# Message Decoder for Solo2, Alto and Deep Solo Floats; v230712
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11342   100  225 000000000881890   695     0   6 2022/09/20, 05:13:40,  59.7329  -44.1449   2      0 6,14,14
SBD 11342   100  340 000000000881890   696     0   5 2022/09/20, 05:13:54,  59.7329  -44.1449   2      0 13
SBD 11342   100  299 000000000881890   697     0   4 2022/09/20, 05:14:06,  59.7329  -44.1449   2      0 8
SBD 11342   100  334 000000000881890   698     0   3 2022/09/20, 05:14:19,  59.7329  -44.1449   9      0 3,0,15,13
SBD 11342   100  268 000000000881890   699     0   2 2022/09/20, 05:14:34,  59.7329  -44.1449   8      0 1,2
SBD 11342   100  282 000000000881890   700     0   1 2022/09/20, 05:14:56,  59.7329  -44.1449   7      0 10,11,7,5,14
SBD 11342   100  330 000000000881890   701     0   0 2022/09/20, 05:15:09,  59.7329  -44.1449   7      0 0,4,14,14,9
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1994           84        2078
# Sensor_ID bytes     % packets Description
          0    48  2.41       2         GPS
          1   100  5.02       1    Pressure
          2   156  7.82       1 Temperature
          3   156  7.82       1    Salinity
          4   198  9.93       1        Fall
          5   138  6.92       1        Rise
          6   125  6.27       1        Pump
          7    35  1.76       1 Drift_times
          8   287 14.39       1     Surface
          9    52  2.61       1  HR/Drift_P
         10    43  2.16       1  HR/Drift_T
         11    35  1.76       1  HR/Drift_S
         13   404 20.26       2  Parameters
         14   151  7.57       5 Engineering
         15    66  3.31       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  100 1 2022/09/19 23:47 +59.69937  -44.13330    0  6 32 37 42  1.1     0
G  100 2 2022/09/20 05:07 +59.72497  -44.15641   70  4 31 37 42  3.4     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   100  135    1     3         0
#      Pres      Temp    Salinity
      1.080     2.2070    31.3680
      2.000     2.2060    31.3680
      3.000     2.2060    31.3680
      4.000     2.2040    31.3730
      5.040     2.1970    31.3850
      6.040     2.1880    31.4000
      7.000     2.1880    31.4010
      8.000     2.1870    31.4010
      9.040     2.1870    31.4010
     10.040     2.1870    31.4020
     11.000     2.1850    31.4030
     12.040     2.1850    31.4040
     13.040     2.1850    31.4030
     14.040     2.1830    31.4050
     15.000     2.1780    31.4090
     16.040     2.1780    31.4100
     17.000     2.1780    31.4090
     18.000     2.1790    31.4080
     19.040     2.1740    31.4130
     20.000     2.1690    31.4190
     20.960     2.1660    31.4230
     21.960     2.1660    31.4240
     23.000     2.1650    31.4240
     24.000     2.1630    31.4260
     24.960     2.1630    31.4270
     26.000     2.1630    31.4270
     27.000     2.1630    31.4270
     28.040     2.1620    31.4280
     29.000     2.1580    31.4340
     30.000     2.1570    31.4370
     31.040     2.1560    31.4400
     32.000     2.1550    31.4420
     33.000     2.1550    31.4430
     34.000     2.1550    31.4430
     35.000     2.1550    31.4440
     36.000     2.1550    31.4440
     37.000     2.1550    31.4440
     38.000     2.1540    31.4480
     39.000     2.1550    31.4520
     40.000     2.1550    31.4540
     41.040     2.1580    31.4600
     42.040     2.1630    31.4690
     43.000     2.1740    31.4780
     44.000     2.1840    31.4880
     45.000     2.1920    31.4930
     46.000     2.2050    31.5050
     47.040     2.1610    31.5800
     48.000     2.0830    31.6330
     49.000     2.0770    31.6360
     50.000     2.0730    31.6370
     51.000     2.0600    31.6400
     52.040     2.0460    31.6450
     53.040     2.0470    31.6520
     54.040     2.0330    31.6710
     55.040     2.0160    31.6810
     56.040     2.0140    31.6850
