# Message Decoder for Solo2, Alto and Deep Solo Floats; v250616
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11342  1352  249 000000000881890  8623     0   6 2025/06/28, 05:36:13,  66.7850  -53.1301   6      0 6,14,14
SBD 11342  1352  340 000000000881890  8624     0   5 2025/06/28, 05:36:23,  66.7850  -53.1301   5      0 13
SBD 11342  1352  280 000000000881890  8625     0   4 2025/06/28, 05:36:36,  66.7586  -53.1846   4      0 3,8
SBD 11342  1352  278 000000000881890  8626     0   3 2025/06/28, 05:36:47,  66.7586  -53.1846   4      0 2
SBD 11342  1352  308 000000000881890  8627     0   2 2025/06/28, 05:36:58,  66.7850  -53.1301   7      0 1,15,13
SBD 11342  1352  306 000000000881890  8628     0   1 2025/06/28, 05:37:10,  66.8048  -53.2029   7      0 11,7,5,14,0
SBD 11342  1352  331 000000000881890  8629     0   0 2025/06/28, 05:37:45,  66.7652  -53.0574   5      0 0,4,14,14,9,10
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         2008           84        2092
# Sensor_ID bytes     % packets Description
          0    48  2.39       2         GPS
          1   154  7.67       1    Pressure
          2   266 13.25       1 Temperature
          3   206 10.26       1    Salinity
          4   103  5.13       1        Fall
          5   133  6.62       1        Rise
          6   125  6.23       1        Pump
          7    48  2.39       1 Drift_times
          8    62  3.09       1     Surface
          9    78  3.88       1  HR/Drift_P
         10    70  3.49       1  HR/Drift_T
         11    70  3.49       1  HR/Drift_S
         13   404 20.12       2  Parameters
         14   175  8.72       5 Engineering
         15    66  3.29       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G 1352 1 2025/06/27 04:15 +66.75016  -53.11805    0  5 41 43 46  9.5     0
G 1352 2 2025/06/28 05:34 +66.75231  -53.11948   20  9 33 41 46  0.7     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0  1352  251    1     3         0
#      Pres      Temp    Salinity
      1.040     4.8600    32.4480
      1.960     4.8280    32.4510
      3.000     4.5840    32.4910
      3.960     4.3840    32.4890
      4.960     4.0780    32.5230
      6.000     3.6020    32.5700
      7.040     3.3850    32.5750
      8.040     3.3670    32.5740
      9.000     3.3720    32.5800
     10.000     3.3520    32.5910
     11.000     3.2730    32.6120
     12.000     3.2100    32.6020
     12.960     3.0980    32.6020
     14.000     2.4600    32.7240
     15.000     1.6510    32.7030
     16.000     1.4000    32.6910
     17.000     1.2660    32.6920
     18.000     1.2060    32.6850
     19.000     1.0720    32.6990
     20.040     0.8990    32.7010
     21.040     0.7830    32.7130
     22.000     0.6790    32.7200
     23.000     0.5980    32.7100
     24.000     0.5990    32.7100
     25.040     0.5540    32.7120
     26.040     0.5080    32.7150
     27.000     0.4350    32.7260
     28.000     0.3200    32.7310
     28.960     0.2680    32.7300
     30.000     0.2100    32.7300
     31.040     0.2110    32.7310
     32.000     0.2270    32.7260
     33.040     0.2710    32.7290
     34.000     0.3010    32.7300
     35.000     0.3260    32.7380
     35.960     0.2060    32.7540
     37.000     0.2110    32.7240
     38.040     0.1300    32.7600
     39.000     0.2560    32.7210
     39.960     0.3550    32.7430
     41.000     0.3650    32.7510
     42.040     0.3830    32.7590
     43.040     0.4000    32.7730
     44.000     0.4120    32.7670
     45.040     0.4560    32.7600
     46.040     0.4920    32.7660
     47.000     0.4460    32.7840
     48.000     0.4190    32.7660
     49.040     0.3760    32.7720
     50.000     0.3890    32.7710
     51.000     0.3900    32.7720
     52.040     0.4060    32.7740
     53.040     0.4590    32.7860
     54.040     0.3240    32.7990
     55.040     0.3040    32.7810
     56.040     0.4950    32.7790
