# Message Decoder for Solo2, Alto and Deep Solo Floats; v230712
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11342   227  225 000000000881890  1582     0   6 2022/10/18, 17:25:33,  61.9638  -50.6730   3      0 6,14,14
SBD 11342   227  340 000000000881890  1583     0   5 2022/10/18, 17:25:50,  61.9638  -50.6730   3      0 13
SBD 11342   227  299 000000000881890  1584     0   4 2022/10/18, 17:26:09,  61.9638  -50.6730   3      0 8
SBD 11342   227  335 000000000881890  1585     0   3 2022/10/18, 17:26:19,  61.9638  -50.6730   3      0 3,0,15,13
SBD 11342   227  283 000000000881890  1586     0   2 2022/10/18, 17:26:30,  61.9638  -50.6730   3      0 14,1,2
SBD 11342   227  333 000000000881890  1587     0   1 2022/10/18, 17:26:42,  61.9638  -50.6730   3      0 9,10,11,7,5
SBD 11342   227  328 000000000881890  1588     0   0 2022/10/18, 17:26:59,  61.9393  -50.7213   3      0 0,4,14,14
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         2059           84        2143
# Sensor_ID bytes     % packets Description
          0    48  2.33       2         GPS
          1    87  4.23       1    Pressure
          2   165  8.01       1 Temperature
          3   157  7.63       1    Salinity
          4   248 12.04       1        Fall
          5   128  6.22       1        Rise
          6   125  6.07       1        Pump
          7    35  1.70       1 Drift_times
          8   287 13.94       1     Surface
          9    54  2.62       1  HR/Drift_P
         10    52  2.53       1  HR/Drift_T
         11    52  2.53       1  HR/Drift_S
         13   404 19.62       2  Parameters
         14   151  7.33       5 Engineering
         15    66  3.21       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  227 1 2022/10/18 12:03 +61.82895  -50.66122    0  5 32 39 46  1.7     0
G  227 2 2022/10/18 17:21 +61.94780  -50.69384  180  4 27 35 44  5.2     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   227  126    1     3         0
#      Pres      Temp    Salinity
      1.040     2.6640    32.5000
      2.000     2.6640    32.5000
      3.000     2.6640    32.5000
      4.040     2.6630    32.5000
      5.000     2.6630    32.5010
      5.960     2.6620    32.5020
      7.000     2.6620    32.5030
      8.000     2.6620    32.5020
      9.000     2.6620    32.5030
     10.040     2.6620    32.5020
     11.040     2.6620    32.5020
     12.040     2.6600    32.4990
     13.000     2.6610    32.4990
     13.960     2.6670    32.5160
     15.000     2.6730    32.5330
     16.000     2.7680    32.6900
     16.960     3.0160    33.0270
     17.960     3.3050    33.4180
     18.960     3.3540    33.4800
     19.960     3.4180    33.5610
     21.000     3.4440    33.5870
     22.000     3.4490    33.5900
     23.000     3.4510    33.5930
     23.960     3.4510    33.5930
     24.960     3.4520    33.5940
     26.000     3.4550    33.5970
     27.040     3.4620    33.6040
     28.040     3.4680    33.6100
     29.040     3.4770    33.6190
     30.000     3.4830    33.6250
     31.000     3.5080    33.6400
     32.000     3.5370    33.6600
     32.960     3.5750    33.6800
     34.000     3.6120    33.6990
     34.960     3.6350    33.7120
     36.000     3.6450    33.7180
     37.000     3.6580    33.7240
     38.000     3.6770    33.7330
     39.040     3.6960    33.7430
     40.000     3.7150    33.7540
     40.960     3.7390    33.7620
     42.000     3.7710    33.7750
     43.000     3.7750    33.7790
     43.960     3.7940    33.7830
     45.000     3.8010    33.7900
     46.040     3.8080    33.7920
     47.000     3.8280    33.8010
     47.960     3.8580    33.8140
     49.040     3.8710    33.8220
     50.040     3.8870    33.8270
     50.960     3.9180    33.8420
     52.000     3.9370    33.8510
     53.040     3.9450    33.8550
     53.960     3.9540    33.8620
     55.000     3.9730    33.8660
     56.080     3.9920    33.8780
