# Message Decoder for Solo2, Alto and Deep Solo Floats; v230712
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11342   166  258 000000000881890  1152     0   6 2022/10/05, 03:23:20,  60.6731  -53.1594 234      2 6,14,14
SBD 11342   166  340 000000000881890  1153     0   5 2022/10/05, 03:23:32,  60.8255  -49.0078   6      0 13
SBD 11342   166  299 000000000881890  1154     0   4 2022/10/05, 03:23:47,  60.8286  -48.9040   5      0 8
SBD 11342   166  337 000000000881890  1155     0   3 2022/10/05, 03:24:00,  60.8492  -48.9595   4      0 3,0,15,13
SBD 11342   166  290 000000000881890  1156     0   2 2022/10/05, 03:24:16,  60.8286  -48.9040   4      0 1,2
SBD 11342   166  297 000000000881890  1157     0   1 2022/10/05, 03:24:27,  60.8255  -49.0078   4      0 10,11,7,5,14
SBD 11342   166  325 000000000881890  1158     0   0 2022/10/05, 03:24:45,  60.8255  -49.0078   4      0 0,4,14,14,9
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         2062           84        2146
# Sensor_ID bytes     % packets Description
          0    48  2.33       2         GPS
          1   103  5.00       1    Pressure
          2   175  8.49       1 Temperature
          3   159  7.71       1    Salinity
          4   193  9.36       1        Fall
          5   143  6.94       1        Rise
          6   158  7.66       1        Pump
          7    35  1.70       1 Drift_times
          8   287 13.92       1     Surface
          9    52  2.52       1  HR/Drift_P
         10    44  2.13       1  HR/Drift_T
         11    44  2.13       1  HR/Drift_S
         13   404 19.59       2  Parameters
         14   151  7.32       5 Engineering
         15    66  3.20       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  166 1 2022/10/04 21:57 +60.80328  -49.00064    0  4 30 41 46  3.0     0
G  166 2 2022/10/05 03:19 +60.81108  -49.00298  190  5 24 36 41  1.1     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   166  144    1     3         0
#      Pres      Temp    Salinity
      1.120     2.4770    31.6220
      2.040     2.4770    31.6240
      3.000     2.4780    31.6250
      4.040     2.4780    31.6250
      5.040     2.4800    31.6260
      6.000     2.4800    31.6260
      7.040     2.4840    31.6320
      8.040     2.4850    31.6320
      9.000     2.4850    31.6330
     10.000     2.4850    31.6330
     11.040     2.4850    31.6330
     12.000     2.4850    31.6330
     13.000     2.4850    31.6330
     14.000     2.4860    31.6360
     15.000     2.4850    31.6370
     15.960     2.4850    31.6370
     16.960     2.4840    31.6390
     17.960     2.4810    31.6470
     19.000     2.4800    31.6510
     20.000     2.4780    31.6530
     20.960     2.4780    31.6540
     21.960     2.4710    31.6640
     23.000     2.4550    31.6930
     24.040     2.4500    31.7070
     25.040     2.4470    31.7160
     26.040     2.4460    31.7190
     27.000     2.4440    31.7260
     28.000     2.4440    31.7290
     29.000     2.4430    31.7360
     30.000     2.4420    31.7440
     31.000     2.4360    31.7600
     32.000     2.4250    31.7860
     32.960     2.4070    31.8320
     34.000     2.4060    31.8360
     35.000     2.4060    31.8370
     36.000     2.4040    31.8420
     37.000     2.3950    31.8910
     38.000     2.3900    31.9140
     38.960     2.3790    31.9720
     39.960     2.3990    32.0430
     40.960     2.4160    32.1390
     42.000     2.3750    32.1700
     43.000     2.3510    32.1920
     44.000     2.3720    32.2310
     45.000     2.3950    32.2560
     45.960     2.4480    32.2880
     47.000     2.5730    32.3860
     48.040     2.5770    32.4460
     49.000     2.5730    32.4930
     50.000     2.6210    32.5370
     51.000     2.6490    32.5930
     52.000     2.6610    32.6100
     53.000     2.6690    32.6260
     54.000     2.6680    32.6560
     55.000     2.6720    32.7150
     56.000     2.6810    32.7260
     57.000     2.6760    32.7600
     58.000     2.6530    32.8270
     59.000     2.6450    32.8490
