# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11339   155  214 000000000932510  1072     0   6 2022/09/29, 00:12:08,  63.2054  -52.5575   2      0 6,14,14
SBD 11339   155  340 000000000932510  1073     0   5 2022/09/29, 00:12:19,  63.2054  -52.5575   2      0 13
SBD 11339   155  299 000000000932510  1074     0   4 2022/09/29, 00:12:30,  63.2054  -52.5575   2      0 8
SBD 11339   155  304 000000000932510  1075     0   3 2022/09/29, 00:12:47,  63.2054  -52.5575   2      0 3,15,13
SBD 11339   155  296 000000000932510  1076     0   2 2022/09/29, 00:12:59,  63.2054  -52.5575   2      0 1,2
SBD 11339   155  336 000000000932510  1077     0   1 2022/09/29, 00:13:11,  63.2054  -52.5575   2      0 9,10,11,7,5
SBD 11339   155  331 000000000932510  1078     0   0 2022/09/29, 00:13:21,  63.2054  -52.5575   2      0 0,4,14,14,14,0
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         2036           84        2120
# Sensor_ID bytes     % packets Description
          0    48  2.36       2         GPS
          1   106  5.21       1    Pressure
          2   178  8.74       1 Temperature
          3   150  7.37       1    Salinity
          4   208 10.22       1        Fall
          5   148  7.27       1        Rise
          6   114  5.60       1        Pump
          7    35  1.72       1 Drift_times
          8   287 14.10       1     Surface
          9    54  2.65       1  HR/Drift_P
         10    44  2.16       1  HR/Drift_T
         11    43  2.11       1  HR/Drift_S
         13   404 19.84       2  Parameters
         14   151  7.42       5 Engineering
         15    66  3.24       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  155 1 2022/09/28 18:45 +63.16863  -52.61498    0  4 35 41 45  3.0     0
G  155 2 2022/09/29 00:06 +63.19145  -52.56412   60  5 38 41 43  1.7     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   155  155    1     3         0
#      Pres      Temp    Salinity
      1.080     3.3300    33.3440
      2.000     3.3280    33.3440
      3.000     3.3300    33.3440
      4.000     3.3260    33.3440
      5.040     3.2980    33.3480
      5.960     3.2870    33.3510
      7.000     3.2850    33.3640
      8.040     3.3220    33.3820
      9.000     3.3250    33.3930
     10.000     3.3100    33.3930
     11.040     3.2830    33.3930
     12.040     3.2570    33.3920
     12.960     3.2420    33.3930
     13.960     3.2030    33.3960
     14.920     3.1510    33.3920
     15.960     3.1580    33.3950
     17.000     3.1510    33.4120
     18.040     3.1280    33.4140
     19.080     3.1280    33.4160
     20.040     3.1290    33.4180
     21.040     3.1320    33.4190
     22.000     3.1550    33.4210
     22.960     3.1920    33.4290
     24.000     3.2270    33.4350
     25.000     3.2500    33.4380
     26.040     3.3340    33.4450
     27.000     3.3040    33.4620
     27.960     3.2730    33.4630
     28.960     3.2570    33.4650
     30.000     3.2550    33.4660
     31.040     3.2560    33.4700
     32.040     3.2580    33.4710
     33.000     3.2650    33.4770
     34.000     3.2700    33.4840
     35.040     3.2660    33.4860
     36.000     3.2610    33.4860
     36.960     3.2600    33.4870
     37.960     3.2580    33.4870
     39.000     3.2570    33.4890
     40.000     3.2590    33.4930
     41.000     3.3280    33.5120
     42.000     3.4510    33.5560
     43.040     3.5140    33.5770
     44.000     3.5530    33.5890
     45.000     3.5960    33.6000
     45.960     3.6290    33.6140
     46.960     3.6530    33.6170
     48.000     3.7270    33.6340
     49.000     3.7540    33.6470
     50.000     3.7620    33.6530
     51.000     3.7580    33.6540
     51.960     3.7580    33.6540
     52.960     3.7790    33.6600
     54.000     3.7950    33.6670
     54.960     3.7950    33.6700
     55.960     3.7900    33.6740
     57.040     3.7750    33.6760
     58.000     3.7670    33.6750
     59.000     3.7600    33.6760
     60.000     3.7580    33.6770
     60.960     3.7580    33.6780
     61.960     3.7610    33.6780
     62.960     3.7660    33.6790
