# Message Decoder for Solo2, Alto and Deep Solo Floats; v230712
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11342   328   88 000000000881890  2222     0   6 2022/11/11, 00:37:53,  64.3152  -53.9240   4      0 14
SBD 11342   328  340 000000000881890  2223     0   5 2022/11/11, 00:38:02,  64.3152  -53.9240   4      0 13
SBD 11342   328  324 000000000881890  2224     0   4 2022/11/11, 00:38:49,  64.3342  -53.9909   2      0 15,13,6,14
SBD 11342   328  299 000000000881890  2225     0   3 2022/11/11, 00:39:02,  64.3342  -53.9909   2      0 8
SBD 11342   328  320 000000000881890  2226     0   2 2022/11/11, 00:39:58,  64.3152  -53.9240   8      0 2,3
SBD 11342   328  317 000000000881890  2227     0   1 2022/11/11, 00:40:12,  64.2996  -53.9240   7      0 11,7,5,14,1,0
SBD 11342   328  313 000000000881890  2228     0   0 2022/11/11, 00:40:27,  64.3080  -54.0395   7      0 0,4,14,14,9,10
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1917           84        2001
# Sensor_ID bytes     % packets Description
          0    48  2.50       2         GPS
          1   100  5.22       1    Pressure
          2   166  8.66       1 Temperature
          3   142  7.41       1    Salinity
          4   133  6.94       1        Fall
          5    83  4.33       1        Rise
          6   158  8.24       1        Pump
          7    35  1.83       1 Drift_times
          8   287 14.97       1     Surface
          9    54  2.82       1  HR/Drift_P
         10    46  2.40       1  HR/Drift_T
         11    44  2.30       1  HR/Drift_S
         13   404 21.07       2  Parameters
         14   151  7.88       5 Engineering
         15    66  3.44       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  328 1 2022/11/10 19:12 +64.28937  -54.02770    0  5 38 41 46  1.9     0
G  328 2 2022/11/11 00:36 +64.30087  -53.95119   90  4 32 37 47  4.4     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   328  135    1     3         0
#      Pres      Temp    Salinity
      1.040     2.1970    32.8000
      2.000     2.1980    32.8000
      3.040     2.1970    32.8000
      4.000     2.1960    32.8000
      5.040     2.1960    32.8000
      5.960     2.1960    32.8000
      7.000     2.1960    32.8000
      8.040     2.1980    32.8010
      9.000     2.1980    32.8010
     10.000     2.1980    32.8000
     11.040     2.1960    32.8000
     12.040     2.1930    32.7980
     13.040     2.1910    32.7990
     13.960     2.1910    32.7990
     14.920     2.1930    32.7990
     16.000     2.1930    32.7990
     17.040     2.1940    32.7990
     17.960     2.1910    32.7990
     19.000     2.1920    32.7980
     20.000     2.1910    32.7980
     21.040     2.1900    32.7990
     23.000     2.1880    32.7980
     23.960     2.1880    32.7980
     25.000     2.1880    32.7990
     26.000     2.1890    32.7990
     27.000     2.1890    32.7990
     28.000     2.1890    32.7990
     29.040     2.1890    32.7990
     30.000     2.1880    32.7990
     31.040     2.1910    32.7990
     32.040     2.1890    32.8000
     33.000     2.1890    32.7990
     34.000     2.1880    32.8000
     35.000     2.1880    32.8000
     36.080     2.1900    32.8000
     37.000     2.1920    32.8010
     38.000     2.2040    32.7960
     39.080     2.2830    32.8350
     40.000     2.2670    32.8320
     41.000     2.2640    32.8270
     42.000     2.2730    32.8320
     43.000     2.3360    32.8600
     44.000     2.3670    32.8740
     45.000     2.3750    32.8700
     46.040     2.4460    32.9120
     47.000     2.6200    32.9740
     48.000     2.7520    33.0360
     49.040     2.6520    32.9930
     50.000     2.7140    33.0240
     51.000     2.8340    33.0660
     52.000     2.8620    33.0890
     52.960     2.8680    33.0940
