# Message Decoder for Solo2, Alto and Deep Solo Floats; v230712
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11342    89  264 000000000881890   617     0   6 2022/09/18, 02:09:17,  59.9709  -41.8061  79      2 6,14,14
SBD 11342    89  340 000000000881890   618     0   5 2022/09/18, 02:09:28,  59.9130  -43.1785  12      0 13
SBD 11342    89  299 000000000881890   619     0   4 2022/09/18, 02:09:39,  59.9709  -41.8061  76      2 8
SBD 11342    89  323 000000000881890   620     0   3 2022/09/18, 02:09:53,  59.9502  -41.8595  83      2 2,0,15
SBD 11342    89  322 000000000881890   621     0   2 2022/09/18, 02:10:05,  59.8903  -43.1296  12      0 1,3
SBD 11342    89  310 000000000881890   622     0   1 2022/09/18, 02:10:18,  59.9502  -41.8595  74      2 7,5,14,13
SBD 11342    89  315 000000000881890   623     0   0 2022/09/18, 02:10:49,  59.9709  -41.8061  82      2 0,4,14,14,9,10,11
# Delayed Transmission - (7/7) Messages transmitted   0.3 days after surfacing on 2022/09/17 20:06:38 (refer to 1st Rise scan < 3dbar)
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         2089           84        2173
# Sensor_ID bytes     % packets Description
          0    48  2.30       2         GPS
          1   121  5.79       1    Pressure
          2   221 10.58       1 Temperature
          3   189  9.05       1    Salinity
          4   103  4.93       1        Fall
          5   168  8.04       1        Rise
          6   158  7.56       1        Pump
          7    35  1.68       1 Drift_times
          8   287 13.74       1     Surface
          9    44  2.11       1  HR/Drift_P
         10    44  2.11       1  HR/Drift_T
         11    44  2.11       1  HR/Drift_S
         13   404 19.34       2  Parameters
         14   157  7.52       5 Engineering
         15    66  3.16       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   89 1 9999/99/99 99:99 +99.99000 +999.99000    0  0  0  0  0  0.0     0
G   89 2 9999/99/99 99:99 +99.99000 +999.99000    0  0  0  0  0  0.0     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0    89  170    1     3         0
#      Pres      Temp    Salinity
      0.360     1.3970    32.6290
      0.960     1.3980    32.6310
      1.960     1.3990    32.6330
      2.960     1.4000    32.6330
      4.040     1.3980    32.6320
      5.040     1.4010    32.6320
      5.920     1.4000    32.6320
      7.040     1.4000    32.6330
      8.000     1.3990    32.6330
      9.000     1.3980    32.6320
     10.040     1.3990    32.6320
     11.040     1.3980    32.6320
     12.080     1.3940    32.6300
     13.040     1.3930    32.6290
     13.960     1.3940    32.6290
     15.000     1.3930    32.6300
     16.080     1.3930    32.6300
     17.000     1.3990    32.6330
     18.120     1.4050    32.6380
     19.000     1.4010    32.6360
     19.960     1.4000    32.6340
     20.960     1.4000    32.6330
     22.000     1.4040    32.6350
     22.960     1.4010    32.6340
     24.000     1.4100    32.6370
     25.000     1.4150    32.6410
     26.040     1.4120    32.6420
     27.000     1.4140    32.6410
     27.960     1.4210    32.6450
     29.000     1.4410    32.6570
     30.000     1.4390    32.6570
     31.000     1.4380    32.6560
     31.960     1.4210    32.6470
     33.000     1.4180    32.6450
     34.040     1.4340    32.6510
     35.000     1.4380    32.6570
     36.000     1.4380    32.6580
     37.000     1.4390    32.6560
     38.040     1.4660    32.6680
     39.040     1.6010    32.7500
     40.080     1.5980    32.7520
     41.040     1.6010    32.7630
     42.040     1.6110    32.7570
     43.000     1.6560    32.7950
     44.000     1.6730    32.8060
     45.000     1.6940    32.8150
     46.040     1.6790    32.8120
     47.040     1.6840    32.8100
     48.000     1.6790    32.8080
     49.000     1.7160    32.8200
     49.960     1.7170    32.8370
     51.000     1.7350    32.8460
     52.040     1.7510    32.8510
     53.040     1.7580    32.8630
     54.000     1.7520    32.8580
     54.960     1.7790    32.8700
     56.040     1.8030    32.8830
     57.080     1.8100    32.8910
     57.960     1.8160    32.8950
     59.000     1.8170    32.8920
     60.040     1.8400    32.9060
