# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11339    58  236 000000000932510   399     0   6 2022/09/07, 03:00:13,  60.1263  -46.3748   4      0 6,14,14
SBD 11339    58  340 000000000932510   400     0   5 2022/09/07, 03:00:53,  60.1263  -46.3748   4      0 13
SBD 11339    58  299 000000000932510   401     0   4 2022/09/07, 03:01:21,  60.1048  -46.3228   4      0 8
SBD 11339    58  287 000000000932510   402     0   3 2022/09/07, 03:01:31,  60.1048  -46.3228   4      0 3,15,13
SBD 11339    58  270 000000000932510   403     0   2 2022/09/07, 03:02:06,  60.1048  -46.3228   3      0 1,2
SBD 11339    58  308 000000000932510   404     0   1 2022/09/07, 03:02:17,  60.1048  -46.3228   3      0 10,11,7,5,14,0
SBD 11339    58  295 000000000932510   405     0   0 2022/09/07, 03:02:42,  60.1263  -46.3748   3      0 0,4,14,14,9
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1951           84        2035
# Sensor_ID bytes     % packets Description
          0    48  2.46       2         GPS
          1    83  4.25       1    Pressure
          2   175  8.97       1 Temperature
          3   133  6.82       1    Salinity
          4   168  8.61       1        Fall
          5   123  6.30       1        Rise
          6   136  6.97       1        Pump
          7    36  1.85       1 Drift_times
          8   287 14.71       1     Surface
          9    47  2.41       1  HR/Drift_P
         10    47  2.41       1  HR/Drift_T
         11    47  2.41       1  HR/Drift_S
         13   404 20.71       2  Parameters
         14   151  7.74       5 Engineering
         15    66  3.38       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   58 1 2022/09/06 21:32 +60.12019  -46.33936    0  5 22 32 41  1.2     0
G   58 2 2022/09/07 02:53 +60.10804  -46.36309   20  6 30 38 45  1.3     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0    58  118    1     3         0
#      Pres      Temp    Salinity
      1.040     4.4620    30.5790
      1.960     4.4760    30.5780
      2.960     4.4770    30.5780
      4.000     4.4770    30.5780
      5.000     4.4760    30.5770
      6.040     4.4730    30.5750
      7.000     4.4720    30.5720
      7.920     4.4560    30.5690
      9.000     4.2000    30.6540
     10.040     3.7810    30.7480
     11.040     3.5670    30.7990
     12.000     3.5090    30.8650
     13.000     3.5060    30.8700
     13.960     3.5010    30.8720
     15.000     3.4600    30.9120
     16.000     3.3360    30.9810
     17.000     3.2490    30.9980
     18.000     3.2410    30.9980
     19.000     3.2380    30.9980
     20.000     3.2380    30.9980
     21.000     3.2780    31.0340
     22.000     3.2920    31.1320
     23.000     3.2400    31.1720
     24.000     3.1690    31.2240
     24.960     3.0770    31.3020
     26.000     3.0230    31.3350
     27.040     2.9580    31.3980
     28.000     2.9360    31.4190
     28.960     2.9560    31.4330
     30.000     2.9340    31.5620
     31.040     2.6840    31.6800
     31.960     2.4300    31.7440
     32.920     2.1300    31.8010
     34.000     2.0210    31.8380
     35.040     1.8350    31.8900
     36.000     2.1270    31.9440
     36.960     2.2360    31.9950
     38.040     2.0540    32.0110
     39.040     1.9640    32.0370
     40.000     1.8440    32.0480
     40.960     1.8990    32.0900
     42.000     1.7540    32.1060
     43.040     1.8310    32.1430
     44.040     1.9030    32.1910
     44.960     1.9520    32.2850
     45.960     1.8150    32.3540
     47.000     1.7810    32.3550
     47.960     1.7880    32.3630
     48.920     1.7850    32.3700
     50.040     1.7850    32.3710
     51.080     1.7850    32.3710
     52.000     1.7830    32.3730
     52.960     1.7790    32.3750
     53.960     1.7780    32.3770
