# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11339    65  214 000000000932510   448     0   6 2022/09/08, 17:29:52,  60.2666  -46.8410   3      0 6,14,14
SBD 11339    65  340 000000000932510   449     0   5 2022/09/08, 17:30:04,  60.2666  -46.8410   3      0 13
SBD 11339    65  299 000000000932510   450     0   4 2022/09/08, 17:30:17,  60.2666  -46.8410   3      0 8
SBD 11339    65  320 000000000932510   451     0   3 2022/09/08, 17:30:46,  60.2453  -46.7883   2      0 3,0,15,13
SBD 11339    65  271 000000000932510   452     0   2 2022/09/08, 17:31:33,  60.2453  -46.7883   2      0 1,2
SBD 11339    65  293 000000000932510   453     0   1 2022/09/08, 17:31:46,  60.2453  -46.7883   2      0 10,11,7,5,14
SBD 11339    65  306 000000000932510   454     0   0 2022/09/08, 17:32:01,  60.2666  -46.8410   2      0 0,4,14,14,9
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1959           84        2043
# Sensor_ID bytes     % packets Description
          0    48  2.45       2         GPS
          1    86  4.39       1    Pressure
          2   173  8.83       1 Temperature
          3   142  7.25       1    Salinity
          4   168  8.58       1        Fall
          5   113  5.77       1        Rise
          6   114  5.82       1        Pump
          7    36  1.84       1 Drift_times
          8   287 14.65       1     Surface
          9    58  2.96       1  HR/Drift_P
         10    58  2.96       1  HR/Drift_T
         11    55  2.81       1  HR/Drift_S
         13   404 20.62       2  Parameters
         14   151  7.71       5 Engineering
         15    66  3.37       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   65 1 2022/09/08 12:04 +60.22199  -46.79466    0  4 31 39 43  1.5     0
G   65 2 2022/09/08 17:24 +60.24283  -46.83537   70  4 30 39 43  1.7     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0    65  111    1     3         0
#      Pres      Temp    Salinity
      1.120     3.8220    30.9780
      2.040     3.5130    31.0930
      3.000     3.2250    31.1980
      4.000     3.2000    31.2210
      5.000     3.1460    31.2500
      6.000     3.1220    31.2640
      7.040     3.1180    31.2670
      8.080     3.1170    31.2660
      9.000     3.1100    31.2740
     10.000     3.0960    31.3040
     10.960     3.1070    31.3500
     12.040     3.1720    31.3950
     12.960     3.2120    31.4720
     14.000     3.1850    31.5610
     15.040     2.8850    31.7770
     15.920     2.7850    31.7970
     17.000     2.6580    31.8430
     18.000     2.5170    31.8910
     19.040     2.5330    31.9410
     20.040     2.5650    31.9780
     21.040     2.5930    31.9980
     22.000     2.6150    32.0180
     22.960     2.6420    32.0790
     24.040     2.5950    32.1440
     25.040     2.5800    32.1750
     26.000     2.5770    32.1950
     27.000     2.5130    32.3200
     28.000     2.5260    32.4050
     29.000     2.5550    32.4450
     30.040     2.5950    32.4950
     31.040     2.6590    32.5270
     32.000     2.6810    32.5350
     32.920     2.6870    32.5380
     34.000     2.6860    32.5390
     35.000     2.6610    32.5380
     36.000     2.6380    32.5340
     37.000     2.5630    32.5440
     38.040     2.4850    32.5510
     38.960     2.4720    32.5650
     39.960     2.4030    32.5720
     41.040     2.3860    32.5680
     42.000     2.3970    32.5670
     43.040     2.3960    32.5690
     44.000     2.4410    32.5620
     45.000     2.5140    32.5810
     46.040     2.3790    32.6000
     46.960     2.2630    32.5950
     48.000     2.0930    32.6090
     48.960     2.0170    32.6130
     50.000     1.9950    32.6270
     50.960     2.0170    32.6420
     52.040     2.1170    32.6740
     53.040     2.1160    32.6790
     54.040     2.1230    32.6810
     55.040     2.1170    32.7150
     55.960     2.0380    32.7230
     57.000     1.9140    32.7400
     57.960     1.9070    32.7460
     58.960     1.8990    32.7540
     59.960     1.9090    32.7660
     60.960     1.8910    32.7750
     61.960     1.8760    32.7810
     63.000     1.8810    32.7990
     64.040     1.9000    32.8120
     65.000     1.9330    32.8320
     66.000     1.9990    32.8740
     66.960     2.0250    32.8950
     68.000     2.0430    32.9140
     69.000     1.9800    32.9350
     70.000     1.9120    32.9430
     71.000     1.8980    32.9420
     72.000     1.9150    32.9400
     73.040     1.9090    32.9440
     74.000     1.9340    32.9560
     75.040     1.9400    32.9450
     76.000     1.9350    32.9420
     77.040     1.9160    32.9470
     78.040     1.8970    32.9550
     79.000     1.8940    32.9510
     79.960     1.9180    32.9540
     81.000     1.8190    32.9680
     82.000     1.7510    32.9730
     83.000     1.6130    32.9950
     84.000     1.5410    32.9950
     84.960     1.5170    33.0100
