# Message Decoder for Solo2, Alto and Deep Solo Floats; v250409
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11334    35  279 000000000000009  3544     0   0 2025/06/11, 00:23:21,  41.5334  -70.6398   3      0 0,4,14,14,5,14,0
SBD 11334    35  237 000000000000009  3550     0   4 2025/06/11, 03:06:18,  41.5334  -70.6398   5      0 13,6,14,14
SBD 11334    35  339 000000000000009  3551     0   3 2025/06/11, 03:06:42,  41.5334  -70.6398   5      0 13
SBD 11334    35  275 000000000000009  3552     0   2 2025/06/11, 03:07:21,  41.5334  -70.6398   4      0 8,15
SBD 11334    35  335 000000000000009  3553     0   1 2025/06/11, 03:07:37,  41.5334  -70.6398   3      0 8
# Delayed Transmission - (4/5) Messages transmitted   0.1 days after surfacing on 2025/06/11 00:09:15 (refer to 1st Rise scan < 3dbar)
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      15         1405           60        1465
# Sensor_ID bytes     % packets Description
          0    48  3.42       2         GPS
          4   138  9.82       1        Fall
          5    18  1.28       1        Rise
          6    37  2.63       1        Pump
          8   520 37.01       2     Surface
         13   415 29.54       2  Parameters
         14   163 11.60       5 Engineering
         15    66  4.70       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   35 1 2025/06/10 22:22 +41.53427  -70.64766    0  4  0  0  0  2.6     0
G   35 2 2025/06/11 00:09 +41.53435  -70.64755   30  6  0  0  0  1.8     0
# Surface Pressure Time Series
#    C npts n_var               Start
S   35    56     6 2025/06/11 00:09:32
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
     0      0.130     17.106      0.009      14 2025/06/11 00:09:32
    11      0.130     17.096      0.009      14 2025/06/11 00:09:43
    22      0.120     17.131      0.009      14 2025/06/11 00:09:54
    34      0.130     17.144      0.009      14 2025/06/11 00:10:06
    45      0.140     17.147      0.009      14 2025/06/11 00:10:17
    56      0.120     17.139      0.009      14 2025/06/11 00:10:28
    66      0.140     17.168      0.009      14 2025/06/11 00:10:38
    77      0.130     17.173      0.009      14 2025/06/11 00:10:49
    88      0.130     17.168      0.009      14 2025/06/11 00:11:00
    99      0.130     17.152      0.009      14 2025/06/11 00:11:11
   109      0.130     17.145      0.009      14 2025/06/11 00:11:21
   120      0.130     17.161      0.009      14 2025/06/11 00:11:32
   131      0.130     17.146      0.009      14 2025/06/11 00:11:43
   142      0.130     17.137      0.009      14 2025/06/11 00:11:54
   152      0.120     17.147      0.009      14 2025/06/11 00:12:04
   163      0.130     17.132      0.009      14 2025/06/11 00:12:15
   175      0.130     17.132      0.009      14 2025/06/11 00:12:27
   185      0.140     17.135      0.009      14 2025/06/11 00:12:37
   196      0.140     17.143      0.009      14 2025/06/11 00:12:48
   207      0.130     17.134      0.009      14 2025/06/11 00:12:59
   217      0.130     17.136      0.009      14 2025/06/11 00:13:09
   228      0.120     17.139      0.009      14 2025/06/11 00:13:20
   239      0.130     17.139      0.009      14 2025/06/11 00:13:31
   250      0.130     17.139      0.009      14 2025/06/11 00:13:42
   261      0.130     17.129      0.009      14 2025/06/11 00:13:53
   272      0.120     17.147      0.009      14 2025/06/11 00:14:04
   282      0.140     17.140      0.009      14 2025/06/11 00:14:14
   293      0.130     17.157      0.009      14 2025/06/11 00:14:25
   304      0.130     17.170      0.009      14 2025/06/11 00:14:36
   315      0.130     17.175      0.009      14 2025/06/11 00:14:47
