# Message Decoder for Solo2, Alto and Deep Solo Floats; v230712
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11342   160  247 000000000881890  1110     0   6 2022/10/03, 18:40:55,  60.7462  -48.8417   4      0 6,14,14
SBD 11342   160  340 000000000881890  1111     0   5 2022/10/03, 18:41:06,  60.7432  -48.9452   4      0 13
SBD 11342   160  299 000000000881890  1112     0   4 2022/10/03, 18:41:19,  60.7432  -48.9452   5      0 8
SBD 11342   160  315 000000000881890  1113     0   3 2022/10/03, 18:41:31,  60.7432  -48.9452   5      0 3,0,15,13
SBD 11342   160  255 000000000881890  1114     0   2 2022/10/03, 18:41:44,  60.7432  -48.9452   6      0 1,2
SBD 11342   160  287 000000000881890  1115     0   1 2022/10/03, 18:41:58,  60.7257  -48.9452   7      0 10,11,7,5,14
SBD 11342   160  322 000000000881890  1116     0   0 2022/10/03, 18:42:22,  60.7257  -48.9452   8      0 0,4,14,14,9
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1981           84        2065
# Sensor_ID bytes     % packets Description
          0    48  2.42       2         GPS
          1    90  4.54       1    Pressure
          2   153  7.72       1 Temperature
          3   137  6.92       1    Salinity
          4   188  9.49       1        Fall
          5   123  6.21       1        Rise
          6   147  7.42       1        Pump
          7    35  1.77       1 Drift_times
          8   287 14.49       1     Surface
          9    54  2.73       1  HR/Drift_P
         10    54  2.73       1  HR/Drift_T
         11    44  2.22       1  HR/Drift_S
         13   404 20.39       2  Parameters
         14   151  7.62       5 Engineering
         15    66  3.33       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  160 1 2022/10/03 13:18 +60.74137  -48.86585    0  5 32 35 44  1.3     0
G  160 2 2022/10/03 18:37 +60.74572  -48.89766  210  3 41 43 44  1.7     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   160  115    1     3         0
#      Pres      Temp    Salinity
      1.080     2.7250    31.3980
      1.960     2.7260    31.3970
      2.960     2.7240    31.3970
      4.000     2.7230    31.3940
      5.040     2.7220    31.3940
      6.000     2.7200    31.3950
      7.000     2.7130    31.3980
      8.040     2.6590    31.4340
      9.040     2.6250    31.4580
     10.040     2.5990    31.4900
     11.000     2.5250    31.6190
     12.000     2.5050    31.6570
     13.040     2.4980    31.6770
     14.040     2.4880    31.6920
     15.040     2.4780    31.7020
     16.040     2.4580    31.7210
     17.000     2.4350    31.7540
     17.960     2.4210    31.7740
     18.960     2.4150    31.7840
     19.920     2.3980    31.8030
     20.960     2.3780    31.8260
     22.000     2.3550    31.8550
     23.000     2.3370    31.8750
     24.040     2.3080    31.9040
     25.040     2.2880    31.9260
     26.000     2.2690    31.9430
     26.960     2.2080    32.0270
     28.000     2.1560    32.1400
     28.960     2.1290    32.2030
     29.960     2.1130    32.2480
     31.000     2.1000    32.2770
     32.000     2.0810    32.3140
     32.960     2.0800    32.3320
     33.960     2.0750    32.3600
     35.000     2.0560    32.4530
     36.000     2.0470    32.5300
     36.960     2.0470    32.5550
     37.960     2.0460    32.5600
     39.000     2.0520    32.5630
     40.000     2.0620    32.5730
     41.000     2.0770    32.5840
     42.040     2.0840    32.5910
     43.040     2.0890    32.5940
     44.040     2.0930    32.5980
     45.000     2.0970    32.6080
     46.000     2.0990    32.6180
     47.040     2.0990    32.6260
     48.000     2.1320    32.6990
     49.000     2.1780    32.7770
     50.000     2.1970    32.7950