     57.040     1.9920    31.7150
     57.960     1.9320    31.7720
     59.000     1.9150    31.7770
     60.040     1.8940    31.7830
     61.000     1.8720    31.7970
     62.000     1.8660    31.8020
     63.000     1.8640    31.8110
     64.000     1.8630    31.8240
     65.000     1.8710    31.8290
     65.960     1.8990    31.8390
     66.960     1.9420    31.8800
     68.000     1.9320    31.8970
     69.080     1.9290    31.9010
     70.000     1.9280    31.9010
     71.040     1.9370    31.9050
     72.040     1.9510    31.9130
     73.000     1.9660    31.9280
     74.000     1.9700    31.9460
     75.000     1.9670    32.0250
     76.040     1.9330    32.1260
     76.960     1.9060    32.2230
     77.960     1.8860    32.2970
     79.000     1.8190    32.5210
     80.000     1.7790    32.5820
     81.000     1.7670    32.5990
     82.040     1.7490    32.6260
     83.040     1.7400    32.6390
     84.000     1.7300    32.6510
     85.000     1.7030    32.6650
     86.000     1.6820    32.6670
     87.040     1.6870    32.6690
     88.000     1.6780    32.6740
     88.960     1.6600    32.6820
     90.000     1.6390    32.7090
     91.000     1.6600    32.7430
     92.000     1.6910    32.7940
     93.000     1.6920    32.8040
     94.000     1.6930    32.8080
     95.000     1.6960    32.8180
     96.000     1.7010    32.8280
     97.000     1.7100    32.8410
     98.000     1.7150    32.8480
     98.960     1.7200    32.8510
     99.960     1.7420    32.8600
    101.000     1.7500    32.8660
    102.000     1.7580    32.8670
    103.040     1.7680    32.8700
    104.040     1.7820    32.8820
    105.040     1.7660    32.8950
    105.960     1.7180    32.8980
    107.040     1.7060    32.8980
    108.040     1.7150    32.9040
    109.040     1.7160    32.9120
    110.040     1.7020    32.9160
    111.000     1.7170    32.9250
    111.960     1.7840    32.9330
    113.000     1.8460    32.9770
    114.040     1.8530    32.9940
    115.000     1.9250    33.0200
    116.000     1.9830    33.0470
    117.000     2.0130    33.1020
    117.960     2.1190    33.1360
    120.000     2.4460    33.2760
    121.000     2.4560    33.2900
    122.000     2.4530    33.2990
    123.000     2.4520    33.3050
    124.000     2.4810    33.3190
    125.040     2.5380    33.3400
    126.000     2.6150    33.3810
    126.960     2.6550    33.4070
    128.000     2.7010    33.4350
    129.080     2.7290    33.4600
    130.040     2.7380    33.4640
    131.000     2.7460    33.4680
    132.040     2.7520    33.4720
    133.040     2.7520    33.4730
    134.000     2.7520    33.4720
    135.000     2.7530    33.4730
    136.040     2.7530    33.4730
# Drift Data
#      C Npts comp n_var
I0   100   21    1     4
#      Pres       Temp   Salinity     Time
    130.360     2.7990    33.4970        6
    129.120     2.7980    33.4960      611
    132.320     2.7960    33.4960     1217
    130.640     2.7960    33.4960     1822
    133.400     2.7950    33.4950     2427
    132.240     2.7910    33.4920     3033
    131.360     2.7950    33.4950     3638
    134.160     2.7940    33.4940     4244
    129.760     2.7910    33.4950     4849
    130.960     2.7850    33.4910     5455
    128.360     2.7790    33.4900     6060
    134.440     2.7790    33.4880     6666
    136.200     2.7770    33.4870     7271
    135.560     2.7740    33.4850     7877
    135.880     2.7720    33.4840     8482
    135.520     2.7690    33.4820     9088
    134.920     2.7660    33.4800     9693
    135.520     2.7630    33.4760    10298
    136.200     2.7600    33.4750    10904
    136.920     2.7570    33.4740    11509
    136.880     2.7550    33.4730    12115
# Surface Pressure Time Series
#    C npts n_var               Start
S  100    31     6 2022/09/20 05:06:34
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    71      0.200      2.215   -999.000      14 2022/09/20 05:07:45