     61.000     1.8540    32.9110
     61.960     1.8870    32.9250
     62.960     1.8960    32.9280
     64.000     1.9130    32.9350
     65.000     1.9230    32.9400
     66.000     1.9350    32.9440
     67.000     1.9330    32.9440
     68.000     1.9380    32.9450
     69.000     1.9360    32.9450
     70.000     1.9330    32.9450
     71.000     1.9400    32.9480
     72.040     1.9370    32.9490
     73.040     1.9370    32.9490
     74.000     2.0060    32.9620
     74.960     2.1050    32.9860
     76.000     2.1440    33.0010
     77.040     2.1580    33.0060
     78.000     2.1930    33.0110
     79.000     2.2010    33.0140
     80.000     2.2030    33.0160
     81.000     2.2140    33.0190
     82.000     2.2080    33.0220
     83.000     2.2990    33.0290
     84.000     2.3570    33.0460
     85.000     2.4270    33.0690
     86.000     2.4770    33.0820
     87.000     2.5140    33.1020
     88.000     2.5150    33.1120
     89.000     2.5160    33.1280
     90.040     2.4800    33.1410
     91.040     2.4300    33.1430
     92.000     2.3900    33.1450
     93.000     2.3790    33.1470
     94.000     2.3140    33.1470
     95.000     2.1880    33.1460
     96.000     2.1570    33.1430
     97.000     2.1320    33.1380
     98.000     2.1180    33.1380
     99.000     2.1000    33.1390
    100.040     2.0680    33.1320
    101.000     2.0670    33.1400
    102.000     2.0380    33.1310
    103.040     2.0210    33.1330
    104.040     2.0280    33.1300
    105.040     2.0230    33.1310
    106.040     1.9870    33.1290
    107.040     1.8740    33.1270
    108.040     1.7780    33.1160
    109.040     1.7240    33.1050
    110.000     1.4530    33.1010
    111.000     1.2900    33.1120
    112.000     0.8180    33.0660
    113.000     0.4990    33.0530
    114.000     0.1880    33.0660
    115.000     0.0360    33.0790
    116.000     0.0400    33.0860
    117.040     0.0610    33.0920
    118.000     0.0700    33.0980
    119.000     0.0720    33.1040
    120.040     0.1160    33.1160
    121.000     0.1980    33.1510
    122.000     0.2370    33.1770
    123.000     0.3390    33.2000
    124.000     0.5350    33.2480
    125.000     0.5060    33.2800
    125.960     0.7540    33.3040
    126.960     1.6840    33.4650
    128.000     2.1660    33.5450
    129.000     2.4810    33.5970
    130.000     2.8530    33.6750
    131.000     3.1230    33.7120
    132.000     3.3860    33.7960
    132.960     3.5830    33.8280
    134.000     3.7400    33.8640
    135.000     3.8270    33.9090
    136.000     3.8970    33.9120
    137.040     4.1710    33.9680
    138.040     4.2540    33.9860
    139.000     4.3100    34.0040
    140.000     4.3100    34.0250
    140.960     4.2810    34.0500
    142.000     4.1530    34.0630
    143.040     4.0050    34.0360
    144.040     3.9150    34.0680
    145.040     3.6780    34.0160
    146.040     3.5020    34.0250
    147.000     2.9650    33.9650
    148.040     2.9820    33.9630
    149.000     2.9350    33.9630
    150.000     2.9420    33.9610
    151.040     2.8930    33.9560
    152.040     2.8700    33.9520
    153.000     2.8440    33.9520
    153.960     2.8110    33.9480
    154.960     2.7960    33.9530
    156.040     2.7870    33.9620
    157.040     2.7810    33.9720
    158.000     2.7870    33.9820
    159.040     2.7860    33.9820
    159.960     2.7850    33.9800
    160.960     2.7850    33.9820
    162.000     2.7890    33.9860
    163.000     2.7940    33.9920
    164.000     2.7970    33.9960
    166.040     2.8060    34.0040
    166.960     2.8080    34.0070
    168.000     2.8100    34.0070
    169.200     2.8180    34.0140
    169.520     2.8200    34.0150
# Drift Data
#      C Npts comp n_var
I0    89   21    1     4
#      Pres       Temp   Salinity     Time
    170.560     2.9220    34.0700        5
    170.520     2.9080    34.0570      611
    170.480     2.8970    34.0500     1216
    170.480     2.8890    34.0430     1822
    170.400     2.8740    34.0330     2427
    170.440     2.8630    34.0260     3033
    171.080     2.8560    34.0210     3638
    170.680     2.8520    34.0180     4243
    168.760     2.8430    34.0120     4849