     51.040     2.2140    32.8160
     52.000     2.2290    32.8290
     53.000     2.2490    32.8470
     54.000     2.2620    32.8620
     55.040     2.2660    32.8720
     56.040     2.2800    32.8850
     57.000     2.3240    32.9040
     57.960     2.3340    32.9180
     59.000     2.3510    32.9430
     60.000     2.3640    32.9700
     61.000     2.3760    32.9960
     61.960     2.3890    33.0260
     63.000     2.3980    33.0440
     64.040     2.4150    33.0630
     65.040     2.4360    33.0910
     66.000     2.4510    33.1100
     66.960     2.4590    33.1230
     68.000     2.4940    33.1610
     69.000     2.5180    33.1890
     70.000     2.5430    33.2120
     71.040     2.5630    33.2350
     72.000     2.5680    33.2450
     73.000     2.5680    33.2510
     74.000     2.5690    33.2530
     75.000     2.5680    33.2560
     76.000     2.5680    33.2620
     78.000     2.5920    33.2850
     79.000     2.6170    33.3040
     80.000     2.6550    33.3350
     81.040     2.6890    33.3670
     82.040     2.7120    33.3820
     83.000     2.7870    33.4230
     84.000     2.9140    33.4570
     85.000     3.1350    33.5300
     86.000     3.4110    33.5800
     87.000     3.6000    33.6360
     88.040     3.7380    33.6830
     89.040     4.2980    33.7620
     90.000     4.9060    33.9150
     91.040     5.1980    33.9950
     92.040     5.2540    34.0130
     93.000     5.3010    34.0270
     94.000     5.3270    34.0330
     95.000     5.3430    34.0410
     96.000     5.4150    34.0550
     96.960     5.5580    34.0910
     98.000     5.5990    34.1270
     99.000     5.6020    34.1480
    100.000     5.5940    34.1990
    100.960     5.7190    34.2200
    101.960     5.9490    34.2890
    103.000     5.9000    34.2970
    104.040     6.0970    34.3350
    105.040     6.1520    34.3490
    106.000     6.1630    34.3530
    107.000     6.1500    34.3560
    108.000     6.0080    34.3430
    109.000     5.4990    34.3250
    110.000     5.2280    34.2900
    111.040     5.2820    34.2910
    112.000     5.2550    34.3060
    113.040     5.1780    34.2980
    114.000     5.1380    34.3180
    115.040     5.0020    34.3140
    115.960  -999.0000  -999.0000
# Drift Data
#      C Npts comp n_var
I0   160   21    1     4
#      Pres       Temp   Salinity     Time
    108.920     6.2290    34.3780        6
    110.160     6.2180    34.3890      612
    104.520     6.2440    34.3820     1217
    105.520     6.1980    34.3870     1822
    109.760     5.6600    34.3450     2428
    117.840     4.9930    34.2820     3033
    115.040     4.9660    34.2930     3639
    116.440     4.9690    34.2990     4244
    119.520     4.9380    34.3040     4850
    119.360     4.9470    34.3040     5455
    123.920     4.9690    34.3010     6061
    118.240     4.9890    34.2950     6666
    121.720     4.9700    34.3030     7272
    117.680     4.9750    34.2990     7877
    124.760     4.9540    34.3040     8483
    123.280     5.0030    34.2790     9088
    131.600     5.0040    34.2780     9693
    109.280     4.9920    34.2800    10299
    121.120     5.2350    34.3320    10904
    122.000     5.3010    34.3430    11510
    116.200     5.2190    34.2470    12115
# Surface Pressure Time Series
#    C npts n_var               Start
S  160    31     6 2022/10/03 18:33:51
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    71      0.280      2.718   -999.000      14 2022/10/03 18:35:02
    83      0.170      2.693   -999.000      14 2022/10/03 18:35:14
    94      0.220      2.716   -999.000      14 2022/10/03 18:35:25
   106      0.230      2.701   -999.000      14 2022/10/03 18:35:37
   117      0.210      2.713   -999.000      14 2022/10/03 18:35:48
