# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11339    83  286 000000000932510   571     0   5 2022/09/12, 19:33:27,  60.4785  -48.7593   2      0 13,6,14,14
SBD 11339    83  340 000000000932510   572     0   4 2022/09/12, 19:33:38,  60.4785  -48.7593   2      0 13
SBD 11339    83  299 000000000932510   573     0   3 2022/09/12, 19:33:51,  60.4785  -48.7593   2      0 8
SBD 11339    83  305 000000000932510   574     0   2 2022/09/12, 19:34:03,  60.4785  -48.7593   2      0 2,3,15
SBD 11339    83  293 000000000932510   575     0   1 2022/09/12, 19:34:17,  60.4785  -48.7593   2      0 11,7,5,14,0,1
SBD 11339    83  333 000000000932510   576     0   0 2022/09/12, 19:34:43,  60.4785  -48.7593   3      0 0,4,14,14,9,10
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1784           72        1856
# Sensor_ID bytes     % packets Description
          0    48  2.69       2         GPS
          1    71  3.98       1    Pressure
          2   119  6.67       1 Temperature
          3   108  6.05       1    Salinity
          4   163  9.14       1        Fall
          5    93  5.21       1        Rise
          6   103  5.77       1        Pump
          7    33  1.85       1 Drift_times
          8   287 16.09       1     Surface
          9    49  2.75       1  HR/Drift_P
         10    41  2.30       1  HR/Drift_T
         11    41  2.30       1  HR/Drift_S
         13   404 22.65       2  Parameters
         14   158  8.86       5 Engineering
         15    66  3.70       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   83 1 2022/09/12 14:30 +60.46842  -48.58379    0  4 40 42 44  4.3     0
G   83 2 2022/09/12 19:27 +60.47886  -48.71169   30  5 21 33 40  2.2     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0    83   88    1     3         0
#      Pres      Temp    Salinity
      1.080     4.5780    31.4540
      1.960     3.9280    31.5480
      2.960     3.5660    31.7290
      4.000     3.2360    31.9770
      4.960     2.9950    32.1470
      6.040     2.8590    32.2450
      7.080     2.6630    32.4180
      8.040     2.5880    32.4780
      9.040     2.5590    32.5100
     10.040     2.5210    32.5840
     11.000     2.5140    32.6520
     12.000     2.5110    32.6720
     13.080     2.3760    32.8390
     14.040     2.2220    32.9180
     15.040     2.1190    32.9830
     16.040     2.0450    33.0300
     17.000     1.9330    33.0980
     18.040     1.8430    33.1670
     19.000     1.7960    33.2000
     19.960     1.7700    33.2290
     21.000     1.7520    33.2520
     22.040     1.7220    33.2800
     23.040     1.7500    33.3060
     24.040     1.7640    33.3170
     25.000     1.7230    33.3330
     25.960     1.6980    33.3540
     27.000     1.7190    33.3680
     28.000     1.7260    33.3820
     28.960     1.8210    33.4280
     29.960     1.9700    33.4680
     30.960     2.1040    33.5110
     31.960     2.2160    33.5680
     33.000     2.3230    33.6050
     34.040     2.3940    33.6300
     35.040     2.5630    33.6720
     36.040     2.7320    33.7150
     37.040     2.9530    33.7580
     38.040     3.1630    33.8200
     39.040     3.2770    33.8440
     40.040     3.3300    33.8580
     41.000     3.3610    33.8690
     42.040     3.3930    33.8770
     43.040     3.4230    33.8820
     44.040     3.4640    33.8920
     45.040     3.4370    33.9010
     46.000     3.3810    33.8960
     47.000     3.3900    33.8940
     48.000     3.4090    33.9000
     49.000     3.3980    33.9010
     50.000     3.3910    33.9000
     51.080     3.3820    33.9000
     52.000     3.3660    33.8990
     53.040     3.3650    33.8980
     54.000     3.3720    33.8990
     55.000     3.3730    33.8990
     55.960     3.3800    33.8990
     56.960     3.3560    33.9010
     58.000     3.2850    33.8990
     59.000     3.2630    33.8940
     59.960     3.2660    33.8940
     61.000     3.2650    33.8950
     61.960     3.2710    33.8970
     63.000     3.3010    33.9010
     64.040     3.3400    33.9090
     65.040     3.3870    33.9210
     66.040     3.4210    33.9330
     67.000     3.4570    33.9390
     68.040     3.4800    33.9450
     69.040     3.4950    33.9490
     70.000     3.4910    33.9510
     70.960     3.4900    33.9530
     72.040     3.5010    33.9560
     73.040     3.5110    33.9600
     74.000     3.5160    33.9640