     57.040     3.9940    33.8790
     58.040     3.9950    33.8800
     59.040     4.0190    33.8850
     60.040     4.0280    33.8930
     61.040     4.0280    33.8950
     61.960     4.0330    33.8940
     62.960     4.0350    33.8970
     64.000     4.0440    33.8980
     65.040     4.0730    33.9140
     66.040     4.0780    33.9150
     67.000     4.0800    33.9160
     68.000     4.0820    33.9180
     69.000     4.0820    33.9180
     70.040     4.0830    33.9170
     71.000     4.0840    33.9190
     72.040     4.0860    33.9190
     73.000     4.0870    33.9210
     74.040     4.0880    33.9210
     75.000     4.0880    33.9210
     76.000     4.0880    33.9210
     77.040     4.0890    33.9220
     78.000     4.0910    33.9220
     79.040     4.0970    33.9270
     80.000     4.0980    33.9270
     80.960     4.1100    33.9320
     82.040     4.1290    33.9380
     82.960     4.1540    33.9500
     83.960     4.1660    33.9570
     85.000     4.1770    33.9630
     86.040     4.1790    33.9640
     87.000     4.1860    33.9680
     87.960     4.2170    33.9760
     89.000     4.2260    33.9870
     90.040     4.2280    33.9880
     91.040     4.2350    33.9920
     92.040     4.2500    33.9990
     93.000     4.2670    34.0070
     94.040     4.2750    34.0110
     95.040     4.2930    34.0210
     96.000     4.3030    34.0230
     96.960     4.3380    34.0420
     98.000     4.3930    34.0630
     99.000     4.4250    34.0830
    100.000     4.4390    34.0930
    101.000     4.4560    34.0980
    102.000     4.5050    34.1190
    102.960     4.5380    34.1340
    104.000     4.6020    34.1640
    105.040     4.6360    34.1810
    106.000     4.6300    34.1830
    107.000     4.6280    34.1810
    108.000     4.6420    34.1860
    109.000     4.6560    34.1890
    110.040     4.6710    34.1940
    111.000     4.6810    34.1990
    111.960     4.6830    34.2010
    113.000     4.6840    34.2030
    114.000     4.6880    34.2050
    115.000     4.7060    34.2130
    116.000     4.7460    34.2260
    117.000     4.7640    34.2380
    118.000     4.7940    34.2450
    119.000     4.8560    34.2630
    120.000     4.8930    34.2800
    121.000     4.9190    34.2950
    122.000     4.9300    34.3040
    123.000     4.9350    34.3100
    124.040     4.9390    34.3140
    125.080     4.9380    34.3150
    125.920     4.9400    34.3170
# Drift Data
#      C Npts comp n_var
I0   227   21    1     4
#      Pres       Temp   Salinity     Time
    119.960     5.0020    34.3480        6
    121.200     4.9940    34.3440      611
    113.360     4.9910    34.3380     1217
    118.680     4.9960    34.3430     1822
    121.360     4.9880    34.3340     2428
    110.720     4.9940    34.3480     3033
    136.240     4.9960    34.3450     3638
    130.920     4.9900    34.3400     4244
    136.880     4.9810    34.3330     4849
    116.440     4.9630    34.3200     5455
    118.720     4.9850    34.3340     6060
    136.600     4.9770    34.3310     6666
    126.080     4.9690    34.3230     7271
    141.960     4.9670    34.3220     7877
    136.480     4.6630    34.1830     8482
    124.400     4.8780    34.2570     9088
    144.720     4.9220    34.2920     9693
    158.360     4.8800    34.2610    10299
    128.040     4.8980    34.2880    10904
    115.120     4.9240    34.3030    11509
    128.920     4.9350    34.3140    12115
# Surface Pressure Time Series
#    C npts n_var               Start
S  227    31     6 2022/10/18 17:18:26
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    70      0.500      2.671   -999.000      14 2022/10/18 17:19:36
    83      0.390      2.676   -999.000      14 2022/10/18 17:19:49
    94      0.440      2.677   -999.000      14 2022/10/18 17:20:00
   105      0.510      2.675   -999.000      14 2022/10/18 17:20:11
   116      0.370      2.679   -999.000      14 2022/10/18 17:20:22