     63.960     3.7720    33.6800
     65.000     3.7760    33.6850
     66.000     3.7570    33.6920
     67.000     3.7540    33.7000
     68.000     3.8100    33.7200
     69.000     3.9000    33.7620
     70.040     3.9040    33.7870
     71.000     3.8530    33.7860
     72.000     3.8140    33.7840
     73.000     3.8050    33.7830
     74.000     3.8000    33.7830
     75.000     3.7970    33.7840
     76.040     3.7930    33.7850
     77.040     3.7920    33.7850
     78.000     3.7920    33.7850
     78.960     3.7930    33.7860
     80.000     3.7930    33.7860
     81.040     3.7920    33.7870
     82.040     3.7890    33.7870
     83.000     3.7900    33.7880
     84.000     3.7890    33.7880
     85.040     3.8020    33.7920
     86.040     3.8320    33.8040
     87.040     3.8180    33.8110
     88.040     3.8130    33.8120
     89.040     3.8040    33.8130
     90.000     3.7970    33.8140
     90.960     3.7790    33.8250
     91.960     3.7730    33.8310
     93.000     3.7810    33.8350
     94.040     3.7840    33.8360
     95.000     3.7870    33.8370
     95.960     3.7870    33.8370
     97.000     3.7880    33.8380
     97.960     3.7890    33.8380
     98.920     3.7910    33.8390
     99.960     3.7950    33.8400
    101.000     3.8060    33.8430
    102.000     3.8370    33.8510
    103.040     3.8630    33.8600
    104.040     3.8830    33.8660
    105.000     3.8950    33.8710
    106.040     3.9070    33.8740
    107.040     3.9160    33.8780
    108.040     3.9250    33.8820
    109.040     3.9290    33.8840
    110.000     3.9310    33.8850
    111.000     3.9360    33.8870
    112.000     3.9470    33.8900
    112.960     3.9510    33.8920
    114.000     3.9540    33.8930
    115.040     3.9600    33.8960
    116.040     3.9690    33.8980
    117.040     3.9800    33.9030
    118.000     3.9890    33.9030
    119.000     4.0110    33.9100
    120.000     4.0130    33.9120
    121.040     4.0120    33.9120
    122.000     4.0120    33.9130
    123.000     4.0130    33.9140
    124.000     4.0180    33.9170
    125.000     4.0260    33.9210
    126.000     4.0340    33.9260
    127.000     4.0440    33.9320
    128.000     4.0790    33.9420
    129.000     4.0950    33.9510
    130.000     4.1020    33.9550
    131.000     4.1100    33.9590
    131.960     4.1160    33.9620
    132.960     4.1320    33.9680
    133.960     4.1620    33.9830
    135.000     4.1810    33.9920
    136.040     4.2290    34.0090
    137.000     4.2500    34.0200
    138.000     4.2660    34.0250
    139.000     4.2840    34.0330
    140.040     4.2860    34.0340
    141.000     4.2900    34.0350
    141.920     4.2980    34.0390
    142.960     4.2990    34.0400
    143.960     4.3050    34.0430
    145.000     4.3140    34.0450
    146.000     4.3190    34.0480
    147.000     4.3480    34.0560
    148.000     4.3630    34.0640
    149.000     4.3710    34.0680
    149.960     4.3720    34.0680
    150.960     4.3730    34.0690
    151.960     4.3740    34.0690
    152.960     4.3760    34.0700
    154.040     4.3990    34.0770
    154.800     4.4070    34.0790
# Drift Data
#      C Npts comp n_var
I0   155   21    1     4
#      Pres       Temp   Salinity     Time
    162.240     4.3370    34.0560        6
    163.560     4.3530    34.0640      612
    165.520     4.3430    34.0550     1217
    159.960     4.3470    34.0570     1823
    159.840     4.3460    34.0590     2428
    160.000     4.3500    34.0610     3034
    161.000     4.3530    34.0610     3640
    166.440     4.3550    34.0620     4245
    162.600     4.3670    34.0660     4851
    163.480     4.3610    34.0630     5456
    163.000     4.3580    34.0630     6062
    163.400     4.3660    34.0650     6668
    160.720     4.3620    34.0630     7273
    158.640     4.3610    34.0640     7879
    153.760     4.3560    34.0650     8484
    160.040     4.3680    34.0630     9090
    163.520     4.3660    34.0660     9696
    158.480     4.3670    34.0650    10301
    160.600     4.3680    34.0660    10907
    160.360     4.3710    34.0680    11512
