# Message Decoder for Solo2, Alto and Deep Solo Floats; v230712
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11342    98  247 000000000881890   681     0   6 2022/09/19, 18:16:04,  59.6703  -44.0460   3      0 6,14,14
SBD 11342    98  340 000000000881890   682     0   5 2022/09/19, 18:16:15,  59.6703  -44.0460   3      0 13
SBD 11342    98  299 000000000881890   683     0   4 2022/09/19, 18:16:26,  59.6703  -44.0460   4      0 8
SBD 11342    98  340 000000000881890   684     0   3 2022/09/19, 18:16:38,  59.6703  -44.0460   4      0 3,0,15,13
SBD 11342    98  272 000000000881890   685     0   2 2022/09/19, 18:16:59,  59.6703  -44.0460   4      0 1,2
SBD 11342    98  307 000000000881890   686     0   1 2022/09/19, 18:17:11,  59.6703  -44.0460   5      0 10,11,7,5,14
SBD 11342    98  335 000000000881890   687     0   0 2022/09/19, 18:17:33,  59.6298  -43.9967   6      0 0,4,14,14,9
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         2056           84        2140
# Sensor_ID bytes     % packets Description
          0    48  2.33       2         GPS
          1    98  4.77       1    Pressure
          2   162  7.88       1 Temperature
          3   162  7.88       1    Salinity
          4   203  9.87       1        Fall
          5   153  7.44       1        Rise
          6   147  7.15       1        Pump
          7    35  1.70       1 Drift_times
          8   287 13.96       1     Surface
          9    52  2.53       1  HR/Drift_P
         10    44  2.14       1  HR/Drift_T
         11    44  2.14       1  HR/Drift_S
         13   404 19.65       2  Parameters
         14   151  7.34       5 Engineering
         15    66  3.21       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   98 1 2022/09/19 12:48 +59.66141  -43.90825  110  4 35 38 44  2.3     0
G   98 2 2022/09/19 18:10 +59.67648  -44.06479   80  3 35 39 42  1.3     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0    98  147    1     3         0
#      Pres      Temp    Salinity
      1.040     2.2340    31.3950
      1.960     2.2040    31.4030
      3.000     2.1960    31.4070
      4.000     2.1880    31.4110
      5.040     2.1800    31.4170
      6.040     2.1800    31.4160
      6.960     2.1720    31.4220
      8.000     2.1610    31.4310
      9.000     2.1370    31.4490
      9.960     2.1260    31.4580
     10.960     2.1250    31.4590
     12.000     2.1080    31.4720
     13.000     2.1080    31.4730
     14.000     2.1020    31.4770
     15.040     2.1000    31.4790
     16.000     2.1000    31.4790
     17.000     2.0980    31.4810
     18.000     2.0970    31.4810
     19.000     2.0970    31.4820
     20.040     2.0980    31.4810
     21.000     2.0930    31.4850
     22.040     2.0800    31.4980
     23.040     2.0680    31.5100
     24.040     2.0670    31.5110
     25.000     2.0660    31.5130
     26.000     2.0640    31.5140
     27.000     2.0630    31.5160
     28.040     2.0600    31.5190
     29.000     2.0560    31.5260
     30.000     2.0550    31.5270
     31.000     2.0490    31.5360
     32.000     2.0310    31.5640
     33.040     2.0250    31.5730
     34.000     2.0270    31.5690
     35.000     2.0150    31.5900
     36.000     2.0110    31.5980
     37.000     1.9910    31.6330
     38.000     1.9840    31.6450
     38.960     1.9710    31.6660
     40.000     1.9340    31.7220
     41.040     1.8690    31.8300
     42.040     1.8590    31.8480
     43.000     1.8520    31.8600
     44.000     1.8420    31.8730
     45.000     1.8210    31.9100
     46.000     1.8140    31.9200
     47.000     1.7880    31.9570
     48.040     1.7320    32.0420
     49.040     1.6340    32.1970
     50.000     1.6170    32.2250
     51.000     1.6000    32.2640
     52.000     1.5870    32.2860
     52.960     1.5900    32.2940
     54.000     1.5970    32.3040
     55.000     1.6070    32.3200
     56.040     1.6170    32.3260
     57.000     1.6260    32.3330
     58.000     1.6380    32.3480
     58.960     1.6530    32.3700
     59.960     1.6570    32.3990
     61.000     1.6550    32.4150
     62.000     1.6520    32.4300
