# Message Decoder for Solo2, Alto and Deep Solo Floats; v230712
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11342   217  301 000000000881890  1514     0   6 2022/10/16, 09:54:50,  61.2026  -51.3286   6      0 13,6,14,14
SBD 11342   217  340 000000000881890  1515     0   5 2022/10/16, 09:55:10,  61.2272  -51.2820   6      0 13
SBD 11342   217  299 000000000881890  1516     0   4 2022/10/16, 09:55:21,  61.2223  -51.3867   7      0 8
SBD 11342   217  273 000000000881890  1517     0   3 2022/10/16, 09:55:30,  61.2223  -51.3867   7      0 3,0,15
SBD 11342   217  274 000000000881890  1518     0   2 2022/10/16, 09:55:42,  61.2223  -51.3867   7      0 1,2
SBD 11342   217  295 000000000881890  1519     0   1 2022/10/16, 09:55:53,  61.2026  -51.3286   2      0 10,11,7,5,14
SBD 11342   217  340 000000000881890  1520     0   0 2022/10/16, 09:56:32,  61.1830  -51.2705   6      0 0,4,14,14,9
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         2038           84        2122
# Sensor_ID bytes     % packets Description
          0    48  2.36       2         GPS
          1    99  4.86       1    Pressure
          2   163  8.00       1 Temperature
          3   171  8.39       1    Salinity
          4   208 10.21       1        Fall
          5   143  7.02       1        Rise
          6   125  6.13       1        Pump
          7    35  1.72       1 Drift_times
          8   287 14.08       1     Surface
          9    52  2.55       1  HR/Drift_P
         10    43  2.11       1  HR/Drift_T
         11    43  2.11       1  HR/Drift_S
         13   404 19.82       2  Parameters
         14   151  7.41       5 Engineering
         15    66  3.24       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  217 1 2022/10/16 04:28 +61.11005  -51.29400    0  6 22 37 46  1.5     0
G  217 2 2022/10/16 09:49 +61.17108  -51.32426  120  4 35 40 48  2.3     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   217  140    1     3         0
#      Pres      Temp    Salinity
      1.080     2.4060    32.3240
      2.000     2.4070    32.3230
      3.040     2.4070    32.3240
      3.960     2.4120    32.3250
      5.000     2.4160    32.3250
      5.960     2.4320    32.3310
      7.000     2.4670    32.3560
      8.000     2.4780    32.3890
      8.960     2.4810    32.4130
      9.960     2.4890    32.4450
     11.000     2.5290    32.4740
     12.080     2.5510    32.4840
     13.040     2.5680    32.4980
     14.000     2.5650    32.5010
     15.000     2.5520    32.5040
     16.000     2.5410    32.5040
     17.040     2.5460    32.5060
     18.040     2.5570    32.5090
     19.040     2.5730    32.5120
     20.000     2.6110    32.5230
     21.000     2.6210    32.5300
     22.000     2.6290    32.5320
     23.000     2.6340    32.5350
     24.000     2.6490    32.5370
     25.040     2.6890    32.5870
     26.000     2.7220    32.6490
     26.960     2.7510    32.6770
     27.960     2.8450    32.8610
     29.000     2.9400    33.0910
     30.000     2.9600    33.1340
     31.000     3.0290    33.2320
     32.040     3.0620    33.2770
     33.040     3.0870    33.3110
     34.040     3.0870    33.3220
     35.000     3.0940    33.3300
     36.000     3.0990    33.3400
     37.000     3.1080    33.3630
     38.000     3.1880    33.4230
     39.000     3.3580    33.5310
     40.040     3.4500    33.5850
     41.000     3.5870    33.6560
     42.000     3.6990    33.7160
     42.960     3.7360    33.7420
     44.000     3.7450    33.7460
     45.000     3.7550    33.7530
     46.000     3.7770    33.7600
     47.000     3.8240    33.7820
     48.000     3.8350    33.7880
     49.040     3.8350    33.7890
     50.000     3.8370    33.7900
     50.960     3.8360    33.7900
     52.000     3.8360    33.7900
     53.000     3.8350    33.7910
     54.000     3.8320    33.7920
     55.000     3.8480    33.8000
     56.000     3.8810    33.8120
     56.960     3.9200    33.8250
     57.960     3.9680    33.8470