    171.760     2.8490    34.0150     5454
    170.920     2.8500    34.0180     6060
    167.000     2.8500    34.0240     6665
    167.120     2.8450    34.0220     7271
    167.600     2.8460    34.0260     7876
    167.720     2.8460    34.0280     8482
    168.400     2.8450    34.0340     9087
    169.360     2.8400    34.0310     9693
    169.040     2.8360    34.0290    10298
    167.640     2.8300    34.0230    10904
    167.600     2.8290    34.0230    11509
    167.920     2.8140    34.0090    12114
# Surface Pressure Time Series
#    C npts n_var               Start
S   89    31     6 2022/09/17 20:06:39
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    75      0.180      1.463   -999.000      14 2022/09/17 20:07:54
    88      0.290      1.567   -999.000      14 2022/09/17 20:08:07
    99      0.210      1.485   -999.000      14 2022/09/17 20:08:18
   110      0.310      1.479   -999.000      14 2022/09/17 20:08:29
   121      0.420      1.466   -999.000      14 2022/09/17 20:08:40
   132      0.240      1.524   -999.000      14 2022/09/17 20:08:51
   143      0.190      1.527   -999.000      14 2022/09/17 20:09:02
   154      0.470      1.434   -999.000      14 2022/09/17 20:09:13
   165      0.720      1.415   -999.000      14 2022/09/17 20:09:24
   176      1.040      1.420   -999.000      14 2022/09/17 20:09:35
   187      1.040      1.419   -999.000      14 2022/09/17 20:09:46
   198      1.310      1.412   -999.000      14 2022/09/17 20:09:57
   210      2.050      1.405   -999.000      14 2022/09/17 20:10:09
   221      3.140      1.399   -999.000      14 2022/09/17 20:10:20
   232      4.290      1.401   -999.000      14 2022/09/17 20:10:31
   243      5.400      1.403   -999.000      14 2022/09/17 20:10:42
   254      6.120      1.404   -999.000      14 2022/09/17 20:10:53
   265      6.840      1.405   -999.000      14 2022/09/17 20:11:04
   276      7.750      1.405   -999.000      14 2022/09/17 20:11:15
   287      8.870      1.403   -999.000      14 2022/09/17 20:11:26
   298     10.680      1.400   -999.000      14 2022/09/17 20:11:37
   309     12.960      1.403   -999.000      14 2022/09/17 20:11:48
   320     15.490      1.401   -999.000      14 2022/09/17 20:11:59
   332     18.040      1.402   -999.000      14 2022/09/17 20:12:11
   343     20.780      1.401   -999.000      14 2022/09/17 20:12:22
   354     23.430      1.400   -999.000      14 2022/09/17 20:12:33
   365     25.500      1.402   -999.000      14 2022/09/17 20:12:44
   376     27.560      1.403   -999.000      14 2022/09/17 20:12:55
   387     29.500      1.404   -999.000      14 2022/09/17 20:13:06
   398     31.550      1.403   -999.000      14 2022/09/17 20:13:17
   409     33.300      1.405   -999.000      14 2022/09/17 20:13:28
# Fall Time Series
#  Cycle npts     Start                Date
F00   89   19 716741481 2022/09/17 14:51:21
# time     Pres phase                Date
     0  170.920     1 2022/09/17 14:51:21
     7  170.960     2 2022/09/17 14:51:28
   304  170.920    10 2022/09/17 14:56:25
   604  170.880    10 2022/09/17 15:01:25
   904  170.800    10 2022/09/17 15:06:25
  1204  170.800    10 2022/09/17 15:11:25
  1504  170.800    10 2022/09/17 15:16:25
  1804  170.720    10 2022/09/17 15:21:25
  2104  170.720    10 2022/09/17 15:26:25
  2404  170.720    10 2022/09/17 15:31:25
  2705  170.680    10 2022/09/17 15:36:26
  3005  170.680    10 2022/09/17 15:41:26
  3305  170.680    10 2022/09/17 15:46:26
  3605  170.600    10 2022/09/17 15:51:26
  3886  170.600     3 2022/09/17 15:56:07
  3905  170.600    10 2022/09/17 15:56:26
  4206  170.600    10 2022/09/17 16:01:27
  4506  170.560    10 2022/09/17 16:06:27
  4529  170.520    10 2022/09/17 16:06:50
# Rise Time Series
#  Cycle npts     Start                Date
R00   89   32 716758619 2022/09/17 19:36:59
# time     Pres phase                Date
     0  169.080     7 2022/09/17 19:36:59
   156  164.760    13 2022/09/17 19:39:35
   217  159.400    13 2022/09/17 19:40:36
   279  153.320    13 2022/09/17 19:41:38
   340  147.480    13 2022/09/17 19:42:39
   402  142.320    13 2022/09/17 19:43:41
   463  136.680    13 2022/09/17 19:44:42