     63.040     1.6450    32.4570
     64.000     1.6390    32.4990
     65.000     1.6320    32.5350
     66.000     1.6260    32.5660
     67.000     1.6200    32.5790
     68.040     1.6170    32.5840
     69.000     1.6120    32.5950
     69.960     1.6010    32.6200
     70.960     1.5940    32.6400
     72.000     1.5920    32.6430
     73.080     1.5870    32.6520
     74.040     1.5880    32.6530
     75.000     1.5870    32.6540
     76.000     1.5860    32.6750
     77.040     1.5890    32.6780
     78.040     1.5940    32.6870
     79.040     1.5940    32.6900
     80.040     1.6000    32.6940
     81.040     1.6060    32.7020
     82.040     1.6150    32.7100
     83.040     1.6200    32.7150
     84.000     1.6230    32.7180
     85.000     1.6250    32.7230
     86.000     1.6350    32.7290
     87.000     1.6370    32.7330
     88.000     1.6350    32.7320
     89.000     1.6380    32.7340
     90.040     1.6430    32.7390
     91.000     1.6500    32.7480
     92.000     1.6660    32.7590
     93.000     1.6780    32.7700
     94.040     1.6900    32.7810
     95.040     1.7120    32.7950
     96.000     1.7340    32.8110
     97.000     1.8060    32.8450
     97.960     1.9910    32.9460
     99.000     2.2040    33.0500
    100.000     2.3550    33.1360
    101.000     2.3970    33.1630
    101.960     2.4620    33.2070
    103.000     2.5410    33.2790
    104.040     2.5850    33.3230
    105.040     2.6760    33.3950
    106.040     2.6940    33.4220
    106.960     2.6930    33.4230
    108.000     2.6920    33.4190
    109.000     2.7110    33.4310
    110.040     2.7130    33.4330
    111.000     2.7060    33.4310
    111.960     2.7020    33.4250
    113.000     2.7060    33.4280
    114.040     2.7070    33.4280
    115.000     2.7100    33.4330
    115.960     2.7120    33.4340
    117.000     2.7190    33.4400
    118.000     2.7190    33.4400
    119.000     2.7190    33.4390
    120.000     2.7180    33.4400
    121.000     2.7170    33.4370
    122.000     2.7230    33.4390
    123.000     2.7390    33.4550
    124.000     2.7410    33.4560
    125.000     2.7410    33.4570
    126.000     2.7420    33.4570
    127.000     2.7470    33.4610
    127.960     2.7470    33.4610
    128.960     2.7530    33.4640
    130.000     2.7550    33.4640
    131.000     2.7450    33.4630
    132.000     2.7460    33.4610
    133.000     2.7480    33.4620
    134.040     2.7500    33.4630
    135.000     2.7560    33.4650
    135.960     2.7570    33.4660
    137.000     2.7540    33.4650
    138.000     2.7480    33.4620
    139.000     2.7510    33.4630
    140.000     2.7490    33.4630
    141.000     2.7450    33.4610
    142.000     2.7420    33.4600
    143.080     2.7440    33.4600
    144.040     2.7470    33.4610
    145.000     2.7410    33.4580
    146.040     2.7320    33.4550
    147.040     2.7310    33.4540
# Drift Data
#      C Npts comp n_var
I0    98   21    1     4
#      Pres       Temp   Salinity     Time
    151.600     2.7690    33.4740        5
    148.760     2.7690    33.4740      611
    151.080     2.7640    33.4720     1216
    150.080     2.7550    33.4670     1822
    148.320     2.7550    33.4640     2427
    151.800     2.7530    33.4640     3033
    153.520     2.7410    33.4590     3638
    153.800     2.7500    33.4640     4244
    147.560     2.7480    33.4610     4849
    146.560     2.7410    33.4600     5454
    144.640     2.7440    33.4610     6060
    141.720     2.7370    33.4570     6665
    138.360     2.7210    33.4500     7271
    123.680     2.7010    33.4380     7876
    126.640     2.7140    33.4460     8482
    134.640     2.7220    33.4510     9087
    145.080     2.7190    33.4500     9693
    147.280     2.7210    33.4490    10298
    146.880     2.7200    33.4490    10904
    143.800     2.7400    33.4580    11509
    143.920     2.7470    33.4610    12114
# Surface Pressure Time Series
#    C npts n_var               Start
S   98    31     6 2022/09/19 18:08:58
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    73      0.170      2.300   -999.000      14 2022/09/19 18:10:11