     75.000     3.5750    33.9760
     75.960     3.6570    34.0040
     77.040     3.6750    34.0160
     78.000     3.6830    34.0190
     78.960     3.7020    34.0220
     80.000     3.7260    34.0320
     81.040     3.7440    34.0350
     82.040     3.7670    34.0400
     83.040     3.7800    34.0430
     84.040     3.7920    34.0440
     85.000     3.8060    34.0470
     86.000     3.8300    34.0510
     87.120     3.8430    34.0550
     87.520     3.8450    34.0560
# Drift Data
#      C Npts comp n_var
I0    83   18    1     4
#      Pres       Temp   Salinity     Time
     97.200     3.7320    34.0320        5
     92.040     3.7600    34.0400      611
     94.480     3.7770    34.0380     1217
     90.240     3.7800    34.0390     1822
     91.920     3.7830    34.0350     2428
     91.200     3.7790    34.0400     3033
     88.880     3.7920    34.0520     3639
     88.720     3.8320    34.0470     4245
     89.680     3.8290    34.0520     4850
     89.240     3.8140    34.0540     5456
     88.600     3.8050    34.0480     6061
     88.000     3.8200    34.0510     6667
     89.240     3.8390    34.0520     7273
     89.080     3.8220    34.0500     7878
     88.120     3.8260    34.0490     8484
     90.400     3.8260    34.0500     9089
     85.760     3.8260    34.0470     9695
     84.280     3.8420    34.0540    10301
# Surface Pressure Time Series
#    C npts n_var               Start
S   83    31     6 2022/09/12 19:26:25
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    67      0.400      5.092   -999.000      14 2022/09/12 19:27:32
    79      0.390      5.123   -999.000      14 2022/09/12 19:27:44
    90      0.390      5.163   -999.000      14 2022/09/12 19:27:55
   102      0.380      5.114   -999.000      14 2022/09/12 19:28:07
   113      0.390      5.141   -999.000      14 2022/09/12 19:28:18
   124      0.350      5.166   -999.000      14 2022/09/12 19:28:29
   135      0.310      5.174   -999.000      14 2022/09/12 19:28:40
   146      0.310      5.233   -999.000      14 2022/09/12 19:28:51
   157      0.340      5.300   -999.000      14 2022/09/12 19:29:02
   168      0.290      5.271   -999.000      14 2022/09/12 19:29:13
   179      0.320      5.295   -999.000      14 2022/09/12 19:29:24
   190      0.330      5.274   -999.000      14 2022/09/12 19:29:35
   201      0.360      5.276   -999.000      14 2022/09/12 19:29:46
   212      0.260      5.319   -999.000      14 2022/09/12 19:29:57
   224      0.350      5.359   -999.000      14 2022/09/12 19:30:09
   235      0.270      5.400   -999.000      14 2022/09/12 19:30:20
   246      0.290      5.433   -999.000      14 2022/09/12 19:30:31
   257      0.310      5.463   -999.000      14 2022/09/12 19:30:42
   268      0.340      5.475   -999.000      14 2022/09/12 19:30:53
   279      0.270      5.142   -999.000      14 2022/09/12 19:31:04
   290      0.340      5.208   -999.000      14 2022/09/12 19:31:15
   301      0.320      5.290   -999.000      14 2022/09/12 19:31:26
   312      0.290      5.361   -999.000      14 2022/09/12 19:31:37
   323      0.350      5.419   -999.000      14 2022/09/12 19:31:48
   334      0.310      5.373   -999.000      14 2022/09/12 19:31:59
   346      0.310      5.370   -999.000      14 2022/09/12 19:32:11
   357      0.310      5.403   -999.000      14 2022/09/12 19:32:22
   368      0.260      5.445   -999.000      14 2022/09/12 19:32:33
   379      0.300      5.310   -999.000      14 2022/09/12 19:32:44
   390      0.290      5.336   -999.000      14 2022/09/12 19:32:55
   401      0.290      5.336   -999.000      14 2022/09/12 19:33:06
# Fall Time Series
#  Cycle npts     Start                Date
F00   83   31 716308252 2022/09/12 14:30:52
# time     Pres phase                Date
     0    0.360     1 2022/09/12 14:30:52
   151    5.200    10 2022/09/12 14:33:23
   190   10.320    10 2022/09/12 14:34:02
   226   15.120    10 2022/09/12 14:34:38
   265   20.280    10 2022/09/12 14:35:17
   304   25.400    10 2022/09/12 14:35:56
   339   30.240    10 2022/09/12 14:36:31
   375   35.200    10 2022/09/12 14:37:07
   386   36.680     2 2022/09/12 14:37:18
   411   40.040    10 2022/09/12 14:37:43
   453   45.400    10 2022/09/12 14:38:25
   492   50.280    10 2022/09/12 14:39:04
   534   55.320    10 2022/09/12 14:39:46
