# Message Decoder for Solo2, Alto and Deep Solo Floats; v230712
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11342   265  231 000000000881890  1837     0   5 2022/10/27, 08:11:24,  63.2001  -53.0665  12      0 6,14,14
SBD 11342   265  340 000000000881890  1838     0   4 2022/10/27, 08:11:52,  63.2129  -53.2415  15      0 13
SBD 11342   265  299 000000000881890  1839     0   3 2022/10/27, 08:12:16,  63.2386  -53.1938  18      0 8
SBD 11342   265  314 000000000881890  1840     0   2 2022/10/27, 08:12:29,  63.1937  -53.1778  12      0 3,0,15,13
SBD 11342   265  229 000000000881890  1841     0   1 2022/10/27, 08:12:44,  63.1745  -53.1142  11      0 1,2
SBD 11342   265  325 000000000881890  1842     0   0 2022/10/27, 08:13:07,  63.2001  -53.0665   9      0 0,4,14,14,7,5,14
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      18         1666           72        1738
# Sensor_ID bytes     % packets Description
          0    48  2.88       2         GPS
          1    81  4.86       1    Pressure
          2   136  8.16       1 Temperature
          3   136  8.16       1    Salinity
          4   133  7.98       1        Fall
          5    68  4.08       1        Rise
          6   125  7.50       1        Pump
          7    25  1.50       1 Drift_times
          8   287 17.23       1     Surface
         13   404 24.25       2  Parameters
         14   157  9.42       5 Engineering
         15    66  3.96       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  265 1 2022/10/27 06:09 +63.14665  -53.00325    0  4 43 45 48  2.9     0
G  265 2 2022/10/27 08:07 +63.18273  -53.07403  210  6 26 37 44  1.5     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   265  113    1     3         0
#      Pres      Temp    Salinity
      1.120     2.8060    32.5240
      2.000     2.8060    32.5240
      2.960     2.8060    32.5230
      4.000     2.8050    32.5240
      5.000     2.8050    32.5230
      6.040     2.8040    32.5230
      7.000     2.8040    32.5230
      7.960     2.8040    32.5230
      9.000     2.8040    32.5230
     10.040     2.8040    32.5230
     11.000     2.8040    32.5230
     12.040     2.8050    32.5230
     13.040     2.8050    32.5240
     13.960     2.8050    32.5240
     15.000     2.8070    32.5240
     16.000     2.8150    32.5280
     17.000     2.8160    32.5290
     18.000     2.8140    32.5290
     18.960     2.8130    32.5280
     19.960     2.8160    32.5290
     20.960     2.8180    32.5300
     22.000     2.8160    32.5290
     23.000     2.8150    32.5280
     24.000     2.8180    32.5300
     24.960     2.8160    32.5290
     26.000     2.8280    32.5360
     27.000     2.8270    32.5360
     28.000     2.8240    32.5340
     28.960     2.8290    32.5360
     29.960     2.8380    32.5410
     31.000     2.8380    32.5410
     32.000     2.8740    32.5640
     33.000     2.8990    32.5870
     34.000     2.9270    32.6300
     35.040     3.0630    32.7980
     36.000     3.3200    33.0120
     36.960     3.3380    33.0400
     37.960     3.3740    33.1000
     39.000     3.5110    33.2300
     40.000     3.6320    33.3070
     41.000     3.6190    33.3230
     42.040     3.6190    33.3350
     43.000     3.5700    33.3480
     44.000     3.7250    33.3770
     45.000     3.8060    33.4030
     46.040     3.8020    33.4230
     47.000     3.8050    33.4260
     47.960     3.8050    33.4360
     48.960     3.8380    33.5150
     50.000     3.8900    33.5550
     51.040     3.8970    33.5810
     52.000     3.8980    33.6070
     53.040     3.8910    33.6320
     54.000     3.9020    33.6400
     55.000     3.9170    33.6540
     56.000     3.9510    33.6840
     56.960     3.9650    33.6900
     57.960     3.9790    33.6960
     59.000     4.0020    33.7040
     59.960     4.0420    33.7220
     61.000     4.0610    33.7350
     62.040     4.0660    33.7390
     63.000     4.0810    33.7510
     64.000     4.1190    33.7750
     65.000     4.1290    33.7860
     66.000     4.1440    33.7960
     67.000     4.1980    33.8230
     67.960     4.2280    33.8480
     69.000     4.1960    33.8660
     70.000     4.2040    33.8760
     71.000     4.2170    33.8790
     72.000     4.2370    33.8820
     73.000     4.2540    33.8970
     74.040     4.2770    33.9170
     75.040     4.2890    33.9290