     54.960     1.7770    32.3770
     56.040     1.7780    32.3760
     56.960     1.7770    32.3770
     58.040     1.7780    32.3770
     58.960     1.7780    32.3810
     60.000     1.7780    32.3820
     61.000     1.7780    32.3830
     62.040     1.7790    32.3830
     63.000     1.7800    32.3860
     64.000     1.7820    32.3870
     65.040     1.7880    32.3890
     66.000     1.7920    32.3940
     67.000     1.8020    32.4010
     68.000     1.8230    32.4120
     68.960     1.8180    32.4160
     69.960     1.8180    32.4180
     70.960     1.8200    32.4190
     71.960     1.8420    32.4250
     73.000     1.8450    32.4330
     73.960     1.8500    32.4400
     75.000     1.8720    32.4440
     75.960     1.9620    32.4560
     77.000     2.0840    32.4780
     78.000     2.0280    32.5060
     79.000     2.0020    32.5010
     80.040     1.9830    32.5030
     80.960     1.9550    32.5040
     82.000     1.9160    32.5060
     83.040     1.8630    32.5060
     84.000     1.8290    32.5040
     85.000     1.7310    32.5100
     86.000     1.6470    32.5080
     87.000     1.6320    32.5100
     87.960     1.6760    32.5150
     89.000     1.9200    32.5230
     90.000     1.9740    32.5480
     91.040     2.0000    32.5500
     92.080     1.9980    32.5580
     93.040     1.9730    32.5610
     94.040     1.9050    32.5650
     95.040     1.7940    32.5620
     96.040     1.6820    32.5560
     97.040     1.7040    32.5550
     98.000     1.7760    32.5650
     99.000     1.8490    32.5700
    100.000     1.9410    32.5860
    101.000     1.9480    32.5920
    102.000     1.9690    32.6020
    103.040     1.9850    32.6150
    104.040     2.0070    32.6370
    105.040     2.0220    32.6500
    106.040     2.0320    32.6560
    107.000     2.0560    32.6800
    108.000     2.0720    32.7060
    109.040     2.0860    32.7280
    110.040     2.1220    32.7570
    111.000     2.1610    32.7940
    112.040     2.2540    32.8750
    113.000     2.3790    32.9750
    114.000     2.4220    33.1040
    115.000     2.2310    33.2590
    116.000     2.1500    33.3030
    117.040     2.1090    33.3220
    118.120     2.1040    33.3250
# Drift Data
#      C Npts comp n_var
I0    58   24    1     4
#      Pres       Temp   Salinity     Time
    119.440     1.7790    33.4070        6
    119.120     1.7950    33.4040      612
    118.760     1.7850    33.4090     1217
    118.200     1.8300    33.3920     1823
    118.680     1.8090    33.4010     2428
    118.200     1.8210    33.3990     3034
    118.320     1.8580    33.3920     3640
    118.680     1.8790    33.3910     4245
    118.800     1.8930    33.3900     4851
    118.200     1.9190    33.3850     5456
    118.040     1.9210    33.3860     6062
    117.320     1.9290    33.3830     6668
    117.440     1.9420    33.3810     7273
    117.560     1.9540    33.3790     7879
    117.480     1.9580    33.3780     8484
    117.600     1.9790    33.3730     9090
    117.880     1.9960    33.3690     9696
    118.000     2.0250    33.3590    10301
    118.560     2.0320    33.3580    10907
    118.200     2.0420    33.3550    11512
    118.840     2.0670    33.3440    12118
    118.400     2.0650    33.3460    12724
    118.280     2.0750    33.3400    13329
    117.600     2.1030    33.3250    13935
# Surface Pressure Time Series
#    C npts n_var               Start
S   58    31     6 2022/09/07 02:53:07
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    70      0.400      4.426   -999.000      14 2022/09/07 02:54:17
    83      0.320      4.429   -999.000      14 2022/09/07 02:54:30
    94      0.340      4.438   -999.000      14 2022/09/07 02:54:41
   105      0.390      4.442   -999.000      14 2022/09/07 02:54:52
