# Message Decoder for Solo2, Alto and Deep Solo Floats; v230712
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11342   110  236 000000000881890   765     0   6 2022/09/22, 11:54:20,  59.6938  -44.2964   2      0 6,14,14
SBD 11342   110  340 000000000881890   766     0   5 2022/09/22, 11:54:33,  59.6938  -44.2964   6      0 13
SBD 11342   110  299 000000000881890   767     0   4 2022/09/22, 11:54:47,  59.6938  -44.2964   2      0 8
SBD 11342   110  314 000000000881890   768     0   3 2022/09/22, 11:55:02,  59.6938  -44.2964   2      0 3,15,13
SBD 11342   110  284 000000000881890   769     0   2 2022/09/22, 11:55:14,  59.6938  -44.2964   2      0 1,2
SBD 11342   110  315 000000000881890   770     0   1 2022/09/22, 11:55:24,  59.6938  -44.2964   5      0 10,11,7,5,14,0
SBD 11342   110  322 000000000881890   771     0   0 2022/09/22, 11:55:35,  59.6938  -44.2964   5      0 0,4,14,14,9
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         2026           84        2110
# Sensor_ID bytes     % packets Description
          0    48  2.37       2         GPS
          1    96  4.74       1    Pressure
          2   176  8.69       1 Temperature
          3   160  7.90       1    Salinity
          4   198  9.77       1        Fall
          5   138  6.81       1        Rise
          6   136  6.71       1        Pump
          7    35  1.73       1 Drift_times
          8   287 14.17       1     Surface
          9    44  2.17       1  HR/Drift_P
         10    44  2.17       1  HR/Drift_T
         11    43  2.12       1  HR/Drift_S
         13   404 19.94       2  Parameters
         14   151  7.45       5 Engineering
         15    66  3.26       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  110 1 2022/09/22 06:29 +59.68250  -44.22611   60  5 29 38 45  1.9     0
G  110 2 2022/09/22 11:47 +59.68900  -44.28094   40  7 26 38 45  3.4     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   110  137    1     3         0
#      Pres      Temp    Salinity
      1.080     2.3390    31.2420
      2.000     2.2740    31.2420
      3.040     2.2650    31.2420
      4.000     2.2650    31.2460
      4.960     2.2660    31.2530
      5.960     2.2530    31.2620
      7.000     2.2400    31.2730
      8.000     2.2410    31.2770
      8.960     2.2400    31.2820
     10.000     2.2380    31.2890
     11.000     2.2350    31.2960
     12.000     2.2260    31.3050
     13.000     2.2060    31.3110
     14.000     2.1890    31.3230
     15.000     2.1810    31.3560
     15.960     2.1810    31.3580
     16.960     2.1810    31.3590
     18.000     2.1800    31.3950
     19.040     2.1830    31.4260
     20.040     2.1710    31.4460
     21.000     2.1540    31.4600
     22.000     2.1450    31.4720
     23.000     2.1400    31.4830
     24.000     2.1380    31.4850
     24.960     2.1380    31.4890
     26.000     2.1380    31.4960
     27.000     2.1390    31.5090
     28.000     2.1430    31.5260
     29.000     2.1410    31.5530
     30.000     2.0340    31.6780
     31.000     1.9140    31.8240
     31.960     1.8820    31.8440
     33.000     1.8340    31.8520
     34.000     1.8040    31.8560
     35.000     1.7690    31.8590
     35.960     1.7640    31.8590
     37.000     1.7560    31.8600
     38.040     1.7530    31.8600
     39.040     1.7380    31.8690
     40.000     1.7270    31.8970
     41.000     1.7460    31.9130
     42.040     1.7620    31.9230
     43.000     1.7810    31.9330
     44.000     1.7690    31.9340
     45.040     1.7510    31.9390
     46.040     1.7620    31.9410
     47.000     1.7620    31.9430
     48.000     1.7620    31.9420
     49.000     1.7650    31.9420
     50.000     1.7630    31.9420
     50.960     1.7590    31.9440
     52.000     1.7550    31.9470
     53.000     1.7380    31.9610
     53.960     1.7160    31.9750
     54.960     1.7070    31.9910
     55.960     1.7150    32.0670
     57.040     1.6670    32.1140