   326      0.120     17.177      0.009      14 2025/06/11 00:14:58
   337      0.130     17.166      0.009      14 2025/06/11 00:15:09
   347      0.140     17.149      0.009      14 2025/06/11 00:15:19
   358      0.130     17.149      0.009      14 2025/06/11 00:15:30
   370      0.130     17.127      0.009      14 2025/06/11 00:15:42
     0      0.140     17.109      0.009      14 2025/06/11 00:15:52
    11      0.130     17.117      0.009      14 2025/06/11 00:16:03
    22      0.130     17.127      0.009      14 2025/06/11 00:16:14
    33      0.120     17.126      0.009      14 2025/06/11 00:16:25
    43      0.130     17.144      0.009      14 2025/06/11 00:16:35
    55      0.130     17.152      0.009      14 2025/06/11 00:16:47
    66      0.120     17.133      0.009      14 2025/06/11 00:16:58
    76      0.130     17.139      0.009      14 2025/06/11 00:17:08
    87      0.130     17.146      0.009      14 2025/06/11 00:17:19
    98      0.130     17.142      0.009      14 2025/06/11 00:17:30
   109      0.130     17.150      0.009      14 2025/06/11 00:17:41
   120      0.130     17.151      0.009      14 2025/06/11 00:17:52
   131      0.130     17.145      0.009      14 2025/06/11 00:18:03
   141      0.130     17.152      0.009      14 2025/06/11 00:18:13
   152      0.130     17.151      0.009      14 2025/06/11 00:18:24
   163      0.130     17.143      0.009      14 2025/06/11 00:18:35
   174      0.140     17.158      0.009      14 2025/06/11 00:18:46
   184      0.130     17.146      0.009      14 2025/06/11 00:18:56
   195      0.130     17.168      0.009      14 2025/06/11 00:19:07
   206      0.130     17.145      0.009      14 2025/06/11 00:19:18
   217      0.130     17.151      0.009      14 2025/06/11 00:19:29
# Fall Time Series
#  Cycle npts     Start                Date
F00   35   26 802909361 2025/06/10 22:22:41
# time     Pres phase                Date
     0    0.040     1 2025/06/10 22:22:41
   302    0.080    10 2025/06/10 22:27:43
   604    0.120    10 2025/06/10 22:32:45
   904    0.120    10 2025/06/10 22:37:45
  1206    0.120    10 2025/06/10 22:42:47
  1506    0.120    10 2025/06/10 22:47:47
  1804    0.120     2 2025/06/10 22:52:45
  1807    0.120    10 2025/06/10 22:52:48
  2107    0.120    10 2025/06/10 22:57:48
  2407    0.120    10 2025/06/10 23:02:48
  2707    0.120    10 2025/06/10 23:07:48
  3007    0.120    10 2025/06/10 23:12:48
  3307    0.120    10 2025/06/10 23:17:48
  3607    0.120    10 2025/06/10 23:22:48
  3627    0.120     3 2025/06/10 23:23:08
  3907    0.120    10 2025/06/10 23:27:48
  4207    0.120    10 2025/06/10 23:32:48
  4507    0.120    10 2025/06/10 23:37:48
  4558    0.120     3 2025/06/10 23:38:39
  4807    0.120    10 2025/06/10 23:42:48
  5107    0.120    10 2025/06/10 23:47:48
  5407    0.120    10 2025/06/10 23:52:48
  5707    0.120    10 2025/06/10 23:57:48
  6007    0.120    10 2025/06/11 00:02:48
  6307    0.120    10 2025/06/11 00:07:48
  6391    0.120    10 2025/06/11 00:09:12
# Rise Time Series
#  Cycle npts     Start                Date
R00   35    2 802915755 2025/06/11 00:09:15
# time     Pres phase                Date
     0    0.120     7 2025/06/11 00:09:15
     2    0.120     8 2025/06/11 00:09:17
# Pump Time Series
#   Cycle npts
B02    35    3
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
     0.120   22   15.25     185    550    -9     2
     0.120    1   15.31     223    550    -9     7
     0.000   61   15.29     178    592    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    35  2.4    4  2000    10   600    31   180  2880      5     2       2   15  2.5841
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17   35 2025/06/10 22:22:40