   127      0.430      2.679   -999.000      14 2022/10/18 17:20:33
   139      0.420      2.670   -999.000      14 2022/10/18 17:20:45
   150      0.520      2.683   -999.000      14 2022/10/18 17:20:56
   161      0.460      2.677   -999.000      14 2022/10/18 17:21:07
   172      0.530      2.674   -999.000      14 2022/10/18 17:21:18
   183      0.420      2.685   -999.000      14 2022/10/18 17:21:29
   194      0.370      2.681   -999.000      14 2022/10/18 17:21:40
   205      0.430      2.695   -999.000      14 2022/10/18 17:21:51
   216      0.370      2.688   -999.000      14 2022/10/18 17:22:02
   227      0.480      2.725   -999.000      14 2022/10/18 17:22:13
   238      0.350      2.699   -999.000      14 2022/10/18 17:22:24
   249      0.380      2.685   -999.000      14 2022/10/18 17:22:35
   260      0.450      2.687   -999.000      14 2022/10/18 17:22:46
   271      0.370      2.683   -999.000      14 2022/10/18 17:22:57
   282      0.400      2.711   -999.000      14 2022/10/18 17:23:08
   294      0.380      2.700   -999.000      14 2022/10/18 17:23:20
   305      0.530      2.690   -999.000      14 2022/10/18 17:23:31
   316      0.410      2.684   -999.000      14 2022/10/18 17:23:42
   327      0.390      2.683   -999.000      14 2022/10/18 17:23:53
   338      0.400      2.690   -999.000      14 2022/10/18 17:24:04
   349      0.430      2.696   -999.000      14 2022/10/18 17:24:15
   360      0.390      2.693   -999.000      14 2022/10/18 17:24:26
   371      0.470      2.699   -999.000      14 2022/10/18 17:24:37
   382      0.440      2.693   -999.000      14 2022/10/18 17:24:48
   393      0.410      2.689   -999.000      14 2022/10/18 17:24:59
   404      0.450      2.687   -999.000      14 2022/10/18 17:25:10
# Fall Time Series
#  Cycle npts     Start                Date
F00  227   48 719409802 2022/10/18 12:03:22
# time     Pres phase                Date
     0    0.360     1 2022/10/18 12:03:22
   171    5.160    10 2022/10/18 12:06:13
   207   10.200    10 2022/10/18 12:06:49
   245   15.160    10 2022/10/18 12:07:27
   291   20.080    10 2022/10/18 12:08:13
   327   25.200    10 2022/10/18 12:08:49
   352   30.360    10 2022/10/18 12:09:14
   375   35.280    10 2022/10/18 12:09:37
   401   40.480    10 2022/10/18 12:10:03
   427   45.600    10 2022/10/18 12:10:29
   435   47.120     2 2022/10/18 12:10:37
   450   50.000    10 2022/10/18 12:10:52
   483   55.520    10 2022/10/18 12:11:25
   515   60.400    10 2022/10/18 12:11:57
   564   65.200    10 2022/10/18 12:12:46
   864   69.320    10 2022/10/18 12:17:46
   914   74.200    10 2022/10/18 12:18:36
   953   79.080    10 2022/10/18 12:19:15
   993   84.120    10 2022/10/18 12:19:55
  1035   89.200    10 2022/10/18 12:20:37
  1080   94.120    10 2022/10/18 12:21:22
  1132   99.160    10 2022/10/18 12:22:14
  1193  104.080    10 2022/10/18 12:23:15
  1265  109.000    10 2022/10/18 12:24:27
  1359  114.080    10 2022/10/18 12:26:01
  1474  119.040    10 2022/10/18 12:27:56
  1689  124.000    10 2022/10/18 12:31:31
  1990  128.000    10 2022/10/18 12:36:32
  2290  129.120    10 2022/10/18 12:41:32
  2426  124.960    10 2022/10/18 12:43:48
  2571  120.000    10 2022/10/18 12:46:13
  2871  117.720    10 2022/10/18 12:51:13
  2955  122.040    10 2022/10/18 12:52:37
  3026  127.040    10 2022/10/18 12:53:48
  3091  132.080    10 2022/10/18 12:54:53
  3159  137.040    10 2022/10/18 12:56:01
  3253  142.080    10 2022/10/18 12:57:35
  3548  137.880    10 2022/10/18 13:02:30
  3616  132.880    10 2022/10/18 13:03:38
  3682  127.920    10 2022/10/18 13:04:44
  3752  122.920    10 2022/10/18 13:05:54
  3832  117.960    10 2022/10/18 13:07:14
  3940  113.000    10 2022/10/18 13:09:02
  4240  109.880    10 2022/10/18 13:14:02
  4310  110.840     3 2022/10/18 13:15:12
  4521  114.000    10 2022/10/18 13:18:43
  4821  116.920    10 2022/10/18 13:23:43
  4948  119.800    10 2022/10/18 13:25:50