     57.000     0.3500    32.8050
     58.000     0.2200    32.8020
     59.040     0.2120    32.7920
     60.040     0.1900    32.7940
     61.000     0.1840    32.7960
     62.000     0.1880    32.7940
     63.040     0.1890    32.7940
     64.040     0.1870    32.7950
     65.000     0.1880    32.7940
     66.000     0.1910    32.7950
     67.000     0.1750    32.7960
     68.000     0.1890    32.7940
     69.000     0.2430    32.7940
     70.000     0.2610    32.8080
     71.000     0.2540    32.8060
     72.000     0.2620    32.8070
     73.000     0.2490    32.8110
     73.960     0.2430    32.8110
     75.000     0.2440    32.8120
     76.040     0.2510    32.8100
     77.000     0.2510    32.8110
     78.000     0.2410    32.8120
     79.000     0.2430    32.8110
     79.960     0.2220    32.8150
     80.960     0.1950    32.8110
     82.000     0.1950    32.8130
     83.040     0.1960    32.8140
     84.000     0.1970    32.8160
     85.040     0.1970    32.8170
     86.040     0.1960    32.8170
     87.000     0.1960    32.8170
     88.040     0.1960    32.8160
     89.040     0.1940    32.8160
     90.040     0.1940    32.8160
     91.040     0.1950    32.8160
     92.040     0.1950    32.8160
     93.000     0.1940    32.8160
     94.000     0.1950    32.8160
     95.000     0.2060    32.8190
     96.000     0.2100    32.8230
     97.040     0.2070    32.8220
     98.080     0.2060    32.8200
     99.000     0.2080    32.8240
     99.960     0.2090    32.8260
    101.000     0.2130    32.8240
    102.040     0.2570    32.8370
    103.040     0.2700    32.8450
    104.040     0.2730    32.8480
    105.000     0.2740    32.8490
    106.000     0.2770    32.8510
    107.000     0.2780    32.8520
    108.000     0.2780    32.8520
    109.000     0.2790    32.8520
    109.960     0.2860    32.8610
    110.960     0.2860    32.8620
    111.960     0.2870    32.8620
    112.960     0.2870    32.8620
    114.000     0.2870    32.8630
    115.040     0.2880    32.8630
    116.000     0.2880    32.8630
    116.960     0.2890    32.8640
    118.000     0.2900    32.8660
    118.960     0.2910    32.8660
    120.000     0.2910    32.8670
    121.040     0.2920    32.8670
    122.000     0.2920    32.8680
    123.000     0.2910    32.8670
    124.000     0.2930    32.8680
    125.000     0.2940    32.8690
    126.000     0.3000    32.8720
    127.000     0.3040    32.8760
    127.960     0.3040    32.8770
    128.960     0.3040    32.8760
    130.000     0.3050    32.8770
    131.040     0.3050    32.8780
    132.040     0.3050    32.8780
    133.000     0.3040    32.8780
    134.000     0.3040    32.8780
    135.040     0.3040    32.8790
    136.040     0.3040    32.8790
    137.000     0.3030    32.8790
    137.960     0.3020    32.8790
    138.960     0.3020    32.8790
    140.000     0.3030    32.8790
    141.000     0.3040    32.8800
    142.000     0.3050    32.8800
    143.040     0.3070    32.8800
    144.040     0.3050    32.8810
    145.000     0.3060    32.8800
    145.960     0.2980    32.8810
    147.000     0.2940    32.8800
    148.040     0.2930    32.8800
    149.000     0.2930    32.8810
    149.960     0.2920    32.8810
    151.000     0.2920    32.8810
    152.000     0.2910    32.8820
    153.000     0.2920    32.8820
    154.000     0.2940    32.8830
    155.000     0.2990    32.8820
    156.000     0.3100    32.8840
    157.000     0.3190    32.8850
    158.040     0.3280    32.8860
    159.040     0.3310    32.8860
    159.960     0.3360    32.8870
    160.960     0.3410    32.8870
    162.000     0.3400    32.8890
    163.040     0.3380    32.8890
    164.040     0.3370    32.8890
    165.000     0.3360    32.8890
    166.000     0.3340    32.8900
    166.960     0.3350    32.8900
    168.000     0.3350    32.8900
    169.040     0.3340    32.8900
    170.000     0.3340    32.8900