     59.040     4.0020    33.8640
     60.040     4.0590    33.8830
     61.000     4.1000    33.9000
     61.960     4.1170    33.9100
     62.960     4.1380    33.9230
     63.960     4.1590    33.9320
     65.040     4.1790    33.9450
     66.080     4.2010    33.9650
     67.040     4.2160    33.9790
     68.000     4.2160    33.9800
     69.000     4.2240    33.9820
     69.960     4.2570    34.0010
     71.000     4.2680    34.0060
     72.080     4.2870    34.0110
     73.040     4.2990    34.0180
     74.040     4.3030    34.0190
     75.000     4.3050    34.0200
     75.960     4.3080    34.0210
     76.960     4.3110    34.0220
     78.000     4.3130    34.0220
     79.000     4.3170    34.0230
     80.040     4.3190    34.0240
     81.040     4.3260    34.0260
     82.000     4.3310    34.0300
     82.960     4.3380    34.0310
     83.960     4.3580    34.0390
     84.960     4.3680    34.0410
     85.960     4.3880    34.0480
     87.000     4.4140    34.0540
     88.000     4.4670    34.0720
     89.000     4.5080    34.0850
     90.000     4.5380    34.0990
     91.000     4.5450    34.1010
     92.000     4.5520    34.1050
     93.000     4.5890    34.1170
     94.040     4.6350    34.1450
     95.000     4.6690    34.1650
     95.960     4.6900    34.1740
     97.000     4.7040    34.1800
     98.040     4.7280    34.1920
     99.040     4.7560    34.2070
    100.040     4.7610    34.2100
    101.040     4.7650    34.2120
    102.000     4.7780    34.2150
    103.000     4.8100    34.2270
    103.960     4.8550    34.2480
    105.000     4.8950    34.2700
    106.040     4.9190    34.2930
    107.040     4.9320    34.3050
    108.000     4.9320    34.3060
    108.960     4.9360    34.3060
    110.000     4.9500    34.3180
    111.040     4.9450    34.3240
    112.040     4.9530    34.3320
    113.040     4.9610    34.3380
    114.000     4.9790    34.3530
    115.000     4.9830    34.3680
    116.040     5.0020    34.3920
    117.000     5.0090    34.4030
    118.000     5.0140    34.4110
    118.960     5.0170    34.4140
    120.000     5.0190    34.4150
    121.000     5.0330    34.4240
    122.000     5.0420    34.4310
    123.000     5.0440    34.4350
    123.960     5.0440    34.4400
    124.960     5.0440    34.4440
    126.000     5.0450    34.4450
    127.000     5.0460    34.4470
    128.000     5.0480    34.4480
    129.000     5.0560    34.4550
    130.000     5.0680    34.4720
    131.000     5.0740    34.4840
    132.000     5.0780    34.4940
    133.000     5.0630    34.5070
    134.000     5.0650    34.5140
    135.000     5.0640    34.5190
    136.000     5.0890    34.5360
    137.040     5.1010    34.5480
    138.040     5.1070    34.5550
    139.080     5.1090    34.5630
    139.560     5.1090    34.5650
# Drift Data
#      C Npts comp n_var
I0   217   21    1     4
#      Pres       Temp   Salinity     Time
    126.040     5.0800    34.6060        6
    128.360     5.0950    34.6000      611
    128.480     5.0950    34.5990     1217
    129.400     5.0910    34.6030     1822
    127.920     5.0950    34.5970     2428
    134.320     5.0960    34.5980     3033
    129.160     5.1070    34.5770     3639
    130.600     5.0950    34.6010     4244
    132.080     5.1090    34.5750     4849
    133.440     5.0980    34.5910     5455
    136.080     5.1070    34.5760     6060
    138.080     5.1090    34.5770     6666
    144.840     5.1070    34.5770     7271
    141.120     5.1060    34.5500     7877
    134.800     5.1080    34.5680     8482
    141.560     5.1030    34.5810     9088
    143.920     5.1040    34.5490     9693
    143.960     5.1070    34.5510    10299
    141.880     5.1090    34.5540    10904
    141.160     5.1100    34.5600    11510
    144.360     5.1090    34.5600    12115
# Surface Pressure Time Series
#    C npts n_var               Start
S  217    31     6 2022/10/16 09:47:47
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    72      0.520      2.401   -999.000      14 2022/10/16 09:48:59