     59.960     2.6450    32.8590
     61.000     2.6420    32.8600
     62.040     2.6410    32.8640
     63.040     2.6390    32.8900
     64.000     2.6800    32.9510
     65.000     2.7310    32.9920
     66.000     2.7580    33.0180
     67.000     2.7820    33.0470
     68.040     2.8390    33.0890
     69.000     2.9680    33.2130
     69.960     2.9790    33.2210
     71.000     3.0050    33.2380
     72.000     3.0670    33.2660
     73.000     3.1190    33.2930
     74.040     3.1240    33.3050
     75.040     3.1610    33.3510
     76.040     3.1990    33.4210
     76.960     3.2430    33.4390
     77.960     3.2700    33.4610
     79.000     3.2630    33.4640
     80.000     3.2560    33.4700
     81.040     3.2660    33.4790
     82.040     3.2610    33.4850
     83.000     3.2730    33.4890
     84.000     3.2680    33.4940
     85.040     3.2630    33.5010
     86.040     3.2580    33.5080
     87.000     3.2620    33.5120
     87.960     3.2710    33.5140
     89.000     3.2790    33.5170
     90.000     3.2910    33.5210
     91.000     3.3240    33.5270
     91.960     3.3570    33.5370
     93.000     3.3950    33.5400
     94.040     3.4910    33.5590
     95.000     3.5560    33.5830
     96.000     3.5530    33.5950
     97.000     3.5610    33.6000
     98.000     3.9200    33.6770
     99.000     4.1400    33.7310
    100.000     4.3060    33.7800
    101.000     4.4020    33.8390
    102.000     4.4220    33.8660
    103.000     4.4320    33.8740
    103.960     4.4460    33.8930
    104.960     4.4290    33.9330
    106.000     4.3360    33.9550
    107.000     4.3200    33.9550
    107.960     4.3120    33.9570
    109.000     4.3090    33.9630
    110.040     4.3370    33.9760
    111.040     4.3810    33.9860
    112.000     4.4460    34.0110
    113.000     4.4970    34.0310
    114.000     4.5410    34.0380
    115.000     4.6130    34.0520
    116.040     4.6510    34.0730
    117.040     4.6280    34.0810
    118.000     4.6050    34.0790
    119.000     4.6670    34.0940
    120.000     4.6990    34.1080
    121.000     4.7130    34.1150
    122.000     4.7310    34.1300
    123.000     4.7430    34.1380
    124.000     4.7620    34.1460
    125.000     4.7710    34.1470
    126.000     4.7940    34.1520
    126.960     4.7990    34.1550
    127.960     4.8050    34.1570
    129.000     4.8150    34.1610
    129.960     4.8090    34.1610
    130.960     4.8200    34.1600
    132.000     4.8320    34.1690
    133.000     4.8290    34.1730
    134.000     4.8270    34.1750
    134.960     4.8630    34.1890
    136.000     4.8760    34.1950
    137.040     4.8790    34.1990
    138.000     4.8880    34.2030
    139.000     4.8940    34.2050
    140.040     4.9110    34.2120
    141.000     4.9210    34.2190
    142.040     4.9230    34.2210
    143.080     4.9260    34.2230
    143.520     4.9280    34.2240
# Drift Data
#      C Npts comp n_var
I0   166   21    1     4
#      Pres       Temp   Salinity     Time
    135.320     4.9580    34.2410        6
    135.760     4.9670    34.2460      611
    135.320     4.9600    34.2440     1217
    133.960     4.9620    34.2440     1822
    134.560     4.9640    34.2450     2428
    136.320     4.9630    34.2460     3033
    136.520     4.9590    34.2430     3639
    137.680     4.9610    34.2420     4244
    136.440     4.9520    34.2360     4850
    139.040     4.9350    34.2310     5455
    141.880     4.9420    34.2340     6061
    140.720     4.9410    34.2340     6666
    139.520     4.9480    34.2340     7272
    133.800     4.9490    34.2350     7877
    135.240     4.9540    34.2400     8482
    137.840     4.9550    34.2400     9088
    141.000     4.9540    34.2390     9693
    141.560     4.9260    34.2240    10299
    143.720     4.9310    34.2210    10904
    142.920     4.9310    34.2210    11510
    142.040     4.9350    34.2260    12115
# Surface Pressure Time Series
#    C npts n_var               Start
S  166    31     6 2022/10/05 03:16:16