    171.000     0.3350    32.8910
    172.000     0.3350    32.8910
    173.000     0.3350    32.8910
    174.000     0.3350    32.8910
    175.000     0.3360    32.8910
    176.000     0.3370    32.8920
    177.000     0.3380    32.8930
    178.040     0.3380    32.8930
    179.000     0.3380    32.8940
    179.960     0.3390    32.8970
    181.000     0.3380    32.8990
    182.000     0.3380    32.9000
    183.000     0.3380    32.9010
    184.040     0.3380    32.9000
    185.040     0.3380    32.9030
    185.960     0.3390    32.9010
    187.000     0.3390    32.9040
    187.960     0.3390    32.9040
    188.960     0.3380    32.9040
    190.000     0.3350    32.9040
    191.000     0.3370    32.9040
    191.960     0.3370    32.9040
    193.000     0.3370    32.9040
    194.040     0.3360    32.9040
    195.040     0.3300    32.9060
    196.040     0.3240    32.9060
    197.040     0.3220    32.9070
    198.000     0.3180    32.9090
    199.040     0.3180    32.9090
    200.040     0.3180    32.9100
    201.040     0.3190    32.9100
    202.040     0.3190    32.9100
    203.000     0.3200    32.9110
    204.000     0.3280    32.9120
    205.000     0.3380    32.9160
    206.000     0.3440    32.9150
    207.040     0.3590    32.9200
    208.040     0.3560    32.9200
    208.960     0.3570    32.9200
    210.000     0.3570    32.9200
    211.000     0.3550    32.9200
    211.960     0.3440    32.9180
    213.000     0.3430    32.9170
    214.000     0.3480    32.9170
    215.040     0.3540    32.9190
    216.040     0.3540    32.9200
    217.040     0.3530    32.9180
    218.000     0.3520    32.9180
    218.960     0.3500    32.9180
    220.000     0.3470    32.9170
    221.000     0.3480    32.9180
    221.960     0.3510    32.9180
    223.000     0.3510    32.9180
    224.000     0.3520    32.9180
    224.960     0.3530    32.9190
    225.960     0.3530    32.9190
    227.000     0.3550    32.9190
    228.000     0.3610    32.9210
    228.960     0.3640    32.9220
    230.000     0.3670    32.9230
    231.080     0.3680    32.9240
    232.040     0.3690    32.9240
    233.000     0.3720    32.9240
    233.960     0.3730    32.9250
    234.960     0.3740    32.9250
    235.960     0.3780    32.9250
    237.000     0.3760    32.9250
    238.000     0.3790    32.9250
    239.000     0.3800    32.9250
    240.040     0.3780    32.9250
    241.040     0.3770    32.9260
    242.040     0.3880    32.9280
    243.000     0.3970    32.9300
    244.000     0.4010    32.9310
    245.040     0.3960    32.9300
    246.000     0.3960    32.9300
    247.000     0.3930    32.9310
    248.000     0.3910    32.9310
    249.040     0.3920    32.9310
    250.240     0.3950    32.9310
    250.520     0.3970    32.9320
# Drift Data
#      C Npts comp n_var
I0  1352   47    1     4
#      Pres       Temp   Salinity     Time
    164.400     0.3000    32.8940        5
    211.440     0.3480    32.9160     1811
    251.920     0.3640    32.9230     3617
    251.600     0.3640    32.9230     5422
    251.480     0.3650    32.9220     7228
    252.080     0.3680    32.9240     9033
    252.400     0.3700    32.9250    10839
    252.560     0.3820    32.9310    12645
    252.920     0.3770    32.9290    14450
    253.160     0.3780    32.9280    16256
    253.280     0.3860    32.9320    18062
    253.200     0.3840    32.9320    19867
    253.480     0.3780    32.9290    21673
    253.640     0.3830    32.9310    23479
    253.800     0.3790    32.9300    25284
    253.800     0.3770    32.9280    27090
    253.720     0.3790    32.9290    28895
    253.520     0.3790    32.9290    30701
    253.280     0.3780    32.9290    32507
    253.080     0.3760    32.9290    34312
    252.760     0.3770    32.9280    36118
    252.400     0.3720    32.9260    37924
    252.120     0.3830    32.9290    39729
    252.400     0.3910    32.9340    41535