   116      0.350      4.443   -999.000      14 2022/09/07 02:55:03
   127      0.410      4.395   -999.000      14 2022/09/07 02:55:14
   138      0.400      4.427   -999.000      14 2022/09/07 02:55:25
   149      0.320      4.395   -999.000      14 2022/09/07 02:55:36
   161      0.380      4.411   -999.000      14 2022/09/07 02:55:48
   172      0.320      4.393   -999.000      14 2022/09/07 02:55:59
   183      0.360      4.338   -999.000      14 2022/09/07 02:56:10
   194      0.340      4.399   -999.000      14 2022/09/07 02:56:21
   205      0.380      4.364   -999.000      14 2022/09/07 02:56:32
   216      0.300      4.304   -999.000      14 2022/09/07 02:56:43
   227      0.280      4.247   -999.000      14 2022/09/07 02:56:54
   238      0.270      4.206   -999.000      14 2022/09/07 02:57:05
   249      0.330      4.172   -999.000      14 2022/09/07 02:57:16
   260      0.280      4.143   -999.000      14 2022/09/07 02:57:27
   271      0.350      4.115   -999.000      14 2022/09/07 02:57:38
   282      0.300      4.098   -999.000      14 2022/09/07 02:57:49
   294      0.340      4.083   -999.000      14 2022/09/07 02:58:01
   305      0.340      4.068   -999.000      14 2022/09/07 02:58:12
   316      0.360      4.063   -999.000      14 2022/09/07 02:58:23
   327      0.300      4.059   -999.000      14 2022/09/07 02:58:34
   338      0.290      4.052   -999.000      14 2022/09/07 02:58:45
   349      0.320      4.043   -999.000      14 2022/09/07 02:58:56
   360      0.320      4.035   -999.000      14 2022/09/07 02:59:07
   371      0.350      4.030   -999.000      14 2022/09/07 02:59:18
   382      0.360      4.026   -999.000      14 2022/09/07 02:59:29
   393      0.320      4.025   -999.000      14 2022/09/07 02:59:40
   404      0.350      4.024   -999.000      14 2022/09/07 02:59:51
# Fall Time Series
#  Cycle npts     Start                Date
F00   58   32 715815149 2022/09/06 21:32:29
# time     Pres phase                Date
     0    0.320     1 2022/09/06 21:32:29
   151    5.160    10 2022/09/06 21:35:00
   184   10.120    10 2022/09/06 21:35:33
   219   15.320    10 2022/09/06 21:36:08
   252   20.160    10 2022/09/06 21:36:41
   284   25.040    10 2022/09/06 21:37:13
   317   30.040    10 2022/09/06 21:37:46
   349   35.320    10 2022/09/06 21:38:18
   360   37.000     2 2022/09/06 21:38:29
   382   40.400    10 2022/09/06 21:38:51
   414   45.240    10 2022/09/06 21:39:23
   450   50.280    10 2022/09/06 21:39:59
   485   55.120    10 2022/09/06 21:40:34
   524   60.320    10 2022/09/06 21:41:13
   563   65.120    10 2022/09/06 21:41:52
   609   70.160    10 2022/09/06 21:42:38
   657   75.160    10 2022/09/06 21:43:26
   723   80.000    10 2022/09/06 21:44:32
   805   85.120    10 2022/09/06 21:45:54
   906   90.040    10 2022/09/06 21:47:35
  1016   95.120    10 2022/09/06 21:49:25
  1185  100.000    10 2022/09/06 21:52:14
  1370  105.000    10 2022/09/06 21:55:19
  1511  110.040    10 2022/09/06 21:57:40
  1682  115.000    10 2022/09/06 22:00:31
  1982  118.640    10 2022/09/06 22:05:31
  2282  119.360    10 2022/09/06 22:10:31
  2582  119.320    10 2022/09/06 22:15:31
  2784  119.160     3 2022/09/06 22:18:53
  2882  119.120    10 2022/09/06 22:20:31
  3182  118.920    10 2022/09/06 22:25:31
  3421  119.440    10 2022/09/06 22:29:30
# Rise Time Series
#  Cycle npts     Start                Date
R00   58   23 715832979 2022/09/07 02:29:39
# time     Pres phase                Date
     0  118.000     7 2022/09/07 02:29:39
   155  112.720    13 2022/09/07 02:32:14
   237  107.120    13 2022/09/07 02:33:36
   299  101.720    13 2022/09/07 02:34:38
   360   96.040    13 2022/09/07 02:35:39
   422   89.920    13 2022/09/07 02:36:41
   483   83.880    13 2022/09/07 02:37:42