    85      0.520      2.409   -999.000      14 2022/10/16 09:49:12
    96      0.540      2.407   -999.000      14 2022/10/16 09:49:23
   107      0.500      2.410   -999.000      14 2022/10/16 09:49:34
   118      0.550      2.430   -999.000      14 2022/10/16 09:49:45
   129      0.450      2.409   -999.000      14 2022/10/16 09:49:56
   140      0.480      2.414   -999.000      14 2022/10/16 09:50:07
   151      0.480      2.413   -999.000      14 2022/10/16 09:50:18
   162      0.510      2.420   -999.000      14 2022/10/16 09:50:29
   173      0.460      2.436   -999.000      14 2022/10/16 09:50:40
   184      0.450      2.427   -999.000      14 2022/10/16 09:50:51
   195      0.500      2.437   -999.000      14 2022/10/16 09:51:02
   206      0.470      2.427   -999.000      14 2022/10/16 09:51:13
   217      0.490      2.426   -999.000      14 2022/10/16 09:51:24
   229      0.470      2.454   -999.000      14 2022/10/16 09:51:36
   240      0.480      2.469   -999.000      14 2022/10/16 09:51:47
   251      0.450      2.451   -999.000      14 2022/10/16 09:51:58
   262      0.350      2.450   -999.000      14 2022/10/16 09:52:09
   273      0.450      2.506   -999.000      14 2022/10/16 09:52:20
   284      0.460      2.487   -999.000      14 2022/10/16 09:52:31
   295      0.470      2.441   -999.000      14 2022/10/16 09:52:42
   306      0.500      2.429   -999.000      14 2022/10/16 09:52:53
   317      0.480      2.474   -999.000      14 2022/10/16 09:53:04
   328      0.450      2.454   -999.000      14 2022/10/16 09:53:15
   339      0.570      2.445   -999.000      14 2022/10/16 09:53:26
   350      0.480      2.445   -999.000      14 2022/10/16 09:53:37
   362      0.460      2.551   -999.000      14 2022/10/16 09:53:49
   373      0.490      2.470   -999.000      14 2022/10/16 09:54:00
   384      0.430      2.436   -999.000      14 2022/10/16 09:54:11
   395      0.430      2.438   -999.000      14 2022/10/16 09:54:22
   406      0.510      2.472   -999.000      14 2022/10/16 09:54:33
# Fall Time Series
#  Cycle npts     Start                Date
F00  217   40 719209688 2022/10/16 04:28:08
# time     Pres phase                Date
     0    0.360     1 2022/10/16 04:28:08
   210    5.120    10 2022/10/16 04:31:38
   245   10.440    10 2022/10/16 04:32:13
   278   15.400    10 2022/10/16 04:32:46
   310   20.400    10 2022/10/16 04:33:18
   343   25.480    10 2022/10/16 04:33:51
   372   30.400    10 2022/10/16 04:34:20
   401   35.440    10 2022/10/16 04:34:49
   430   40.400    10 2022/10/16 04:35:18
   460   45.480    10 2022/10/16 04:35:48
   467   46.760     2 2022/10/16 04:35:55
   489   50.280    10 2022/10/16 04:36:17
   521   55.160    10 2022/10/16 04:36:49
   557   60.160    10 2022/10/16 04:37:25
   596   65.120    10 2022/10/16 04:38:04
   641   70.040    10 2022/10/16 04:38:49
   700   75.080    10 2022/10/16 04:39:48
   793   80.040    10 2022/10/16 04:41:21
   903   85.040    10 2022/10/16 04:43:11
  1020   90.040    10 2022/10/16 04:45:08
  1184   95.000    10 2022/10/16 04:47:52
  1388  100.040    10 2022/10/16 04:51:16
  1599  105.000    10 2022/10/16 04:54:47
  1737  110.000    10 2022/10/16 04:57:05
  1876  115.000    10 2022/10/16 04:59:24
  2176  118.000    10 2022/10/16 05:04:24
  2476  117.960    10 2022/10/16 05:09:24
  2668  122.000    10 2022/10/16 05:12:36
  2853  127.000    10 2022/10/16 05:15:41
  3153  131.200    10 2022/10/16 05:20:41
  3453  127.000    10 2022/10/16 05:25:41
  3753  129.160    10 2022/10/16 05:30:41
  3906  134.040    10 2022/10/16 05:33:14
  4206  134.120    10 2022/10/16 05:38:14
  4321  129.960    10 2022/10/16 05:40:09
  4339  129.200     3 2022/10/16 05:40:27
  4445  124.880    10 2022/10/16 05:42:13
  4646  120.000    10 2022/10/16 05:45:34
  4909  124.040    10 2022/10/16 05:49:57
  4977  125.880    10 2022/10/16 05:51:05
# Rise Time Series
#  Cycle npts     Start                Date
R00  217   27 719227274 2022/10/16 09:21:14