     58.040     1.6570    32.1250
     59.000     1.6520    32.1320
     60.040     1.6510    32.1430
     60.960     1.6410    32.1540
     61.960     1.6240    32.1610
     63.000     1.6070    32.1700
     64.000     1.6060    32.1750
     65.000     1.5980    32.1920
     66.000     1.5710    32.2160
     67.000     1.5610    32.2420
     68.000     1.5510    32.2670
     69.000     1.5430    32.2780
     69.960     1.5460    32.2860
     70.960     1.5450    32.2870
     71.960     1.5440    32.2890
     73.000     1.5450    32.3050
     74.040     1.5410    32.3280
     75.000     1.5420    32.3440
     76.000     1.5220    32.3750
     77.000     1.5170    32.3910
     78.000     1.5010    32.4000
     79.000     1.4790    32.4150
     80.000     1.4830    32.4510
     81.040     1.5390    32.5780
     82.000     1.6090    32.6600
     83.000     1.7750    32.8050
     83.960     1.8010    32.8340
     85.000     1.8480    32.8620
     86.040     1.9540    32.9340
     87.000     1.9700    32.9520
     87.960     1.9930    32.9660
     89.000     2.0190    32.9830
     90.040     2.0390    32.9920
     91.040     2.0620    33.0020
     92.040     2.0840    33.0170
     93.000     2.0920    33.0260
     94.000     2.0980    33.0310
     95.040     2.1070    33.0410
     96.040     2.1160    33.0530
     97.000     2.1150    33.0640
     97.960     2.1140    33.0670
     99.000     2.1100    33.0790
    100.000     2.1100    33.0760
    101.000     2.0940    33.0810
    102.040     2.0870    33.0810
    103.000     2.0810    33.0800
    104.000     2.0740    33.0810
    105.000     2.0730    33.0810
    106.000     2.0640    33.0830
    107.000     2.0550    33.0800
    108.000     2.0380    33.0830
    108.960     2.0270    33.0870
    110.000     2.0230    33.0910
    111.040     2.0120    33.0930
    112.040     2.0010    33.0960
    113.040     1.9890    33.0970
    114.000     1.9870    33.0960
    114.960     1.9850    33.0980
    115.960     1.9690    33.0980
    117.000     1.9380    33.0990
    118.000     1.9410    33.1000
    119.000     1.9200    33.1030
    120.000     1.9160    33.1100
    121.000     2.0350    33.1350
    122.000     2.3250    33.2350
    123.000     2.3480    33.2650
    124.000     2.3410    33.2690
    124.960     2.3530    33.2880
    125.960     2.4920    33.3570
    127.000     2.6090    33.4090
    128.000     2.6580    33.4250
    128.960     2.7080    33.4440
    130.000     2.7270    33.4540
    131.000     2.7380    33.4580
    132.000     2.7730    33.4660
    133.000     2.8050    33.4810
    134.040     2.8140    33.4840
    135.040     2.8130    33.4840
    136.040     2.8150    33.4850
    136.600     2.8150    33.4850
# Drift Data
#      C Npts comp n_var
I0   110   21    1     4
#      Pres       Temp   Salinity     Time
    129.120     2.8400    33.4980        5
    127.720     2.8870    33.5340      611
    128.920     2.9150    33.5400     1216
    128.640     2.8470    33.4910     1822
    130.320     2.8720    33.5130     2427
    129.840     2.8800    33.5100     3032
    130.760     2.8630    33.5090     3638
    131.600     2.8190    33.4850     4243
    130.400     2.8550    33.5070     4849
    131.040     2.8880    33.4970     5454
    132.280     2.8380    33.4990     6060
    135.040     2.8750    33.5170     6665
    134.040     2.8740    33.5140     7271
    134.560     2.8460    33.4970     7876
    134.000     2.8100    33.4830     8482
    135.320     2.8130    33.4830     9087
    137.200     2.8290    33.4910     9693
    136.640     2.8310    33.4920    10298
    135.760     2.8280    33.4880    10903
    135.360     2.8260    33.4860    11509
    134.120     2.8160    33.4850    12114
# Surface Pressure Time Series
#    C npts n_var               Start
S  110    31     6 2022/09/22 11:47:12
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    71      0.380      2.444   -999.000      14 2022/09/22 11:48:23