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    72      0.070      2.487   -999.000      14 2022/10/05 03:17:28
    85      0.160      2.486   -999.000      14 2022/10/05 03:17:41
    96      0.120      2.487   -999.000      14 2022/10/05 03:17:52
   107      0.130      2.489   -999.000      14 2022/10/05 03:18:03
   118      0.150      2.490   -999.000      14 2022/10/05 03:18:14
   129      0.180      2.487   -999.000      14 2022/10/05 03:18:25
   141      0.070      2.485   -999.000      14 2022/10/05 03:18:37
   152      0.130      2.496   -999.000      14 2022/10/05 03:18:48
   163      0.130      2.499   -999.000      14 2022/10/05 03:18:59
   174      0.100      2.494   -999.000      14 2022/10/05 03:19:10
   185      0.080      2.493   -999.000      14 2022/10/05 03:19:21
   196      0.130      2.487   -999.000      14 2022/10/05 03:19:32
   207      0.060      2.519   -999.000      14 2022/10/05 03:19:43
   218      0.060      2.507   -999.000      14 2022/10/05 03:19:54
   229      0.070      2.496   -999.000      14 2022/10/05 03:20:05
   240      0.070      2.507   -999.000      14 2022/10/05 03:20:16
   251      0.020      2.506   -999.000      14 2022/10/05 03:20:27
   262      0.110      2.547   -999.000      14 2022/10/05 03:20:38
   273      0.000      2.533   -999.000      14 2022/10/05 03:20:49
   284      0.100      2.557   -999.000      14 2022/10/05 03:21:00
   295      0.070      2.504   -999.000      14 2022/10/05 03:21:11
   307      0.110      2.568   -999.000      14 2022/10/05 03:21:23
   318      0.090      2.548   -999.000      14 2022/10/05 03:21:34
   329      0.040      2.532   -999.000      14 2022/10/05 03:21:45
   340      0.030      2.506   -999.000      14 2022/10/05 03:21:56
   351      0.100      2.510   -999.000      14 2022/10/05 03:22:07
   362      0.050      2.563   -999.000      14 2022/10/05 03:22:18
   373      0.000      2.523   -999.000      14 2022/10/05 03:22:29
   384      0.160      2.495   -999.000      14 2022/10/05 03:22:40
   395      0.050      2.502   -999.000      14 2022/10/05 03:22:51
   406      0.030      2.506   -999.000      14 2022/10/05 03:23:02
# Fall Time Series
#  Cycle npts     Start                Date
F00  166   37 718235863 2022/10/04 21:57:43
# time     Pres phase                Date
     0    0.080     1 2022/10/04 21:57:43
   151    5.280    10 2022/10/04 22:00:14
   177   10.080    10 2022/10/04 22:00:40
   206   15.040    10 2022/10/04 22:01:09
   239   20.080    10 2022/10/04 22:01:42
   278   25.400    10 2022/10/04 22:02:21
   310   30.360    10 2022/10/04 22:02:53
   343   35.480    10 2022/10/04 22:03:26
   353   37.160     2 2022/10/04 22:03:36
   375   40.320    10 2022/10/04 22:03:58
   411   45.280    10 2022/10/04 22:04:34
   456   50.280    10 2022/10/04 22:05:19
   505   55.120    10 2022/10/04 22:06:08
   560   60.000    10 2022/10/04 22:07:03
   635   65.080    10 2022/10/04 22:08:18
   713   70.080    10 2022/10/04 22:09:36
   792   75.040    10 2022/10/04 22:10:55
   888   80.040    10 2022/10/04 22:12:31
  1005   85.040    10 2022/10/04 22:14:28
  1109   90.040    10 2022/10/04 22:16:12
  1209   95.040    10 2022/10/04 22:17:52
  1329  100.080    10 2022/10/04 22:19:52
  1446  105.080    10 2022/10/04 22:21:49
  1664  110.000    10 2022/10/04 22:25:27
  1877  115.000    10 2022/10/04 22:29:00
  2170  120.040    10 2022/10/04 22:33:53
  2409  125.040    10 2022/10/04 22:37:52
  2629  130.040    10 2022/10/04 22:41:32
  2929  134.640    10 2022/10/04 22:46:32
  3229  136.960    10 2022/10/04 22:51:32
  3529  139.240    10 2022/10/04 22:56:32
  3829  140.800    10 2022/10/04 23:01:32
  4129  142.200    10 2022/10/04 23:06:32
  4230  142.080     3 2022/10/04 23:08:13
  4429  140.160    10 2022/10/04 23:11:32
  4729  136.440    10 2022/10/04 23:16:32
  4870  135.320    10 2022/10/04 23:18:53
# Rise Time Series
#  Cycle npts     Start                Date
R00  166   27 718253342 2022/10/05 02:49:02
# time     Pres phase                Date
     0  143.640     7 2022/10/05 02:49:02