    252.360     0.3940    32.9350    43340
    252.200     0.3960    32.9360    45146
    252.080     0.3930    32.9320    46952
    252.200     0.3970    32.9360    48757
    252.280     0.3970    32.9360    50563
    252.520     0.3950    32.9350    52369
    252.840     0.3940    32.9350    54174
    253.240     0.3940    32.9350    55980
    253.680     0.3920    32.9340    57785
    253.920     0.3880    32.9310    59591
    252.040     0.3920    32.9320    61397
    251.800     0.3940    32.9310    63202
    252.200     0.4730    32.9440    65008
    252.480     0.4540    32.9380    66814
    252.640     0.5070    32.9470    68619
    252.720     0.5120    32.9470    70425
    252.680     0.5100    32.9470    72230
    252.560     0.5100    32.9470    74036
    252.160     0.5090    32.9460    75842
    251.840     0.4440    32.9380    77647
    251.440     0.4520    32.9370    79453
    251.000     0.4400    32.9360    81259
    250.680     0.4570    32.9400    83064
# Surface Pressure Time Series
#    C npts n_var               Start
S 1352     6     6 2025/06/28 05:34:56
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    79      0.130      4.010      0.000      14 2025/06/28 05:34:56
    92      0.120      3.884      0.000      14 2025/06/28 05:35:09
   103      0.120      3.741      0.000      14 2025/06/28 05:35:20
   114      0.110      3.608      0.000      14 2025/06/28 05:35:31
   125      0.110      3.682      0.000      14 2025/06/28 05:35:42
   136      0.110      3.548      0.000      14 2025/06/28 05:35:53
# Fall Time Series
#  Cycle npts     Start                Date
F00 1352   19 804312962 2025/06/27 04:16:02
# time     Pres phase                Date
     0    0.080     1 2025/06/27 04:16:02
   154   10.160    10 2025/06/27 04:18:36
   216   20.520    10 2025/06/27 04:19:38
   275   30.120    10 2025/06/27 04:20:37
   336   40.200    10 2025/06/27 04:21:38
   376   46.840     2 2025/06/27 04:22:18
   398   50.360    10 2025/06/27 04:22:40
   463   60.280    10 2025/06/27 04:23:45
   538   70.240    10 2025/06/27 04:25:00
   622   80.360    10 2025/06/27 04:26:24
   732   90.080    10 2025/06/27 04:28:14
   887  100.040    10 2025/06/27 04:30:49
  1033  110.000    10 2025/06/27 04:33:15
  1211  120.040    10 2025/06/27 04:36:13
  1396  130.080    10 2025/06/27 04:39:18
  1682  140.080    10 2025/06/27 04:44:04
  1880  150.080    10 2025/06/27 04:47:22
  2065  160.080    10 2025/06/27 04:50:27
  2163  164.240    10 2025/06/27 04:52:05
# Rise Time Series
#  Cycle npts     Start                Date
R00 1352   25 804401534 2025/06/28 04:52:14
# time     Pres phase                Date
     0  250.600     7 2025/06/28 04:52:14
   237  239.760    13 2025/06/28 04:56:11
   360  229.600    13 2025/06/28 04:58:14
   462  219.840    13 2025/06/28 04:59:56
   564  209.680    13 2025/06/28 05:01:38
   667  199.120    13 2025/06/28 05:03:21
   769  188.080    13 2025/06/28 05:05:03
   872  177.480    13 2025/06/28 05:06:46
   974  166.520    13 2025/06/28 05:08:28
  1056  157.000    13 2025/06/28 05:09:50
  1159  145.040    13 2025/06/28 05:11:33
  1241  135.840    13 2025/06/28 05:12:55
  1343  124.640    13 2025/06/28 05:14:37
  1446  113.520    13 2025/06/28 05:16:20
  1548  102.160    13 2025/06/28 05:18:02
  1651   91.240    13 2025/06/28 05:19:45
  1753   80.720    13 2025/06/28 05:21:27
  1856   69.560    13 2025/06/28 05:23:10
  1958   59.280    13 2025/06/28 05:24:52
  2061   47.840    13 2025/06/28 05:26:35
  2143   37.960    13 2025/06/28 05:27:57
  2225   27.960    13 2025/06/28 05:29:19
  2327   15.760    13 2025/06/28 05:31:01
  2430    4.800    13 2025/06/28 05:32:44
  2481    0.080     8 2025/06/28 05:33:35
# Pump Time Series
#   Cycle npts
B02  1352   11
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    46.840  396   11.38     224    668    -9     2
   250.600   94   11.83     309    670    -9     7
