# Message Decoder for Solo2, Alto and Deep Solo Floats; v240529
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11334   -10  276 000000000000009  3014     0   3 2024/09/06, 20:31:11,  41.5334  -70.6398   3      0 14
SBD 11334   -10  339 000000000000009  3015     0   2 2024/09/06, 20:32:02,  41.5334  -70.6398   5      0 13
SBD 11334   -10  124 000000000000009  3016     0   1 2024/09/06, 20:34:47,  41.5334  -70.6398   2      0 0,13
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0       4          703           36         739
# Sensor_ID bytes     % packets Description
          0    24  3.41       1         GPS
         13   415 59.03       2  Parameters
         14   264 37.55       1 Engineering
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  -10 5 2024/09/06 20:30 +41.53431  -70.64695   10  8 33 41 47  1.0     0
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   -10  2.4    1   150   150   180   360     0     0     11     1    2160    0  0.2908
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS           25      10      1000       5      1000       1 
# Engineering Data
#       Alto BIT Test
#    cycle software   BIT         Total
#          version    result      time
E011  -10  0x0a090001 0x00000000  0
 # Skyview           BIT_Start             BIT_End
       255 1970/01/01 00:00:00 2024/09/06 20:11:24
# Successful Bit Test
# CTD Status temp    salinity
 0x3c42f838   0.241  25.156
# Pressure offsets 
# before   now     after [dBar]
 518043012298528968933376.000    0.231    0.000
# Pcase vacuum [inHg]
 19366159889262464513920741670912.000   5.240
# Battery Voltages [V]
   5.710  15.330
# Air Pump test
# Air Valve Positions [V]
  15.210   1.190   0.161
# Air Valve Calibrations [V]
   1.248   0.102   0.678
# duration count max P max currents      max voltage
   16172   247 2503081204659994885581963264.00    7.52    0.05    0.03
# Oil Pump test
# Oil Valve Positions [V]
  15.320   1.618
# Oil Valve Calibrations [V]
   1.660   0.716
# duration max currents [mAmp]      max voltage [V]
   16195     -0.00    0.20         0.07
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  -10         0    2024/09/06 20:34:47