# Pcase vacuums
   7.080   5.500
# Oil valve positions
   1.650   0.725   0.670
# Seek attempts and pump time
      2      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18   35 2025/06/11 00:09:12 9999/99/99 99:99:99
# Pcase vacuums
   5.500   5.500
# Average 
#   P      T      S
    0.000   0.000   0.000
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18   35 2025/06/11 00:09:15 9999/99/99 99:99:99
# Pcase vacuums
   5.500   5.500
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20   35 2025/06/11 00:09:18
# number of packets previous packets
      5              62
# CTD Status pres offset
 0x64697361   0.000
# Pcase vacuums
   5.500   7.710
# Air Pump max current time  max pressure
      40       608     6.704
# Air Valve Positions
   1.246   0.000
# Air Valve Calibrations
   1.194   0.160   0.677
# Oil Valve Positions
   1.654   0.717
# Oil Valve Calibrations
   1.628   0.762
# Encountered Error Codes
#    Cycle Errors
EE20    35      2
#               Date   Code Description
 2025/06/11 00:09:13 0x0615 ParkDepthMin
 2025/06/11 00:09:14 0x0615 ParkDepthMin
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00   35  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.040   1802   5.480
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   35       200    2025/06/11 03:06:42
Parameter Table Param_Table-V18.txt |
     IFV = 10.10.0.1	(IFV) |
     VER =       18	(version) |
     STY =        0	(Sensor 0=RBR,1=SBE) |
     MSN =       35	(cycle number) |
     MPS =        2	(phase) |
     MPC =       23	(count) |
  MNM[0] =        6	(count) |
  MNM[1] =        6	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       10	(dBar) |
  DTD[1] =       10	(dBar) |
  DTD[2] =       10	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        2	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        1	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =       20	(dBar) |
  CPN[2] =       10	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =     1000	(ms) |
  CPP[2] =     1000	(ms) |
  CPP[3] =     1000	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =        2	(dBar) |
     MTT =    30000	(ms) |
     MTS =    10800	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        0	(sec) |
    MAT2 =        0	(sec) |
     MAP =        0	(sec) |
     ACA =        0	(gain) |
     ACG =        0	(scale) |
     ACR =        0	(sampling rate) |
     PNB =    43668	(V) |
     PNC =    50958	(inHg) |
     PXC =        0	(inHg) |
     PBL =    25960	(12V cal) |
     PBH =     1790	(14V cal) |
     PP0 =     2089	(oil pump 0 mA cal) |
     PP1 =    61614	(oil pump 1000 mA cal) |
     PVL =    32000	(V12 12V cal) |
     PVH =        0	(V12 14V cal) |
     PV0 =    16600	(vac 0 inHg cal) |
     PV5 =        0	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =     1280	(V12_sig 1000mA cal) |
     ICM =        5	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        1	(ice month mask) |
  ICA[0] =        1	(summer attempts) |
  ICA[1] =       70	(winter attempts) |
  ICB[0] =       70	(summer cycles) |
  ICB[1] =      180	(winter cycles) |
  IMT[0] =      180	(summer message threshold) |
  IMT[1] =       20	(winter message threshold) |
  ILT[0] =       40	(summer time threshold) |
  ILT[1] =    65371	(winter time threshold) |
     ICN =       30	(shallow limit for test dBar) |
     ICX =      514	(deep limit for test dBar) |
     ICT =     2.56	(deg C) |
     ICS =     5120	(sec) |
  ISS[0] =      180	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        0	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     2.56	(PSU) |
     ISO =    25600	(sec) |