    156.360     4.3850    34.0720    12118
# Surface Pressure Time Series
#    C npts n_var               Start
S  155    31     6 2022/09/29 00:05:03
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    72      0.210      3.335   -999.000      14 2022/09/29 00:06:15
    85      0.140      3.337   -999.000      14 2022/09/29 00:06:28
    96      0.180      3.349   -999.000      14 2022/09/29 00:06:39
   107      0.180      3.340   -999.000      14 2022/09/29 00:06:50
   118      0.190      3.342   -999.000      14 2022/09/29 00:07:01
   129      0.190      3.361   -999.000      14 2022/09/29 00:07:12
   140      0.260      3.349   -999.000      14 2022/09/29 00:07:23
   152      0.180      3.361   -999.000      14 2022/09/29 00:07:35
   163      0.150      3.359   -999.000      14 2022/09/29 00:07:46
   174      0.150      3.414   -999.000      14 2022/09/29 00:07:57
   185      0.270      3.399   -999.000      14 2022/09/29 00:08:08
   196      0.210      3.375   -999.000      14 2022/09/29 00:08:19
   207      0.160      3.444   -999.000      14 2022/09/29 00:08:30
   218      0.190      3.400   -999.000      14 2022/09/29 00:08:41
   229      0.140      3.379   -999.000      14 2022/09/29 00:08:52
   240      0.170      3.388   -999.000      14 2022/09/29 00:09:03
   251      0.200      3.388   -999.000      14 2022/09/29 00:09:14
   262      0.140      3.374   -999.000      14 2022/09/29 00:09:25
   274      0.150      3.431   -999.000      14 2022/09/29 00:09:37
   285      0.150      3.375   -999.000      14 2022/09/29 00:09:48
   296      0.120      3.398   -999.000      14 2022/09/29 00:09:59
   307      0.170      3.401   -999.000      14 2022/09/29 00:10:10
   318      0.180      3.445   -999.000      14 2022/09/29 00:10:21
   329      0.170      3.368   -999.000      14 2022/09/29 00:10:32
   340      0.160      3.392   -999.000      14 2022/09/29 00:10:43
   351      0.230      3.387   -999.000      14 2022/09/29 00:10:54
   362      0.140      3.356   -999.000      14 2022/09/29 00:11:05
   373      0.180      3.414   -999.000      14 2022/09/29 00:11:16
   384      0.160      3.497   -999.000      14 2022/09/29 00:11:27
   396      0.160      3.436   -999.000      14 2022/09/29 00:11:39
   407      0.220      3.467   -999.000      14 2022/09/29 00:11:50
# Fall Time Series
#  Cycle npts     Start                Date
F00  155   40 717705914 2022/09/28 18:45:14
# time     Pres phase                Date
     0    0.240     1 2022/09/28 18:45:14
   158    5.040    10 2022/09/28 18:47:52
   193   10.240    10 2022/09/28 18:48:27
   226   15.160    10 2022/09/28 18:49:00
   258   20.040    10 2022/09/28 18:49:32
   294   25.200    10 2022/09/28 18:50:08
   330   30.400    10 2022/09/28 18:50:44
   362   35.120    10 2022/09/28 18:51:16
   376   37.080     2 2022/09/28 18:51:30
   398   40.000    10 2022/09/28 18:51:52
   440   45.320    10 2022/09/28 18:52:34
   479   50.000    10 2022/09/28 18:53:13
   531   55.240    10 2022/09/28 18:54:05
   596   60.200    10 2022/09/28 18:55:10
   697   65.080    10 2022/09/28 18:56:51
   826   70.000    10 2022/09/28 18:59:00
   952   75.000    10 2022/09/28 19:01:06
  1076   80.000    10 2022/09/28 19:03:10
  1219   85.000    10 2022/09/28 19:05:33
  1371   90.080    10 2022/09/28 19:08:05
  1507   95.080    10 2022/09/28 19:10:21
  1643  100.000    10 2022/09/28 19:12:37
  1767  105.080    10 2022/09/28 19:14:41
  1878  110.040    10 2022/09/28 19:16:32
  1999  115.040    10 2022/09/28 19:18:33
  2117  120.040    10 2022/09/28 19:20:31
  2255  125.000    10 2022/09/28 19:22:49
  2393  130.040    10 2022/09/28 19:25:07
  2524  135.040    10 2022/09/28 19:27:18
  2691  140.040    10 2022/09/28 19:30:05
  2895  145.040    10 2022/09/28 19:33:29
  3098  150.000    10 2022/09/28 19:36:52
  3291  155.040    10 2022/09/28 19:40:05
  3520  160.040    10 2022/09/28 19:43:54
  3794  165.040    10 2022/09/28 19:48:28
  4094  166.440    10 2022/09/28 19:53:28
  4265  166.120     3 2022/09/28 19:56:19
  4395  165.560    10 2022/09/28 19:58:29
  4695  163.440    10 2022/09/28 20:03:29