     76.040     4.3120    33.9380
     77.000     4.3370    33.9520
     78.000     4.3430    33.9540
     79.000     4.3530    33.9600
     79.960     4.3730    33.9650
     80.960     4.4190    33.9820
     81.960     4.4400    33.9900
     82.960     4.5040    34.0000
     83.960     4.5590    34.0210
     85.000     4.6060    34.0420
     86.040     4.6270    34.0580
     87.040     4.6300    34.0680
     88.000     4.6480    34.0840
     89.000     4.6810    34.1010
     90.000     4.7080    34.1190
     90.960     4.7150    34.1250
     92.000     4.7350    34.1380
     92.960     4.7800    34.1590
     94.000     4.8070    34.1690
     95.000     4.8580    34.1880
     96.000     4.9080    34.2060
     97.040     4.9510    34.2260
     98.040     4.9560    34.2300
     99.000     4.9610    34.2330
    100.000     4.9800    34.2450
    100.960     4.9810    34.2500
    102.000     4.9830    34.2510
    103.040     4.9840    34.2520
    104.000     4.9840    34.2530
    105.000     4.9860    34.2540
    106.000     4.9880    34.2560
    107.000     4.9910    34.2570
    107.960     4.9980    34.2600
    109.000     5.0040    34.2640
    110.040     5.0110    34.2670
    111.040     5.0180    34.2750
    112.080     5.0040    34.2830
    112.880     5.0070    34.2860
# Drift Data
#      C Npts comp n_var
I0   265    1    1     4
#      Pres       Temp   Salinity     Time
   -999.000  -999.0000  -999.0000 16769922
# Surface Pressure Time Series
#    C npts n_var               Start
S  265    31     6 2022/10/27 08:03:57
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    70      0.340      2.829   -999.000      14 2022/10/27 08:05:07
    83      0.290      2.831   -999.000      14 2022/10/27 08:05:20
    94      0.360      2.859   -999.000      14 2022/10/27 08:05:31
   105      0.300      2.837   -999.000      14 2022/10/27 08:05:42
   116      0.260      2.830   -999.000      14 2022/10/27 08:05:53
   127      0.360      2.842   -999.000      14 2022/10/27 08:06:04
   138      0.340      2.857   -999.000      14 2022/10/27 08:06:15
   150      0.290      2.882   -999.000      14 2022/10/27 08:06:27
   161      0.170      2.846   -999.000      14 2022/10/27 08:06:38
   175      0.260      2.860   -999.000      14 2022/10/27 08:06:52
   186      0.250      2.908   -999.000      14 2022/10/27 08:07:03
   197      0.280      2.864   -999.000      14 2022/10/27 08:07:14
   208      0.260      2.900   -999.000      14 2022/10/27 08:07:25
   217      0.250      2.937   -999.000      14 2022/10/27 08:07:34
   228      0.260      2.932   -999.000      14 2022/10/27 08:07:45
   239      0.300      2.919   -999.000      14 2022/10/27 08:07:56
   250      0.190      2.897   -999.000      14 2022/10/27 08:08:07
   261      0.200      2.919   -999.000      14 2022/10/27 08:08:18
   272      0.170      2.919   -999.000      14 2022/10/27 08:08:29
   284      0.270      2.993   -999.000      14 2022/10/27 08:08:41
   295      0.240      2.897   -999.000      14 2022/10/27 08:08:52
   306      0.250      2.907   -999.000      14 2022/10/27 08:09:03
   317      0.170      2.927   -999.000      14 2022/10/27 08:09:14
   328      0.250      2.943   -999.000      14 2022/10/27 08:09:25
   339      0.340      2.900   -999.000      14 2022/10/27 08:09:36
   350      0.330      2.868   -999.000      14 2022/10/27 08:09:47
   361      0.210      2.897   -999.000      14 2022/10/27 08:09:58
   372      0.230      2.876   -999.000      14 2022/10/27 08:10:09
   383      0.290      2.994   -999.000      14 2022/10/27 08:10:20
   394      0.300      2.916   -999.000      14 2022/10/27 08:10:31
   406      0.310      2.908   -999.000      14 2022/10/27 08:10:43
# Fall Time Series
#  Cycle npts     Start                Date
F00  265   25 720166151 2022/10/27 06:09:11
# time     Pres phase                Date
     0    0.240     1 2022/10/27 06:09:11
   190   10.080    10 2022/10/27 06:12:21
   255   20.040    10 2022/10/27 06:13:26
   320   30.080    10 2022/10/27 06:14:31
   391   40.440    10 2022/10/27 06:15:42
   438   46.640     2 2022/10/27 06:16:29
   466   50.120    10 2022/10/27 06:16:57
   557   60.040    10 2022/10/27 06:18:28
   667   70.080    10 2022/10/27 06:20:18
   903   80.040    10 2022/10/27 06:24:14
  1203   89.520    10 2022/10/27 06:29:14