   525  131.200    13 2022/09/17 19:45:44
   586  126.000    13 2022/09/17 19:46:45
   648  120.920    13 2022/09/17 19:47:47
   709  115.680    13 2022/09/17 19:48:48
   771  110.560    13 2022/09/17 19:49:50
   832  105.200    13 2022/09/17 19:50:51
   873  100.760    13 2022/09/17 19:51:32
   935   94.520    13 2022/09/17 19:52:34
   996   88.400    13 2022/09/17 19:53:35
  1058   82.600    13 2022/09/17 19:54:37
  1119   76.800    13 2022/09/17 19:55:38
  1181   70.720    13 2022/09/17 19:56:40
  1242   64.440    13 2022/09/17 19:57:41
  1304   58.320    13 2022/09/17 19:58:43
  1345   53.480    13 2022/09/17 19:59:24
  1386   48.560    13 2022/09/17 20:00:05
  1430   43.560    13 2022/09/17 20:00:49
  1491   38.280    13 2022/09/17 20:01:50
  1553   32.400    13 2022/09/17 20:02:52
  1594   27.320    13 2022/09/17 20:03:33
  1635   22.520    13 2022/09/17 20:04:14
  1676   17.120    13 2022/09/17 20:04:55
  1717   11.320    13 2022/09/17 20:05:36
  1758    4.440    13 2022/09/17 20:06:17
  1779    0.560     8 2022/09/17 20:06:38
# Pump Time Series
#   Cycle npts
B02    89   14
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
   170.960  276   15.21     283    648    -9     2
   170.560    4   15.28     330    644    -9     3
   169.080   94   15.35     288    652    -9     7
   166.040   21   15.36     294    654    -9    13
   161.320   20   15.35     274    656    -9    13
   144.040   21   15.34     288    657    -9    13
   131.200   20   15.34     291    656    -9    13
   126.000   21   15.34     248    661    -9    13
   120.920   20   15.33     270    667    -9    13
   113.960   20   15.33     257    672    -9    13
   107.160   20   15.33     247    672    -9    13
    42.040   21   15.31     242    673    -9    13
    34.640   21   15.32     223    675    -9    13
    33.280  120   15.28     216    782    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    89  2.4   11    90   125   600    79    10   210      8     1    4020    5  0.2707
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17   89 2022/09/17 14:51:19
# Pcase vacuums
   6.250   6.450
# Oil valve positions
   1.808   0.823   0.633
# Seek attempts and pump time
      1      4
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18   89 2022/09/17 16:06:51 9999/99/99 99:99:99
# Pcase vacuums
   6.450   6.460
# Average 
#   P      T      S
  169.320   2.856  34.029
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18   89 2022/09/17 19:36:59 9999/99/99 99:99:99
# Pcase vacuums
   6.460   6.760
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20   89 2022/09/17 20:06:38
# number of packets previous packets
     14              43
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.760   8.070
# Air Pump max current time  max pressure
      47       230    11.019
# Air Valve Positions
   1.259   0.000
# Air Valve Calibrations
   1.176   0.158   0.667
# Oil Valve Positions
   1.805   0.834
# Oil Valve Calibrations
   1.740   0.876
# Encountered Error Codes
#    Cycle Errors
EE20    89      1
#               Date   Code Description
 2022/09/17 14:51:21 0x0013 Descend
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00   89  1      0
# MC    Pres      Time  Pcase Vacuum
   1   170.960      2   6.260
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   89       197    2022/09/18 02:10:18
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =       89	(cycle number) |
     MPS =        0	(phase) |
     MPC =       89	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      125	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    42104	(vac 0 inHg cal) |
     PV5 =    18698	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =       88	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       36	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      275	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        1	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =       88	(cycle) |
  PTO[0] =    12600	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =       88	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =       88	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      340	(sec) |
     SSP =      120	(sec) |