# Rise Time Series
#  Cycle npts     Start                Date
R00  227   24 719427359 2022/10/18 16:55:59
# time     Pres phase                Date
     0  125.160     7 2022/10/18 16:55:59
   175  120.600    13 2022/10/18 16:58:54
   257  114.600    13 2022/10/18 17:00:16
   319  108.520    13 2022/10/18 17:01:18
   360  104.000    13 2022/10/18 17:01:59
   401   98.960    13 2022/10/18 17:02:40
   442   93.520    13 2022/10/18 17:03:21
   483   87.640    13 2022/10/18 17:04:02
   524   81.560    13 2022/10/18 17:04:43
   565   75.400    13 2022/10/18 17:05:24
   606   69.000    13 2022/10/18 17:06:05
   647   62.800    13 2022/10/18 17:06:46
   688   56.880    13 2022/10/18 17:07:27
   729   51.240    13 2022/10/18 17:08:08
   770   45.840    13 2022/10/18 17:08:49
   811   40.640    13 2022/10/18 17:09:30
   872   33.840    13 2022/10/18 17:10:31
   934   28.240    13 2022/10/18 17:11:33
   995   22.920    13 2022/10/18 17:12:34
  1057   17.960    13 2022/10/18 17:13:36
  1159   12.680    13 2022/10/18 17:15:18
  1241    7.040    13 2022/10/18 17:16:40
  1303    1.360    13 2022/10/18 17:17:42
  1345    0.480     8 2022/10/18 17:18:24
# Pump Time Series
#   Cycle npts
B02   227   11
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    47.120  272   15.21     190    643    -9     2
   125.160   93   15.34     234    639    -9     7
   122.800   21   15.33     236    642    -9    13
   119.360   20   15.33     256    644    -9    13
   114.600   21   15.33     247    646    -9    13
    28.240   20   15.33     208    653    -9    13
    21.280   20   15.31     217    655    -9    13
    16.520   21   15.32     217    657    -9    13
    13.800   20   15.31     210    661    -9    13
     9.960   21   15.31     201    663    -9    13
     0.480  240   15.27     180    685    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   227  2.4   13    90   145   600    79    10   210      8     1    4020    5  0.2707
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  227 2022/10/18 12:03:19
# Pcase vacuums
   6.910   6.340
# Oil valve positions
   1.811   0.816   0.631
# Seek attempts and pump time
      1     -3
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  227 2022/10/18 13:25:50 9999/99/99 99:99:99
# Pcase vacuums
   6.330   6.330
# Average 
#   P      T      S
  127.880   4.947  34.314
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  227 2022/10/18 16:55:59 9999/99/99 99:99:99
# Pcase vacuums
   6.330   6.640
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  227 2022/10/18 17:18:24
# number of packets previous packets
      7               8
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.640   6.850
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.176   0.158   0.667
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.740   0.876
# Encountered Error Codes
#    Cycle Errors
EE20   227      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  227  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.320    430   6.230
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  227       197    2022/10/18 17:26:19
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      227	(cycle number) |
     MPS =        0	(phase) |
     MPC =      227	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      145	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    42104	(vac 0 inHg cal) |
     PV5 =    18698	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      226	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       46	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      270	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      226	(cycle) |
  PTO[0] =    12600	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      226	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      226	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      340	(sec) |
     SSP =      240	(sec) |

