# Message Decoder for Solo2, Alto and Deep Solo Floats; v230712
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11342   256   88 000000000881890  1776     0   6 2022/10/25, 10:17:57,  63.2289  -54.1014 102      2 14
SBD 11342   256  340 000000000881890  1777     0   5 2022/10/25, 10:18:07,  63.2477  -54.1659 100      2 13
SBD 11342   256  311 000000000881890  1778     0   4 2022/10/25, 10:18:43,  63.1994  -52.6689   4      0 8,14
SBD 11342   256  335 000000000881890  1779     0   3 2022/10/25, 10:19:51,  63.2865  -54.5179 151      2 3,0,15,13
SBD 11342   256  302 000000000881890  1780     0   2 2022/10/25, 10:20:18,  63.2528  -54.6753 169      2 2,6
SBD 11342   256  293 000000000881890  1781     0   1 2022/10/25, 10:20:37,  63.2898  -54.8054 185      2 11,7,5,14,1
SBD 11342   256  307 000000000881890  1782     0   0 2022/10/25, 10:21:00,  63.2820  -54.9165 195      2 0,4,14,14,9,10
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1892           84        1976
# Sensor_ID bytes     % packets Description
          0    48  2.54       2         GPS
          1   101  5.34       1    Pressure
          2   176  9.30       1 Temperature
          3   157  8.30       1    Salinity
          4   138  7.29       1        Fall
          5    83  4.39       1        Rise
          6   114  6.03       1        Pump
          7    35  1.85       1 Drift_times
          8   287 15.17       1     Surface
          9    54  2.85       1  HR/Drift_P
         10    35  1.85       1  HR/Drift_T
         11    43  2.27       1  HR/Drift_S
         13   404 21.35       2  Parameters
         14   151  7.98       5 Engineering
         15    66  3.49       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  256 1 2022/10/25 04:48 +63.05160  -52.77807  140  5 28 34 40  1.9     0
G  256 2 2022/10/25 10:17 +63.15681  -52.74117   60  4 32 36 41  2.9     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   256  137    1     3         0
#      Pres      Temp    Salinity
      1.040     2.8820    32.2920
      1.960     2.8810    32.2900
      3.040     2.8840    32.2900
      4.000     2.8870    32.2940
      5.040     2.8870    32.2970
      6.000     2.8870    32.2950
      6.920     2.8880    32.2980
      8.000     2.8840    32.2950
      9.000     2.8810    32.2900
     10.000     2.8800    32.2910
     12.000     2.8980    32.3060
     13.000     2.9040    32.3190
     14.000     2.9390    32.3540
     15.040     3.0090    32.4270
     16.040     3.1040    32.5530
     17.000     3.1610    32.6420
     18.000     3.1890    32.6900
     19.080     3.1990    32.7070
     20.000     3.2200    32.7400
     21.000     3.2470    32.7790
     22.040     3.2830    32.8360
     23.000     3.3210    32.8840
     24.000     3.3470    32.9200
     24.960     3.3700    32.9520
     25.960     3.4020    32.9990
     27.040     3.4580    33.1100
     28.040     3.4820    33.1880
     29.000     3.5000    33.2100
     30.000     3.5070    33.2170
     30.960     3.5230    33.2510
     31.960     3.5320    33.2870
     33.000     3.5370    33.3150
     34.000     3.5460    33.3560
     35.000     3.5790    33.4730
     36.040     3.6170    33.5390
     37.040     3.6960    33.5840
     38.000     3.7790    33.6520
     39.040     3.8760    33.7210
     40.000     3.9080    33.7400
     41.000     3.9280    33.7540
     42.040     3.9780    33.7780
     43.000     4.0070    33.7910
     44.000     4.0380    33.8130
     44.960     4.0410    33.8140
     46.000     4.0600    33.8230
     47.000     4.0750    33.8320
     47.960     4.1150    33.8470
     48.960     4.1570    33.8740
     49.960     4.1800    33.8870
     50.960     4.1950    33.9030
     52.000     4.2030    33.9130
     53.000     4.2070    33.9160
     54.000     4.2110    33.9170
     55.000     4.2270    33.9230
     56.000     4.2510    33.9280
     57.000     4.2670    33.9370
     57.960     4.2660    33.9410
     59.000     4.2610    33.9450
     60.040     4.2810    33.9490
     61.040     4.3450    33.9700
     62.000     4.3860    33.9800
     62.960     4.3880    33.9840
     64.000     4.3910    33.9860
     65.000     4.3910    33.9870
     66.000     4.3890    33.9850
     67.000     4.3940    33.9940
     68.040     4.3990    34.0010
     69.000     4.4040    34.0070
     69.960     4.4050    34.0070
     70.960     4.4060    34.0090
     72.040     4.4060    34.0080
     73.040     4.4110    34.0110
     74.000     4.4120    34.0140
     75.000     4.4200    34.0180
     76.000     4.4380    34.0290
     77.000     4.4420    34.0370
     78.000     4.4510    34.0420
     79.000     4.4850    34.0520
     80.000     4.5330    34.0680
     81.000     4.5510    34.0860
     82.040     4.5590    34.0890
     83.080     4.5920    34.0960
     84.040     4.6180    34.1100
     85.080     4.6280    34.1150
     86.040     4.6380    34.1180
     87.000     4.6980    34.1260
