# Message Decoder for Solo2, Alto and Deep Solo Floats; v221025
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11331    -4   36 000000000000009    16     0   0 2022/10/28, 17:49:20,  41.5334  -70.6398   4      0 0
SBD 11331    -4   78 000000000000009    17     0   0 2022/10/28, 17:49:29,  41.5334  -70.6398   4      0 15
SBD 11331    -4   88 000000000000009    18     0   0 2022/10/28, 17:49:46,  41.5334  -70.6398   3      0 14
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0       3          166           36         202
# Sensor_ID bytes     % packets Description
          0    24 14.46       1         GPS
         14    76 45.78       1 Engineering
         15    66 39.76       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   -4 5 2022/10/28 17:29 +41.53433  -70.64765   20  3 39 43 47  0.7     0
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    -4  2.4    1   500    10   600     6     0     0      5     2       2   15  0.1408
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
#       Alto BIT Test
#    cycle software   BIT         Total
#          version    result      time
E008   -4  0x0a040500 0x00000000  1006
# Successful Bit Test
# CTD Status temp    salinity
 0x706f7473  13.209   0.007
# Pressure offsets 
# before   now     after [dBar]
    0.000    0.000    0.000
# Pcase vacuum [inHg]
   5.700   6.530
# Battery Voltages [V]
  15.450  15.350
# Air Pump test
# Air Valve Positions [V]
   1.196   0.066   0.719
# Air Valve Calibrations [V]
   1.138   0.122   0.630
# duration count max P max currents      max voltage
     360     1    7.51   52.00    0.00   15.42
# Oil Pump test
# Oil Valve Positions [V]
   1.800   0.905
# Oil Valve Calibrations [V]
   1.754   0.950
# duration max currents [mAmp]      max voltage [V]
     600    225.00   82.00         1.53