     86.000     1.5100    33.0270
     87.000     1.7110    33.0670
     88.000     1.7890    33.0840
     89.040     1.9330    33.1300
     90.000     1.8820    33.1600
     90.960     1.7160    33.1740
     92.040     1.5160    33.1760
     93.040     1.4340    33.1740
     94.000     1.4460    33.1740
     95.000     1.4260    33.2050
     96.040     1.4410    33.2390
     97.000     1.7110    33.2960
     98.040     1.9260    33.3820
     98.960     1.9820    33.4000
    100.040     2.0820    33.4280
    101.000     2.1250    33.4400
    101.960     2.2520    33.4810
    102.960     2.2000    33.4700
    104.000     2.2510    33.4920
    105.000     2.4060    33.5320
    106.000     2.4820    33.5590
    107.080     2.5010    33.5680
    108.040     2.5030    33.5680
    109.040     2.4990    33.5670
    110.040     2.4930    33.5660
    110.720     2.4950    33.5660
# Drift Data
#      C Npts comp n_var
I0    65   24    1     4
#      Pres       Temp   Salinity     Time
    116.400     2.2670    33.5350        6
    115.520     2.2360    33.5420      611
    113.200     2.1720    33.4730     1217
    112.200     2.2080    33.4990     1823
    120.680     2.6010    33.6150     2428
    118.360     2.5800    33.5950     3034
    112.040     2.2870    33.5380     3639
    118.840     2.3380    33.5400     4245
    124.680     2.3360    33.5430     4851
    119.760     2.3340    33.5440     5456
    116.560     2.3370    33.5510     6062
    115.960     2.2840    33.5330     6667
    116.720     2.3220    33.5460     7273
    112.800     2.3340    33.5510     7879
    114.160     2.3350    33.5430     8484
    115.080     2.3670    33.5500     9090
    110.760     2.3680    33.5510     9695
    110.840     2.4250    33.5400    10301
    110.080     2.4250    33.5300    10907
    106.200     2.4190    33.5610    11512
    104.600     2.6180    33.5720    12118
    102.600     2.4240    33.5560    12723
    109.680     2.4000    33.5440    13329
    106.400     2.4290    33.5640    13935
# Surface Pressure Time Series
#    C npts n_var               Start
S   65    31     6 2022/09/08 17:22:49
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    69      0.400      3.817   -999.000      14 2022/09/08 17:23:58
    82      0.480      3.822   -999.000      14 2022/09/08 17:24:11
    93      0.450      3.863   -999.000      14 2022/09/08 17:24:22
   104      0.350      3.834   -999.000      14 2022/09/08 17:24:33
   115      0.330      3.818   -999.000      14 2022/09/08 17:24:44
   126      0.310      3.838   -999.000      14 2022/09/08 17:24:55
   137      0.430      3.860   -999.000      14 2022/09/08 17:25:06
   148      0.340      3.849   -999.000      14 2022/09/08 17:25:17
   159      0.300      3.841   -999.000      14 2022/09/08 17:25:28
   171      0.340      3.863   -999.000      14 2022/09/08 17:25:40
   182      0.360      3.888   -999.000      14 2022/09/08 17:25:51
   193      0.340      3.856   -999.000      14 2022/09/08 17:26:02
   204      0.360      3.850   -999.000      14 2022/09/08 17:26:13
   215      0.310      3.868   -999.000      14 2022/09/08 17:26:24
   226      0.310      3.958   -999.000      14 2022/09/08 17:26:35
   237      0.300      3.976   -999.000      14 2022/09/08 17:26:46
   248      0.350      4.142   -999.000      14 2022/09/08 17:26:57
   259      0.340      3.962   -999.000      14 2022/09/08 17:27:08
   270      0.320      3.911   -999.000      14 2022/09/08 17:27:19
   281      0.350      3.890   -999.000      14 2022/09/08 17:27:30
   293      0.280      3.894   -999.000      14 2022/09/08 17:27:42
   304      0.380      3.850   -999.000      14 2022/09/08 17:27:53
   315      0.330      3.885   -999.000      14 2022/09/08 17:28:04
   326      0.330      3.818   -999.000      14 2022/09/08 17:28:15
   337      0.370      3.854   -999.000      14 2022/09/08 17:28:26
   348      0.330      4.051   -999.000      14 2022/09/08 17:28:37
   359      0.360      4.139   -999.000      14 2022/09/08 17:28:48
   370      0.350      3.961   -999.000      14 2022/09/08 17:28:59
   381      0.310      3.866   -999.000      14 2022/09/08 17:29:10
   392      0.280      3.801   -999.000      14 2022/09/08 17:29:21
   403      0.310      4.043   -999.000      14 2022/09/08 17:29:32
# Fall Time Series
#  Cycle npts     Start                Date
F00   65   32 715953875 2022/09/08 12:04:35
# time     Pres phase                Date
     0    0.320     1 2022/09/08 12:04:35
   158    5.120    10 2022/09/08 12:07:13
   190   10.440    10 2022/09/08 12:07:45
   219   15.160    10 2022/09/08 12:08:14
   252   20.400    10 2022/09/08 12:08:47
   281   25.080    10 2022/09/08 12:09:16
   313   30.400    10 2022/09/08 12:09:48
   343   35.160    10 2022/09/08 12:10:18