    86      0.170      2.300   -999.000      14 2022/09/19 18:10:24
    96      0.190      2.299   -999.000      14 2022/09/19 18:10:34
   107      0.220      2.321   -999.000      14 2022/09/19 18:10:45
   119      0.210      2.314   -999.000      14 2022/09/19 18:10:57
   130      0.220      2.320   -999.000      14 2022/09/19 18:11:08
   141      0.160      2.358   -999.000      14 2022/09/19 18:11:19
   152      0.160      2.312   -999.000      14 2022/09/19 18:11:30
   163      0.170      2.373   -999.000      14 2022/09/19 18:11:41
   174      0.190      2.323   -999.000      14 2022/09/19 18:11:52
   185      0.190      2.408   -999.000      14 2022/09/19 18:12:03
   196      0.190      2.375   -999.000      14 2022/09/19 18:12:14
   207      0.170      2.653   -999.000      14 2022/09/19 18:12:25
   218      0.180      2.452   -999.000      14 2022/09/19 18:12:36
   229      0.090      2.630   -999.000      14 2022/09/19 18:12:47
   241      0.160      2.382   -999.000      14 2022/09/19 18:12:59
   252      0.090      2.394   -999.000      14 2022/09/19 18:13:10
   263      0.160      2.611   -999.000      14 2022/09/19 18:13:21
   274      0.160      2.769   -999.000      14 2022/09/19 18:13:32
   285      0.100      2.472   -999.000      14 2022/09/19 18:13:43
   296      0.150      2.658   -999.000      14 2022/09/19 18:13:54
   307      0.180      2.745   -999.000      14 2022/09/19 18:14:05
   318      0.130      2.914   -999.000      14 2022/09/19 18:14:16
   329      0.080      3.156   -999.000      14 2022/09/19 18:14:27
   340      0.130      3.317   -999.000      14 2022/09/19 18:14:38
   351      0.190      3.230   -999.000      14 2022/09/19 18:14:49
   363      0.130      2.952   -999.000      14 2022/09/19 18:15:01
   374      0.160      3.159   -999.000      14 2022/09/19 18:15:12
   385      0.110      3.326   -999.000      14 2022/09/19 18:15:23
   396      0.090      3.482   -999.000      14 2022/09/19 18:15:34
   407      0.170      3.631   -999.000      14 2022/09/19 18:15:45
# Fall Time Series
#  Cycle npts     Start                Date
F00   98   39 716906893 2022/09/19 12:48:13
# time     Pres phase                Date
     0    0.120     1 2022/09/19 12:48:13
   281    5.520    10 2022/09/19 12:52:54
   310   10.240    10 2022/09/19 12:53:23
   342   15.280    10 2022/09/19 12:53:55
   375   20.200    10 2022/09/19 12:54:28
   407   25.520    10 2022/09/19 12:55:00
   433   30.040    10 2022/09/19 12:55:26
   462   35.040    10 2022/09/19 12:55:55
   473   36.880     2 2022/09/19 12:56:06
   495   40.360    10 2022/09/19 12:56:28
   528   45.040    10 2022/09/19 12:57:01
   563   50.200    10 2022/09/19 12:57:36
   602   55.360    10 2022/09/19 12:58:15
   641   60.160    10 2022/09/19 12:58:54
   696   65.080    10 2022/09/19 12:59:49
   788   70.040    10 2022/09/19 13:01:21
   942   75.040    10 2022/09/19 13:03:55
  1020   80.040    10 2022/09/19 13:05:13
  1095   85.040    10 2022/09/19 13:06:28
  1188   90.080    10 2022/09/19 13:08:01
  1256   95.040    10 2022/09/19 13:09:09
  1376  100.040    10 2022/09/19 13:11:09
  1676  104.440    10 2022/09/19 13:16:09
  1760  109.120    10 2022/09/19 13:17:33
  1857  114.040    10 2022/09/19 13:19:10
  2093  119.000    10 2022/09/19 13:23:06
  2335  124.040    10 2022/09/19 13:27:08
  2550  129.040    10 2022/09/19 13:30:43
  2822  134.040    10 2022/09/19 13:35:15
  2923  139.120    10 2022/09/19 13:36:56
  3223  141.880    10 2022/09/19 13:41:56
  3523  144.560    10 2022/09/19 13:46:56
  3765  149.000    10 2022/09/19 13:50:58
  4065  150.760    10 2022/09/19 13:55:58
  4365  151.840    10 2022/09/19 14:00:58
  4381  152.200     3 2022/09/19 14:01:14
  4666  151.080    10 2022/09/19 14:05:59
  4966  153.080    10 2022/09/19 14:10:59
  5022  151.720    10 2022/09/19 14:11:55
# Rise Time Series
#  Cycle npts     Start                Date
R00   98   29 716924524 2022/09/19 17:42:04
# time     Pres phase                Date
     0  147.200     7 2022/09/19 17:42:04
   135  141.440    13 2022/09/19 17:44:19
   197  135.680    13 2022/09/19 17:45:21