   128      0.220      2.678   -999.000      14 2022/10/03 18:35:59
   139      0.320      2.711   -999.000      14 2022/10/03 18:36:10
   150      0.220      2.646   -999.000      14 2022/10/03 18:36:21
   161      0.160      2.682   -999.000      14 2022/10/03 18:36:32
   172      0.130      2.658   -999.000      14 2022/10/03 18:36:43
   183      0.150      2.693   -999.000      14 2022/10/03 18:36:54
   194      0.080      2.677   -999.000      14 2022/10/03 18:37:05
   205      0.240      2.696   -999.000      14 2022/10/03 18:37:16
   216      0.140      2.702   -999.000      14 2022/10/03 18:37:27
   227      0.170      2.697   -999.000      14 2022/10/03 18:37:38
   238      0.130      2.682   -999.000      14 2022/10/03 18:37:49
   249      0.200      2.668   -999.000      14 2022/10/03 18:38:00
   261      0.190      2.651   -999.000      14 2022/10/03 18:38:12
   272      0.110      2.666   -999.000      14 2022/10/03 18:38:23
   283      0.220      2.691   -999.000      14 2022/10/03 18:38:34
   294      0.090      2.702   -999.000      14 2022/10/03 18:38:45
   305      0.270      2.697   -999.000      14 2022/10/03 18:38:56
   316      0.150      2.697   -999.000      14 2022/10/03 18:39:07
   327      0.140      2.681   -999.000      14 2022/10/03 18:39:18
   338      0.170      2.664   -999.000      14 2022/10/03 18:39:29
   349      0.080      2.583   -999.000      14 2022/10/03 18:39:40
   360      0.250      2.696   -999.000      14 2022/10/03 18:39:51
   371      0.120      2.696   -999.000      14 2022/10/03 18:40:02
   383      0.130      2.661   -999.000      14 2022/10/03 18:40:14
   394      0.180      2.683   -999.000      14 2022/10/03 18:40:25
   405      0.110      2.679   -999.000      14 2022/10/03 18:40:36
# Fall Time Series
#  Cycle npts     Start                Date
F00  160   36 718118339 2022/10/03 13:18:59
# time     Pres phase                Date
     0    0.160     1 2022/10/03 13:18:59
   174    5.440    10 2022/10/03 13:21:53
   206   10.200    10 2022/10/03 13:22:25
   239   15.360    10 2022/10/03 13:22:58
   268   20.040    10 2022/10/03 13:23:27
   300   25.160    10 2022/10/03 13:23:59
   333   30.280    10 2022/10/03 13:24:32
   369   35.440    10 2022/10/03 13:25:08
   379   37.000     2 2022/10/03 13:25:18
   401   40.080    10 2022/10/03 13:25:40
   440   45.200    10 2022/10/03 13:26:19
   486   50.280    10 2022/10/03 13:27:05
   537   55.240    10 2022/10/03 13:27:56
   596   60.240    10 2022/10/03 13:28:55
   664   65.000    10 2022/10/03 13:30:03
   739   70.040    10 2022/10/03 13:31:18
   814   75.040    10 2022/10/03 13:32:33
   914   80.080    10 2022/10/03 13:34:13
  1055   85.000    10 2022/10/03 13:36:34
  1200   90.080    10 2022/10/03 13:38:59
  1331   95.040    10 2022/10/03 13:41:10
  1500  100.040    10 2022/10/03 13:43:59
  1711  105.040    10 2022/10/03 13:47:30
  2011  107.160    10 2022/10/03 13:52:30
  2311  107.280    10 2022/10/03 13:57:30
  2470  112.080    10 2022/10/03 14:00:09
  2770  115.480    10 2022/10/03 14:05:09
  3070  112.760    10 2022/10/03 14:10:09
  3370  110.880    10 2022/10/03 14:15:09
  3670  108.040    10 2022/10/03 14:20:09
  3970  109.800    10 2022/10/03 14:25:09
  4256  109.320     3 2022/10/03 14:29:55
  4270  109.280    10 2022/10/03 14:30:09
  4572  109.480    10 2022/10/03 14:35:11
  4872  109.160    10 2022/10/03 14:40:11
  4895  108.960    10 2022/10/03 14:40:34
# Rise Time Series
#  Cycle npts     Start                Date
R00  160   23 718135843 2022/10/03 18:10:43
# time     Pres phase                Date
     0  116.280     7 2022/10/03 18:10:43
   114  111.680    13 2022/10/03 18:12:37
   176  106.960    13 2022/10/03 18:13:39