   353   36.920     2 2022/09/08 12:10:28
   375   40.440    10 2022/09/08 12:10:50
   404   45.280    10 2022/09/08 12:11:19
   437   50.480    10 2022/09/08 12:11:52
   469   55.240    10 2022/09/08 12:12:24
   505   60.160    10 2022/09/08 12:13:00
   547   65.240    10 2022/09/08 12:13:42
   589   70.120    10 2022/09/08 12:14:24
   641   75.240    10 2022/09/08 12:15:16
   744   80.040    10 2022/09/08 12:16:59
   882   85.080    10 2022/09/08 12:19:17
   997   90.040    10 2022/09/08 12:21:12
  1098   95.120    10 2022/09/08 12:22:53
  1220  100.000    10 2022/09/08 12:24:55
  1445  105.000    10 2022/09/08 12:28:40
  1637  110.000    10 2022/09/08 12:31:52
  1912  115.000    10 2022/09/08 12:36:27
  2212  115.200    10 2022/09/08 12:41:27
  2512  114.320    10 2022/09/08 12:46:27
  2784  113.400     3 2022/09/08 12:50:59
  2812  113.960    10 2022/09/08 12:51:27
  3114  116.920    10 2022/09/08 12:56:29
  3414  116.400    10 2022/09/08 13:01:29
  3421  116.400    10 2022/09/08 13:01:36
# Rise Time Series
#  Cycle npts     Start                Date
R00   65   21 715971705 2022/09/08 17:01:45
# time     Pres phase                Date
     0  110.360     7 2022/09/08 17:01:45
   155  104.960    13 2022/09/08 17:04:20
   237   98.400    13 2022/09/08 17:05:42
   299   93.200    13 2022/09/08 17:06:44
   360   87.920    13 2022/09/08 17:07:45
   442   81.680    13 2022/09/08 17:09:07
   503   75.240    13 2022/09/08 17:10:08
   544   70.880    13 2022/09/08 17:10:49
   606   65.360    13 2022/09/08 17:11:51
   667   59.400    13 2022/09/08 17:12:52
   729   53.240    13 2022/09/08 17:13:54
   790   47.040    13 2022/09/08 17:14:55
   852   41.040    13 2022/09/08 17:15:57
   893   36.440    13 2022/09/08 17:16:38
   934   31.720    13 2022/09/08 17:17:19
   975   26.960    13 2022/09/08 17:18:00
  1016   21.920    13 2022/09/08 17:18:41
  1077   15.520    13 2022/09/08 17:19:42
  1159    9.400    13 2022/09/08 17:21:04
  1221    4.120    13 2022/09/08 17:22:06
  1263    0.800     8 2022/09/08 17:22:48
# Pump Time Series
#   Cycle npts
B02    65   10
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    36.920  328   15.04     218   1136    -9     2
   110.360  135   15.19     249   1136    -9     7
   101.800   21   15.19     248   1136    -9    13
    96.640   21   15.19     247   1136    -9    13
    89.600   21   15.18     256   1136    -9    13
    84.600   21   15.18     264   1136    -9    13
    13.920   21   15.16     237   1136    -9    13
     9.400   21   15.17     224   1136    -9    13
     2.400   21   15.17     216   1136    -9    13
     0.280  210   15.12     196   1136    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    65  2.4   11    90   110   600    55    10   240      8     1    4020    5  0.2748
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17   65 2022/09/08 12:04:32
# Pcase vacuums
  11.360  11.360
# Oil valve positions
   1.729   0.903   0.577
# Seek attempts and pump time
      1      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18   65 2022/09/08 13:01:36 9999/99/99 99:99:99
# Pcase vacuums
  11.360  11.360
# Average 
#   P      T      S
  113.480   2.368  33.546
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18   65 2022/09/08 17:01:45 9999/99/99 99:99:99
# Pcase vacuums
  11.360  11.360
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20   65 2022/09/08 17:22:48
# number of packets previous packets
      7               7
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
  11.360  11.360
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.141   0.139   0.640
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.692   0.939
# Encountered Error Codes
#    Cycle Errors
EE20    65      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00   65  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.320    349  11.360
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   65       197    2022/09/08 17:30:46
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =       65	(cycle number) |
     MPS =        0	(phase) |
     MPC =       65	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      110	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    56487	(vac 0 inHg cal) |
     PV5 =    31723	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =       64	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       36	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      326	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     2100	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =       64	(cycle) |
  PTO[0] =    14400	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =       64	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =       64	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      340	(sec) |
     SSP =      210	(sec) |