   258  130.520    13 2022/09/19 17:46:22
   320  124.440    13 2022/09/19 17:47:24
   381  119.320    13 2022/09/19 17:48:25
   443  113.200    13 2022/09/19 17:49:27
   484  108.680    13 2022/09/19 17:50:08
   545  102.120    13 2022/09/19 17:51:09
   607   97.120    13 2022/09/19 17:52:11
   668   91.680    13 2022/09/19 17:53:12
   709   86.800    13 2022/09/19 17:53:53
   750   81.600    13 2022/09/19 17:54:34
   791   76.480    13 2022/09/19 17:55:15
   832   71.320    13 2022/09/19 17:55:56
   873   66.760    13 2022/09/19 17:56:37
   935   61.160    13 2022/09/19 17:57:39
   996   56.160    13 2022/09/19 17:58:40
  1058   50.920    13 2022/09/19 17:59:42
  1119   45.720    13 2022/09/19 18:00:43
  1181   40.960    13 2022/09/19 18:01:45
  1242   34.920    13 2022/09/19 18:02:46
  1304   29.040    13 2022/09/19 18:03:48
  1345   24.680    13 2022/09/19 18:04:29
  1406   18.120    13 2022/09/19 18:05:30
  1447   13.720    13 2022/09/19 18:06:11
  1509    8.400    13 2022/09/19 18:07:13
  1570    2.520    13 2022/09/19 18:08:14
  1612    0.160     8 2022/09/19 18:08:56
# Pump Time Series
#   Cycle npts
B02    98   13
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    36.880  305   15.19     229    660    -9     2
   153.120    2   15.25     348    651    -9     3
   147.200   94   15.34     281    657    -9     7
   134.120   21   15.33     294    659    -9    13
   121.040   20   15.34     263    661    -9    13
   100.480   20   15.33     253    664    -9    13
    95.560   21   15.32     265    666    -9    13
    61.160   20   15.32     251    671    -9    13
    56.160   21   15.32     229    672    -9    13
    47.360   20   15.31     253    674    -9    13
    42.480   21   15.31     220    676    -9    13
     8.400   20   15.31     241    677    -9    13
     0.160  120   15.27     228    784    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    98  2.4   12    90   125   600    80    10   210      8     1    4020    5  0.2714
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17   98 2022/09/19 12:48:10
# Pcase vacuums
   8.880   6.510
# Oil valve positions
   1.785   0.824   0.631
# Seek attempts and pump time
      1      2
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18   98 2022/09/19 14:11:56 9999/99/99 99:99:99
# Pcase vacuums
   6.510   6.510
# Average 
#   P      T      S
  144.760   2.740  33.458
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18   98 2022/09/19 17:42:04 9999/99/99 99:99:99
# Pcase vacuums
   6.510   6.770
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20   98 2022/09/19 18:08:56
# number of packets previous packets
      7               9
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.770   8.710
# Air Pump max current time  max pressure
      44       415     7.462
# Air Valve Positions
   1.260   0.000
# Air Valve Calibrations
   1.176   0.158   0.667
# Oil Valve Positions
   1.806   0.833
# Oil Valve Calibrations
   1.740   0.876
# Encountered Error Codes
#    Cycle Errors
EE20    98      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00   98  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.200    468   6.330
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   98       197    2022/09/19 18:16:38
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =       98	(cycle number) |
     MPS =        0	(phase) |
     MPC =       98	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      125	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    42104	(vac 0 inHg cal) |
     PV5 =    18698	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =       97	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       36	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      305	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        2	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =       97	(cycle) |
  PTO[0] =    12600	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =       97	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =       97	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      340	(sec) |
     SSP =      120	(sec) |