   242.720   21   11.82     323    672    -9    13
   233.200   21   11.80     320    674    -9    13
   217.960   21   11.77     309    676    -9    13
   205.760   21   11.76     288    677    -9    13
   175.520   21   11.74     292    678    -9    13
    91.240   21   11.69     258    681    -9    13
    59.280   21   11.67     251    683    -9    13
     6.920   21   11.64     241    678    -9    13
     0.080   59   11.60     232    676    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2  1352  2.4   30    90   120   600    81    10  1440      9     1    4020   15  1.1317
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17 1352 2025/06/27 04:15:59
# Pcase vacuums
   6.750   6.650
# Oil valve positions
   1.827   0.790   0.631
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18 1352 2025/06/27 04:52:06 9999/99/99 99:99:99
# Pcase vacuums
   6.650   6.640
# Average 
#   P      T      S
  249.840   0.403  32.932
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18 1352 2025/06/28 04:52:14 9999/99/99 99:99:99
# Pcase vacuums
   6.640   6.750
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20 1352 2025/06/28 05:33:36
# number of packets previous packets
      7               7
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.750   6.760
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.176   0.158   0.667
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.740   0.876
# Encountered Error Codes
#    Cycle Errors
EE20  1352      4
#               Date   Code Description
 2025/06/27 04:52:02 0x0113 DescendCtd
 2025/06/27 04:52:03 0x0913 DescendMaxDepth
 2025/06/27 04:52:03 0x0013 Descend
 2025/06/28 04:59:35 0x0114 AscendCtd
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00 1352  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.080    372   6.340
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M 1352       197    2025/06/28 05:36:58
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =     1352	(cycle number) |
     MPS =        0	(phase) |
     MPC =      353	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      120	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    42104	(vac 0 inHg cal) |
     PV5 =    18698	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =     4095	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        0	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        0	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       99	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      999	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       10	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        1	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =      600	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =     1351	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       46	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       10	(dBar) |
     DDT =     1800	(sec) |
  DFT[0] =      395	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        0	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65531	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      900	(sec) |
     DUC =       67	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =     1351	(cycle) |
  PTO[0] =    86400	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  1800000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =     1800	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      100	(dBar) |
     PST =     7200	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =     1351	(cycle number) |
     ARR =        9	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =      100	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =       10	(dBar) |
     ADT =     1800	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =     1345	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =       60	(sec) |
     SSP =       60	(sec) |