  1503   98.800    10 2022/10/27 06:34:14
  1803  103.080    10 2022/10/27 06:39:14
  2103  101.960    10 2022/10/27 06:44:14
  2403  102.840    10 2022/10/27 06:49:14
  2703  107.800    10 2022/10/27 06:54:14
  3003  109.600    10 2022/10/27 06:59:14
  3303  109.800    10 2022/10/27 07:04:14
  3603  109.960    10 2022/10/27 07:09:14
  3903  111.080    10 2022/10/27 07:14:14
  4203  111.360    10 2022/10/27 07:19:14
  4313  112.120     3 2022/10/27 07:21:04
  4503  113.840    10 2022/10/27 07:24:14
  4803  113.720    10 2022/10/27 07:29:14
  4951  113.600    10 2022/10/27 07:31:42
# Rise Time Series
#  Cycle npts     Start                Date
R00  265   12 720171711 2022/10/27 07:41:51
# time     Pres phase                Date
     0  113.120     7 2022/10/27 07:41:51
   175  102.520    13 2022/10/27 07:44:46
   277   92.640    13 2022/10/27 07:46:28
   400   81.600    13 2022/10/27 07:48:31
   523   70.720    13 2022/10/27 07:50:34
   646   60.200    13 2022/10/27 07:52:37
   749   50.840    13 2022/10/27 07:54:20
   872   40.000    13 2022/10/27 07:56:23
   974   30.800    13 2022/10/27 07:58:05
  1097   19.040    13 2022/10/27 08:00:08
  1200    9.200    13 2022/10/27 08:01:51
  1324    0.320     8 2022/10/27 08:03:55
# Pump Time Series
#   Cycle npts
B02   265   11
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    46.640  273   15.22     200    642    -9     2
   113.120   93   15.24     231    640    -9     7
   106.320   20   15.23     229    642    -9    13
    87.240   20   15.25     215    645    -9    13
    76.240   20   15.24     209    647    -9    13
    69.000   20   15.23     207    650    -9    13
    62.000   20   15.23     220    652    -9    13
    47.280   20   15.23     201    655    -9    13
    32.640   20   15.23     198    659    -9    13
     1.480   31   15.23     198    664    -9    13
     0.320  240   15.19     191    686    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   265  2.4   13    90   145   600    79    10   210      8     1    4020    5  0.2707
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  265 2022/10/27 06:09:08
# Pcase vacuums
   6.830   6.340
# Oil valve positions
   1.811   0.815   0.631
# Seek attempts and pump time
      1     -3
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  265 2022/10/27 07:31:42 9999/99/99 99:99:99
# Pcase vacuums
   6.330   6.340
# Average 
#   P      T      S
 9999.000  99.000 -999.000
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  265 2022/10/27 07:41:51 9999/99/99 99:99:99
# Pcase vacuums
   6.330   6.640
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  265 2022/10/27 08:03:53
# number of packets previous packets
      6               7
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.640   6.860
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.176   0.158   0.667
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.740   0.876
# Encountered Error Codes
#    Cycle Errors
EE20   265      1
#               Date   Code Description
 2022/10/27 07:41:42 0x0115 ParkCtd
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  265  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.240    434   6.210
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  265       197    2022/10/27 08:12:29
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      265	(cycle number) |
     MPS =        0	(phase) |
     MPC =      265	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      145	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    42104	(vac 0 inHg cal) |
     PV5 =    18698	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      264	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       46	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       10	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      271	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      264	(cycle) |
  PTO[0] =    12600	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        1	(0=dis 1=en) |
     PSD =      100	(dBar) |
     PST =     7200	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      264	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =       10	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      264	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      340	(sec) |
     SSP =      240	(sec) |