   545   77.800    13 2022/09/07 02:38:44
   606   71.400    13 2022/09/07 02:39:45
   647   66.960    13 2022/09/07 02:40:26
   709   60.040    13 2022/09/07 02:41:28
   750   55.360    13 2022/09/07 02:42:09
   791   50.720    13 2022/09/07 02:42:50
   852   44.200    13 2022/09/07 02:43:51
   914   38.520    13 2022/09/07 02:44:53
   975   32.360    13 2022/09/07 02:45:54
  1037   27.320    13 2022/09/07 02:46:56
  1119   22.080    13 2022/09/07 02:48:18
  1201   16.520    13 2022/09/07 02:49:40
  1262   11.400    13 2022/09/07 02:50:41
  1324    5.520    13 2022/09/07 02:51:43
  1396    0.360    13 2022/09/07 02:52:55
  1406    0.440     8 2022/09/07 02:53:05
# Pump Time Series
#   Cycle npts
B02    58   12
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    37.000  321   15.01     225   1137    -9     2
   118.000  135   15.14     252   1136    -9     7
   111.360   20   15.14     261   1136    -9    13
   107.120   21   15.13     251   1136    -9    13
    99.960   21   15.14     251   1136    -9    13
    40.560   21   15.12     242   1136    -9    13
    28.800   21   15.12     248   1136    -9    13
    24.640   20   15.12     231   1136    -9    13
    21.000   21   15.11     228   1136    -9    13
    16.520   20   15.10     225   1136    -9    13
    11.400   21   15.10     228   1136    -9    13
     0.320  210   15.06     197   1136    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    58  2.4   11    90   110   600    55    10   240      8     1    4020    5  0.2748
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17   58 2022/09/06 21:32:26
# Pcase vacuums
  11.360  11.360
# Oil valve positions
   1.730   0.904   0.587
# Seek attempts and pump time
      1      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18   58 2022/09/06 22:29:30 9999/99/99 99:99:99
# Pcase vacuums
  11.360  11.360
# Average 
#   P      T      S
  118.240   1.936  33.377
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18   58 2022/09/07 02:29:39 9999/99/99 99:99:99
# Pcase vacuums
  11.360  11.360
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20   58 2022/09/07 02:53:06
# number of packets previous packets
      7               7
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
  11.360  11.360
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   1.141   0.139   0.640
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   1.692   0.939
# Encountered Error Codes
#    Cycle Errors
EE20    58      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00   58  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.320    356  11.360
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   58       197    2022/09/07 03:01:31
Parameter Table Param_Table-V17.txt |
     IFV = 10.2.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =       58	(cycle number) |
     MPS =        0	(phase) |
     MPC =       58	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =       99	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      110	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    56487	(vac 0 inHg cal) |
     PV5 =    31723	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =       57	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       36	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      321	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     2100	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        1	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =       57	(cycle) |
  PTO[0] =    14400	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =       57	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =       57	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      340	(sec) |
     SSP =      210	(sec) |