   114  138.800    13 2022/10/05 02:50:56
   176  133.560    13 2022/10/05 02:51:58
   237  128.200    13 2022/10/05 02:52:59
   299  122.400    13 2022/10/05 02:54:01
   360  117.000    13 2022/10/05 02:55:02
   422  111.400    13 2022/10/05 02:56:04
   483  105.840    13 2022/10/05 02:57:05
   545  100.080    13 2022/10/05 02:58:07
   606   94.760    13 2022/10/05 02:59:08
   668   88.840    13 2022/10/05 03:00:10
   729   82.560    13 2022/10/05 03:01:11
   791   76.000    13 2022/10/05 03:02:13
   832   71.960    13 2022/10/05 03:02:54
   893   66.320    13 2022/10/05 03:03:55
   955   61.120    13 2022/10/05 03:04:57
  1016   55.800    13 2022/10/05 03:05:58
  1078   50.240    13 2022/10/05 03:07:00
  1139   44.480    13 2022/10/05 03:08:01
  1201   38.520    13 2022/10/05 03:09:03
  1262   33.040    13 2022/10/05 03:10:04
  1324   27.640    13 2022/10/05 03:11:06
  1385   21.720    13 2022/10/05 03:12:07
  1447   15.760    13 2022/10/05 03:13:09
  1508    9.720    13 2022/10/05 03:14:10
  1570    3.480    13 2022/10/05 03:15:12
  1632    0.200     8 2022/10/05 03:16:14
# Pump Time Series
#   Cycle npts
B02   166   14
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    37.160  273   15.22     216    649    -9     2
   141.920    1   15.26     358    634    -9     3
   143.640   94   15.36     267    640    -9     7
   137.080   20   15.35     268    641    -9    13
   131.880   21   15.35     237    644    -9    13
   120.680   21   15.34     253    646    -9    13
   113.320   21   15.34     249    648    -9    13
    94.760   21   15.33     239    652    -9    13
    66.320   21   15.33     216    657    -9    13
    59.360   21   15.32     220    660    -9    13
    52.160   21   15.32     220    662    -9    13
    44.480   21   15.32     228    665    -9    13
    29.520   21   15.31     225    667    -9    13
     0.000  240   15.27     209    689    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   166  2.4   11    90   125   600    79    10   210      8     1    4020    5  0.2707
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  166 2022/10/04 21:57:40
# Pcase vacuums
   6.880   6.340
# Oil valve positions
   1.806   0.820   0.631
# Seek attempts and pump time
      1      1
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  166 2022/10/04 23:18:53 9999/99/99 99:99:99
# Pcase vacuums
   6.340   6.330
# Average 
#   P      T      S
  138.160   4.950  34.237
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  166 2022/10/05 02:49:02 9999/99/99 99:99:99
# Pcase vacuums
   6.330   6.680
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  166 2022/10/05 03:16:14
# number of packets previous packets
      7               8
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.680   6.890
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.176   0.158   0.667
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.740   0.876
# Encountered Error Codes
#    Cycle Errors
EE20   166      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  166  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.080    349   6.260
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  166       197    2022/10/05 03:24:00
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      166	(cycle number) |
     MPS =        0	(phase) |
     MPC =      166	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      125	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    42104	(vac 0 inHg cal) |
     PV5 =    18698	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      165	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       36	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      273	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      165	(cycle) |
  PTO[0] =    12600	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      165	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      165	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      340	(sec) |
     SSP =      240	(sec) |