     87.960     4.7460    34.1400
     89.000     4.8620    34.1450
     90.040     4.9900    34.1870
     91.040     5.0070    34.1910
     92.040     5.0810    34.2030
     93.040     5.0730    34.2240
     94.040     5.0270    34.2220
     95.000     5.1210    34.2140
     96.000     5.6300    34.2920
     97.000     5.7880    34.3420
     98.000     5.8320    34.3720
     99.000     5.6170    34.3440
    100.000     5.4900    34.3290
    101.000     5.2780    34.3130
    102.040     5.2610    34.3000
    103.040     5.2520    34.2980
    104.000     5.2330    34.3030
    105.000     5.2070    34.2930
    106.040     5.3120    34.3120
    107.000     5.2810    34.3170
    108.000     5.2890    34.3090
    109.000     5.2500    34.3200
    110.000     5.2790    34.3120
    111.000     5.2440    34.3310
    112.000     5.2140    34.3310
    113.000     5.2130    34.3300
    114.040     5.1920    34.3340
    115.040     5.1570    34.3410
    116.000     5.0960    34.3520
    116.960     5.0930    34.3530
    118.000     5.0920    34.3600
    119.000     5.0870    34.3660
    120.000     5.0860    34.3670
    121.000     5.0810    34.3680
    122.000     5.0650    34.3740
    123.000     5.0560    34.3780
    124.000     5.0550    34.3810
    125.000     5.0620    34.3890
    126.000     5.0750    34.4010
    127.000     5.0800    34.4070
    128.000     5.0910    34.4100
    129.000     5.0940    34.4160
    130.000     5.0900    34.4230
    131.000     5.0960    34.4300
    132.000     5.0930    34.4360
    133.000     5.1100    34.4420
    134.040     5.1200    34.4540
    135.040     5.1250    34.4590
    136.040     5.1310    34.4670
    137.040     5.1350    34.4730
    138.080     5.1370    34.4770
# Drift Data
#      C Npts comp n_var
I0   256   21    1     4
#      Pres       Temp   Salinity     Time
    126.360     5.1480    34.5190        6
    132.120     5.1480    34.5120      612
    128.400     5.1460    34.5040     1217
    134.480     5.1470    34.5080     1822
    130.040     5.1450    34.4970     2428
    131.680     5.1460    34.5060     3033
    135.640     5.1450    34.4960     3639
    134.960     5.1460    34.4980     4244
    132.800     5.1450    34.4940     4850
    139.520     5.1470    34.4940     5455
    132.960     5.1460    34.4930     6061
    135.560     5.1460    34.4970     6666
    135.720     5.1460    34.4950     7272
    141.800     5.1470    34.4950     7877
    132.440     5.1460    34.4930     8483
    134.880     5.1470    34.4940     9088
    141.160     5.1470    34.4950     9693
    145.680     5.1470    34.4920    10299
    140.040     5.1420    34.4840    10904
    139.560     5.1420    34.4830    11510
    144.800     5.1410    34.4800    12115
# Surface Pressure Time Series
#    C npts n_var               Start
S  256    31     6 2022/10/25 10:08:01
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    72      0.330      2.794   -999.000      14 2022/10/25 10:09:13
    85      0.430      2.759   -999.000      14 2022/10/25 10:09:26
    96      0.310      2.751   -999.000      14 2022/10/25 10:09:37
   107      0.350      2.836   -999.000      14 2022/10/25 10:09:48
   118      0.440      2.837   -999.000      14 2022/10/25 10:09:59
   129      0.290      2.574   -999.000      14 2022/10/25 10:10:10
   140      0.350      2.627   -999.000      14 2022/10/25 10:10:21
   152      0.310      2.705   -999.000      14 2022/10/25 10:10:33
   163      0.320      2.767   -999.000      14 2022/10/25 10:10:44
   174      0.470      2.716   -999.000      14 2022/10/25 10:10:55
   185      0.390      2.681   -999.000      14 2022/10/25 10:11:06
   196      0.330      2.799   -999.000      14 2022/10/25 10:11:17
   207      0.360      2.726   -999.000      14 2022/10/25 10:11:28
   218      0.250      2.715   -999.000      14 2022/10/25 10:11:39
   229      0.360      2.792   -999.000      14 2022/10/25 10:11:50
   240      0.350      2.704   -999.000      14 2022/10/25 10:12:01
   251      0.400      2.798   -999.000      14 2022/10/25 10:12:12
   262      0.430      2.831   -999.000      14 2022/10/25 10:12:23
   273      0.440      2.772   -999.000      14 2022/10/25 10:12:34
   285      0.390      2.808   -999.000      14 2022/10/25 10:12:46
   296      0.500      2.818   -999.000      14 2022/10/25 10:12:57
   307      0.470      2.845   -999.000      14 2022/10/25 10:13:08
   318      0.450      2.818   -999.000      14 2022/10/25 10:13:19
   329      0.350      2.833   -999.000      14 2022/10/25 10:13:30
   340      0.340      2.723   -999.000      14 2022/10/25 10:13:41
   351      0.490      2.846   -999.000      14 2022/10/25 10:13:52
   362      0.280      2.767   -999.000      14 2022/10/25 10:14:03
   373      0.350      2.684   -999.000      14 2022/10/25 10:14:14
   384      0.450      2.815   -999.000      14 2022/10/25 10:14:25
   395      0.520      2.849   -999.000      14 2022/10/25 10:14:36