     ISP =      100	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =       30	(summer descent depth dBar) |
  ICD[1] =        0	(winter descent depth dBar) |
  ICR[0] =  1966080	(summer minutes) |
  ICR[1] = 16779780	(winter minutes) |
  ICG[0] =      176	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =       30	(attempts) |
     ISD =     1200	(dBar) |
     IAT =       45	(secs) |
     ISV =       45	(secs) |
     ISW =        0	(secs) |
  ICO[0] =        0	(summer secs) |
  ICO[1] =        0	(winter secs) |
     ICF =       34	(summer mask) |
     ICW =  1310723	(winter ice mask) |
     DNS =        3	(cycle number) |
     DRA =        3	(target descent rate dBar/sec) |
  DSD[0] =     1800	(dBar) |
  DSD[1] =     1800	(dBar) |
  DSD[2] =     1800	(dBar) |
  DST[0] =      300	(sec) |
  DST[1] =        0	(sec) |
  DST[2] =        0	(sec) |
     DDF =     2000	(sec) |
     DDC =        1	(sec) |
     DSI = 19660820	(ms) |
     DDD =       20	(dBar) |
     DDT =       20	(sec) |
  DFT[0] =      257	(sec) |
  DFT[1] =     1800	(sec) |
  DFT[2] =     1800	(sec) |
     DCM =        7	(0=dis 1=en) |
     DAL =        8	(0=cal 1=seek) |
  DCT[0] =      512	(sec) |
  DCT[1] =      768	(sec) |
  DCT[2] =        3	(sec) |
     DCA =      252	(max attempts) |
     DRC =        2	(cal rate calculation) |
     DRT =        2	(sec) |
     DCX =      200	(sec) |
     DCP =        0	(sec) |
     DCF =      900	(sec) |
     DCD =       50	(dBar) |
     DCR =       10	(dBar/sec) |
     DSW =       10	(sec) |
     DUC =        5	(under damping factor) |
     DPT =       30	(sec) |
     DUT =        5	(sec) |
     DPD =      900	(sec) |
     DVT =       19	(sec) |
     DVD =       16	(dBar) |
     DVW =        3	(sec) |
     PFR =       60	(cc/min) |
     PFC =       34	(.001 cc/dBar) |
     DDP =        0	(dBar) |
     DDS =    10800	(sec) |
     PNS =        0	(cycle) |
  PTO[0] =        2	(sec) |
  PTO[1] = -1560281088	(sec) |
  PTO[2] =        1	(sec) |
  PTH[0] =  1048576	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =      256	(24 LSB) |
     PTD = 153600000	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI = 16777479	(ms) |
  PFM[0] =        8	(0=dis 1=en) |
  PFM[1] =       42	(0=dis 1=en) |
  PFM[2] =       48	(0=dis 1=en) |
  PFT[0] =      500	(sec) |
  PFT[1] =       20	(sec) |
  PFT[2] =     2000	(sec) |
  PFD[0] =        1	(dBar) |
  PFD[1] =        5	(dBar) |
  PFD[2] =        5	(dBar) |
  PFV[0] =     1800	(sec) |
  PFV[1] =      256	(sec) |
  PFV[2] =      200	(sec) |
     PDT =    10800	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =     7682	(dBar) |
     PST =    22535	(sec) |
     PSV =       34	(sec) |
     PSW =        5	(sec) |
     PSS =        9	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =    24576	(cycle number) |
     ARR =     5120	(target rise rate cm/sec) |
     APM =     2560	(sec) |
     APS =     5120	(sec) |
     APN =       39	(sec) |
     APF =     4096	(sec) |
     ASI = 16777473	(ms) |
     ASD =    11268	(dBar) |
     ADD =    19459	(dBar) |
     ADT =    59392	(sec) |
     ARC =      512	(?) |
     AHR =     8959	(cm/sec) |
     AHD =    62976	(dBar) |
     SFN =     3840	(cycle number) |
     SDC =     7680	(dBar) |
     SGT =      300	(min) |
     SET =        0	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =       39	(sec) |
     SSA =     4096	(0=dis 1=en) |
     SSI = 167845196	(ms) |
     SDS =     6500	(sec) |
     SAM =        4	(0=dis 1=en) |
     SAX =    14338	(.001 psi) |
     SAN =    22528	(.001 psi) |
     SAO =    15360	(sec) |
     SPT =        0	(sec) |
     SSP =        0	(sec) |
     PPT =   0.0000	(1) |
     PPA =   0.0000	(1) |
     PPD =   0.0000	(sec) |