    84      0.430      2.475   -999.000      14 2022/09/22 11:48:36
    95      0.420      2.469   -999.000      14 2022/09/22 11:48:47
   106      0.330      2.477   -999.000      14 2022/09/22 11:48:58
   117      0.350      2.415   -999.000      14 2022/09/22 11:49:09
   128      0.390      2.544   -999.000      14 2022/09/22 11:49:20
   139      0.370      2.478   -999.000      14 2022/09/22 11:49:31
   150      0.370      2.615   -999.000      14 2022/09/22 11:49:42
   161      0.360      2.598   -999.000      14 2022/09/22 11:49:53
   172      0.360      2.534   -999.000      14 2022/09/22 11:50:04
   184      0.380      2.563   -999.000      14 2022/09/22 11:50:16
   195      0.280      2.717   -999.000      14 2022/09/22 11:50:27
   206      0.310      2.546   -999.000      14 2022/09/22 11:50:38
   217      0.330      2.497   -999.000      14 2022/09/22 11:50:49
   228      0.340      2.661   -999.000      14 2022/09/22 11:51:00
   239      0.310      2.765   -999.000      14 2022/09/22 11:51:11
   250      0.290      2.778   -999.000      14 2022/09/22 11:51:22
   261      0.360      2.883   -999.000      14 2022/09/22 11:51:33
   272      0.340      3.039   -999.000      14 2022/09/22 11:51:44
   283      0.330      2.973   -999.000      14 2022/09/22 11:51:55
   294      0.290      3.052   -999.000      14 2022/09/22 11:52:06
   306      0.350      3.180   -999.000      14 2022/09/22 11:52:18
   317      0.310      2.601   -999.000      14 2022/09/22 11:52:29
   328      0.350      2.742   -999.000      14 2022/09/22 11:52:40
   339      0.250      2.857   -999.000      14 2022/09/22 11:52:51
   350      0.290      2.719   -999.000      14 2022/09/22 11:53:02
   361      0.330      2.878   -999.000      14 2022/09/22 11:53:13
   372      0.320      2.690   -999.000      14 2022/09/22 11:53:24
   383      0.270      2.765   -999.000      14 2022/09/22 11:53:35
   394      0.340      2.957   -999.000      14 2022/09/22 11:53:46
   405      0.310      3.190   -999.000      14 2022/09/22 11:53:57
# Fall Time Series
#  Cycle npts     Start                Date
F00  110   38 717143348 2022/09/22 06:29:08
# time     Pres phase                Date
     0    0.320     1 2022/09/22 06:29:08
   155    5.080    10 2022/09/22 06:31:43
   184   10.040    10 2022/09/22 06:32:12
   216   15.240    10 2022/09/22 06:32:44
   249   20.200    10 2022/09/22 06:33:17
   281   25.440    10 2022/09/22 06:33:49
   310   30.280    10 2022/09/22 06:34:18
   339   35.000    10 2022/09/22 06:34:47
   350   36.720     2 2022/09/22 06:34:58
   372   40.360    10 2022/09/22 06:35:20
   401   45.040    10 2022/09/22 06:35:49
   437   50.400    10 2022/09/22 06:36:25
   473   55.280    10 2022/09/22 06:37:01
   512   60.000    10 2022/09/22 06:37:40
   560   65.240    10 2022/09/22 06:38:28
   612   70.000    10 2022/09/22 06:39:20
   688   75.120    10 2022/09/22 06:40:36
   786   80.040    10 2022/09/22 06:42:14
   875   85.080    10 2022/09/22 06:43:43
   964   90.040    10 2022/09/22 06:45:12
  1095   95.040    10 2022/09/22 06:47:23
  1267  100.080    10 2022/09/22 06:50:15
  1398  105.040    10 2022/09/22 06:52:26
  1513  110.080    10 2022/09/22 06:54:21
  1635  115.040    10 2022/09/22 06:56:23
  1794  120.000    10 2022/09/22 06:59:02
  2036  125.000    10 2022/09/22 07:03:04
  2336  128.600    10 2022/09/22 07:08:04
  2636  129.240    10 2022/09/22 07:13:04
  2936  130.240    10 2022/09/22 07:18:04
  3236  132.120    10 2022/09/22 07:23:04
  3536  132.040    10 2022/09/22 07:28:04
  3836  130.680    10 2022/09/22 07:33:04
  4136  129.720    10 2022/09/22 07:38:04
  4265  128.800     3 2022/09/22 07:40:13
  4436  129.800    10 2022/09/22 07:43:04
  4736  129.360    10 2022/09/22 07:48:04
  4902  129.080    10 2022/09/22 07:50:50
# Rise Time Series
#  Cycle npts     Start                Date
R00  110   26 717160859 2022/09/22 11:20:59
