# Message Decoder for Solo2, Alto and Deep Solo Floats; v240919
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11332   154  300 000000000954970   125     0   2 2024/10/21, 07:14:33, -41.3222  174.8315   2      0 13,6,14,14
SBD 11332   154  340 000000000954970   126     0   1 2024/10/21, 07:14:53, -41.3222  174.8315   3      0 13
SBD 11332   154  270 000000000954970   127     0   0 2024/10/21, 07:15:06, -41.3222  174.8315   3      0 0,4,14,14,7,5,14,0,15
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      14          874           36         910
# Sensor_ID bytes     % packets Description
          0    48  5.49       2         GPS
          4    38  4.35       1        Fall
          5    18  2.06       1        Rise
          6    26  2.97       1        Pump
          7    25  2.86       1 Drift_times
         13   421 48.17       2  Parameters
         14   232 26.54       5 Engineering
         15    66  7.55       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  154 1 2024/10/20 10:12 -41.28291 +174.00412    0  5 28 34 40  1.3     0
G  154 2 2024/10/21 07:12 -41.30344 +174.80585   40  5 29 33 38  1.6     0
# Drift Data
#      C Npts comp n_var
I0   154    1    1     4
#      Pres       Temp   Salinity     Time
   -999.000  -999.0000  -999.0000 15337895
# Fall Time Series
#  Cycle npts     Start                Date
F00  154    1 9223372036854 9999/99/99 99:99:99
# time     Pres phase                Date
 9223372036854    0.000     1 9999/99/99 99:99:99
# Rise Time Series
#  Cycle npts     Start                Date
R00  154    1 9223372036854 9999/99/99 99:99:99
# time     Pres phase                Date
 9223372036854    0.000     7 9999/99/99 99:99:99
# Pump Time Series
#   Cycle npts
B02   154    2
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
     0.000  199   15.21     159    555    -9     2
     0.000   11   15.33     161    535    -9     7
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   154  2.4    2  2000  1000   600   483   360 12240      9     3       3  180  9.7177
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  154 2024/10/20 10:13:01
# Pcase vacuums
   8.770   5.340
# Oil valve positions
   1.682   0.808   0.585
# Seek attempts and pump time
      3   -150
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  154 2024/10/21 06:40:53 9999/99/99 99:99:99
# Pcase vacuums
   5.340   5.340
# Average 
#   P      T      S
 9999.000  99.000 -999.000
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  154 2024/10/21 07:11:27 9999/99/99 99:99:99
# Pcase vacuums
   5.340   5.350
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  154 2024/10/21 07:11:49
# number of packets previous packets
    573              48
# CTD Status pres offset
 0x00000000   0.000
# Pcase vacuums
   5.350   5.740
# Air Pump max current time  max pressure
      50        51     2.575
# Air Valve Positions
   1.185   0.000
# Air Valve Calibrations
   1.119   0.117   0.618
# Oil Valve Positions
   0.000   0.824
# Oil Valve Calibrations
   1.666   0.841
# Encountered Error Codes
#    Cycle Errors
EE20   154     10
#               Date   Code Description
 2024/10/20 10:13:12 0x080C CtdRetry
 2024/10/20 10:13:12 0x0113 DescendCtd
 2024/10/20 10:13:24 0x080C CtdRetry
 2024/10/20 10:13:24 0x0113 DescendCtd
 2024/10/20 10:13:35 0x080C CtdRetry
 2024/10/20 10:13:35 0x0113 DescendCtd
 2024/10/20 10:13:49 0x080C CtdRetry
 2024/10/20 10:13:49 0x0113 DescendCtd
 2024/10/20 10:14:02 0x080C CtdRetry
 2024/10/20 10:14:02 0x0113 DescendCtd
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  154  2      0
# MC    Pres      Time  Pcase Vacuum
   1     0.000   1803   5.320
   3     0.000     59   5.480
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  154       201    2024/10/21 07:14:53
Parameter Table Param_Table-V17.txt |
     IFV = 10.6.3.1	(IFV) |
     VER =       17	(version) |
     STY =        0	(Sensor 0=RBR,1=SBE) |
     MSN =      154	(cycle number) |
     MPS =        2	(phase) |
     MPC =      151	(count) |
  MNM[0] =        1	(count) |
  MNM[1] =        2	(count) |
  MNM[2] =     1000	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      500	(dBar) |
  DTD[1] =     1000	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        1	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2010	(dBar) |
  CPB[2] =     1010	(dBar) |
  CPB[3] =      310	(dBar) |
  CPN[1] =       20	(dBar) |
  CPN[2] =       20	(dBar) |
  CPN[3] =       10	(dBar) |
  CPP[1] =     1000	(ms) |
  CPP[2] =     1000	(ms) |
  CPP[3] =     1000	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   600000	(ms) |
     MTS = 15552000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =      800	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    55179	(vac 0 inHg cal) |
     PV5 =    31478	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =    -1.65	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      153	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =      200	(dBar) |
  DSD[1] =       50	(dBar) |
  DSD[2] =       20	(dBar) |
  DST[0] =     3600	(sec) |
  DST[1] =     1800	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      360	(sec) |
     DDC =        0	(sec) |
     DSI =     2000	(ms) |
     DDD =       20	(dBar) |
     DDT =     1800	(sec) |
  DFT[0] =      220	(sec) |
  DFT[1] =      190	(sec) |
  DFT[2] =      190	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     7200	(sec) |
  DCT[1] =    25200	(sec) |
  DCT[2] =    25200	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65386	(sec) |
     DCX =     1020	(sec) |
     DCP =        2	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =    10800	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =      100	(dBar) |
     DDS =     3600	(sec) |
     PNS =      153	(cycle) |
  PTO[0] =    10800	(sec) |
  PTO[1] =   129600	(sec) |
  PTO[2] =   734400	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  1800000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =    10800	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =     1000	(dBar) |
  PFD[1] =     1500	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =      420	(sec) |
  PFV[2] =      600	(sec) |
     PDT =     1800	(sec) |
     PSM =        1	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      153	(cycle number) |
     ARR =        9	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =      100	(sec) |
     ASI =    20000	(ms) |
     ASD =        2	(dBar) |
     ADD =       20	(dBar) |
     ADT =     1800	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      153	(cycle number) |
     SDC =    65526	(dBar) |
     SGT =       15	(min) |
     SET =       15	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =     1000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =       60	(sec) |
     PPT =   0.0000	(1) |
     PPA =   0.0000	(1) |
     PPD =   0.0000	(sec) |
     PPC =   0.0000	(1) |