  4906  162.240    10 2022/09/28 20:07:00
# Rise Time Series
#  Cycle npts     Start                Date
R00  155   28 717723429 2022/09/28 23:37:09
# time     Pres phase                Date
     0  154.880     7 2022/09/28 23:37:09
   135  148.400    13 2022/09/28 23:39:24
   196  142.880    13 2022/09/28 23:40:25
   258  137.720    13 2022/09/28 23:41:27
   319  132.120    13 2022/09/28 23:42:28
   381  126.440    13 2022/09/28 23:43:30
   442  120.600    13 2022/09/28 23:44:31
   504  114.400    13 2022/09/28 23:45:33
   565  108.560    13 2022/09/28 23:46:34
   627  102.720    13 2022/09/28 23:47:36
   688   97.240    13 2022/09/28 23:48:37
   750   91.560    13 2022/09/28 23:49:39
   811   86.280    13 2022/09/28 23:50:40
   873   80.440    13 2022/09/28 23:51:42
   934   74.480    13 2022/09/28 23:52:43
   996   68.360    13 2022/09/28 23:53:45
  1057   62.880    13 2022/09/28 23:54:46
  1119   56.920    13 2022/09/28 23:55:48
  1180   50.760    13 2022/09/28 23:56:49
  1242   44.760    13 2022/09/28 23:57:51
  1303   39.200    13 2022/09/28 23:58:52
  1365   33.720    13 2022/09/28 23:59:54
  1426   27.560    13 2022/09/29 00:00:55
  1488   21.120    13 2022/09/29 00:01:57
  1529   16.720    13 2022/09/29 00:02:38
  1590    9.920    13 2022/09/29 00:03:39
  1652    2.960    13 2022/09/29 00:04:41
  1673    0.840     8 2022/09/29 00:05:02
# Pump Time Series
#   Cycle npts
B02   155   10
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    37.080  286   15.11     210   1137    -9     2
   166.040    2   15.16     336   1137    -9     3
   154.880   94   15.24     263   1137    -9     7
   148.400   20   15.23     260   1137    -9    13
   139.400   21   15.24     267   1137    -9    13
   124.680   21   15.23     260   1137    -9    13
    88.000   20   15.23     231   1137    -9    13
    62.880   21   15.21     228   1137    -9    13
    35.680   21   15.21     221   1137    -9    13
     0.200  240   15.17     191   1137    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   155  2.4   17    90   140   600    79    10   210      8     1    4020    5  0.2707
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  155 2022/09/28 18:45:11
# Pcase vacuums
  11.370  11.370
# Oil valve positions
   1.725   0.897   0.578
# Seek attempts and pump time
      1      2
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  155 2022/09/28 20:07:00 9999/99/99 99:99:99
# Pcase vacuums
  11.370  11.370
# Average 
#   P      T      S
  161.120   4.359  34.063
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  155 2022/09/28 23:37:09 9999/99/99 99:99:99
# Pcase vacuums
  11.370  11.370
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  155 2022/09/29 00:05:02
# number of packets previous packets
      7               7
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
  11.370  11.370
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.141   0.139   0.640
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.692   0.939
# Encountered Error Codes
#    Cycle Errors
EE20   155      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  155  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.200    372  11.370
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  155       197    2022/09/29 00:12:47
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      155	(cycle number) |
     MPS =        0	(phase) |
     MPC =      155	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      140	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    56487	(vac 0 inHg cal) |
     PV5 =    31723	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      154	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       36	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      286	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      154	(cycle) |
  PTO[0] =    12600	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      154	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      154	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      340	(sec) |
     SSP =      240	(sec) |