   407      0.610      2.852   -999.000      14 2022/10/25 10:14:48
# Fall Time Series
#  Cycle npts     Start                Date
F00  256   26 719988520 2022/10/25 04:48:40
# time     Pres phase                Date
     0    0.280     1 2022/10/25 04:48:40
   207   10.360    10 2022/10/25 04:52:07
   265   20.160    10 2022/10/25 04:53:05
   340   30.280    10 2022/10/25 04:54:20
   417   40.240    10 2022/10/25 04:55:37
   470   46.960     2 2022/10/25 04:56:30
   495   50.120    10 2022/10/25 04:56:55
   583   60.160    10 2022/10/25 04:58:23
   687   70.160    10 2022/10/25 05:00:07
   873   80.080    10 2022/10/25 05:03:13
  1173   84.800    10 2022/10/25 05:08:13
  1473   87.880    10 2022/10/25 05:13:13
  1675   97.080    10 2022/10/25 05:16:35
  1907  107.040    10 2022/10/25 05:20:27
  2207  112.280    10 2022/10/25 05:25:27
  2507  117.680    10 2022/10/25 05:30:27
  2807  123.560    10 2022/10/25 05:35:27
  3107  123.320    10 2022/10/25 05:40:27
  3407  124.840    10 2022/10/25 05:45:27
  3707  125.640    10 2022/10/25 05:50:27
  4007  125.040    10 2022/10/25 05:55:27
  4308  126.440    10 2022/10/25 06:00:28
  4343  126.520     3 2022/10/25 06:01:03
  4610  125.800    10 2022/10/25 06:05:30
  4910  125.600    10 2022/10/25 06:10:30
  4980  126.280    10 2022/10/25 06:11:40
# Rise Time Series
#  Cycle npts     Start                Date
R00  256   15 720006109 2022/10/25 09:41:49
# time     Pres phase                Date
     0  138.600     7 2022/10/25 09:41:49
   155  128.960    13 2022/10/25 09:44:24
   278  117.560    13 2022/10/25 09:46:27
   401  107.280    13 2022/10/25 09:48:30
   504   97.000    13 2022/10/25 09:50:13
   606   85.840    13 2022/10/25 09:51:55
   709   75.560    13 2022/10/25 09:53:38
   811   65.960    13 2022/10/25 09:55:20
   914   55.800    13 2022/10/25 09:57:03
  1037   45.000    13 2022/10/25 09:59:06
  1180   33.680    13 2022/10/25 10:01:29
  1283   23.720    13 2022/10/25 10:03:12
  1406   12.840    13 2022/10/25 10:05:15
  1529    1.280    13 2022/10/25 10:07:18
  1570    0.440     8 2022/10/25 10:07:59
# Pump Time Series
#   Cycle npts
B02   256   10
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    46.960  270   15.21     201    638    -9     2
   138.600   94   15.33     242    637    -9     7
   128.960   21   15.33     240    640    -9    13
   114.040   21   15.32     239    642    -9    13
   108.960   20   15.32     257    643    -9    13
    46.640   21   15.32     223    649    -9    13
    42.200   20   15.31     226    651    -9    13
    16.000   20   15.30     190    656    -9    13
    11.320   21   15.30     207    659    -9    13
     0.440  240   15.26     198    682    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   256  2.4   13    90   145   600    79    10   210      8     1    4020    5  0.2707
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  256 2022/10/25 04:48:37
# Pcase vacuums
   6.820   6.320
# Oil valve positions
   1.812   0.815   0.630
# Seek attempts and pump time
      1     -1
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  256 2022/10/25 06:11:40 9999/99/99 99:99:99
# Pcase vacuums
   6.320   6.310
# Average 
#   P      T      S
  135.760   5.146  34.497
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  256 2022/10/25 09:41:49 9999/99/99 99:99:99
# Pcase vacuums
   6.310   6.600
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  256 2022/10/25 10:07:58
# number of packets previous packets
      6               7
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.600   6.820
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.176   0.158   0.667
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.740   0.876
# Encountered Error Codes
#    Cycle Errors
EE20   256      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  256  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.240    466   6.190
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  256       197    2022/10/25 10:19:51
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      256	(cycle number) |
     MPS =        0	(phase) |
     MPC =      256	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      145	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    42104	(vac 0 inHg cal) |
     PV5 =    18698	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      255	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       46	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       10	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      270	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      255	(cycle) |
  PTO[0] =    12600	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      255	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =       10	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      255	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      340	(sec) |
     SSP =      240	(sec) |