# time     Pres phase                Date
     0  139.720     7 2022/10/16 09:21:14
   134  133.960    13 2022/10/16 09:23:28
   216  128.200    13 2022/10/16 09:24:50
   278  123.360    13 2022/10/16 09:25:52
   339  117.880    13 2022/10/16 09:26:53
   401  112.040    13 2022/10/16 09:27:55
   462  105.920    13 2022/10/16 09:28:56
   524  100.160    13 2022/10/16 09:29:58
   585   94.400    13 2022/10/16 09:30:59
   647   88.960    13 2022/10/16 09:32:01
   708   83.240    13 2022/10/16 09:33:02
   770   76.920    13 2022/10/16 09:34:04
   831   70.200    13 2022/10/16 09:35:05
   872   65.720    13 2022/10/16 09:35:46
   934   59.400    13 2022/10/16 09:36:48
   995   53.600    13 2022/10/16 09:37:49
  1057   48.200    13 2022/10/16 09:38:51
  1118   43.320    13 2022/10/16 09:39:52
  1180   38.960    13 2022/10/16 09:40:54
  1241   33.920    13 2022/10/16 09:41:55
  1303   27.120    13 2022/10/16 09:42:57
  1344   22.680    13 2022/10/16 09:43:38
  1405   15.560    13 2022/10/16 09:44:39
  1446   10.840    13 2022/10/16 09:45:20
  1508    5.040    13 2022/10/16 09:46:22
  1580    0.560    13 2022/10/16 09:47:34
  1591    0.560     8 2022/10/16 09:47:45
# Pump Time Series
#   Cycle npts
B02   217   11
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    46.760  270   15.18     201    642    -9     2
   139.720   93   15.32     253    637    -9     7
   133.960   21   15.33     244    640    -9    13
   129.680   21   15.32     262    642    -9    13
   124.960   21   15.32     257    643    -9    13
   119.800   21   15.31     252    645    -9    13
    48.200   21   15.31     235    651    -9    13
    43.320   21   15.30     227    653    -9    13
    38.960   21   15.31     194    656    -9    13
     5.040   21   15.30     211    662    -9    13
     0.400  240   15.26     180    685    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   217  2.4   13    90   145   600    79    10   210      8     1    4020    5  0.2707
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  217 2022/10/16 04:28:05
# Pcase vacuums
   7.040   6.330
# Oil valve positions
   1.810   0.817   0.631
# Seek attempts and pump time
      1     -1
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  217 2022/10/16 05:51:05 9999/99/99 99:99:99
# Pcase vacuums
   6.330   6.310
# Average 
#   P      T      S
  135.800   5.102  34.579
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  217 2022/10/16 09:21:14 9999/99/99 99:99:99
# Pcase vacuums
   6.310   6.630
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  217 2022/10/16 09:47:45
# number of packets previous packets
      7               7
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.630   6.850
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.176   0.158   0.667
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.740   0.876
# Encountered Error Codes
#    Cycle Errors
EE20   217      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  217  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.360    463   6.230
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  217       197    2022/10/16 09:55:10
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      217	(cycle number) |
     MPS =        0	(phase) |
     MPC =      217	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      145	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    42104	(vac 0 inHg cal) |
     PV5 =    18698	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      216	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       46	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      269	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65522	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      216	(cycle) |
  PTO[0] =    12600	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      216	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      216	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      340	(sec) |
     SSP =      240	(sec) |