     54.000     2.9490    33.1260
     55.040     3.1100    33.1940
     56.000     3.2410    33.2690
     57.000     3.3690    33.3220
     58.000     3.5180    33.3790
     59.000     3.6170    33.4390
     59.960     3.6730    33.4670
     60.960     3.8750    33.5500
     62.000     4.0350    33.6390
     63.000     4.0730    33.6690
     64.000     4.0880    33.6800
     65.000     4.0970    33.6880
     66.000     4.1030    33.6910
     67.000     4.1140    33.7000
     67.960     4.1210    33.7070
     69.000     4.1290    33.7140
     70.000     4.1320    33.7170
     71.000     4.1410    33.7230
     72.000     4.1480    33.7280
     73.000     4.1600    33.7340
     74.000     4.1780    33.7450
     75.000     4.1840    33.7500
     76.000     4.1890    33.7530
     77.000     4.2000    33.7580
     78.000     4.2570    33.7750
     79.040     4.3090    33.7930
     80.040     4.3340    33.8030
     81.040     4.3540    33.8130
     82.040     4.3780    33.8260
     83.000     4.4020    33.8390
     84.000     4.4270    33.8520
     85.000     4.4590    33.8700
     86.000     4.4810    33.8870
     87.000     4.5240    33.9100
     87.960     4.5540    33.9230
     89.000     4.5600    33.9310
     90.000     4.5600    33.9350
     91.000     4.5610    33.9380
     92.000     4.5660    33.9420
     93.040     4.5650    33.9460
     94.000     4.5660    33.9470
     94.960     4.5700    33.9520
     95.960     4.5670    33.9530
     96.960     4.5570    33.9580
     98.000     4.5560    33.9600
     99.000     4.5600    33.9650
    100.000     4.5680    33.9770
    101.000     4.5680    33.9790
    102.000     4.5690    33.9820
    103.040     4.5760    33.9890
    104.040     4.5930    34.0030
    105.040     4.6160    34.0380
    106.000     4.6220    34.0470
    107.000     4.6400    34.0560
    107.960     4.6600    34.0690
    109.000     4.6650    34.0740
    110.000     4.6820    34.0880
    111.000     4.7060    34.1040
    112.000     4.7590    34.1370
    113.040     4.8080    34.1600
    114.040     4.8270    34.1690
    115.040     4.9000    34.1850
    116.040     4.9340    34.1970
    117.040     4.9510    34.2000
    118.000     4.9210    34.2090
    119.000     4.8970    34.2100
    120.000     4.8670    34.2120
    121.000     4.8670    34.2130
    122.000     4.8700    34.2160
    123.000     4.9310    34.2420
    124.000     4.9820    34.2600
    125.000     4.9950    34.2720
    126.000     4.9850    34.2760
    127.000     5.0650    34.2850
    128.040     5.3000    34.3440
    129.000     5.3330    34.3590
    130.040     5.3520    34.3670
    131.000     5.3770    34.3780
    131.960     5.3970    34.3850
    132.960     5.4130    34.3950
    134.000     5.4080    34.4030
    135.040     5.3820    34.4040
    136.000     5.3270    34.3960
# Drift Data
#      C Npts comp n_var
I0   328   21    1     4
#      Pres       Temp   Salinity     Time
    129.680     5.3950    34.4110        6
    126.880     5.4060    34.4150      612
    132.920     5.3770    34.4130     1217
    135.160     5.4150    34.4080     1822
    134.880     5.4200    34.4090     2428
    133.320     5.4320    34.4080     3033
    138.680     5.4140    34.3940     3639
    136.880     5.3870    34.3840     4244
    140.440     5.3810    34.3790     4850
    138.960     5.3840    34.3820     5455
    143.960     5.3660    34.3690     6061
    138.400     5.3830    34.3810     6666
    137.440     5.3810    34.3840     7272
    137.680     5.4040    34.3890     7877
    136.360     5.3760    34.3820     8483
    131.680     5.4420    34.4050     9088
    134.240     5.4170    34.4010     9693
    130.160     5.4300    34.4090    10299
    132.960     5.4100    34.4080    10904