# time     Pres phase                Date
     0  136.600     7 2022/09/22 11:20:59
   114  131.800    13 2022/09/22 11:22:53
   176  126.720    13 2022/09/22 11:23:55
   237  121.560    13 2022/09/22 11:24:56
   299  115.960    13 2022/09/22 11:25:58
   360  110.480    13 2022/09/22 11:26:59
   422  104.160    13 2022/09/22 11:28:01
   463   99.840    13 2022/09/22 11:28:42
   524   93.720    13 2022/09/22 11:29:43
   586   87.320    13 2022/09/22 11:30:45
   647   82.000    13 2022/09/22 11:31:46
   729   75.520    13 2022/09/22 11:33:08
   791   69.960    13 2022/09/22 11:34:10
   852   64.440    13 2022/09/22 11:35:11
   914   58.240    13 2022/09/22 11:36:13
   975   52.080    13 2022/09/22 11:37:14
  1037   46.200    13 2022/09/22 11:38:16
  1098   40.840    13 2022/09/22 11:39:17
  1160   35.240    13 2022/09/22 11:40:19
  1221   29.560    13 2022/09/22 11:41:20
  1283   24.680    13 2022/09/22 11:42:22
  1344   19.040    13 2022/09/22 11:43:23
  1406   13.240    13 2022/09/22 11:44:25
  1467    7.360    13 2022/09/22 11:45:26
  1529    1.080    13 2022/09/22 11:46:28
  1570    0.400     8 2022/09/22 11:47:09
# Pump Time Series
#   Cycle npts
B02   110   12
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    36.720  312   15.20     239    660    -9     2
   136.600   94   15.35     277    657    -9     7
   130.080   21   15.35     266    658    -9    13
   123.240   21   15.34     264    661    -9    13
   114.240   21   15.34     267    665    -9    13
    82.000   21   15.33     247    667    -9    13
    77.160   21   15.34     250    670    -9    13
    64.440   21   15.32     241    672    -9    13
    40.840   21   15.32     240    673    -9    13
    27.960   21   15.31     217    675    -9    13
    22.920   21   15.31     231    676    -9    13
     0.240  240   15.28     195    695    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   110  2.4   12    90   125   600    80    10   210      8     1    4020    5  0.2714
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  110 2022/09/22 06:29:05
# Pcase vacuums
   6.960   6.500
# Oil valve positions
   1.807   0.824   0.631
# Seek attempts and pump time
      1      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  110 2022/09/22 07:50:51 9999/99/99 99:99:99
# Pcase vacuums
   6.500   6.500
# Average 
#   P      T      S
  132.520   2.850  33.501
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  110 2022/09/22 11:20:59 9999/99/99 99:99:99
# Pcase vacuums
   6.500   6.750
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  110 2022/09/22 11:47:10
# number of packets previous packets
      7               7
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.760   6.960
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.176   0.158   0.667
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.740   0.876
# Encountered Error Codes
#    Cycle Errors
EE20   110      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  110  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.360    346   6.320
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  110       197    2022/09/22 11:55:02
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      110	(cycle number) |
     MPS =        0	(phase) |
     MPC =      110	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      125	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    42104	(vac 0 inHg cal) |
     PV5 =    18698	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      109	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       36	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      312	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      109	(cycle) |
  PTO[0] =    12600	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      109	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      109	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      340	(sec) |
     SSP =      240	(sec) |