   237  101.320    13 2022/10/03 18:14:40
   299   95.400    13 2022/10/03 18:15:42
   360   90.240    13 2022/10/03 18:16:43
   422   85.360    13 2022/10/03 18:17:45
   483   80.360    13 2022/10/03 18:18:46
   545   74.800    13 2022/10/03 18:19:48
   606   69.000    13 2022/10/03 18:20:49
   668   62.720    13 2022/10/03 18:21:51
   729   56.240    13 2022/10/03 18:22:52
   791   50.120    13 2022/10/03 18:23:54
   852   45.040    13 2022/10/03 18:24:55
   914   39.800    13 2022/10/03 18:25:57
   975   34.240    13 2022/10/03 18:26:58
  1037   29.120    13 2022/10/03 18:28:00
  1098   24.080    13 2022/10/03 18:29:01
  1160   18.440    13 2022/10/03 18:30:03
  1221   13.040    13 2022/10/03 18:31:04
  1283    7.160    13 2022/10/03 18:32:06
  1344    1.400    13 2022/10/03 18:33:07
  1386    0.200     8 2022/10/03 18:33:49
# Pump Time Series
#   Cycle npts
B02   160   13
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    37.000  273   15.26     220    645    -9     2
   116.280   94   15.37     261    638    -9     7
   110.120   20   15.37     260    640    -9    13
   105.240   21   15.36     254    640    -9    13
    93.720   21   15.36     235    642    -9    13
    88.520   20   15.36     216    646    -9    13
    83.680   21   15.35     223    651    -9    13
    46.600   20   15.35     222    659    -9    13
    41.560   21   15.34     241    662    -9    13
    32.600   20   15.34     220    664    -9    13
    27.400   21   15.34     233    666    -9    13
    16.720   21   15.33     202    668    -9    13
     0.160  240   15.28     189    688    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   160  2.4   11    90   125   600    79    10   210      8     1    4020    5  0.2707
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  160 2022/10/03 13:18:56
# Pcase vacuums
   6.870   6.260
# Oil valve positions
   1.809   0.821   0.631
# Seek attempts and pump time
      1     -1
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  160 2022/10/03 14:40:34 9999/99/99 99:99:99
# Pcase vacuums
   6.260   6.320
# Average 
#   P      T      S
  117.000   5.284  34.315
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  160 2022/10/03 18:10:43 9999/99/99 99:99:99
# Pcase vacuums
   6.320   6.670
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  160 2022/10/03 18:33:50
# number of packets previous packets
      7               7
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.670   6.880
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.176   0.158   0.667
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.740   0.876
# Encountered Error Codes
#    Cycle Errors
EE20   160      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  160  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.120    375   6.240
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  160       197    2022/10/03 18:41:31
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      160	(cycle number) |
     MPS =        0	(phase) |
     MPC =      160	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      125	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    42104	(vac 0 inHg cal) |
     PV5 =    18698	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      159	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       36	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      272	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      159	(cycle) |
  PTO[0] =    12600	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      159	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      159	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      340	(sec) |
     SSP =      240	(sec) |