    130.000     5.4010    34.4070    11510
    130.040     5.2450    34.3910    12115
# Surface Pressure Time Series
#    C npts n_var               Start
S  328    31     6 2022/11/11 00:30:36
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    72      0.280      1.971   -999.000      14 2022/11/11 00:31:48
    84      0.120      1.930   -999.000      14 2022/11/11 00:32:00
    95      0.220      1.914   -999.000      14 2022/11/11 00:32:11
   107      0.120      1.929   -999.000      14 2022/11/11 00:32:23
   118      0.040      1.632   -999.000      14 2022/11/11 00:32:34
   129      0.170      1.932   -999.000      14 2022/11/11 00:32:45
   140      0.150      1.831   -999.000      14 2022/11/11 00:32:56
   151      0.160      1.985   -999.000      14 2022/11/11 00:33:07
   162      0.160      1.999   -999.000      14 2022/11/11 00:33:18
   173      0.280      2.109   -999.000      14 2022/11/11 00:33:29
   184      0.160      2.114   -999.000      14 2022/11/11 00:33:40
   195      0.190      2.002   -999.000      14 2022/11/11 00:33:51
   206      0.400      1.919   -999.000      14 2022/11/11 00:34:02
   217      0.090      2.047   -999.000      14 2022/11/11 00:34:13
   229      0.230      1.993   -999.000      14 2022/11/11 00:34:25
   240      0.160      1.931   -999.000      14 2022/11/11 00:34:36
   251      0.120      1.877   -999.000      14 2022/11/11 00:34:47
   262      0.140      1.804   -999.000      14 2022/11/11 00:34:58
   273      0.220      1.968   -999.000      14 2022/11/11 00:35:09
   284      0.260      2.011   -999.000      14 2022/11/11 00:35:20
   295      0.460      2.120   -999.000      14 2022/11/11 00:35:31
   306      0.520      2.174   -999.000      14 2022/11/11 00:35:42
   317      0.210      2.123   -999.000      14 2022/11/11 00:35:53
   328      0.170      1.854   -999.000      14 2022/11/11 00:36:04
   339      0.160      1.979   -999.000      14 2022/11/11 00:36:15
   351      0.180      1.703   -999.000      14 2022/11/11 00:36:27
   361      0.220      1.974   -999.000      14 2022/11/11 00:36:37
   372      0.180      1.981   -999.000      14 2022/11/11 00:36:48
   384      0.190      1.927   -999.000      14 2022/11/11 00:37:00
   395      0.230      2.067   -999.000      14 2022/11/11 00:37:11
   406      0.150      2.048   -999.000      14 2022/11/11 00:37:22
# Fall Time Series
#  Cycle npts     Start                Date
F00  328   25 721422765 2022/11/10 19:12:45
# time     Pres phase                Date
     0    0.200     1 2022/11/10 19:12:45
   174   10.400    10 2022/11/10 19:15:39
   258   20.080    10 2022/11/10 19:17:03
   330   30.360    10 2022/11/10 19:18:15
   398   40.400    10 2022/11/10 19:19:23
   441   46.640     2 2022/11/10 19:20:06
   469   50.160    10 2022/11/10 19:20:34
   567   60.160    10 2022/11/10 19:22:12
   741   70.000    10 2022/11/10 19:25:06
  1027   80.000    10 2022/11/10 19:29:52
  1312   90.080    10 2022/11/10 19:34:37
  1558  100.000    10 2022/11/10 19:38:43
  1858  108.160    10 2022/11/10 19:43:43
  2158  113.040    10 2022/11/10 19:48:43
  2458  118.640    10 2022/11/10 19:53:43
  2758  121.480    10 2022/11/10 19:58:43
  3058  123.280    10 2022/11/10 20:03:43
  3358  127.720    10 2022/11/10 20:08:43
  3658  131.600    10 2022/11/10 20:13:43
  3958  130.560    10 2022/11/10 20:18:43
  4258  127.480    10 2022/11/10 20:23:43
  4314  127.120     3 2022/11/10 20:24:39
  4560  126.760    10 2022/11/10 20:28:45
  4860  129.120    10 2022/11/10 20:33:45
  4951  129.680    10 2022/11/10 20:35:16
# Rise Time Series
#  Cycle npts     Start                Date
R00  328   15 721440325 2022/11/11 00:05:25
# time     Pres phase                Date
     0  136.160     7 2022/11/11 00:05:25
