040   20   15.28     216    645    -9    13
    41.680   21   15.29     235    647    -9    13
    36.280   20   15.28     214    650    -9    13
    17.240   21   15.28     208    657    -9    13
    13.560   20   15.27     187    662    -9    13
     7.160   20   15.27     191    665    -9    13
     0.280  180   15.25     184    683    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   377  2.4   14    90   160   600    79    10   210      8     1    4020    5  0.2707
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  377 2022/11/22 08:08:09
# Pcase vacuums
   6.830   6.300
# Oil valve positions
   1.814   0.811   0.631
# Seek attempts and pump time
      1     -3
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  377 2022/11/22 09:30:40 9999/99/99 99:99:99
# Pcase vacuums
   6.300   6.290
# Average 
#   P      T      S
  136.040   5.349  34.526
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  377 2022/11/22 13:00:49 9999/99/99 99:99:99
# Pcase vacuums
   6.290   6.670
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  377 2022/11/22 13:28:21
# number of packets previous packets
      7               8
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.660   6.840
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.176   0.158   0.667
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.740   0.876
# Encountered Error Codes
#    Cycle Errors
EE20   377      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  377  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.240    440   6.200
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  377       197    2022/11/22 13:36:35
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      377	(cycle number) |
     MPS =        0	(phase) |
     MPC =      377	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      160	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    42104	(vac 0 inHg cal) |
     PV5 =    18698	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      376	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       46	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       10	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      262	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      376	(cycle) |
  PTO[0] =    12600	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        1	(0=dis 1=en) |
     PSD =      100	(dBar) |
     PST =     7200	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      376	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =       10	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      376	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      340	(sec) |
     SSP =      180	(sec) |