    84      0.230      2.209   -999.000      14 2022/09/20 05:07:58
    95      0.260      2.210   -999.000      14 2022/09/20 05:08:09
   106      0.220      2.209   -999.000      14 2022/09/20 05:08:20
   117      0.250      2.212   -999.000      14 2022/09/20 05:08:31
   128      0.280      2.215   -999.000      14 2022/09/20 05:08:42
   139      0.200      2.214   -999.000      14 2022/09/20 05:08:53
   150      0.230      2.219   -999.000      14 2022/09/20 05:09:04
   161      0.250      2.213   -999.000      14 2022/09/20 05:09:15
   173      0.120      2.215   -999.000      14 2022/09/20 05:09:27
   184      0.260      2.212   -999.000      14 2022/09/20 05:09:38
   195      0.140      2.233   -999.000      14 2022/09/20 05:09:49
   206      0.160      2.215   -999.000      14 2022/09/20 05:10:00
   217      0.190      2.272   -999.000      14 2022/09/20 05:10:11
   228      0.180      2.223   -999.000      14 2022/09/20 05:10:22
   239      0.150      2.221   -999.000      14 2022/09/20 05:10:33
   250      0.190      2.257   -999.000      14 2022/09/20 05:10:44
   261      0.180      2.259   -999.000      14 2022/09/20 05:10:55
   272      0.110      2.224   -999.000      14 2022/09/20 05:11:06
   283      0.160      2.230   -999.000      14 2022/09/20 05:11:17
   295      0.180      2.225   -999.000      14 2022/09/20 05:11:29
   306      0.140      2.231   -999.000      14 2022/09/20 05:11:40
   317      0.220      2.221   -999.000      14 2022/09/20 05:11:51
   328      0.210      2.251   -999.000      14 2022/09/20 05:12:02
   339      0.090      2.249   -999.000      14 2022/09/20 05:12:13
   350      0.190      2.264   -999.000      14 2022/09/20 05:12:24
   361      0.210      2.229   -999.000      14 2022/09/20 05:12:35
   372      0.180      2.258   -999.000      14 2022/09/20 05:12:46
   383      0.240      2.228   -999.000      14 2022/09/20 05:12:57
   394      0.150      2.236   -999.000      14 2022/09/20 05:13:08
   405      0.140      2.248   -999.000      14 2022/09/20 05:13:19
# Fall Time Series
#  Cycle npts     Start                Date
F00  100   38 716946437 2022/09/19 23:47:17
# time     Pres phase                Date
     0    0.120     1 2022/09/19 23:47:17
   248    5.440    10 2022/09/19 23:51:25
   277   10.080    10 2022/09/19 23:51:54
   310   15.320    10 2022/09/19 23:52:27
   339   20.280    10 2022/09/19 23:52:56
   368   25.160    10 2022/09/19 23:53:25
   401   30.040    10 2022/09/19 23:53:58
   433   35.400    10 2022/09/19 23:54:30
   444   37.160     2 2022/09/19 23:54:41
   463   40.080    10 2022/09/19 23:55:00
   498   45.400    10 2022/09/19 23:55:35
   531   50.120    10 2022/09/19 23:56:08
   566   55.160    10 2022/09/19 23:56:43
   605   60.280    10 2022/09/19 23:57:22
   648   65.120    10 2022/09/19 23:58:05
   696   70.240    10 2022/09/19 23:58:53
   748   75.200    10 2022/09/19 23:59:45
   816   80.040    10 2022/09/20 00:00:53
   902   85.000    10 2022/09/20 00:02:19
  1038   90.080    10 2022/09/20 00:04:35
  1146   95.040    10 2022/09/20 00:06:23
  1277  100.040    10 2022/09/20 00:08:34
  1409  105.000    10 2022/09/20 00:10:46
  1577  110.040    10 2022/09/20 00:13:34
  1718  115.040    10 2022/09/20 00:15:55
  2009  120.000    10 2022/09/20 00:20:46
  2309  124.520    10 2022/09/20 00:25:46
  2562  129.040    10 2022/09/20 00:29:59
  2862  129.680    10 2022/09/20 00:34:59
  3162  130.600    10 2022/09/20 00:39:59
  3462  127.880    10 2022/09/20 00:44:59
  3762  126.440    10 2022/09/20 00:49:59
  4062  127.040    10 2022/09/20 00:54:59
  4355  129.000     3 2022/09/20 00:59:52
  4362  129.040    10 2022/09/20 00:59:59
  4662  129.840    10 2022/09/20 01:04:59
  4962  130.720    10 2022/09/20 01:09:59
  4993  130.440    10 2022/09/20 01:10:30
# Rise Time Series
#  Cycle npts     Start                Date
R00  100   26 716964039 2022/09/20 04:40:39