   176  126.480    13 2022/11/11 00:08:21
   299  115.760    13 2022/11/11 00:10:24
   401  105.920    13 2022/11/11 00:12:06
   504   95.200    13 2022/11/11 00:13:49
   606   85.520    13 2022/11/11 00:15:31
   729   75.440    13 2022/11/11 00:17:34
   852   65.040    13 2022/11/11 00:19:37
   996   54.400    13 2022/11/11 00:22:01
  1119   45.000    13 2022/11/11 00:24:04
  1221   34.000    13 2022/11/11 00:25:46
  1324   21.600    13 2022/11/11 00:27:29
  1406   10.160    13 2022/11/11 00:28:51
  1498    0.120    13 2022/11/11 00:30:23
  1509    0.200     8 2022/11/11 00:30:34
# Pump Time Series
#   Cycle npts
B02   328   14
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    46.640  269   15.21     220    642    -9     2
   136.160   94   15.31     239    636    -9     7
   130.080   20   15.30     249    638    -9    13
   121.040   21   15.31     248    641    -9    13
    85.520   21   15.29     254    645    -9    13
    80.560   20   15.30     240    647    -9    13
    75.440   21   15.28     236    649    -9    13
    68.320   21   15.29     234    651    -9    13
    63.400   20   15.28     213    654    -9    13
    59.120   21   15.27     212    656    -9    13
    54.400   20   15.27     209    660    -9    13
    49.240   21   15.28     210    663    -9    13
    45.000   20   15.26     208    666    -9    13
     0.120  181   15.24     186    685    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   328  2.4   14    90   160   600    79    10   210      8     1    4020    5  0.2707
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  328 2022/11/10 19:12:42
# Pcase vacuums
   6.770   6.310
# Oil valve positions
   1.811   0.812   0.631
# Seek attempts and pump time
      1     -3
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  328 2022/11/10 20:35:16 9999/99/99 99:99:99
# Pcase vacuums
   6.310   6.300
# Average 
#   P      T      S
  134.800   5.394  34.396
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  328 2022/11/11 00:05:25 9999/99/99 99:99:99
# Pcase vacuums
   6.300   6.690
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  328 2022/11/11 00:30:34
# number of packets previous packets
      6              11
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.690   6.850
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.176   0.158   0.667
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.740   0.876
# Encountered Error Codes
#    Cycle Errors
EE20   328      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  328  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.240    437   6.260
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  328       197    2022/11/11 00:38:49
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      328	(cycle number) |
     MPS =        0	(phase) |
     MPC =      328	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      160	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    42104	(vac 0 inHg cal) |
     PV5 =    18698	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      327	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       46	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       10	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      269	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      327	(cycle) |
  PTO[0] =    12600	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        1	(0=dis 1=en) |
     PSD =      100	(dBar) |
     PST =     7200	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      327	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =       10	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      327	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      340	(sec) |
     SSP =      180	(sec) |