   580   60.200    10 2022/09/12 14:40:32
   635   65.240    10 2022/09/12 14:41:27
   728   70.040    10 2022/09/12 14:43:00
  1028   72.840    10 2022/09/12 14:48:00
  1328   73.800    10 2022/09/12 14:53:00
  1628   73.640    10 2022/09/12 14:58:00
  1928   75.680    10 2022/09/12 15:03:00
  2228   77.080    10 2022/09/12 15:08:00
  2528   76.640    10 2022/09/12 15:13:00
  2828   76.040    10 2022/09/12 15:18:00
  3128   76.720    10 2022/09/12 15:23:00
  3428   78.480    10 2022/09/12 15:28:00
  3730   78.640    10 2022/09/12 15:33:02
  4030   79.000    10 2022/09/12 15:38:02
  4295   78.320     3 2022/09/12 15:42:27
  5243   93.200    10 2022/09/12 15:58:15
  5543   96.960    10 2022/09/12 16:03:15
  5843   97.240    10 2022/09/12 16:08:15
# Rise Time Series
#  Cycle npts     Start                Date
R00   83   17 716324905 2022/09/12 19:08:25
# time     Pres phase                Date
     0   87.400     7 2022/09/12 19:08:25
   134   82.000    13 2022/09/12 19:10:39
   196   76.280    13 2022/09/12 19:11:41
   257   70.560    13 2022/09/12 19:12:42
   319   64.840    13 2022/09/12 19:13:44
   380   59.240    13 2022/09/12 19:14:45
   442   53.160    13 2022/09/12 19:15:47
   503   47.040    13 2022/09/12 19:16:48
   565   41.200    13 2022/09/12 19:17:50
   626   35.400    13 2022/09/12 19:18:51
   688   30.040    13 2022/09/12 19:19:53
   749   24.160    13 2022/09/12 19:20:54
   811   18.760    13 2022/09/12 19:21:56
   872   13.320    13 2022/09/12 19:22:57
   934    8.760    13 2022/09/12 19:23:59
  1016    2.960    13 2022/09/12 19:25:21
  1078    0.440     8 2022/09/12 19:26:23
# Pump Time Series
#   Cycle npts
B02    83    9
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    36.680  305   15.12     212   1137    -9     2
    87.400  134   15.24     245   1136    -9     7
    61.240   20   15.24     237   1137    -9    13
    33.840   20   15.24     231   1137    -9    13
    18.760   20   15.23     224   1136    -9    13
    11.800   20   15.22     217   1137    -9    13
     7.240   21   15.23     217   1137    -9    13
     2.960   20   15.21     215   1136    -9    13
     0.320  240   15.17     198   1137    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    83  2.4   12    90   125   600    80    10   180      8     1    4020    5  0.2505
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17   83 2022/09/12 14:30:49
# Pcase vacuums
  11.360  11.370
# Oil valve positions
   1.727   0.902   0.566
# Seek attempts and pump time
      1     -4
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18   83 2022/09/12 16:08:16 9999/99/99 99:99:99
# Pcase vacuums
  11.360  11.370
# Average 
#   P      T      S
   89.840   3.805  34.046
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18   83 2022/09/12 19:08:25 9999/99/99 99:99:99
# Pcase vacuums
  11.360  11.360
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20   83 2022/09/12 19:26:22
# number of packets previous packets
      6               7
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
  11.360  11.370
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.141   0.139   0.640
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.692   0.939
# Encountered Error Codes
#    Cycle Errors
EE20    83      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00   83  2      0
# MC    Pres      Time  Pcase Vacuum
   1     0.320    382  11.370
   3    78.160      3  11.370
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   83       197    2022/09/12 19:33:38
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =       83	(cycle number) |
     MPS =        0	(phase) |
     MPC =       83	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      125	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    56487	(vac 0 inHg cal) |
     PV5 =    31723	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =       82	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       36	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      304	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =       82	(cycle) |
  PTO[0] =    10800	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =       82	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =       82	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      340	(sec) |
     SSP =      240	(sec) |