# time     Pres phase                Date
     0  136.440     7 2022/09/20 04:40:39
   114  131.840    13 2022/09/20 04:42:33
   196  125.640    13 2022/09/20 04:43:55
   257  121.000    13 2022/09/20 04:44:56
   339  114.920    13 2022/09/20 04:46:18
   401  109.840    13 2022/09/20 04:47:20
   462  104.040    13 2022/09/20 04:48:21
   506   99.720    13 2022/09/20 04:49:05
   568   93.640    13 2022/09/20 04:50:07
   629   87.600    13 2022/09/20 04:51:08
   691   81.280    13 2022/09/20 04:52:10
   752   75.080    13 2022/09/20 04:53:11
   814   69.680    13 2022/09/20 04:54:13
   875   63.560    13 2022/09/20 04:55:14
   937   57.560    13 2022/09/20 04:56:16
   998   51.840    13 2022/09/20 04:57:17
  1060   46.360    13 2022/09/20 04:58:19
  1121   40.840    13 2022/09/20 04:59:20
  1183   34.680    13 2022/09/20 05:00:22
  1244   28.080    13 2022/09/20 05:01:23
  1285   23.920    13 2022/09/20 05:02:04
  1347   17.800    13 2022/09/20 05:03:06
  1408   11.880    13 2022/09/20 05:04:07
  1470    6.000    13 2022/09/20 05:05:09
  1542    0.240    13 2022/09/20 05:06:21
  1553    0.240     8 2022/09/20 05:06:32
# Pump Time Series
#   Cycle npts
B02   100   11
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    37.160  308   15.20     256    660    -9     2
   136.440   93   15.34     266    657    -9     7
   130.240   21   15.35     275    660    -9    13
   125.640   20   15.34     261    661    -9    13
   121.000   21   15.33     261    664    -9    13
   116.480   20   15.34     275    666    -9    13
   109.840   20   15.33     254    669    -9    13
    73.400   20   15.32     261    671    -9    13
    50.080   20   15.31     270    673    -9    13
    42.760   20   15.31     240    674    -9    13
     0.120  240   15.27     201    695    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   100  2.4   12    90   125   600    80    10   210      8     1    4020    5  0.2714
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  100 2022/09/19 23:47:14
# Pcase vacuums
   8.840   6.510
# Oil valve positions
   1.786   0.824   0.631
# Seek attempts and pump time
      1      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  100 2022/09/20 01:10:30 9999/99/99 99:99:99
# Pcase vacuums
   6.510   6.510
# Average 
#   P      T      S
  133.360   2.780  33.488
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  100 2022/09/20 04:40:39 9999/99/99 99:99:99
# Pcase vacuums
   6.510   6.750
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  100 2022/09/20 05:06:32
# number of packets previous packets
      7               7
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.750   6.950
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.176   0.158   0.667
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.740   0.876
# Encountered Error Codes
#    Cycle Errors
EE20   100      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  100  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.120    439   6.330
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  100       197    2022/09/20 05:14:19
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      100	(cycle number) |
     MPS =        0	(phase) |
     MPC =      100	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      125	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    42104	(vac 0 inHg cal) |
     PV5 =    18698	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =       99	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       36	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      307	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =       99	(cycle) |
  PTO[0] =    12600	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =       99	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =       99	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      340	(sec) |
     SSP =      240	(sec) |

